Carbon Dioxide CO2 Detector Monitor Meter Tester Sensor Temperature Humidity Indoor 9999ppm NDIR Air Quality IAQ Handheld Portable Digital Wet Bulb
Brand: TEKCOPLUS
Color: Hand Held CO2 Meter

Shows CO2 level, temp. (°C/°F), Humidity, Dew Point & Wet Bulb. Auto/Manual power off, PC connection loggin and data analysis. Backlight, Hold data, Min/Max

Features:

1.Triple displays of CO2 level, temperature (°C/°F) and Humidity
2.Stable NDIR sensor for CO2 detection

Statistics of weighted averages

TWA (8 hours weighted average)

STEL (15 minutes-weighted average)

3.Updates reading every second on the meter, output to PC every 2 seconds

Specifications:

Measuring Range

CO2: 0~9999ppm

(2001~9999 out of scale)

Temperature : -10~60°C (14~140°F)

Humidity : 0.1% ~ 99.9%RH

Dew Point : -20.0~59.9°C

Wet Bulb : -5.0~59.9°C

Resolution : 1ppm,0.1°C/°F, 0.1%RH

Accuracy :

C02 : ±50ppm ±5% of reading (0~2000ppm) other ranges are not specified

Temp : ±0.6°C / ±0.9°F

Humidity : ±3%RH (at 25°C, 10~90% RH); ±5%RH (at 25°C, others)

Warm-up time : 30 seconds

Response time

CO2 : <30 seconds (90% step change)

Temp : <2 mins (90% step change)

RH : <10 mins (90% step change)

NO RELAY

Operating condition : 0~50°C, 0~95%RH (avoid condensation)

Storage Condition : -20~50°C, 0~95%RH (avoid condensation)

Set Includes:

1 x Portable CO2 meter
4 x AA batteries
1 x Power Adapter
English Operation Manual
Hard Carrying Case

MONITORING & ANALYSIS Allows continuous monitoring of CO2, RH & Temperature Level. Data output to PC every 2 seconds.
MULTIPLE MEASUREMENT DISPLAYS AT ONCE- CO2, Temperature, and Relative Humidity readings are all shown on the LCD display at the same time
NORMAL/ MAX/ MIN/ AVERAGE (15mins/ 8hrs) MODE- It can show not only the normal, maximum and minimum CO2 readings but also the weighted average CO2 readings in the past 15mins (STEL mode) and 8 hours (TWA mode).
AUDIBLE CO2 SOUND- The alarm beeps when the CO2 Level exceeds the preset value. The alarm threshold is adjustable.
NDIR SENSOR for CO2 DETECTION- Non-dispersive infrared technology is used to ensure the reliability and long-term stability. Measures 0~9999ppm while 0~2000ppm can maintain a high accuracy at +/- 50ppm