Sealless rotary blood pump

Abstract: An implantable rotary sealless blood pump (11) is provided. The pump (11) includes a housing (12) having an inlet tube (13) on one end and an impeller casing (14) on the other end. A rotor (17) is mounted for rotation within the housing (12), with the rotor (17) having an elongated shaft portion (18) and an impeller (19) attached to the shaft portion (18). The impeller (19) is located within the impeller casing (14). Radial magnetic bearings (28,29) are carried by the shaft portion (18) and radial magnetic bearings (23,24) are carried by the housing (12) for maintaining the shaft portion (18) of the rotor (17) within the inlet tube (13) of the housing (12). A rotor motor (53) includes a plurality of permanent magnets (54) carried by the impeller (19) and a motor stator (56) including an electrically conductive coil (57) located within the housing (12). A ring of back iron (58) is carried by the impeller casing (14) to aid in completing a flux return path for the permanent magnets (54). A plurality of hydrodynamic thrust bearings are located outside of the axis of rotation of the rotor (17). The impeller (19) uses large axially thick blade sectors (49) with narrow blood channels (51) extending through the impeller (19), to minimize hemolysis and to increase the working surface of the blades (49).

Description: Field Of The Invention The invention relates generally to the field of blood pumps. More specifically, the invention pertains to continuous flow pumps of rotary design, suitable for permanent implantation in humans, for use as chronic ventricular assist devices. Background Of The Invention Thousands of heart patients who suffer from severe left ventricular heart failure could benefit from cardiac transplantation. However, owing to a shortage of donor hearts, most of these patients face a foreshortened life span characterized by frequent hospitalizations, severe physical disability, and death from congestive failure or cardiogenic shock. If a left ventricular assist device ("LVAD") were available for chronic use, many of these patients could be returned to prolonged and productive lives. Prior art LVADs, now in clinical trials, provide a cyclic or pulsating delivery of blood, designed to emulate the natural pulsatile blood flow through the heart. This design approach has resulted in a variety of anatomic and engineering problems. Cyclic delivery systems tend to be physically large, making implantation difficult or impossible for some patients. Cyclic delivery systems also employ artificial valves, having special material, longevity, and performance requirements. All of these characteristics make cyclic blood pumping devices both complex and expensive. It is apparent that if the requirement of pulsatile blood flow is eliminated, the LVAD could be much smaller, and less expensive. Rotary pumps, whether of centrifugal or axial flow design, provide substantially continuous liquid flow, and potentially enjoy a number of the listed advantages over cyclic delivery systems. However, the prior art has not developed a durable rotary blood pump, owing to unique problems with the rotary pump's driveshaft seal. In a blood environment, such driveshaft seals have a short life, and contribute to a premature failure of the pump. Prior art driveshaft seals may also cause embolisms, resulting in a stroke or even death for the patient. One such rotary pump is discussed in document WO 97/29795, which shows the state of the art according to Article 54(3) EPC. Here, a rotary blood pump is disclosed as including a mounted impeller for rotation which is located within a pump housing. The pump further comprises a motor, which incorporates a plurality of permanent magnets carried by the impeller, and a motor stator. The motor stator is fashioned from an electrically conductive coil which is located on one side of the impeller and is contained within the housing "This document, however, does not disclose a second motor stator positioned on the opposite side of the Impeller, nor does it disclose a plurality of wedge-shaped hydrodynamic thrust bearings located outside the axis of rotation of the rotor, nor does it disclose that during rotation of the impeller those hydrodynamic bearings are separated from the housing by a fluid film and are not in direct mechanical contact with the housing." Accordingly, it is an object of the present invention to provide an improved rotary blood pump, by eliminating the necessity for a driveshaft seal; It is a further object of the present invention to provide a pump impeller which is effective pumping viscous liquids, such as blood, at low flow rates, and which minimizes hemolysis of the blood by using only a few pump impeller blades Summary of invention In accordance with illustrative embodiments of the present invention, a rotary blood pump is provided which is incorporated within a pump housing. A rotor is mounted within this housing, and has an impeller. There is provided a motor for driving the rotor, which includes a plurality of permanent magnets which are carried by the impeller of the rotor. Either side of the impeller are positioned first and second motor stators, each of which includes a plurality of electrically conductive coils and pole pieces. These coils and pole pieces making up the motor stators are located within the pump housing. A plurality of rectangularly-shaped or wedge-shaped hydrodynamic thrust bearings are located outside the axis of rotation of the rotor. During rotation of the impeller, these hydrodynamic bearings are separated from the housing of the pump by a fluid film, and do not make a direct mechanical contact with the housing. Further, it is possible for the hydrodynamic bearings to be arcuate in form, and are located on the forward side of the impeller. Furthermore, it is possible for some of these hydrodynamic thrust bearings to be carried by the impeller. A more detailed explanation of the invention is provided in the following description and claims, and is illustrated in the accompanying drawings. Brief Description Of The Drawings Figure 1 is a left front perspective of the blood pump of the present invention; Figure 2 is a fragmentary, cross-sectional view of the pump of Figure 1, showing a plurality of ring magnets comprising part of the magnetic bearing assembly; Figure 3 is a fragmentary, cross-sectional view of the pump of Figure 1, showing the shaft and an impeller; Figure 4 is a view as in Figure 1, but with the shaft and impeller shown removed from the housing; Figure 5 is a simplified, fragmentary, representation of a human heart, showing the pump implanted within the left ventricle of the heart; Figure 6 is a transverse, cross-sectional view of the housing, impeller, and impeller chamber, taken along the line 6-6, shown in Figure 1; Figure 7 is a longitudinal, cross-sectional view of the pump, taken along the line 7-7, shown in Figure 1; Figure 8 is a longitudinal, cross-sectional view of a simplified, schematic representation of the pump, showing respective polarities of the magnets and the pole pieces of the passive radial magnetic bearings, and the elements of the pump motor, including rotor magnets and a motor stator; Fig. 8a is a schematic view, similar to Fig. 8, but showing another embodiment of the present invention; Fig. 8b is a schematic view, similar to Fig. 8a, but showing another embodiment of the present invention. Figure 9 is a longitudinal, cross-sectional view of an impeller constructed in accordance with the principles of the present invention; Figure 10 is an end view thereof, taken from the right side of Figure 9; Figure 11 is a longitudinal, cross-sectional view of a simplified, schematic representation of another embodiment of the pump; Figure 11a is an enlarged view of the circled portion 11a from Figure 11; Figure 12 is a cross-sectional end view of the Figure 11 pump with the end of the housing and casing removed for clarity; Figure 13 is a perspective view, partially broken for clarity, of the blood pump of Figure 11; Fig. 13a is a perspective view of a portion of Fig. 13, showing the slotted motor stator; Fig. 13b is a perspective view, similar to Fig. 13a but showing a slotless motor stator. Figure 14 is another perspective view, partially broken for clarity, of the blood pump of Figure 11; Figure 15 is a longitudinal, cross-sectional view of another embodiment of the pump; Figure 15a is an enlarged view of the circled portion 15a from Figure 15; ; Figure 16 is a cross-sectional end view of the Figure 15 pump, with the end of the housing and casing removed for clarity; Figure 17 is a longitudinal, cross-sectional view of another embodiment of a blood pump; Figure 17a is an enlarged view of the circled portion 17a from Figure 17; Figure 18 is a cross-sectional end view of the Figure 17 pump, with the end of the housing and casing removed for clarity; Figure 19 is a longitudinal, cross-sectional view of another embodiment of the present invention; Figure 19a is an enlarged view of the circled portion 19a from Figure 19; and Figure 20 is a cross-sectional end view of the Figure 19 pump, with the end of the housing and casing removed for clarity. Detailed Description Of The Preferred Embodiment Turning now to Figures 1-8 of the drawings, a sealless rotary blood pump 11 includes a housing 12, having an elongated inlet tube 13 and an impeller casing or volute 14. A discharge tube 16 extends through the housing to communicate with the interior periphery of casing 14. Tube 16 has a tangential orientation with respect to a radius of the casing, for effectively channeling the blood output from the pump. A pump rotor 17 is located within housing 12, within casing 14, and includes an elongated, right-circular cylindrical support shaft or spindle 18, attached to a disc-shaped impeller 19. Rotor 17 is mounted for rotation about a longitudinal axis which extends both through shaft 18 and impeller 19. It should be noted that the preferred embodiment disclosed herein includes an impeller and a casing of centrifugal design. However, many of the structural features and aspects of operation of the present invention may also be adapted advantageously to rotary blood pumps of axial flow design. The pump 11 of the present invention includes a forward magnetic bearing 21 and a rearward magnetic bearing 22 to levitate rotor 17 and maintain it in proper radial alignment with respect to its longitudinal axis. A radial magnetic bearing construction is shown in U.S. Patent No. 4,072,370, issued to Wasson. The forward magnetic bearing 21 herein may be constructed entirely in accordance with the teachings of the '370 Patent. However, several simplifications and improvements to the construction shown in the '370 Patent are disclosed herein. For example, it has been determined that the radially polarized ring magnets (numerals 44 and 46) of the '370 device, are not necessary for successful practice of the invention herein. In addition, as will be explained below, the axially magnetized ring magnets (numeral 22) of the '370 device may advantageously be replaced with axially magnetized disc magnets for purposes of the present invention. Accordingly, the forward magnetic bearing 21 includes a plurality of rings, comprising ferromagnetic pole pieces 23 and axially polarized permanent magnets 24. As shown most clearly in Figures 7 and 8, pole pieces 23 and magnets 24 are arranged in contingent, alternating fashion, and are located between outer sidewall 26 and inner sidewall 27 of inlet tube 13. The polarization of opposing magnets is the same, inducing an identical polarization into a respective pole piece therebetween. A combination of high strength adhesive and surrounding tube sidewalls, maintains the arrangement of magnets and pole pieces in contingent relation, despite strong magnet forces attempting to urge the rings apart. Forward magnetic bearing 21 also includes a plurality of discs, comprising ferromagnetic pole pieces 28 and axially polarized permanent magnets 29. Pole pieces 28 and magnets 29 are also arranged in contingent, alternating fashion, so as to form a magnetic structure which mirrors the polarity and axial position of respective pieces and magnets of the surrounding rings. This magnetic structure is first assembled and secured together using high strength adhesive, and is then installed within the hollow volume of shaft or spindle 17. The magnetic polarizations and repulsive forces produced by the magnets and the pole pieces of forward magnetic bearing 21 are such that magnetic levitation of support shaft 18 results. To provide additional radial restraint for rotor 17, rearward magnetic bearing 22 is also provided. Bearing 22 includes a first ring magnet 31 mounted on an outer wall of casing 14, and a second ring magnet 32 imbedded within a circular casing base 33. The bottom portion of casing 14 is attached and sealed to base 33, to form a fluid impervious enclosure for impeller 19 (see Figure 7). Both magnets 31 and 32 are axially polarized, but each has a different polarization facing impeller 19. Bearing 22 also includes a plurality of rod magnets 34, transversely extending from an upper face portion 36 to a lower face portion 37 of impeller 19. Rod magnets 34 are arranged in spaced, circular fashion, adjacent an outer periphery 38 of impeller 19. The polarizations between the ends of magnets 34 and the adjacent surfaces of magnets 31 and 32 are respectively opposite, creating attractive, but equal and opposite magnetic forces acting on the impeller. It can be seen that radial movement of the impeller (deflection from the axis of rotation) will result in a restoring force due to the attraction between the magnets 34 towards magnets 31 and 32. The magnetic force in the axial direction will largely be counterbalanced to the opposing magnetic attraction of magnets 34 to magnet 31 and magnets 34 to magnet 32. However, the action of the magnetic force in the axial direction would not be restoring. It should also be noted that other configurations, locations, numbers, and polarization orientations may be used for the components forming rearward magnetic bearing 22. For example, magnets 34 may be arcuate segments, rather than rods. Also, the polarizations of the magnets 31, 32, and 34 may be arranged to effect respective repulsive forces, rather than the attractive forces specifically disclosed herein. In this manner, referring to Figs. 8a and 8b, the south pole of magnets 34 would be adjacent the south pole of magnet 31 and the north pole of magnets 34 would be adjacent the north pole of magnet 32. For the magnets to be restoring in the radial direction, the magnets would have to be offset. To this end, in the Fig. 8a embodiment magnets 34 would be more outward radially than magnets 31 and 32. Alternatively, in the Fig. 8b embodiment magnets 34 are radially inside the radial dimension of magnets 31 and 32. If a repulsive configuration is used, as illustrated in Figs. 8a and 8b, the action of the magnetic force would be restoring in both the radial and axial direction. Although the drawings show magnets 32 and 34 as if portions thereof are directly immersed in blood, in actual practice, a thin-walled non-magnetic jacket or a plastic coating would be placed over these portions, to prevent contact between the magnets and the blood. Such contact, if it were allowed, would likely cause an undesirable chemical reaction, to the detriment of the blood. However, for clarity, the referenced jacket or coating, is not shown in the drawings. To provide mechanical limitations on axial, translational excursions of the rotor, a first thrust bearing 39 and a second thrust bearing 41 are provided. First thrust bearing 39 includes a threaded plug 42, installed within casing base 33. Plug 42 is screw adjustable along the longitudinal axis of rotor 17, and includes a recessed bearing surface 43. Surface 43 is contoured to accommodate a corresponding bearing tip 44, in the lower face portion of impeller 19. It should be noted that the particular configuration of bearing 39 is not critical, and planar bearing surfaces may alternatively be used in this application. Second thrust bearing 41 is secured within the blood entry end of inlet tube 13, and includes a spider 46, adjustment knob 47, and ball 48. Rotation of knob 47 will translate ball 48 along the longitudinal axis of rotor 17. Alternative locations and constructions for second thrust bearing 41 are also contemplated. For example, an annular thrust bearing surface could be provided on the inner wall of casing 14, adjacent the upper face portion 36 of impeller 19. In this arrangement, portion 36 would slidably contact the annular thrust bearing surface. By eliminating spider 46 and the associated components of the upstream thrust bearing, the possibility of blood deposits forming on these structures would be eliminated. It will be appreciated that thrust bearings 39 and 41 are effective not only to provide limit stops to axial movement of rotor 17, but also to adjust certain operational aspects of the pump. In the drawings, the upstream end of support shaft 18 is shown in contact with ball 48. However, this will not always be the case during the course of operating the pump. For example, it is desirable for the two thrust bearings to be adjusted so that the distance between them, is slightly greater than the overall length of the rotor. This will allow the rotor to "shuttle", back and forth between the axial constraints provided by the thrust bearings with each cardiac cycle of the user. Each such cycle will produce a pumping action, bringing fresh blood into the touchdown, or thrust bearing area. The present invention does not use a journal bearing to restrain the rotor. Of necessity, a journal bearing radially encases at least a portion of the rotor's support shaft or spindle. It is within this thin, annular volume between the shaft and the bearing surface, where thrombosis can occur in prior art devices as a consequence of heat and excessive residence time within the bearing. The bi-stable operation of the pump and rotor of the present invention, continuously flushes the blood around each thrust bearing, avoiding thrombosis effects of prior art journal bearings. There is also an important physical relationship which exists between the rotor and the magnetic bearings of the device disclosed herein. This relationship is established and maintained by proper axial placement of the adjustable thrust bearings. In operation of the pump, the pressure gradient produced by the rotating impeller imparts an upstream axial force on the rotor. This force needs to be substantially counterbalanced, to ensure that cardiac pulses will create sufficient pressure variances through the pump, to effect bi-stable operation. By adjusting the axial relationship of the pole pieces 23 and the magnets 24 with respect to the pole pieces 28 and magnets 29, a downstream axial force will be produced. Since the forces within forward magnetic bearing 21 are repulsive, the desired downstream loading or bias will be effected when the magnets and pole pieces within the shaft are translated slightly downstream from the magnets and pole pieces in the inlet tube (See, Figures 7 and 8). Thus, second thrust bearing 41 is effective to shift, or offset the rotor downstream a sufficient amount so the resultant, repulsive magnetic forces substantially counterbalance the hydrodynamic axial force produced by the rotating pump impeller. We can now turn to the special design considerations and operational characteristics of impeller 19. As will be noted particularly in Figure 6, the impeller includes a plurality of large blade sectors 49. Owing to its relatively high viscosity and susceptibility to damage from heat and mechanical action, blood is a uniquely difficult liquid to pump. It is generally preferable in a large centrifugal pump, to have a substantial number of thin, sharp impeller blades with relatively large voids or passages, between the blades, for the passage of low viscosity liquid. However, such a conventional design is not desirable, for a small centrifugal pump which has to pump a viscous liquid, such as blood. When blood flows axially into the leading edges of impeller blades it tends to be damaged by the mechanical action and turbulence associated with the impeller blades. Thus, one of the design considerations of the present invention is to reduce such hemolysis, by minimizing the number of impeller blades and leading edges. To maintain efficiency in a small pump with so few blades, the effective working area of the blades needs to be increased. This was accomplished in the present design by modifying the size and configuration of conventional blades in two significant aspects. First, blade sectors 49 are made relatively wide or expansive through a rotational aspect (see Figure 6). In other words, the outer periphery of each blade sector 49 assumes approximately 80 to 85 degrees of rotation. It should be noted that an alternative design contemplated herein includes only two blade sectors, each of which assumes approximately 175 degrees of rotation. In either case, the width of the impeller blade sectors of the present invention differ significantly from known prior art blades. The second modification pertains to the thickness or height of the blade sectors. As shown particularly in Figures 4 and 7, blade sectors 49 are relatively thick in an axial direction. As a consequence of these modifications, a narrow and deep impeller blood flow path or passageway 51 is defined between adjacent edges of blade sectors 49. By increasing the thickness of the blade sectors and narrowing the blood passageway, the ratio between the area of working surface of the blades and the volume of the passageway is increased. Also, the average distance of the liquid in the passageway from the working surface of the blades is decreased. Both of these beneficial results provide a small pump for blood which has few blades for damaging blood, yet maintains acceptable efficiency. The size and configuration of the impeller blades also allows the structural integration of a number of features directly within the impeller 19. For example, the previously discussed rearward magnetic bearing 22 includes a plurality of rod magnets 34 of considerable length. Owing to the thickness of the blade sectors, these magnets are readily accommodated within the sectors. The sectors may also be provided with respective hollow chambers 52, to reduce the mass of the impeller and the gravity induced loads on the thrust bearings (see, Figure 6). Lastly, a brushless rotor motor 53 includes arcuate magnetic segments 54, imbedded within the upper face portion 36 of blade sectors 49. As discussed above, the portions of segments 54 which would otherwise be in fluid communication with the pumped blood, are encased in a jacket or a coating (not shown) to prevent any chemical reaction between the blood and the magnetic segments. Making reference to Figures 6 and 8, segments 54 have alternating orientations in their polarities, and are directed toward an adjacent motor stator 56. Included within stator 56 are windings 57 and a circular pole piece or back iron 58, mounted on the outer surface of impeller casing 14. Windings 57 are interconnected by means of percutaneous wires to a controller 59 and a power supply 61, as shown in Figure 5. Alternative to using wires, transcutaneous power transmission could be used. It is contemplated that controller 59 and power supply 61 may be worn externally by the user, or alternatively, they may be completely implanted in the user. Controller 59 may include circuitry as simple as a variable voltage or current control, manually adjusted or programmed to determine the running rate of pump. However, controller 59 may also have interactive and automatic capabilities. For example, controller 59 may be interconnected to sensors on various organs of the user, automatically and instantaneously to tailor operation of the pump to the user's physical activity and condition. The windings 57 are energized by the electrical output of controller 59 to produce an electromagnetic field. This field is concentrated by pole piece 58, and is effective to drive magnets 54 and the rotor 17, in rotary fashion. The back EMF resulting from the magnets 54 passing by the windings is detected by the controller. The controller uses this back EMF voltage to continue generation of the electromagnetic field in synchronism with further rotation of the rotor. Brushless operation of the motor 53 is effected, then, by electromagnetic interaction between the stator and magnets imbedded within the pump's impeller blades. Motor 53, with windings 57 and pole piece 58, together with magnets 54, function not only to transmit torque but also provide a restoring radial magnetic force that acts as a radial bearing. As illustrated in Figures 7 and 8, magnets 54 are carried by blade sectors 49 and are positioned in radial alignment with pole piece 58. The magnets 54 have attraction with the iron pole piece 58 of the stator. Any attempt to deflect the impeller radially produces an increasing restoring force between the pole piece 58 and the magnets 54 which would cause the impeller to return to a neutral position. Rotation of the rotor 17, including shaft 18 and impeller 19, causes blood to flow through inlet tube 13 in the direction of arrows 62. The blood continues its path from the upper edge of passage 51 to the interior of casing 14. Discharge tube 16 allows the blood to be expelled from the casing an into the user's cardiovascular system. Anatomical placement of the pump 11 is shown in Figure 5. The simplified representation of a human heart 63, includes a left ventricle 64 and an aorta 67. The inlet tube 16 serves as the inflow cannula and is placed into the apex of the left ventricle 64. An arterial vascular graft 66 is connected on one end to tube 16 and on the other end to the aorta 67 through an end to side anastomosis. The centrifugal design of the pump allows a considerable amount of flexibility during implantation. Owing to the axial inflow and radial outflow of the pump, a 90 degree redirection of the blood is effected without the necessity of a flow-restrictive elbow fitting. Moreover, the pump can be rotated on its longitudinal axis to adjust the orientation of the discharge tube and minimize kinking and hydraulic losses in the vascular graft. Good anatomic compatibility is possible since the pump casing is compact and disc-shaped, fitting well between the apex of the heart and the adjacent diaphragm. In a specific example although no limitation is intended, referring to Figure 7, blood flow path 62a is 1,52 mm to 2,54 mm (.06 inch to 0.1 inch) in thickness. The fluid gap 70 comprising the clearance between the impeller and the housing is 0,127 mm to 0,5 mm (.005 inch to .02 inch). The impeller diameter is 25,4 mm to 38,1 mm (1.0 inch to 1.5 inch). The rotor diameter is 0,635 mm to 10,2 mm (.025 inch to 0.4 inch). The outside diameter of the flow annulus is 8,9 mm to 13,9 mm (.35 inch to .55 inch). The outer diameter of the housing adjacent the forward end of the pump is 21,6 mm to 31,8 mm (.85 inch to 1.25 inch). The axial length of the entire pump is 44,5 mm to 76,2 mm (1.75 inch to 3.0 inch). The axial length of the rotor spindle is 25,4 mm to 38,1 mm (1.0 inch to 1.5 inch) and the axial length of the impeller is 5,1 mm to 12.7 mm (0.2 inch to 0.5 inch). By using a thick impeller (having a long axial length) the fluid gap 70 can be larger and still provide a highly efficient pumping action. Enlarged views of an impeller used in the pump of the present invention are set forth in Figures 9 and 10. Referring to Figures 9 and 10, an impeller 74 is shown therein having a number of blade sectors 76, 78 and 80. Blade sectors 76 and 78 are separated by slot 82; blade sectors 78 and 80 are separated by slot 84; and blade sectors 80 and 76 are separated by slot 86. By utilizing blade sectors 76, 78 and 80 that are relatively thick in the axial direction, narrow and deep impeller blood flow paths are formed by slots 82, 84 and 86 between the adjacent edges of the blade sectors. By increasing the thickness of the blade sectors and narrowing the blood passageway, the ratio between the area of working surface of the blades and the volume of the passageway is increased. Also, the average distance of the liquid in the passageway from the working surface of the blades is decreased. Both of these beneficial results allow a small pump for blood which has less blades for potentially damaging blood, yet the small pump maintains acceptable efficiency. As a specific example although no limitation is intended, the diameter of the impeller is 25,4 mm to 38,1 mm (1 inch to 1.5 inch), the blade depth bd (Fig. 9) is 5,1 mm to 12,7 mm (0.2 inch to 0.5 inch), the magnet width mw (Fig. 9) is 3,81 mm to 7,6 mm (.15 inch to 0.3 inch), the spindle diameter sd (Fig. 9) is 6,4 mm to 12,7 mm (.25 inch to 0.5 inch), and the inner diameter id (Fig. 9) of the impeller inlet is 11,4 mm to 15,2 mm (.45 inch to .6 inch). The width w of the slots (see Fig. 10) is approximately 1.9 mm (.075 inch) and preferably ranges from 1,3 mm to 5,1 mm (.05 inch to 0.2 inch). The outlet angle a (Fig. 10) preferably ranges between 30° and 90°. Another benefit of the thick impeller is the ability to utilize magnetic pieces 88 that are inserted in a manner enabling the stators to be on opposite sides of the impeller. Referring to Figures 11, 11a, 12, 13 and 14, the blood pump 11' shown therein is similar in many respects to blood pump 11 illustrated in Figures 1-8, and includes housing 12 having an elongated inlet tube 13 and a scroll-shaped impeller casing or volute 14. A discharge tube 16 extends through the housing to communicate with the interior periphery of casing 14. Tube 16 has a tangential orientation with respect to a radius of the casing, for effectively channeling the blood output from the pump. Pump rotor 17 is located within housing 12, within casing 14, and includes an elongated, right-circular cylindrical support shaft or spindle 18, attached to impeller 74. Rotor 17 is mounted for rotation about an longitudinal axis which extends both through shaft 18 and impeller 74. The magnetic bearings for levitating rotor 17 and maintaining it in proper radial alignment with respect to its longitudinal axis are not specifically shown but may be identical to those illustrated in the pump embodiment of Figures 1-8 and described above. In the Figures 11-14 embodiment, a first motor stator 90, comprising conductive coils or motor windings 91, is located at the rear of impeller 74. A ring of back iron 92 is located behind windings 91 and, as illustrated in Figure 9, first motor stator 90 and back iron 92 are fixed between housing 12 and casing 14. A second motor stator 94, comprising windings 95, is positioned on the forward side of impeller 74. As illustrated in Figure 11, windings 95 are fixed to casing 14 and a ring of back iron 96 is positioned forward of windings 95. As illustrated in Figures 13, 13A and 14, back iron 92 and back iron 96 have teeth 98 which extend into the stator windings to form the stator iron. Thus the windings 95 wrap around the teeth 98 in the intervening slots 99 (See Fig. 13a). In the Fig. 13A embodiment, a slotless motor stator is illustrated. In that embodiment, the windings 91 are fixed to the back iron 96 and there are no teeth extending into the stator windings. It can be seen that the motor stators 90 and 94 are placed on opposite sides of casing 14 such that each is adjacent to the pole faces of the motor rotor magnets 98. Back iron 92 and back iron 96 serve to complete a magnetic circuit. The windings 91 and 95 of the stators 90, 94 can be in series or each stator 90, 94 can be commutated independent of the other. There are several advantages to this approach: First, as long as the pole faces of the motor rotor magnets are centered between the faces of the motor stators, the net axial force will be relatively low. Second, the radial restoring force which results from the attractive force of the motor rotor magnets to the motor stators will be nearly twice as large as the restoring force with only one stator. The total volume and weight of the motor will be smaller than a single stator design. Third, the dual stator design is adapted to provide system redundancy for a fail safe mode, since each stator can be made to operate independently of the other in the case of a system failure. Fourth, hydrodynamic bearings are located on the surface of the impeller to constrain axial motion and to provide radial support in the case of eccentric motion or shock on the device. Referring to Figures 11 and 11a in particular, hydrodynamic bearings in the form of raised pads 100, 101 and contact surfaces 102 and 103 are illustrated. Such hydrodynamic bearings are symmetrically located about the impeller as illustrated in Figure 13, in which raised pads 100 are shown. The raised pads could be rectangularly-shaped or wedge-shaped and are preferably formed of hardened or wear resistant materials such as ceramics, diamond coatings or titanium nitride. Alternatively, the raised pads may be formed of a different material having an alumina or other ceramic coating or insert. The raised pads are carried by either the impeller or the casing, or an attachment to the casing. In the Figures 11 and 11a embodiment, the raised pads 100 are carried by the impeller and the raised pads 101 are carried by a cup-shaped member 104 that is fastened to the casing. Cup-shaped member 104 is utilized as a reinforcement for the casing which would not be structurally stable enough to carry the raised pads itself. The hydrodynamic bearings are formed by a raised pad spaced from a contact surface by the blood gap. Although at rest there may be contact between the impeller and the casing, once rotation begins each hydrodynamic bearing is structured so that during relative movement between the raised pad and the contact surface the hydrodynamic action of the fluid film produces increased pressure within the bearing gap which forces the raised pad and the contact surface apart. Depending upon the location of the hydrodynamic bearings, they can aid in axial support, radial support or both axial and radial support. For example, if the bearings are perpendicular to the rotational axis, they aid primarily in axial support but if they are at an angle with respect to the rotational axis, they aid in both radial and axial support. In the embodiment of Figures 11-14, the hydrodynamic bearings are positioned outside the axis of rotation, as illustrated. In the Figures 15-16 embodiment, there is a single axial motor and the stator 90 is located at the rear end of impeller 74. Stator 90 comprises windings 91, and a ring of back iron 92 is located downstream of windings 91. The motor stator 90 and back iron are fixed between casing 14 and housing 12. In the Figures 15-16 embodiment, a ring of back iron 106 is placed in the impeller, in axial alignment with the magnets, such that it completes the flux return path for the motor rotor magnets in the impeller. Thus while motor stator 90 and back iron 92 are located downstream of the impeller and outside of casing 12, back iron 106 is located within the impeller and within the casing 12. Using back iron to complete the magnetic circuit in this manner increases the overall efficiency of the motor. Referring to the embodiment of Figures 17-18, a motor stator 90 and back iron 92 are provided at the rear end of impeller 74 as with the Figures 9-14 embodiments, but another ring of back iron 108 is placed outside pump casing 12 on the front side of the impeller and is fixed to the casing. Back iron ring 108 serves two purposes. First, it serves to help complete the flux return path for the motor rotor magnets. Second, the attractive force between the motor rotor magnets and the ring of back iron 108 substantially reduces the net axial force produced by the attraction of the motor rotor magnets for the stator iron. Third, the ring of back iron significantly increases the radial restoring force compared to just the interaction between the motor rotor magnets and the stator iron. Although the Figures 1-18 embodiments utilize an axial flux gap motor, in the Figures 19-20 embodiment a radial flux gap motor is utilized. To this end, a ring-shaped structure is placed on either side of the impeller to house a series of motor rotor magnets (an even number) oriented such that the magnetic poles of the motor rotor magnets are radially, and alternately, aligned. The inner diameter of the magnets is located on the surface of a ring of back iron to provide a flux return path. On the opposite end of the impeller, passive radial magnetic bearings are used. It can be seen that in the Figure 19-20 embodiment the motor rotor magnets 110 are radially aligned. Radially within the motor rotor magnets 110 is a ring of back iron 112. The inner diameter of magnets 110 are located on the surface of back iron ring 112 (see Fig. 20) to provide a flux return path. The motor rotor magnets 110 and ring of back iron 112 are carried by the impeller, within the casing 14. Outside of the casing 14 there is radially positioned a ring-shaped stator 114 with motor windings 116. A number of axial permanent magnets 120 are carried by the impeller, at its rear end. A number of axial permanent magnets 122 are fixed to the casing 14 and housing 12, downstream of and partially offset from, magnets 120. Magnets 120 and 122 serve as passive magnetic bearings for the impeller. There are two significant differences from axial flux gap motors by using the radial flux gap motor. First, there is very little axial force produced by the interaction between the motor rotor magnets and the stator. Second, there is no restoring force with the radial flux gap motor. Radial support is provided by mechanical bearings or dedicated radial magnet bearings. It will be appreciated, then, that I have provided an improved sealless blood pump including magnetic bearings and thrust bearing suspension to minimize thrombosis, and an impeller having a blood flow path therethrough which is calculated to minimize hemolysis. Various elements from the Figures 1-8 embodiment can be used in the Figures 11-20 embodiments. For example, magnets 34 illustrated in Figures 3 and 4 could be used in impeller 74 of the Figures 11-20 embodiments. Also, rotor 18 of the Figures 11-20 embodiments could be supported using front thrust bearings such as thrust bearing 41 of the Figures 1-8 embodiment. Various other elements may be employed in the Figures 11-20 embodiments from the Figures 1-8 embodiment. Although illustrative embodiments of the invention have been shown and described, it is to be understood that various modifications and substitutions may be made by those skilled in the art without departing from the scope of the present invention.

Claims: A rotary blood pump (11), comprising: a pump housing (12); a rotor mounted for rotation within said housing, said rotor having an impeller (74); a rotor motor, said motor including a plurality of permanent magnets carried by said impeller (74); a first motor stator positioned on one side of said impeller (74) and a second motor stator positioned on an opposite side of said impeller (74); said motor stators each including a plurality of electrically conductive coils and pole pieces located within said housing; a plurality of rectangularly-shaped or wedge-shaped hydrodynamic thrust bearings (100, 101) located outside the axis of rotation of said rotor; and during rotation of the impeller the hydrodynamic bearings are separated from the housing by a fluid film and are not in direct mechanical contact with the housing. A rotary blood pump as defined in claim 1, in which said hydrodynamic bearings are are arcuate and are located on the forward side of said impeller. A rotary blood pump (11) as defined in claim 1, in which at least some of said hydrodynamic thrust bearings are carried by said impeller.

Inventors: WAMPLER RICHARD K, WAMPLER, RICHARD K.
Applicants: HEARTWARE INC, HEARTWARE, INC.
Application year: 1998
Main classification: A61M1
IPC codes: F16C39, A61M1