Abstract:
Aqueous solutions containing buffers and electrolytes are adjusted to levels required for calibration of both blood gas analyzers and ion selective electrolyte analyzers. A control material, composed of similar matrix, is adjusted to three levels of blood gas and electrolyte conditions. These quality control materials are used to monitor blood gas/electrolyte laboratory instrumentation.

Description:
This is a continuation of application Ser. No. 07/359,250, filed May 31, 1989, now abandoned, which is a division of application Ser. No. 07/121,499, filed Nov. 17, 1987, now abandoned. 
    
    
     FIELD OF THE INVENTION 
     This invention relates to aqueous solutions containing buffers and electrolytes adjusted to specific levels for calibration or quality control of chemical analyzers. In particular, this invention relates to a solution which has the specific parameters for the calibration and quality control of both blood gas and ion selective electrode instrumentation. 
     BACKGROUND OF THE INVENTION 
     Analytical devices capable of determining various parameters in whole blood with the aid of ion-selective electrodes have been used in intensive care units and in the field of anaesthetics. Blood gases, such as carbon dioxide and oxygen, can be measured using ion selective electrodes. An ion selective electrode consists of an electrochemical half-cell (an internal electrolyte solution and an internal reference electrode) and a sensing membrane. These types of electrodes are used to measure one type of ion in the presence of other ions in the bodily fluids. To obtain accurate measurements, a set of calibration solutions are used to calibrate the electrodes. 
     In addition to calibration, the tests run on clinical chemical analyzers are verified by specific control solutions. Under ideal conditions, control materials used in the verification of instrumentation performance should have essentially the same constituents as those present in the fluid which is to be subjected to analysis. 
     Most blood gas instruments are currently calibrated using a buffer for the pH and humidified gases for carbon dioxide and oxygen. Some instruments use tonometer buffer inside the instrument and use this liquid for equilibration. Blood gas instruments that also measure electrolytes have separate solutions for calibration and control of the ion selective electrodes. The electrolyte calibrator and control solutions do not currently contain all the blood gas parameters. 
     BRIEF SUMMARY OF THE INVENTION 
     The present invention relates to calibrating and control solutions for instrumentation which measures both blood gas and electrolytes and a method of using such solutions with blood gas instruments. This invention relates to aqueous solutions containing buffers and electrolytes adjusted to specific levels for calibration or quality control blood gas instruments. In particular, the present invention relates to a novel blood gas-electrolyte instrument control solution having three levels. The three levels simulate three different patient conditions: one normal and two abnormal. Level one has low electrolyte parameters, acidosis pH and gas parameters which represent respiratory acidosis, i.e. low O 2  and high CO 2  concentrations. Level two has normal conditions in all parameters. Level three has high electrolyte parameters, alkalosis pH, and gas parameters which represent respiratory alkalosis, i.e. low CO 2  and high O 2  concentrations. 
     Additionally, the present invention relates to a novel blood gas electrolyte instrument calibrating solution that contains parameters necessary for the calibration of both blood gas and ion selective electrode instrumentation. In particular, the present calibrating solution contains two levels having the necessary parameters. Level one has normal pH, pCO 2  and pO 2  and electrolytes. Level two is the zero oxygen calibrator. It has physiologically high concentrations of CO 2 , physiologically low electrolyte levels and zero O 2 . 
     DESCRIPTION OF THE PREFERRED EMBODIMENTS 
     The blood gas-electrolyte calibrating solutions of this invention are prepared by conventional manufacturing techniques using readily available materials. A typical calibrating solution with the contemplation of this invention is prepared by simply combining in appropriate relative proportions buffer, electrolytes, and other standard auxiliary reagents. 
    
    
     EXAMPLE I 
     A Diamond Sensor System GEM 6 Blood Gas Electrolyte monitor is calibrated with a calibrating solution having two levels. The calibrating solution is comprised of an aqueous biological buffer with a preservative, electrolytes, and blood gas components. A detergent is added for use as a wetting agent. 
     Level one has normal pH, pCO 2 , pO 2 . Sulfate ions are used to keep the calcium in solution. Level two is the &#34;0&#34; calibrator. It has low pH, physiologically high concentrations of CO 2  and zero O 2  and physiologically low electrolyte levels. The O 2  is kept at zero, even in the presence of room air, with sufficient sulfite and nitrogen. 
     
                       TABLE I______________________________________           LEVEL 1 LEVEL 2______________________________________NaCL (mMol/L)     --        46CaCl.sub.2 (mMol/L)             3.8       0.3KCL (mMol/L)      6.9       2.2Sulfite (mMol/L)  --        40Sulfate (mMol/L)  26        --CO.sub.2 (mmHg)   35        65O.sub.2 (mmHg)    115-170   03-(N-Morpholino) propane             100       100sulfonic acid (mMol/L)pH (37° C.)             7.4       6.9Na.sup.+  (mMol/L)             135       150K.sup.+  (mMol/L) 6.0       2Ca.sup.2+  (mMol/L)             2.0       0.2CL.sup.-  (mMol/L)             120       80Ionic Strength    160       160______________________________________ 
    
     Table I lists the composition of the two levels of the blood gas electrolyte calibrating solution together with concentrations and pH values. The pH of the solution is adjusted and sodium bicarbonate is added to reach the desired carbon dioxide level. Albumin may be added if a protein matrix is desired. 
     The electrolytes used in this control solution may be: sodium, potassium, calcium, and chloride. Lithium is not added to the calibrating solutions or controls for the DSS GEM 6 instrument due to interference with the sensors, although, with other instruments lithium may be used. The buffer used is designed to give the appropriate PH value by combining the acid and salt forms. This buffer was selected due to the very small temperature effect demonstrated by the buffer. Various solutions having the above discussed levels can be prepared for as many calibration points as necessary for the instrumentation. Host instrumentation use a one or two point calibration for sloping electrodes. 
     The final solution is sterile filtered and is stored in gas tight containers without a gaseous headspace. In the final container no further equilibrium with gases will take place and the final solution is therefore not affected by pressure and temperature. 
     The addition of calcium as an electrolyte and bicarbonate as a buffer for pH and carbon dioxide, creates conditions for the formation of calcium carbonate. Calcium carbonate is an insoluble precipitate, which will eventually remove all the calcium from solution and therefore make the calcium no longer measurable as an ion. To prevent this from occurring sulfate ions are added which protect the calcium ion from contact with the carbonate ion. This protection is necessary when high levels of both calcium and carbon dioxide are desired. 
     The Level I solution is tonometered with 100% nitrogen until the oxygen level is less than 10 mmHg. Sulfite ions are then added to scavenge the remaining oxygen. The sulfite is added in excess so that residual sulfite is available to scavenge oxygen that may contact the solution. This enables the solution to maintain a zero oxygen value. The bicarbonate is then added to create the desired pCO 2  value. 
     The solutions are then tonometered using the appropriate gases needed to reach the requested oxygen and carbon dioxide values (the final solutions are sterile filtered and stored in gas tight containers without headspace). No further equilibrium with gases takes place and the final solution is not affected by pressure and temperature. The solution is stable in the final container and is ready for use with a blood gas/electrolyte analyzer. 
     EXAMPLE II 
     The blood gas-electrolyte controls of this invention are prepared by conventional manufacturing techniques using readily available materials. A typical control within the contemplation of this invention is prepared by simply combining in appropriate relative proportions, buffer, electrolytes, foaming agent and other standard auxiliary reagents and chemicals; the pH and concentration of dissolved carbon dioxide thereof being adjusted to the appropriate value, depending on the natures, the control, an aliquot thereof placed in a sealable receptacle; the receptacle flushed with gas; and the receptacle sealed so as to create a headspace above control fluids which is occupied by the gas. Albumin may be added if a protein matrix is desired. Glycine is added to bind a portion of the calcium to provide a total calcium value that reflects normal values for Level II and abnormal values for Levels I and III. The glycine also prevents precipitation of the remaining ionized calcium. 
     The performance of a Diamond Sensor Gem 6 Blood Gas-Electrolyte monitor is verified using a blood gas-electrolyte control. The controls are comprised o&amp; three levels: normal and two abnormal conditions. Level one, an abnormal control, having low electrolyte levels, acidosis pH and gas levels which represent respiratory, acidosis, level two, normal parameters and level three, high levels of electrolytes, alkalosis pH and gas levels which represent respiratory alkalosis, were formed in the presence of the biological buffer N-2-hydroxyethylpiperazine-N 1  -2-ethane sulfonic acid. This buffer was selected due to the temperature effect demonstrated by the buffer. The temperature coefficient (change in pKa per degree C=0.014), enables the control to indicate a malfunction of the instrument temperature regulation. The material is packaged in glass ampules which have been flushed with the appropriate gases. This ampule must be shaken prior to use in a blood gas-electrolyte analyzer. The shaking equilibrates the gas with the liquid and also creates a foam layer. The foam layer will protect the solution from contamination with room air gases for 4 minutes after opening the ampule. 
     
                       TABLE II______________________________________        LEVEL 1 LEVEL 2   LEVEL 3______________________________________NaCL (mMol/L)  41        48        79CaCl.sub.2 (mMol/L)          2.1       2.9       3.4KCL (mMol/L)   2.5       4.2       6.8Glycine (mMol/L)          639       599       532O.sub.2 (mmHg) 40        100       150CO.sub.2 (mmHg)          60        40        13N-2-hydroxyethylpiper-          50        50        50azine-N&#39;-2-ethane sulfonicacid (mMol/L)pH (37° C.)          7.16      7.4       7.6Na.sup.+  (mMol/L)          120       138       150K.sup.+  (mMol/L)          2.5       4         6.8Ca.sup.2+  (mMol/L)          0.9       1.15      1.5Total Ca (mg/dl)          8.5       11.5      13.5CL.sup.-  (mMol/L)          86        102       116Tonic Strength 160       160       160______________________________________ 
    
     Table II lists the composition of the three levels of the blood gas-electrolyte control solution together with concentrations and pH values. 
     The calibrator and control solutions are currently used to calibrate and control the Diamond Sensor Systems GEM-6 blood gas electrolyte monitor. The two level calibrator solutions are used to perform two calibration points on the instrumentation. The calibration points at 37 C. are: 
     
         ______________________________________                      HIGHPARAMETER*  LOW CALIBRATOR CALIBRATOR______________________________________pH          6.9 ± 0.02  7.4 ± 0.02pCO.sub.2   65 mmHg ± 3.0                      35 mmHg ± 3.0pO.sub.2    0 mmHg ± 5.0                      115 ± 5.0K.sup.+     2.0 mM/L ± 0.3                      6.0 mM/L ± 0.3Ca.sup.++   0.2 mM/L ± 0.1                      2.0 mM/L ± 0.1______________________________________ *Stated values are at atmospheric pressure, temperature coefficient of pH = -.01 pH unit/C. 
    
     EXAMPLE 3 
     The control formulations by typical blood gas electrolyte instrumentation. Table III shows the results of some of these tests. 
     
                                           TABLE II__________________________________________________________________________Typical Values ExpectedAqueous Blood Gas/Electrolyte ControlParameter    Level        Instrument                Mean                    Expected Range                             1SD__________________________________________________________________________pH       I   Il 1303 7.16                    7.14-7.18                             0.002pH       I   Corning 178                7.17                    7.15-7.19                             0.006pH       I   NOVA Stat                7.15                    7.13-7.17                             0.001pH       II  Il 1303 7.39                    7.37-7.41                             0.004pH       II  Corning 178                7.41                    7.39-7.43                             0.002pH       II  NOVA Stat                7.40                    7.38-7.42                             0.006pH       III Il 1303 7.63                    7.61-7.65                             0.002pH       III Corning 178                7.63                    7.61-7.65                             0.001pH       III NOVA Stat                7.64                    7.62-7.66                             0.003pCO.sub.2 mmHg    I   Il 1303 64  59-69    1.0pCO.sub.2 mmHg    I   Corning 178                67  62-72    2.8pCO.sub.2 mmHg    I   NOVA Stat                66  61-71    1.0pCO.sub.2 mmHg    II  Il 1303 36  32-40    0.2pCO.sub.2 mmHg    II  Corning 178                36  32-40    0.4pCO.sub.2 mmHg    II  NOVA Stat                36  32-40    0.8pCO.sub.2 mmHg    III Il 1303 11.0                     9-13    0.1pCO.sub.2 mmHg    III Corning 178                8.0  6-10    0.1pCO.sub.2 mmHg    III NOVA Stat                11.0                     9-13    0.1pO.sub.2 mmHg    I   Il 1303 42  34-50    1.4pO.sub.2 mmHg    I   Corning 178                41  33-49    2.0pO.sub.2 mmHg    I   NOVA Stat                42  34-50    1.3pO.sub.2 mmHg    II  Il 1303 104  96-112  2.0pO.sub.2 mmHg    II  Corning 178                106  98-114  3.0pO.sub.2 mmHg    II  NOVA Stat                102  94-110  3.9pO.sub.2 mmHg    III Il 1303 180 172-188  2.5pO.sub.2 mmHg    III Corning 178                183 175-191  3.1pO.sub.2 mmHg    III NOVA Stat                189 183-199  2.5Sodium mM/L    III NOVA 6  139 134-144  0.8Sodium mM/L    III NOVA Stat                141 136-146  1.0Sodium mM/L    III NOVA 10 144 139-149  1.0Potassium mM/L    I   NOVA 6  2.4 2.1-2.7  0.1Potassium mM/L    I   NOVA Stat                2.5 2.2-2.8  0.1Potassium mM/L    I   NOVA 10 2.4 2.1-2.7  0.1Potassium mM/L    II  NOVA 6  4.0 3.7-4.3  0.1Potassium mM/L    II  NOVA Stat                3.8 3.5-4.1  0.1Potassium mM/L    II  NOVA 10 3.8 3.5-4.1  0.1Potassium mM/L    III NOVA 6  6.4 6.1-6.7  0.1Potassium mM/L    III NOVA Stat                6.9 6.6-7.2  0.1Potassium mM/L    III NOVA 10 6.5 6.2-6.8  0.1iCa mM/L I   NOVA 6  0.8 0.6-1.0  0.1iCa mM/L I   NOVA Stat                0.8 0.6-1.0  0.1iCa mM/L II  NOVA 6  1.1 0.9-1.3  0.1iCa mM/L II  NOVA Stat                1.1 0.9-1.3  0.1iCa mM/L III NOVA 6  1.6 1.4-1.8  0.1iCa mM/L III NOVA Stat                1.6 1.4-1.8  0.1Chloride mM/L    I   NOVA Stat                96  93-99    0.4Chloride mM/L    I   NOVA 10 96  93-99    0.2Chloride mM/L    II  NOVA Stat                108 104-112  1.0Chloride mM/L    II  NOVA 10 108 104-112  0.1Chloride mM/L    III NOVA Stat                124 120-128  0.3Chloride mM/L    III NOVA 10 124 121-129  0.1Total Ca mg/L    I   NOVA 10 0.9 0.7-1.1  0.2Total Ca mg/L    II  NOVA 10 1.0 0.8-1.2  0.2Total Ca mg/L    III NOVA 10 1.8 1.6-2.0  0.2Lithium mM/L    I   Corning Flame                0.6 0.4-0.8  0.3Lithium mM/L    II  Corning Flame                1.7 1.5-1.1  0.3Lithium mM/L    III Corning Flame                3.0 2.8-3.2  0.3__________________________________________________________________________ 
    
     While these calibrating and control systems are particularly advantageous in biological fluid analysis systems such as Diamond Sensor Systems GEM®-6 it will be apparent that other biological fluid analysis systems can use the presently described calibrating and control solutions. Therefore, while particular embodiments of the invention have been shown and described, various modifications will be apparent to those skilled in the art and therefore is not intended that the invention be limited to the disclosed embodiments or to details thereof and departures may be made there from within the spirit and scope of the invention.