Abstract:
A wound treatment apparatus that ventilates a wound dressing with warm humidified oxygen and/or air at a very low flow rate while under a sub-atmospheric pressure. A controlled flow of filtered air and/or oxygen is selected or blended under atmospheric pressure. This gas is pulled by a downstream vacuum source through a flow limiting device and exits at a sub-atmospheric pressure equal to the vacuum source. The gas flow is then humidified and heated to prescribed conditions before entering a substantially airtight negative pressure type wound dressing which is adapted for ventilation with an added inlet connection. This conditioned gas ventilates and nourishes the wound environment exiting the dressing outlet, through a fluid trap, to the vacuum source.

Description:
CROSS-REFERENCE TO RELATED APPLICATION 
       [0001]    Not applicable. 
       STATEMENT REGARDING FEDERALLY SPONSORED RESEARCH OR DEVELOPMENT 
       [0002]    Not applicable. 
       BACKGROUND OF THE INVENTION 
       [0003]    Most wounds are small and heal quickly. This invention, however, is concerned with healing chronic wounds. A chronic wound is a wound that fails to heal in an orderly set of stages and in a reasonable amount of time. These wounds are difficult to treat, requiring much skill and many resources, and which may take months or even years to heal. Wounds that don&#39;t heal properly can lead to amputation and even loss of life. Common types of chronic wounds are leg ulcers, pressure ulcers (bedsores) and diabetic foot ulcers. 
         [0004]    Leg ulcers occur due to impairment of the circulation in the leg. When veins are not functioning properly, the tissue surrounding those veins no longer receives an adequate supply of fresh blood. Oxygen cannot reach those tissues, leading to an open wound. 
         [0005]    Pressure ulcers can occur in people of any age and are typically caused by staying in the same position for a prolonged period of time. Lying in bed or sitting in a wheelchair can put sustained pressure on the skin over a bony prominence. This pressure can cut off the blood supply and lead to a pressure ulcer or bedsore. 
         [0006]    Diabetic foot ulcers are a common complication of diabetes. A diabetic can be unaware of a minor foot injury due to the loss of pain sensation associated with neuropathy. Continued walking on an unnoticed injury causes increased trauma and can lead to more serious injury that can become chronic. 
         [0007]    Several million people are afflicted with chronic wounds every year. Elderly people have thinner skin which is more easily damaged; bed-ridden elderly people are at a particularly high risk. Chronic wounds that take months to heal or those that never heal, cause severe pain and hardship with much diminished quality of life. Complex wounds such as surgical wounds with complications, and burn wounds, also require additional procedures to achieve wound closure. All of these wounds represent major health burdens and drains on healthcare resources. 
         [0008]    There are many known methods to treat chronic or complex wounds. Those relevant to this invention are: topical oxygen therapy; negative pressure wound therapy; moist wound therapy; and localized warming therapy. 
       Topical Oxygen Therapy 
       [0009]    Topical oxygen therapy uses pure oxygen applied directly to the surface of the wound at a pressure slightly greater than atmospheric. The oxygen is usually supplied to an inflatable, see-through plastic bag with edges that adhere to the skin. The air-tight devices can be in the shape of a sleeve, boot or pouch. Oxygen is both a nutrient and an antibiotic, and an adequate supply is vital for each of the steps required for wound healing. Oxygen facilitates healing by suppressing the proliferation of bacteria, promoting tissue granulation, and accelerating epithelialization. 
         [0010]    Stivala U.S. Pat. No. 4,224,941 and Loori U.S. Pat. No. 4,801,291 are good examples of topical oxygen therapy. Products using topical oxygen include O2Boot® and O2Sacral® by GWR Medical, Inc. and O2Misly™, by IYIA Technologies. Topical oxygen is not administered continuously as excess may cause other problems. Therapy is usually administered 90 minutes per day for four consecutive days, followed by a three day break. This protocol is then repeated as necessary until the wound is healed. Other oxygen-based therapies include hyperbaric oxygen chamber treatment, in which the oxygen is inspired; and the use of self-contained oxygen producing dressings. 
       Negative Pressure Wound Therapy 
       [0011]    Negative pressure wound therapy applies localized negative pressure (vacuum) to the wound bed, promoting healing by sucking air and exudate from the wound. Excess fluids or exudate are known to be detrimental to the healing process. According to Kinetic Concepts, Inc., the other benefits of negative pressure wound therapy are: promoting the reduction in local edema; reducing infection risk; and stimulating the growth of healthy, vascularized granulation tissue. 
         [0012]    Kinetic Concepts, Inc. markets the V.A.C.® (Vacuum Assisted Closure) System. This system uses a foam dressing cut to fit the size and shape of the wound, which is inserted into the wound. One end of a plastic tube is inserted into the foam dressing, while the other end is connected to a vacuum pump. An occlusive drape covers the wound and exiting tube, adhering to the skin surrounding the wound, making the dressing air-tight. When the pump is turned on, the negative pressure draws the edges of the wound together, and aspirates excess fluids from the foam dressing. The fluids are collected in a canister. Kinetic Concepts, Inc. is the assignee on numerous negative pressure wound therapy patents. One of the more recent is Argenta et al. U.S. Pat. No. 7,216,651. 
         [0013]    Quisenberry et al. U.S. Pat. Application Publication No. US2008/0071330 A1 discloses a negative pressure wound therapy dressing which is combined with a pad comprising a plurality of light emitting diodes for treating the wound area. Karpowicz et al. U.S. Pat. Application No. US 2007/0219532A1 discloses a negative pressure pump system with a reference airflow. 
       Moist Wound Therapy 
       [0014]    Moist wound therapy has been accepted as a major advance in the treatment of chronic wounds for more than 40 years. Any therapy that dehydrates the wound, such as dry gauze dressings, heat lamps, or exposure to air, is detrimental to chronic wound healing. When wounds dry out they form a scab which slows down the healing process. The principal function of a wound dressing is to provide a moist healing environment. Healing is stimulated and wounds heal faster if moisture is provided or maintained at the wound. Saline-soaked gauze that is kept moist is an effective wound dressing. There may be a delicate balance between keeping a wound moist and removing/absorbing excessive fluid from the wound. Excess fluid can delay healing and contribute to a maceration or breakdown of the skin. 
         [0015]    Hundreds of dressings that help to create a moist wound environment are currently available. Examples of technologies that use moist wound healing are hydrogels, hydrocolloids, alginates, foams and films. Bogart U.S. Pat. No. 5,512,041 entitled “Wound Dressing for Promoting Moist Wound Healing” is one such example. This dressing is semi-occlusive, meaning it is vapor permeable and impermeable to liquids and bacteria. This dressing also has a pad which absorbs excess wound fluids (wound exudate). Henley et al. U.S. Pat. No. 7,276,051 entitled “Wound Treatment Apparatus” discloses a fluid supply and a fluid drainage in communication with a bandage cavity. 
       Localized Warming Therapy 
       [0016]    Localized warming therapy, the use of mild heat for the treatment of chronic wounds, is known to have beneficial effects. A chronic wound is lacking in blood supply and typically becomes hypothermic. Even healthy extremities can be cool and their wounds more hypothermic. Hypothermia slows cellular function and biochemical reactions and is detrimental to healing. Mild heating dilates blood vessels, stimulating local circulation, increasing oxygen perfusion in tissues and accelerating wound healing. 
         [0017]    People have used hot water bottles, heat lamps and heating pads for years, but accurate temperature control is lacking with these devices. Normal body temperature is 37 degrees C. Hyperthermia, higher than normal body temperatures, can be detrimental to the wound. Temperatures above 45 degrees C. can seriously damage healthy cells. Accurate control of temperature is very important to insure that no harm is done. 
         [0018]    Augustine U.S. Pat. No. 6,241,697 discloses an electrically powered noncontact radiant heater enclosed within a wound dressing. An alternate embodiment uses heated air which passes through a dressing. 
         [0019]    There are hundreds of companies, producing thousands of wound care products, using over a dozen different technologies. Besides those therapies cited above, other technologies include: antimicrobial silver dressings; collagen dressings; growth factors; skin substitutes; gene therapy; electrical stimulation; therapeutic ultrasound; electromagnetic therapy; low level laser therapy; nanotechnology products; honey-based dressings; and even maggot therapy. Many of these technologies and the products used are not approved for payment by insurance carriers, as their use is still considered to be experimental, investigational or unproven. 
         [0020]    There are presently too many disparate technologies for the wound therapy marketplace. The currently available technologies with the best healing efficacy are still terribly ineffective in many cases. What is needed is higher healing efficacy and a method to significantly decrease the time needed to close these wounds. We need to help more people and improve their lives. 
       SUMMARY OF THE INVENTION 
       [0021]    Years ago, wounds were thought to heal faster if they were left exposed to the air. Researchers then discovered that wounds so exposed were more susceptible to infection and would also dry out, forming a scab, which inhibited the healing process. Most all modern wound healing methods or technologies, including the four mentioned above, are designed to keep a wound in a closed environment preventing exposure to the air. 
         [0022]    This invention teaches that a flow of fresh air can be therapeutic when done under controlled conditions. The environment within a wound dressing may become stale and depleted of oxygen. A flow of fresh air replenishes the oxygen and removes stagnant gasses, vapors, and odors which otherwise would accumulate under the wound dressing. A bacterial filter and active humidity control address potential problems. 
         [0023]    The background of this inventor is in the heating, ventilating and air conditioning (H.V.A.C.) industry. This invention is a tiny specialized H.V.A.C. system to condition fresh air for use within a wound dressing. For years, H.V.A.C. systems have provided healthy environments for our bodies. Properly designed, an H.V.A.C. system can do the same for our wounds. This apparatus is designed to ventilate an airtight wound dressing with filtered warm humidified fresh air and oxygen while under a negative pressure. Unlike trying to combine the above four therapies, the ventilations system is the enabler to combine benefits attributed to these therapies. This new therapy is herein referred to as WOUND VENTILATION THERAPY. 
         [0024]    Most ventilation systems use fans to create the air movement. This ventilation system uses the suction force from a therapeutic vacuum pump to provide the air movement. 
         [0025]    The controlled variables of Wound Ventilation Therapy are: air flow, oxygen content, temperature, humidity, and negative pressure. This invention is designed to ventilate a wound dressing with these variables. It provides means to change these variables to suit an individual situation and the ability to administer all variables simultaneously. 
         [0026]    It is the intent of Wound Ventilation Therapy that the healing benefits of each controlled variable will complement each other and become cumulative. It is also the intent of this invention that the efficacies of the individual controlled variables will also be cumulative. This comprehensive care would greatly improve the cure rates, speed up the time to cure, increasing the effectiveness of this new therapy and thereby decreasing the total costs involved. 
         [0027]    A greenhouse is used to mimic the best of nature by providing plants with ideal conditions of temperature, humidity, sunshine and carbon dioxide content. Plants respond by being healthy and growing rapidly. It is the intent of this invention to mimic nature and accelerate the body&#39;s own natural healing process by engineering a protective wound environment. This environment would nourish the wound with precisely controlled natural conditions of fresh air, temperature, humidity, oxygen, and negative pressure. 
         [0028]    The use of negative pressure wound therapy (N.P.W.T.) has become widespread in recent years because of its proven effectiveness. Enhancing N.P.W.T. with fresh air, oxygen, warmth and controlled humidity can only improve on this effectiveness. 
         [0029]    Hyperbaric and topical oxygen wound therapies have been used successfully for years. This simple and novel apparatus is able to provide a hyperoxic wound environment without the need for elevated pressures. This oxygen treatment under negative pressure could be called “hypobaric oxygen” wound therapy. It is important that oxygen is not being inducted into a total vacuum, only a partial vacuum. A partial vacuum allows the oxygen to be therapeutic. 
         [0030]    Almost all wound dressings are designed to provide a moist wound environment, but lack a control means; this can be problematic. This is the first known wound treatment apparatus to operate with active humidity control. All others are passive. The wound therapist sets a precise humidity level for the ventilation flow entering the wound. This setting is to be at a high enough level to prevent drying and low enough to prevent maceration of the skin. As an option, intentional condensate could be created to momentarily flush the wound. 
         [0031]    A healthy human core body temperature is 37 degrees Centigrade. A wound on an extremity is often cooler which inhibits healing. Mild heating promotes healing by dilating the blood vessels at the wound site. This ventilation system allows the wound therapist to set a precise temperature, such as 37° C., for the ventilation flow entering the wound dressing. 
         [0032]    Using different individual therapies on a wound usually requires costly dressing changes and exposes the wound to possible infection. This invention eliminates these problems as the therapies may be done simultaneously within the same dressing. 
         [0033]    Therapies would not always have to be done simultaneously. When a patient with a portable wound vacuum unit desires to be ambulatory, the dressing inlet tube is removed and the open connection port is blocked, leaving only negative pressure therapy in use. 
         [0034]    Another benefit of this invention is that the ventilation within the dressing is powered by an existing wound vacuum unit which may be selected based on price, suitability or preference. Likewise, this invention uses existing negative pressure wound dressings. There are several manufacturers and they offer many types and sizes for different applications. Again, the user has the choice based on suitability or preference. 
         [0035]    The rule is “First, do no harm.” The existing individual therapies referred to in this invention have been well-studied and their parameters are well known. It is therefore anticipated that safety precautions for these individual therapies will also be cumulative for this invention. The settings, limits, and safety considerations are to be based on clinical experience. 
         [0036]    The primary components of this wound ventilation system are: a controllable flow of oxygen; a free flow of air; a ventilation flow rate controller; a heated humidifier; a re-heater; a negative pressure type of wound dressing; and a controlled source of vacuum. 
         [0037]    Oxygen and filtered air are selected or blended under atmospheric pressure. This is accomplished by controlling the flow rate of the oxygen and allowing a free flow of air to make up the balance of the total flow. The total flow rate (liters/min.) is established by the ventilation flow rate controller. This provides the pressure differential necessary to create flow within the system. Flow enters this device at atmospheric pressure and exits at the negative pressure of the vacuum source. This ventilation flow rate is intended to be very low, so as to provide sufficient heat to warm most wounds to normal body temperature. The ventilation flow rate must also be well below the capacity of the vacuum source. 
         [0038]    The purpose of the heated humidifier is to both heat the ventilation gasses and to fully saturate these gasses with sterile water vapor. The purpose of the re-heater is to minimize condensation in the insulated tubing that delivers these gasses to the negative pressure type wound dressing. To be therapeutic, these gasses are to enter the dressing inlet connection at a precisely controlled temperature and humidity. The temperature is to be at the normal body temperature of 37 C. (adjustable). The relative humidity should be adjustable within a range of about 80 to 100% R.H. The temperature set-point is maintained by controlling the operation of the re-heater. The relative humidity set-point is maintained by controlling the operation of the humidifier&#39;s heater. 
         [0039]    The purpose of the controlled vacuum source is to both provide the negative pressure therapy and to provide the power source for the wound ventilation system. 
         [0040]    The embodiment of this invention herein described, consists of the ventilation unit which is to be located bedside and which may be pole mounted. All of the ventilation unit components, with an internal flow of ventilation gasses are designed for single patient use. Other components such as a temperature control system for both the heated humidifier, and the re-heater, the operating controls, and monitoring controls are designed to be easily sanitized. This unit connects externally with an electrical power source, a tubing input from the oxygen source, an insulated tubing output to the wound dressing inlet, and an electronic input from a temperature sensor located at the wound dressing inlet connection. 
         [0041]    This invention provides the means to set/control variables such as: oxygen content, ventilation flow rate, temperature, humidity, and negative pressure. The ability to independently control these variables along with their duration, allows a wound therapist to develop an individualized healing program for each patient. 
     
    
     
       BRIEF DESCRIPTION OF THE DRAWINGS 
         [0042]      FIG. 1  is a schematic flow diagram of the ventilation system showing how it connects to the other wound healing components. Relative size of the ventilation system is much smaller than shown; 
           [0043]      FIG. 2  is the ventilation system of  FIG. 1  installed in an enclosure and comprising a ventilation system device. This drawing also shows the relationship of the ventilation device to the other wound healing components; 
           [0044]      FIG. 3  is a schematic flow diagram of the preferred embodiment of the ventilation system device; and 
           [0045]      FIG. 4  is a front view of the preferred embodiment of the ventilation system device. 
       
    
    
     DETAILED DESCRIPTION OF THE INVENTION 
       [0046]      FIG. 1  is a schematic flow diagram of the vacuum powered ventilation system of this invention. This drawing shows the relationship between the ventilation system, the wound dressing, and the wound vacuum unit. This drawing also shows how these components are connected with each other as well as being connected with an oxygen source. 
         [0047]    Referring to  FIG. 1 , there is shown a schematic of the ventilation system  1  in accordance with principles of this invention. This wound ventilation system is powered by the suction force created by a wound vacuum unit  22 . Air used for ventilation is always available to be freely drawn in through the air intake  2  after passing through a bacterial filter  3 . Oxygen, which is not administered continuously, is available for O 2  treatment sessions. Oxygen used for ventilation is sourced by either wall outlet oxygen or bottled oxygen  4 . An adjustable pediatric (lower available flows) flow meter or other adjustable flow limiting means is used as the oxygen flow rate controller  5 . The sum of the air and oxygen flow rates is the ventilation flow rate and is measured in liters/min. It is governed by a flow limiting device called the ventilation flow rate controller  8 . During oxygen therapy sessions, the oxygen flow rate must be equal to or slightly greater than the ventilation flow rate to achieve a 100% oxygen flow. A 50% oxygen to air mixture would be achieved by setting the oxygen flow rate controller  5  to half that of the ventilation flow rate controller  8 . Oxygen flow  6  through tubing, or other conduit, mixes with free air in the mixture plenum  7  under atmospheric pressure. An imbalance or a malfunction with oxygen flow is unable to build up pressure as it would flow freely out the air intake and maintain atmospheric pressure. The air/oxygen mixture flows freely as the suction force draws it into the ventilation flow rate controller  8 . This device may be adjustable, but once a desired flow rate is known, a constant flow rate makes component design and control simpler. A good constant flow rate device is a filtered orifice flow restrictor. A filter protects the tiny orifice from plugging up. The ventilation flow rate is intended to be very low, at perhaps 0.2 to 0.5 liters/min. This flow rate needs to be high enough to sufficiently warm most wounds and low enough to be well below the capacity of the vacuum source. The ventilation flow rate controller  8  creates the pressure differential that sets a boundary between atmospheric pressure and the negative pressure created by the wound vacuum unit  22 . 
         [0048]    The purpose of the humidifier  11  and its heater  9  is to raise the wet bulb temperature of the ventilation gasses exiting the humidifier to a user-adjustable set point slightly below normal body temperature, such as 37° C.  FIG. 1  shows the heater for humidifier  9  and the humidifier  11  as being separate components, with the humidifier shown as a simple diffusion bubble type. This was chosen for ease of explanation of this patent and is not meant to be a requirement. There are many types of heaters and humidifiers or heated humidifiers that could be used for the above-stated purpose. Some types use an IV type sterile water bag that drips the water by gravity into a humidifier cartridge. Heated humidifiers are typically used when supplying respiratory oxygen to a patient. They are available in many different designs that would be appropriate in this application, providing they were downsized for the much lower flow rates. The humidifier must be designed to produce water vapor only. Unlike mist or droplets, molecules of water vapor are too small to transport bacteria. The heated ventilation gasses  10  enter the humidifier  11  and pass through a diffuser which breaks the ventilation flow into tiny bubbles before passing through the sterile water disposed in the humidifier  11 . These heated bubbles absorb water vapor before leaving the humidifier  11  at wet bulb temperature sensor  13 . This sensor is to be fitted with a woven sock (not shown) which extends down into the water supply. This sock wicks up water which then evaporates in the gas at the sensor, thereby sensing actual wet bulb temperature. Preferably, if the humidifier design was such that complete saturation was all but assured, actual wet bulb sensing could be assumed with using only a dry bulb sensor at the same sensor location  13 . This is because at 100% relative humidity or saturation, wet bulb and dry bulb temperatures are the same. Sensor  13  inputs a wet bulb temperature controller which controls the operation of the heater for humidifier  9 . This wet bulb temperature controller maintains a user-adjustable set point slightly below normal body temperature, such as approximately 37° C. 
         [0049]    Generally speaking, precise measurement of relative humidity is very difficult. The control scheme embodiment described here uses the relationship of wet bulb and dry bulb temperatures to indirectly control the relative humidity. A specific wet bulb and dry bulb temperature has a specific corresponding relative humidity. This can be plotted on a psychrometric chart and the relative data can be transferred to a simpler table for the wound therapist to use. 
         [0050]    The heated and humidified ventilation gasses  12  enter the re-heater  14  and leave at location  15 . The re-heater  14  is used to minimize any condensate that may form caused by the ambient cooling of the saturated ventilation gasses as they are transported to the wound dressing  18 . Thermal insulation  16  on the dressing delivery tubing is used to reduce heat loss and to add control stability. An alternate embodiment to using this re-heater  14  and the thermal insulation  16  on the dressing delivery tubing would be to use a neonatal heated-wire breathing circuit typical to those used in respiratory care. This device uses a coiled heating element along the entire length of the dressing delivery tubing and would perform a similar re-heat function to minimize condensation. Either way, the dressing delivery tubing should be as short as possible. 
         [0051]    Dry bulb temperature sensor  17  inputs the dry bulb temperature controller which controls the operation of the re-heater  14 . This dry bulb temperature controller maintains a user-adjustable set point of 37 C at the sensor  17  which is located as close as possible or within the dressing inlet connection  19 . Temperature sensors  13  and  17  are to be very precise, preferably to tenths of a degree, over a narrow range of normal body temperatures. The wound therapist is able to maintain the desired temperature of the gasses entering the dressing by adjusting the dry bulb temperature. The desired relative humidity of the gasses entering the dressing is maintained by adjusting the wet bulb temperature of the gasses leaving the humidifier. The dry bulb is usually set at normal body temperature of 37 C and the wet bulb is set slightly lower which lowers the relative humidity. If both were set the same, the relative humidity would be 100%. 
         [0052]    Until the wound site warms up, there may be condensate forming within the dressing  18 . This could be minimized with thermal insulation or a blanket placed over the dressing. 
         [0053]    The wound dressing  18  used may be chosen from among the many available existing negative pressure dressings. Most likely it will be necessary to add an inlet tube connection  19  to the typical dressing. This connection should be added in a location non-adjacent to the existing outlet tube connection  20 . 
         [0054]    Dressing exit tubing  21  extends to the remotely located wound vacuum unit  22 . This unit may be chosen from the many types available. The operation of this unit is to be as per the manufacturer&#39;s instructions. The added ventilation system is not meant to change any of the operational or safety requirements of the typical negative pressure wound therapy as recommended by the vacuum unit manufacturer. 
         [0055]    While  FIG. 1  is used to show the basic components and the principles of operation of this invention,  FIGS. 2 ,  3  and  4  are used to show how these components can be used together in a preferred embodiment of a ventilation system device.  FIG. 2  shows the relationship of the ventilation device to the O 2  source, wound dressing, and the wound vacuum unit.  FIG. 3  and  FIG. 4  show the ventilation device in more detail. This preferred embodiment shown is one of many possible embodiments that could be constructed using these same principles of operation. 
         [0056]      FIG. 2  shows the ventilation device  23  and the location of the ambient air intake  2 . The O 2  inlet connection  24  and the ventilation device outlet connection  25  are also shown. Components  2  through  22  operate as described in  FIG. 1 . The ventilation device  23  may be mounted on an IV pole, mounted on a bed rail, or placed on a bed side table. It should be located close to the wound site. It requires an electrical power source from either an internal battery or a 120V power cord. The temperature sensor  17  either wired or wireless, needs to communicate with the ventilation device  23 . If the ventilation device is able to communicate with the wound vacuum unit  22 , a wireless or a wired connection must be provided. 
         [0057]      FIG. 3  is the schematic flow diagram of the ventilation system of  FIG. 1 , installed inside of an enclosure and referred to as the ventilation device  23 . Components  2  through  14  operate as described in  FIG. 1 . Again referring to  FIG. 3 , optional components not shown in  FIG. 1  are an O 2  stop valve  26 , a ventilation stop valve  27 , and a programmable timer/display unit. These components are used to create the automated operation of the ventilation device in five different therapy modes: 
         [0058]    A. ‘Ventilate with O 2 ” 
         [0059]    B. ‘Ventilate with Air” 
         [0060]    C. ‘Ventilate with Flush” 
         [0061]    D. “Negative Pressure Only” 
         [0062]    E. “Negative Pressure Off” 
         [0063]    The programmable timer/display unit and stop valves automate the sequence and duration of each therapy mode (A through E) to create an individualized patient program. Once programmed, the display unit would be able to graphically display the time and duration of the occurring therapy modes. 
         [0064]    Referring again to  FIG. 2 , the O 2  stop valve  26  must be open for Mode A, ‘Ventilate with O 2 ”, to occur, and closed for all other therapy modes. Ventilation stop valve  27  must be open for one of the ventilate modes (A, B, or C) to occur and closed for Mode D, “Negative Pressure Only”. To facilitate tubing replacement, for single patient use, both stop valves could be electrically actuated solenoid pinch valves. Mode C, ‘Ventilate with Flush”, is used to ventilate the wound with combined air and condensate at 37 C. During this mode, the re-heater  14  is shut off and the dry bulb sensor  17  is switched over to allow it to input the wet bulb controller and operate the heater for humidifier  9 . The duration of this mode must be short as prolonged use could lead to maceration of the skin. 
         [0065]    If the wound vacuum unit is operating on an intermittent cycle, Mode E, “Negative Pressure Off”, will occur regardless of programming. To program Mode E, “Negative Pressure Off’, it is necessary for the ventilation device to communicate with the wound vacuum unit. 
         [0066]    Inside the ventilation device and not shown on the drawings, are the wet bulb temperature controller and the dry bulb temperature controller. 
         [0067]      FIG. 4  is a front view of the ventilation device  23 . The programmable timer/display  28  is a touch screen device that is used to program the sequence and duration of the desired therapy modes (A through E as previously described). The display screen also graphically displays time and duration of each occurring therapy mode. This display may also monitor and display such variables as dry bulb temperature entering the dressing, wet bulb temperature leaving the humidifier, temperature leaving the dressing, and relative humidity leaving the dressing. Dry bulb temperature adjustment  29  is used to enter the desired set-point for the temperature of the gasses entering the dressing. Wet bulb temperature adjustment  30  is used to enter the desired set-point for the wet bulb temperature of the gasses leaving the humidifier, which determines the relative humidity of the gasses entering the dressing. Both set-points are to be highly accurate to tenths of a degree over a very narrow range. Green L.E.D.&#39;s  35 - 39  may be used to indicate the occurring therapy mode (A through E). The on/off switch  31  is used to turn on the electrical power for this ventilation device. A red L.E.D.  32  indicates an alarm condition for high temperature entering the dressing. Another red L.E.D.  33  is used to indicate an alarm condition for low relative humidity leaving the dressing. Push button  34  is used to silence the audible alarm. For safety, all heat sources should be disabled upon either an alarm condition or a loss of vacuum as sensed with a vacuum pressure switch. 
         [0068]    The above described embodiment is only one of many possibilities. All described therapy modes may not be necessary. This ventilation device may be simplified with non-automated therapy modes using manual pinch valves and eliminating the programmable timer/display unit at least during early prototype testing. 
         [0069]    Another embodiment of this invention could be designed to administer oxygen continuously. This would require that oxygen be at a low and safe concentration, perhaps 25, 30, or 35%. This would eliminate the need for separate therapy modes A and B. Since therapy modes C, D, and E are not necessary, another embodiment of this invention could be designed to operate in one constant mode. This would be a lot simpler and allow for a less costly ventilation device. 
         [0070]    This described embodiment also shows a stand-alone ventilation device with the optional ability to communicate with the wound vacuum unit. Other embodiments could be created depending on the desired degree of integration of both ventilation and vacuum unit devices, up to and including designing all functions into a single device.