HiLetgo USB Logic Analyzer Device With EMI Ferrite Ring USB Cable 24MHz 8CH 24MHz 8 Channel UART IIC SPI Debug
Brand: HiLetgo
Color: Gray

HiLetgo USB Logic Analyzer Device With USB Cable 24MHz 8CH 24MHz 8 Channel UART IIC SPI Debug
Sampling rate up to: 24 MHz , can be 24MHz. 16MHz, 12MHz, 8MHz, 4MHz, 2MHz, 1MHz, 500KHz, 250KHz, 200KHz, 100KHz, 50KHz, 25KHz;
The logic for each channel sampling rate of 24M/s. General applications around 10M, enough to cope with a variety ofoccasions;
8-channel;
Voltage range:
Input voltage range: -0.5V to 5.25V
Input Low Voltage: -0.5V to 0.8V
Input High Voltage: 2.0V to 5.25V
Works with 5V, 3.3V, 2.5V, 2.0V systems. May work with 1.8V but not recommended.
Input Impedance: 1Mohm || 10pF (typical, approximate)
Crystal: +/-20ppm, 24MHz
Error/Accuracy: pulse-width measurement: +/- 42ns (at 24MHz).

Packing Included:
1 * Logic Analyser
1 * USB Cable
1 * Dupont Cable

The logic for each channel sampling rate of 24M/s. General applications around 10M, enough to cope with a variety ofoccasions; 8-channel
Sampling rate up to: 24 MHz , can be 24MHz. 16MHz, 12MHz, 8MHz, 4MHz, 2MHz, 1MHz, 500KHz, 250KHz, 200KHz, 100KHz, 50KHz, 25KHz;
The logic for each channel sampling rate of 24M/s. General applications around 10M, enough to cope with a variety ofoccasions;
Input voltage range: -0.5V to 5.25V; Input Low Voltage: -0.5V to 0.8V; Input High Voltage: 2.0V to 5.25V
Input Impedance: 1Mohm || 10pF (typical, approximate); Crystal: +/-20ppm, 24MHz