Abstract:
A scanning microscope having a laser outputting an excitation laser beam and a fiber member having a first core and a second core. The second core is generally disposed within the first core and is operable to receive the excitation laser beam from the laser and transmit the excitation laser beam to a sample to be tested. A moveable stage supports an end of the fiber member and/or a sample to be tested and is operable to move the end of the fiber member and the sample to be tested relative to each other.

Description:
CROSS-REFERENCE TO RELATED APPLICATIONS  
       [0001]     This application claims the benefit of U.S. Provisional Application No. 60/474,113, filed on May 29, 2003. The disclosure of the above application is incorporated herein by reference. 
     
    
     GOVERNMENT RIGHTS  
       [0002]     This invention was made with government support under Contract No. N01-CO-27173 awarded by the National Cancer Institute, National Institutes of Health. The Government has certain rights in the invention. 
     
    
     FIELD OF THE INVENTION  
       [0003]     The present invention relates to scanning microscopes and, more particularly, to a scanning microscope using a double-clad fiber for improved detection efficiency.  
       BACKGROUND AND SUMMARY OF THE INVENTION  
       [0004]     Confocal microscopy was first invented by Marvin Minsky in 1957. Initially, stage scanning was employed to obtain an image by moving a specimen in a raster pattern across a focused point of a stationary light beam. An exit pinhole is placed in the image plane conjugated to the spot being scanned so that only the signal originating from the focused spot is transmitted through the pinhole, while out-of-focus signals are prevented from reaching a detection system. A confocal microscope therefore generally has a higher resolution than a wide-field microscope, and more importantly, it has sectioning capability to achieve a 3-D image. The benefit of stage scanning is that the field of view is vastly expanded, because the area that can be imaged is determined by the travel range of the scanning stage rather than the optics in the microscope. However, there are drawbacks that limit its application. The scanning rate is extremely slow because it requires time to translate the massive stage precisely. In addition, a moving stage causes vibration problems to samples, especially liquid-bathed biological samples. These problems were circumvented when laser scanning confocal microscope was developed into a practical instrument in the late 1980s, where beam scanning was controlled by two galvanometer mirrors that are imaged onto the entrance pupil of an objective lens. Thus, only the direction of the incident excitation light rays is deviated at the entrance plane, while the pupil remains fully illuminated throughout the scanning. Accordingly, the focus spot of the laser beam out of the objective scans across the sample to be imaged. The beam scanning with the galvanometer mirrors has much higher scanning rates, and samples are not disturbed by vibrations, because there is no movement of sample stage. However, the beam scanning also has its own shortcomings. Because the incident angle of the laser beam on the entrance pupil of an objective lens has to vary in a certain range, associated aberrations are inevitable even with an expensive high quality objective lens. In addition, the field of view is severely limited by the acceptable angle of the objective. Although the invention of multiphoton confocal microscopes enhanced the detection efficiency by omitting the exit pinholes, the basic scanning mechanism remains the same as previous confocal microscopes.  
         [0005]     As briefly mentioned above, both stage- and beam-scanning confocal microscopes have their own disadvantages, despite the fact that they are indispensable tools in many research fields, especially in biological studies. In the present invention, a novel double-clad fiber based scanning confocal microscope, which possesses the advantages of both stage- and beam-scanning configurations, while overcomes all the main disadvantages of conventional confocal microscopes. In addition, the present invention further provides important new features, such as increased flexibility and low cost.  
         [0006]     According to the principles of the present invention, a scanning microscope having a laser outputting laser energy is provided. A fiber member having a first core and a second core is coupled to the laser. The second core of the fiber member is generally disposed within the first core, which also acts as the first cladding for the second core. The second core is sized smaller than the first core. The first core is surrounded by a second cladding. An opposing end of the fiber member is mounted to a moveable stage for movement therewith.  
         [0007]     Further areas of applicability of the present invention will become apparent from the detailed description provided hereinafter. It should be understood that the detailed description and specific examples, while indicating the preferred embodiment of the invention, are intended for purposes of illustration only and are not intended to limit the scope of the invention. 
     
    
     BRIEF DESCRIPTION OF THE DRAWINGS  
       [0008]     The present invention will become more fully understood from the detailed description and the accompanying drawings, wherein:  
         [0009]      FIG. 1  is a graph illustrating that the detected two-photon fluorescence power through a double-clad fiber in comparison with a single-mode fiber;  
         [0010]      FIG. 2  is a schematic view illustrating a double-clad fiber scanning microscope of the present invention; and  
         [0011]      FIG. 3  illustrates the calculated results when a GRIN lens is used to focus excitation beam and collect flourescence or other resultant signal. The spot size of the collected fluorescence is about 49 μm in radius, which is larger than the output excitation beam at the fiber tip;  
         [0012]      FIG. 4  is a perspective view of a double-clad fiber array used as a scanning head. 
     
    
     DETAILED DESCRIPTION OF THE PREFERRED EMBODIMENT  
       [0013]     The following description of the preferred embodiment is merely exemplary in nature and is in no way intended to limit the invention, its application, or uses.  
         [0014]     In contrast to conventional beam scanning, which includes changing the angle of an incident beam at an objective lens, the beam scanning of the present invention can be achieved by moving an optical fiber, which delivers a laser beam for excitation and collects signals back along the same fiber. Conventional fibers, either single-mode or multimode fibers, cannot be practically used in this way. Although a single-mode fiber (SMF) has an acceptable mode for excitation, the numerical aperture (NA) is typically only about 0.1, which results in a very inefficient signal collection. On the other hand, although a multimode fiber multimode fiber has a larger numerical aperture that is good for collecting signals, the output mode is unable to be tightly focused, thus resulting in inefficient excitation and low resolution. In addition, in case of multiphoton excitation, the multimode fiber leads to further lower excitation rate, because an ultra short laser pulse is severely deformed during propagating through a multimode fiber.  
         [0015]     In order to address this trade-off issue for biosensing, a double-clad fiber may be used for enhancing both excitation and collection efficiency for through-fiber biosensing as described in U.S. Provisional Application No. 60/434,604. This application is incorporated herein by reference. In that application, two-photon fluorescence detection sensitivity, represented by line A, is increased by a factor of 40 using a photonic crystal double-clad fiber in comparison with a conventional SMF, represented by line B (see  FIG. 1 ).  
         [0016]     Referring to  FIG. 2 , a schematic diagram of a double-clad fiber scanning microscope, generally indicated at  10 , is illustrated. Although it should be understood that alternative configurations might also be possible based on this double-clad fiber scanning mechanism. Double-clad fiber scanning microscope  10  is illustrated having a laser  12  capable of outputting a laser beam  14 , which will also be referenced as excitation laser beam. Laser  12  is operably coupled to a double-clad fiber or fiber member  16  via a fiber coupler  18 . More specification, double-clad fiber  16  includes an inner core  20 , an outer core  22 , and an outer cladding  24 . Inner core  20  is illustrated being coaxial with each of outer core  22  and outer cladding  24 ; however, it should be understood that this is not required. It should be noted that outer core  22  also serves as an inner cladding to inner core  20  and, thus, serves a dual purpose. As will be seen in  FIG. 3 ( a ), it should be understood that double-clad fiber  16  may be a fiber member system  16 ′ comprised of a plurality of fibers  16 .  
         [0017]     Laser  12  is coupled with double-clad fiber  16  through fiber coupler  18  such that laser beam  14  is introduced into inner core  20  at a proximal end  26  of double-clad fiber  16 . A distal end  28  of double-clad fiber  16  is coupled to a 3-D rapid scanning stage  30  that is operable to move laser beam  14 , exiting distal end  28  of double-clad fiber  16 , across a sample of interest  32 . A micro-lens  34 , such as a GRIN lens, may be attached to distal end  28  of double-clad fiber  16  to focus laser beam  14  to an even smaller spot to achieve higher resolution. Resultant signals, such as, but not limited to, flourescence signals, Raman signals, back reflection of the laser beam  14 , and the like), emitted from sample of interest  32  are then collected back through both inner core  20  and outer core  22  of double-clad fiber  16  and separated from excitation laser beam  14  using an optical separation system  36 , such as a dichroic mirror, before reaching an optical detection system  38 . A filter  40  may also be used for filtering undesirable signals from reaching optical detection system  38 .  
         [0018]     With respect to double-clad fiber  16 , the numerical apertures of the inner core and outer core (inner clad) can be adjusted independently. The outer core numerical aperture can be as large as about 0.8 or even just in air, which is comparable with most high magnification objective lenses. Furthermore, when a lens, such as a gradient index (GRIN) lens, is connected with double-clad fiber  16  to further focus excitation light, the collection efficiency of fluorescence signals received back from the lens to the double-clad fiber is high, because the larger outer core can efficiently collect fluorescence even if chromatic aberration of the lens exists. The resultant signal collection efficiency is low if a conventional fiber is used in this case.  
         [0019]     FIGS.  3 ( a )-( e ) illustrate that the collected fluorescence from a GRIN lens forms a large spot on distal end  28  of double-clad fiber  16 . That is, as seen in  FIG. 3 ( a ), fiber member system  16 ′ includes the aforementioned plurality of inner cores  20  and outer cores  22 . As seen in  FIG. 3 ( b ), when an excitation beam  300  exits double-clad fiber  16  it passes through a lens  34 , such as a GRIN lens, and is focused on sample  32 . The excitation beam  300  causes a resultant signal  302  to be produced from sample  32  generally indicated in  FIG. 3 ( c ). This resultant signal  302  may, for example, have a radius of about 1 μm. However, as seen in  FIG. 3 ( d ), resultant signal  302  then passes back through lens  34 . Ideally, resultant signal  302  would be focused perfectly on distal end  28  of double-clad fiber  16 . However, due to chromatic aberration and/or other anomalies, a larger footprint of resultant signal  302  is produced and may have a radius of about 49 μm. In conventional collection, this larger footprint would not be collected and thus would reduce the efficiency of the system. However, in the present invention, outer core  22 , having a high numerical aperture, is capable of collecting more of resultant signal  302 , thereby providing improved detection efficiency.  
         [0020]     As should be appreciated, double-clad fiber scanning microscope  10  of the present invention provides a number of advantages over conventional scanning microscopes. For example, as described above, double-clad fiber scanning microscope  10  has extremely simple structure.  
         [0021]     However, it has revolutionary and fundamental changes of the scanning mechanism, which ensures many unique features of this new type of scanning microscope.  
         [0000]     Excellent Flexibility  
         [0022]     Double-clad fiber scanning microscope  10  of the present invention is extremely flexible. More particularly, double-clad fiber scanning microscope  10  can be freely adjusted without affecting the excitation source and the detection, because the scanning head containing distal end  28  of double-clad fiber  16  is controlled by small translation (i.e. x-y or x-y-z) of scanning stage  30  through a single fiber. Thus, scan, imaging can be performed in either upright or inverted configurations, or at an arbitrary angle, if needed. Scanning stage  30  can also easily achieve any scanning pattern on a sample of interest. Still further, scanning stage  30  can be used to construct a stand-alone microscope together with an excitation. source and detection system. It can also be used as a unit to be incorporated into a conventional light microscope. For instance, scanning stage  30  can be made as a standard component to be screwed in a nosepiece. Thus, one can easily convert a conventional microscope into a scanning microscope with the beneficial functions as described herein.  
         [0000]     Large Scanning Range  
         [0023]     Unlike conventional beam scanning microscope, the scanning range of double-clad fiber scanning microscope  10  is determined by the travel range of scanning stage  30  used to control distal end  28  of double-clad fiber  16 . In fact, it has been found that this travel range may be increased to millimeters or larger while maintaining high resolution, such as less than a micron. This feature allows one to obtain a whole image of a large sample. For example, a conventional beam-scanning microscope has a scanning range only on a cellular scale due to the limited field of view of the objective lens. In contrast, the new beam-scanning mechanism based on double-clad fiber  16  makes it possible to image a whole organism or a tumor with a single scan.  
         [0000]     Fast Scanning  
         [0024]     Fast scan rate is required. for constructing a practical instrument. For conventional stage-scanning microscope, the scan rate is normally very slow, because it takes time to translate a massive stage together with a sample and sample holder. The scanning mechanism described herein only involves moving a lightweight fiber tip. Similar to scanner mirrors used in beam scanning, the fiber tip can scan in a fast rate with a rapid scanner.  
         [0000]     No Vibrations to the Sample  
         [0025]     Despite the fast scan rate noted above, there is no vibration disturbing the imaging sample, because the sample remains stationary during the scanning process, which is in contrast to stage scanning. Beside the light weight of the fiber tip, this is another practical reason that fast scan rate is allowed here. In addition, far field excitation from a fiber tip is utilized here to achieve a quiet beam scan, which avoids an inevitable problem in near field scanning optical microscopy where interaction between a scanning tip and samples is generally a serious problem.  
         [0000]     Aberration-free Scanning  
         [0026]     In conventional beam scanning, two scanner mirrors are used to change the incident angle of excitation light at the entrance pupil of an objective lens, which causes severe off-axis aberrations. It is very difficult and costly to design and fabricate an objective lens that is corrected for the off-axis aberrations. Moreover, even with a lot of effort, one still must compromise between the field of view and the image quality, because the off-axis aberration is hard to be fully compensated, especially for a relatively large fields of view. The scanning of excitation beam with flexible double-clad fiber  16  fundamentally solved the problem of aberrations associated with conventional beam scanning. In double-clad fiber scanning microscope  10 , each scanned point of a sample is equally illuminated and signal collection remains the same throughout the entire scanning range. This feature ensures a high quality image of a large sample of interest.  
         [0000]     Low Cost  
         [0027]     The cost for constructing double-clad fiber scanning microscope  10  is much lower than a conventional beam-scanning microscope with a scan unit based on- scanner mirrors. As described above, the requirement of an objective lens is important in order to achieve a relatively large flat field of view and to compensate for off-axis aberrations. In addition, an imaging system with high optical quality is also needed to image the scanner mirrors onto the entrance pupil of the objective lens. These factors make a conventional beam-scanning microscope very expensive.  
         [0028]     In contrast, in double-clad fiber scanning microscope  10 , the objective lens used in fiber coupler  18  solely focuses light onto proximal end  26  of double-clad fiber  16 . Thus, the objective lens in fiber coupler  18  satisfies the requirements, yet may be manufactured relatively inexpensively. The beam scanning is achieved by controlling distal end  28  of fiber double-clad fiber  16  with a scanning stage  30 , which replaces the expensive scanning unit composed of scanner mirrors and a high quality imaging system used in conventional beam-scanning microscope. Therefore, the new scanning mechanism based on double-clad fiber  16  makes it possible to construct a low cost, high performance microscope.  
         [0000]     C. Double-clad Fiber Array Scanning Microscope  
         [0029]     In the above, a double-clad fiber scanning microscope  10  utilizing a single double-clad fiber  16  is discussed. However, it has been determined that the scanning rate can be further enhanced by using a 1-D or 2-D array, generally indicated at  200 , of double-clad fibers  16 , as illustrated in  FIG. 4 .  
         [0030]     Excitation light can be coupled into double-clad fiber array  200  utilizing existing techniques, such as a MEMS switch. When double-clad fiber array  200  scans simultaneously instead of scanning a single fiber, the scan rate increases by a factor of the number of double-clad fibers in the array. For example, employing five double-clad fibers  16  aligned with 1 mm spacing between each other and mounted on a single translation stage  30 , a 5-mm line to be scanned only requires a translation of 1 mm. Thus, the scan rate increases by five times compared with a single fiber scanning. If a 2-D array of double-clad fibers is used, one should be able to maintain a high scan rate even for a large imaging area.  
         [0031]     A novel mechanism for a new generation of scanning microscopes based on double-clad fiber scanning is provided. This microscope overcomes the drawbacks of conventional stage- and beam-scanning microscopes, and possesses many advantages as described above, i.e., excellent flexibility, large scanning range, fast scan rate, quiet scanning, aberration-free scanning, and low cost. With all these benefits integrated into one microscope, a wide range of potential applications is anticipated.  
         [0032]     The description of the invention is merely exemplary in nature and, thus, variations that do not depart from the gist of the invention are intended to be within the scope of the invention. Such variations are not to be regarded as a departure from the spirit and scope of the invention.