Owon 100mhz Osciloscopio Sds7102V 1g/s Larga pantalla Lcd 8" con 3 a?os de garant¨ªa
Brand: OWONQsweet
Color: white-45

VGA+ LAN 1G/s Sample rate, Large 8" LCD, 10M points Record length, FFT free firmware upgradeBrand new with 3 yrs warranty, manufacturer pay for shipping back for warranty problem Special offer: For limited time: we offer 2 free BNC test leads ( BNC to Alligator clip leads + BNC to mini grabber leads ) for all SDS 7 and 8 series oscilloscopes order; we match competitor’s price, make your offer to us Now! Details: 1.100MHz Bandwidth 2.1GS/S Real-sample Rate 3.10M points Record length 4.8 inch color LCD(800X600 Pixels) 5. Autoscale function 6. Display Grid: 10div, Horizontal: 15 div 7. Waveform Storage: 15 waveforms 8. Battery(Optional) 9. Size: 340mmx155mmx70mm(L*H*W) 10. Light Weight: 1.8kg(without battery) Accessories: 1) 2 units of 100Mhz probes; 2) Power cord for your country; 3) Driver CD; 4) USB cable; 5) Chargable Battery ( optional, add EUR45.00 for battery) 6) VGA version with LAN module; SDS7102V: Channels 2 Real-time sample rate 1 GSa/s real time Bandwidth 100 MHz Display 8" TFT LCD with 65535 colors Resolution 800 × 600 Pixel Interfaces USB PC Connection USB Host for USB-Sticks VGA Interface LAN Port Max. Input voltage 400 V (peak) Real-time sample rate 10 Sa/s to 1 GSa/s real time Time base 2 ns/Div. to 5 s/Div. (Steps 1 - 2 - 5) Memory 10 Mpts per channel Analog digital converter (A/D) With the resolution of 8 bits, make sampling on both channels synchronously. Sensitivity range (V/div) 2 mV/Div to 10 V/Div. Displacement range ±1 V (2 mV - 100 mV) ±10 V (200 mV - 1 V) ±100 V (2 V - 10 V) Trigger Channel 1, 2 and external Trigger Trigger displacement depends on record lengh and selected time base Signal system and line/field frequency (Video triggering mode) Support the NTSC, PAL and SECAM broadcasting systems of any field or line frequency. Cursor measurement Voltage difference (∆V) and time difference (∆T) between cursors

Nuevo ultra fino, gran pantalla LCD de 8 "(800 x 600) 1 g/S velocidad de muestreo con LAN + VGA + bolsa
Puntos de 10 m Longitud de registro, FFT, autoscale functionm Dual canales
Display Grid: 10div, almacenamiento horizontal: 15 div, 15 las ondas.
Escala horizontal (S/div) 2 ns/div ~ 100s/div, paso por 1 ~ 2 ~ 5
Tiempo de subida (en la entrada, típica) & # x2264; 3.5nsm, sensibilidad vertical: 2 mV/div ~ 10 V entrada/div (a)