IN-BED PATIENT SCALE

Abstract: A scale disposable on a bed frame for measuring the weight of a person in the bed on a mattress, the scale comprising a relatively thin scale frame assembly having overall length and width dimensions related to the dimensions of the bed frame. A plurality of load cells are carried by the scale frame assembly at selected locations in the assembly, each cell producing a signal representative of the weight applied to the frame assembly in its vicinity. A scale control and display device receives the several load cell signals and produces a scale output signal indicative of the weight of a person supported on the mattress.

Description: IN-BED PATIENT SCALE Field of the Invention The present invention pertains to the field of bed scales, and more particularly to a bed scale which is positionable between a bed frame and at least part of a mattress. The scale is used for weighing a person supported on the mattress. Background of the Invention There are currently three types of bed scales available for use in weighing a person lying in a bed. The bed scales are either "built-in," "under-bed," or "bedside." "Built-in" scales are designed into the bed frame by the bed manufacturer and are usually encountered in higher priced, critical-care bed systems. They can be substantially more expensive than other types of scales, including scales according to the present invention. "Built-in" scales inherently are unable to be transported from bed to bed. They require the purchase of an entire bed frame specially designed with the "built-in" scale installed. They are not retrofittable to existing bed frames. A patient must be in one of the specially designed "built-in" scale bed frames in order to be weighed. If not, the patient must be moved to that specific bed frame. As the study that is discussed below indicates, when a patient's weight is most desired, movement of the patient is least likely to be in the patient's best interest. Moving the patient is least desirable when the patient is critically ill, and yet this is when the patient is most likely in need of having his/her weight monitored. The relatively high expense of these scales can result in the hospital or care center being unable to afford many of such beds. Another difficulty with "built-in" scales is that anything hanging on the bed frame can cause error in most of the weighing systems. The error is caused when the scale is initially zeroed out and then items such as urine bags (Folley bags), drug pumps, CPR equipment and the like are hung on the bed frame. The equipment weight causes the scale to include this in the patient's weight, resulting in an incorrect reading of weight. A scale according to the present invention can be independent of both the mattress and the frame of the bed. Therefore, anything hanging on the frame of the bed will not affect the scale or its reading of the patient's weight. The "under-bed" scales are designed as a ramp for the bed to be rolled onto. The "under-bed" scales can provide continuous monitoring of the total combined weights of the bed, bedding, and patient. These scales typically come as two part systems. One section is used at the head of die bed while the second section is used at the foot of the bed. The installation of these scale systems requires proper ramp placement and alignment so that each of the four bed wheels can be rolled onto the weighing platforms in unison and thereafter be properly positioned on the weighing mechanisms. These "under-bed" scales are not conveniently transportable, and they are expensive. These "under-bed" scales are relatively difficult to maintain, and often require specially trained personnel to operate them. If these scales are ever moved for use with another bed, they usually require re-calibration by specially trained personnel. Therefore, "under-bed" scales are often both difficult and expensive to install and to maintain. Finally, there are two types of "bedside" scales. The first type is known alternately as a sling, hammock, or stretcher scale. This scale consists of a canvas type material that is slipped under the patient in the manner of a stretcher. The canvas is then attached to an overhead hoist which lifts the patient from the bed for weighing. The second type of "bedside" scale, sometimes referred to as a litter scale, operates similarly. However, instead of having a canvas sling, this type of scale has a rigid board which lies parallel to the bed, requiring patient transfer before and after weighing. The need for accurate monitoring of fluctuations in patient body weight has become increasingly important to the medical community in recent years. This fact is being continually reinforced through comments by clinicians, nurses, and in scientific journals. Doctors must constantly tailor medications and specialty treatments according to a patient's weight or weight change. This type of customized health care is essential to providing safe and effective wound or disease management and prevention. One study in the New England Journal of Medicine describes a desire by medical personnel to observe a patient's weight within plus or minus one pound. For the complete text, see Kay Schlegel-Pratt & William D. Heizer, How Accurate are Hospital Scales?. 320 NEW ENG. J. MED. 189 (Jan. 19, 1989). Another study, in the Journal of Parenteral and Enteral Nutrition, states that "[djeterminations of body weight are important in most hospitalized patients and especially in those individuals receiving nutritional support to monitor fluid balance and to measure nutritional therapeutic efficacy. Actual tissue gains and losses of greater than 0.5 pounds/day are almost impossible. Weight changes greater than this reflect changes in total body water, a clinically relevant fact with therapeutic implications." For the complete text, see Peggi A. Guenter, et al., Body Weight Measurement of Patients Receiving Nutritional Support. J. PARENTERAL AND ENTERAL NUTRITION 441-43 (1982). Part of the Guenter study included eliciting "what factors the nurses felt needed to be changed in order to improve obtainment of weights." One of the three most important items that the nurses felt could be changed to increase weight measurement was "scales broken or unreliable." In addition, a number of nurses expressed a concern which was "focused on equipment problems such as litter scales being unavailable, broken, too cumbersome or uncomfortable for the patient." Other reasons for not obtaining patient weights included "patient refuses" and "nursing staffing shortages." The article postulated that "Patient cooperation may increase if they are not awakened . . . [and required to be moved] to be weighed." The patients were required to either get out of bed and use a standup scale or had to be moved in order to use a litter scale. In use of the present invention, there is no need to move the patient or to require the patient to get out of bed to use a standup scale. Also, it does not require extensive use of nursing staff. The medical personnel need only briefly look at the scale readout unit and note die weight on the patient's chart. Medical personnel do not have to move or wake up the patient. The Guenter article also stated that " [measurements of height and weight are considered by some to be indicators of nutritional status. Serial weight measurements are clinically useful in determining a patient's fluid status and therefore in monitoring the nutritional regimen." The study then examined the likelihood that a patient would have their weight monitored regularly. The weights were least often-obtained for patients that were the most critically ill. However, even though the importance of weight measurement was expressed to the medical personnel, the "[l]ack of weights clustered in the intensive care units with patients in whom weight obtainment may be most clinically useful." The weights of intensive care patients were often not obtained, even tiiough weight measurement was required for these patients. The two reasons most often cited for not obtaining patient weight by the nursing staff were "patient too critically ill" and "other priorities of care exist." Both of these reasons exist primarily because the existing technology that is economically viable usually requires movement of the patient to a scale unit. "Almost by definition, if a patient were in an intensive care unit, he/she was too sick to be weighed by either the conventional standing or the litter scale method." Problems in weight monitoring arise, as indicated above, including when a 100-pound nurse is confronted with moving a 200-pound bedridden patient to a scale. It is seen, therefore, that a need exists for simple, reliable, effective and relatively low cost equipment. This equipment should be able to be installed conveniently in an existing hospital bed for weighing a patient supported by the bed. Summary of the Invention This invention addresses the need which has been identified above. It does so by providing an effective, efficient, reliable, and easily usable scale assembly which can be supported on a bed frame either under a mattress or as a bottom part of a scale and mattress combination. The scale assembly is useful to accurately weigh a patient lying or otherwise supported on the mattress and to display information descriptive of die weight of the patient. The assembly can be constructed so that it is portable for ready transfer between existing bed frames. It is readily mountable on a range of bed frames and, when so mounted, does not detract from die comfort of a patient using the bed. The scale assembly preferably is comprised of a plurality of separate weighing units. These weighing units are so sized and interrelated that the assembly can be used on an articulated hospital bed frame without interference with the articulatability of the bed frame. The scale assembly also provides useful information about the weight of a patient in such a bed. The scale assembly is readily and efficiently useable, without extensive special training, by health care personnel of differing skills and backgrounds. Generally speaking, according to one of its aspects, the invention provides a scale system for measuring the weight of a person supported on a bed frame. The scale comprises a load assembly which is mountable on a bed frame between die frame and a mattress. The load assembly has length and width dimensions which are related to the length and width dimensions of die bed frame. The load assembly is comprised of at least one load receiver which preferably is substantially planar and substantially rigid. The load assembly has a height dimension which is small relative to its width and lengd so that, when placed between a bed frame and a mattress, it does not appreciably raise die mattress above the frame. The scale also includes a plurality of load sensors which are disposed at selected locations in die load assembly. Each load sensor has an output signal which is representative of die load carried by die load assembly in die vicinity of the sensor. The scale also includes a system output device which operates to receive die several sensor output signals and to produce a system output signal which is representative of die total load carried by die load assembly. Brief Description of the Drawings The above-mentioned and odier features and characteristics of this invention are more fully set forth in the following description of die presently preferred and odier forms and embodiments of die invention, which description is presented widi reference to the accompanying drawings wherein: FIG. 1 is a perspective view, widi certain parts broken away, of an articulated bed of die kind commonly used in hospitals and other health care facilities, on the frame of which are disposed four scale load receiver units according to a presently preferred embodiment of die present invention; FIG. 2 is an exploded perspective view of one individual scale load receiver unit shown in FIG. 1; FIG. 3 A is a free-body diagram representative of a load sensitive beam element which is a component of die scale unit shown in FIG. 2; FIG. 3B is a free-body diagram of an alternate load sensitive beam element of die style shown in FIG. 7; FIG. 3C is a free-body diagram of anodier load sensitive beam arrangement of die style shown in FIGS. 5 and 6; FIG. 4A is a simplified electrical schematic diagram illustrative of die electrical circuitry of a scale unit of the kind shown in FIG. 2; FIG. 4B is a simplified electrical schematic diagram for die complete scale system shown in FIG. 1; FIG. 5 is perspective view of another scale load assembly according to the present invention; FIG. 6 is an enlarged fragmentary cross-section view taken along line 6-6 in FIG. 5; FIG. 7 is a fragmentary cross-sectional side elevation view of another embodiment of a load sensitive beam element and attachment arrangement according to die present invention; FIGs. 8A and 8B are fragmentary cross-sectional side elevation views of further mounting arrangements useful in the practice of the present invention; FIG. 9A is a top plan view of the presently preferred arrangement for mounting a scale assembly to a bed frame in the practice of the present invention; FIG. 9B is a fragmentary side view of the mounting arrangement shown in FIG. 9A; FIG. 10 is a top plan view of a clamping bracket useful for mounting a scale control and display module for the presently preferred scale system; FIG. 11 is a side elevation view of a control and display module and the clamping bracket showing die movability of the module in the bracket; FIG. 12 is a view of the operator input key pad portion of the control and display module shown in FIG. 11. FIG. 13 is a top plan view of a mattress and scale set according to this invention; FIG. 14 is a side elevation view of the mattress/scale set shown in FIG. 13; and FIG. 15 is a view of an auxiliary electrical cable useful to interconnect a scale load assembly and a scale control and display module. Detailed Description As shown in FIG. 1, this invention provides a portable in-bed patient scale system 10 which includes a load assembly mat preferably is composed of a plurality of cooperating scale load receiver units 20A-20D. The scale 10 can be placed on a horizontal articulated frame 12 of a hospital bed 11. The scale 10 is configured to be located between die bed frame 12 and a mattress 13 on which a patient to be weighed can be and preferably is supported. The scale load assembly can remain in place in die bed 11, under die mattress 13, as long as necessary. The patient normally is unaware of the presence of the scale in die bed. This is because the scale's presence has no effect upon the comfort of the patient, and because of the very low profile of the scale 10 under the mattress. The mattress can be a conventional mattress as shown in FIG. 1, or it can be an air flotation mattress system of the kind described in PCT International Publication No. WO 92/07541, or it can be some other pneumatic mattress arrangement or pressure management laying surface. Further, as shown in FIG. 13, a scale load assembly can be incorporated into a mattress envelope, if desired. A common form of articulated hospital bed has a horizontal frame comprised of four load receiver sections 14A-D serially hinged together. The presently preferred in-bed scale 10 is comprised of four load receiver units 20A-D tiiat can be placed atop die respective bed frame sections 14A-D and under a mattress 13. The several scale units 20A-D substantially cover the horizontal area of the respective bed frame sections 14A-D. Because d e scale units 20A-D are physically separate from each other, except for flexible electrical interconnections, the load assembly of scale 10 follows and does not interfere with die articulation of the bed frame sections 14A-D. All four scale units 20A-D, as shown in FIG. 1, are structurally and functionally similar to each odier. However, each scale unit's preferably planar and rigid structure 21 has a length defined to substantially match its respective articulatable bed frame section 14A-D. FIG. 2 shows scale unit 20C diat corresponds to articulated bed frame section 14C as shown in FIG. 1. Because die scale units 20A-D are structurally and functionally similar, only one load receiver unit need be shown and described in detail. It will be understood mat there are similar, if not identical, devices and circuitry, as described below, provided in die odier three scale units 20A, 20B, 20D. However, the lengths of the rigid structures 21 of d ose other load receiver units vary to correspond to the length of die respective articulated bed frame sections. Each scale unit 20A-D is a generally flat rectangular construction which preferably is about one and one-quarter inch high when supported by die rubber support feet 35. One rubber foot 35 is located approximately as shown in FIG. 2, at each of die four corners of each of die scale units 20A-D. Each rubber foot 35 is vertically adjustable in the unit via the threaded post 34 that is threadably engaged with a beam end 42. For example, the rubber foot is adjusted by screwing it in or out of the beam end for height adjustment to eliminate rocking on an uneven bed frame. Each of die scale units 20A-D have lengdi (along the bed from end to end, as in "head" 15 to "foot" 16) and width (across the bed, from side to side) dimensions chosen with reference to the length and width of a corresponding bed frame section 14A-D. Lightening holes 26 are provided which preferably form a lightening hole pattern 25 in the load receiver 20A-D. These lightening holes 26 allow for a high strengdi-to-weight ratio for the scale units. It will be understood that the different length units 20A-D will have their respective pattern of lightening holes modified as compared to die lightening hole pattern 25 in FIG. 2. The other units 20A, 20B, 20D will have more, less, or the same number of lightening holes as shown in FIG. 2, dependent upon the length of each unit. The lightening hole pattern 25 is defined, as by machining or casting, in die preferably aluminum scale unit platforms 21. These lightening holes cause each platform to be substantially rigid and of low mass as compared to similar structures without lightening holes. The load sensitive beams 40 are also made as low in mass as practicable. Even with the lightening holes 26, the platform structures 21 are substantially rigid as compared to die transverse cross-sectional areas 44 of each beam 40 shown in FIG. 2. This rigidity is required in order to transmit the applied load of the patient on die scale load receiver to die load sensors 50. The load is transmitted from the top, or mattress, surface of the platform 21 to the beam flexure zones 44 via the beam's center rigid portion 41. If die scale platform 21 and die beam's center rigid portion 41 were not adequately rigid, dien the sensors 50 would not adequately sense the deflection caused by die load. By not adequately sensing die deflection, die scale's accuracy of patient weight would deteriorate from the presently preferred accuracy. Presently, die scale provides die desired accuracy if constructed in the preferred manner. Of course, though not presently preferred, a greater accuracy can be achieved if die scale's relatively rigid components are constructed to be more rigid. As shown in FIG. 2, scale unit 20C includes a rigid load-receiving platform 21, a pair of load sensitive beams 40 disposed respectively along each cross-bed edge 22 of die platform, and rubber feet 35 or other support devices at each corner of the platform 21. The rubber feet 35 are used for mounting the scale unit 20C releasably to a hospital bed frame unit and for supporting the adjacent end 42 of a load sensitive beam 40. Each beam 40 is supported at its end 42 in a manner approximating the end supports of a simply supported beam as shown in FIG. 3A. Fig. 3 A shows a simply supported representational beam 5. This representational beam 5 is die functional equivalent of a beam 40. The beam 5 is supported at its respective ends 6 by die simple end supports 1. The beam 5 has a relatively large central cross-sectional area 3 mat is transverse to the beam. The large cross-sectional area 3 is nearer to the linear midpoint between die beam's ends, i.e., die beam's center, as compared to the location of die relatively small cross-sectional area 4 that is transverse to die beam nearer die beam's ends. A force 2, representing the mass of die object to be weighed by die scale 10 as shown in FIG. 1, is applied as shown in FIG. 3A to die top surface 7 of die representational beam. The force 2 is the functional equivalent of the patient to be weighed by die current invention. The relatively large cross-section area 3 substantially transmits, without significant flexing, the applied force 2 to the relatively small cross-section area 4 of die beam 5. At die relatively small cross-section 4, die transmitted force causes die small cross-section to flex. Thus, each beam end portion 4 defines a flexure zone of beam 5. Flexing (deflection) of die beam is a result of strain, i.e., relative elongation, in the beam, and mat deflection is concentrated in die small cross-sectional area 4. Suitably mounted sensors are collocated in the vicinity of the small cross-section 4 to sense this strain. The strain measurement is men converted by a control and display module 170, as shown in FIGs. 11-12 and described below, into a representation of patient weight. Each scale load receiver unit 20A-D can be viewed overall as a single load cell, or load sensor, for sensing the weight carried by it and for generating a signal having a value indicative of mat weight. Alternatively, each scale unit can be viewed as a group of individual load cells, or load sensors. These individual load cells are so distributed diroughout the unit tiiat together they sense die weight carried by d e unit. The individual load cells generate plural signals d at together are indicative of that weight. In the following description, the term "load cell," or "load sensor," is used in both senses, with the context of use making it apparent in which sense die term is used. Each load sensitive beam 40, over a major central portion 41 of its lengdi, is fixedly, yet releasably, attached (e.g., by screws 30, nuts 31, and shims) to die scale unit platform 21. Relatively short end portions 43 of the beams 40 are unconnected to die platform 21 and extend to the adjacent rubber feet 35 to which the ends 42 of die beam are connected. The beams 40 can be pinned, hinged, hung, screwed, or otherwise affixed to die support devices mat, in scale 10, are rubber feet 35. As shown in FIG. 2, deflection occurs in the end portions 43 of die beams 40 in response to the loads applied to the platform. Each beam end portion 43 preferably has a section, as at 44, of beam from which material has been removed. The material is removed such mat the beam's cross-sectional area, that is transverse to die beam, is substantially less than the transverse cross-sectional area of the beam 40 at other places along its length. In another arrangement, this reduced transverse cross-sectional area 43 can be a wasp-waist contour in plan view. The purpose of this reduced transverse cross-sectional area 43 is to concentrate stresses and deflections (strain) in diat part of the beam length that functions to increase the stress/deflection sensitivity of die beam. That part of die beam which is strain sensitive constitutes a beam flexure zone. A load sensor 50, preferably a bonded strain gage, is suitably mounted to die beam end portion 43 at its section 44 of reduced cross-sectional area. The sensor is mounted in a manner and place selected to be most responsive to strain in e beam end portion. Thus, each beam end portion is, in effect, a sensitive load cell. Preferably, diere is one pair of load sensors 50 on each beam, widi each scale unit 20A-D comprised of two beams, resulting in a total of 8 pairs of load sensors in scale 10. Preferably, each beam's pair of load sensors are interconnected widi die odier beams such mat, overall, each beam behaves as a single variable resistor. The resistances of the several beams are effectively connected in series in the overall scale. Preferably, each beam is composed of two load sensors 50, with one mounted near each end of a beam as shown in FIG. 2. At one end of a beam, each load sensor 50 defines a pair of strain sensitive resistors. One sensor 50 at one end of die beam defines variable resistors Rl and R2, and die sensor at die odier and of die beam defines variable resistors R3 and R4. Resistors R1-R4 are interconnected in a relation as shown in FIG. 4A. As shown in FIG. 4A, resistors R1-R4 are arranged to define a bridge circuit 51 which itself behaves as a variable resistance. The output of the bridge circuit is a signal representative of die total load carried by die relevant beam 40. It will be understood mat the strain sensitive resistors R1-R4 are substantially similar. The strain sensitive resistors essentially vary only in their relative orientation of mounting to die beam. For example, one beam end's strain sensitive resistors Rl, R2 could be mounted 180 degrees in rotation relative to the other beam end's strain sensitive resistors R3, R4. This rotation is in the mathematical plane sense comprised of the top plan view of FIG. 2. The resistors R1-R4 of each beam are interconnected widi die scale's wire cabling set 101 of FIG. 2, diat is composed of a six wire circuit as is described below. Therefore, FIG. 4A is representative, in the electrical sense, of preferably one beam in d e practice of this invention. Each Wheatstone bridge circuit 51 has a connection 55 with the scale system cabling. This cabling provides die communication path for sensor information to and from die control and display module 170. The scale system's approximate electrical schematic functional representation is shown in FIG. 4B, where each beam is represented as one variable resistor. Essentially, die scale system provides a series connection of variable resistance values RVl-RVn and a fixed resistance value RF. The current invention can be represented as composed schematically of eight variable resistors, i.e., the beams, and an equivalent fixed resistance RF d at may be collocated widi a control and display module 170 diat is described below. Essentially, die voltage V is measured across an equivalent fixed resistance value RF. The voltage measurement is men converted by a control and display module, using software that is also described below, into die weight of die patient. As shown in FIG. 4B, die sum of die variable resistances provided by die several beams is indicative of die patient's weight. It will also be understood diat die entire scale system can be viewed essentially as only one variable resistance. This variable resistance, which is interpreted by the control and display module, is representative of patient weight. Preferably, die connection of the control and display module to die scale load sensors is accomplished by using a six wire circuit. The six wire circuit functions on substantially die same principles as a "Kreuzer Circuit" and "Extended Kreuzer Circuit." The details of this circuitry can be found and are more fully described in KARL HOFFMANN, AN INTRODUCTION TO MEASUREMENTS USING STRAIN GAGES 193-97 (1989). Also, preferably the scale units and the control and display circuitry are temperature stabilized and compensated using die six wire circuit, so that ambient temperature changes have little or no effect upon scale measurement accuracy. In die practice of this invention, care is taken in manufacturing the load sensitive beams 40, with load sensors gages 50 attached, to cause the beams to be as identical as possible. Care is also taken to conform as closely as possible to a stringent specification. This control of manufacture of the beams allows for precalibration of the scales and dieir individual units prior to sending them into service. This precalibration makes it possible for scale units 20A-D, and load sensitive beam assemblies 40 within scale units, to be replaced and interchanged in die field with minimum effect upon overall scale sensitivity, accuracy and calibration. The scale is readily portable and readily installed on a conventional hospital or odier bed frame. The scale does not care with what kind of mattress it is used. When in place on a bed frame, it is unobtrusive. It can be interrogated, as described below, at any time for an accurate measurement of the weight of a person in the bed. Preferably die truss-like platforms 21 of scale system 10 are powder coated for durability. The durable powder coating is preferably a substantially high visibility color, for example, orange. This provides medical staff with the ability to identify more easily ose beds d at have a scale installed and those that do not. This allows medical personnel to easily locate die scale if it is desired to move it or to allocate that bed to a specific patient. While die brightly colored and durable powder coating is currently preferred, die platforms can be left unpainted, or painted or coated with any color or colors, as functionally or aesthetically desired. In order to further immobilize the rubber feet 35 of the current invention, when diey are engaged with a bed frame, a self-adhering attachment apparatus 38 is used as shown in FIG. 9A and 9B. The attachment apparatus 38 is comprised of a shallow block 36. The shallow block 36 is attached to die bed frame 14C, for example, by means of an adhesive 37, in association with each scale unit foot 35. Each block is comprised of a relatively thick layer, as compared to the adhesive layer, of stiff material, such as a semi-resilient plastic. However, die stiff material 36 can be made of metal, composites, or odier material as desired. Each block 36 is recessed, as at 39, along one of its edges as shown in FIG. 9A. The blocks associated widi die scale unit feet at each end of die scale unit are applied to die top surface of the adjacent bed frame section. Preferably, the blocks are applied such diat die recesses of the block, located on the same side of the bed, face in opposite directions along d e bed frame section, and so diat die notches provide laterally open receptacles for the respective scale unit feet. The scale unit is placed on die bed frame so that its feet 35 engage die frame. The blocks cooperate to constrain and confine the feet from moving laterally or longitudinally of die frame. A minimum of two blocks per scale unit can be used in cooperation with each scale unit of the scale system, with one block being opposite die other in die width sense of die bed. If desired, one block for each scale foot can be used. It is preferred that the adhesive 37 used to affix the blocks 36 to a bed frame be of a semi-permanent character. The semi-permanent adhesive is preferred so diat the blocks can be removed from the bed frame sections. This removability is provided so diat if die scale is removed from die bed, e.g., for use in a different bed, the bed may be removed from service with the scale, if desired. The blocks are also removable so that the scale can be position adjusted on the same bed frame with new blocks, if desired. It may be desired to use scale 10, for example, in a healdi care facility having several beds, widi the scale being movable from time to time from one bed to another. Each bed, or selected beds, in the facility can be made ready to receive die scale before actual use of the scale in the bed. Making the beds ready for a scale is accomplished by applying an appropriate number of blocks 36 to each bed at the proper places and in die proper relative orientations. The blocks are unobtrusive and do not interfere widi normal use of a bed, even when no scale is in die bed. By replacement of blocks 36 on a number of beds, scale 10 can be moved from one of those beds to any of the other beds and placed quickly into patient- weighing use. The flexible electrical circuitry 101 that interconnects the scale units is shown in FIG. 2. For each load sensitive beam 40 mere are two wire cabling sets 101, i.e. wire harnesses, emanating from the load sensitive beam 40. Each wire cabling set is a part of the scales unit's six wire circuit, as described above, that is interconnected with die scale sensors as shown in FIG. 4A. A first wire cabling has a male connector 102 and a second wire cabling has a female connector 103. By providing both a male and female connection on each load sensitive beam 40, die load sensitive beam is not orientation specific. To connect die scale units 20A-D, the male wire cabling connector 102 is eid er connected to die corresponding female wire cabling connector of die odier load sensitive beam mat is attached to die same scale unit, or to die corresponding female connector of die nearest adjoining load sensitive beam, of another scale unit. For example, male connector 102 may be connected to die corresponding female connector emanating from die load sensitive beam attached to the edge of die scale unit 20D diat is adjacent the illustrated scale unit 20C. In die preferred arrangement, one male and one female connector are preconnected widiin each scale unit. This preconnection consists of identifying a male connector from one beam, and a female connector from the odier beam of die same scale unit. These two connectors are en preconnected to each other. By this preconnection, diere need only be one connection made for each additional scale unit that is used to form the scale system mat is installed in a bed. For example, if one scale unit is used alone in a bed, die unit is merely connected to the control and display module, requiring only one connection. However, if multiple scale units are used, dien the number of connections need only be one for die control and display module to eidier the "head" or "foot" scale unit, plus one connection between each scale unit diat is used in addition to the first unit. Therefore, the total connections required to be completed for installation of a scale system will equal die number of scale units used in a bed. In die presently preferred arrangement, after the scale has been preconnected, only four connections need be made to install the scale in a bed. These four bed-installation connections consist of diree connections between the four scale units and a connection between the control and display module 170 and eidier the relatively "head" 20A or the "foot" 20D scale unit. In die current arrangement, die connector remaining at d e "head" or "foot" scale unit to be used for the control and display module is preferably a male connector. It will be understood diat d e remaining unused connector diat is not connected to the control and display module will remain open and unconnected. This unused connector will be at d e relative "head" or "foot" end location that is opposite the end of the bed at which the control and display module is connected to die scale system. While only four connections are required to install the scale system into the bed, due to preconnection, it will also be understood diat die connections, including die preconnections, will total eight serial connections. These serial connections are preferably from the control and display module to die relative "head" or "foot" scale unit, and dien in a continuous line along the scale, such diat each beam is connected to its adjacent beam along the length of the scale system. To evaluate and process die signals from die scale units, die scale system also includes a control and display module 170 as shown in FIG. 11. This module has die attributes which are described below. The control and display module 170 of FIG. 11 has on its face 172 a control panel as shown in FIG. 12. The module is comprised of a housing 179 that has a projection 171 for removably mounting it on a bed frame. The module has a face 172, as shown in FIG. 12, that contains the functions available to the user when in USER mode. These functions are WEIGH, LBSJKG., HOLD, and ZERO. In addition to the USER mode, diere are two odier modes, named CALIBRATION and CONTINUOUS. The control and display module 170 is battery operated. Preferably, 6 size AA alkaline batteries are used widi die module. This provides approximately one year of use for weight interrogations before the batteries need to be replaced when the module is primarily used in USER mode. The module is a microprocessor-controlled, 24-bit resolution, four function tactile membrane keypad device when in USER mode. Preferably, die control and display module 170, wi iout the holding projection 171, fits within an envelope of approximately diree inches by seven inches by diree inches. The control and display module 170 performs background calibration to remove any electronic drift, and is digitally filtered. The module also provides for a six wire transducer connection to eliminate the effect of cable resistance variation. The software that is used in die central processing unit ("CPU") is preferably firmware or software at can be upgraded as desired by the manufacturer of the scale. The firmware is implemented in high-level C language diat facilitates its ease of use and reduces its development costs for any future modifications and improvements to the software as desired by die users. The overall construction of the control and display module is primarily of surface mount technology, electronic circuitry, and patented gate array design technology. This compact and reliable circuitry insures the long term accuracy and stability of die module and die scale system. The module is primarily a high precision transducer amplifier along widi a microprocessor at performs the calibration and filtering procedures described above. In addition to die batteries diat power the control and display module, a backup lidiium battery is provided to store all system configuration parameters in nonvolatile RAM. Preferably, die firmware or software data will be able to be kept for approximately ten years, even if mere is no power applied to die module from the batteries or other source, other man the lidiium battery. The module's four digit numerical liquid crystal display ("LCD") 180 has a display range from -99.9 to 999.9 when die WEIGH function is used. Once die medical staff has obtained die weight, die control and display module will automatically shut off, if mere is no keypad activity for approximately ten to fifteen seconds. If any key is pressed while die weight is being displayed, die automatic unit shut off period will automatically be extended for anodier ten to fifteen seconds, approximately. The keypad with d e four keys is a static control tactile feed back membrane switch keypad. While the keys are usually used for the functions as labeled in FIG. 12, diey can also be used for other special purposes such as when the module is placed in CALIBRATION mode. When in USER mode, the module will automatically check itself, including the functionality of the input stage, during each power up sequence. If a circuit malfunction is detected, die module will display an error or warning code and ti en shut itself down. The four keypad USER mode functions are WEIGH, LBSJKG., HOLD, and ZERO. A lbs./kg. light emitting diode ("LED") 177 is provided to signal the user whether pounds or kilograms is being displayed. The LED is turned on, causing it to emit light, when pound units are displayed, and is turned off when kilogram units are displayed. In addition, a low battery LED 178 is turned on to provide a signal to die user when batteries are in need of replacement. Preferably, the weight and other data are displayed on a four digit liquid crystal display 180, which is located just above the keypad, on the control panel face 172. In order to obtain a patient's weight, the bed is first moved into a completely flat position. Next, die scale operator, i.e. a medical staff person, presses the WEIGH key 173 to power up the module and take an initial weight reading. In actually weighing die patient, after the WEIGH key 173 is pressed the first time, it must actually be pressed a second time in order to update die weight on display from die previous weight taken. Eidier pound or kilogram units may be read. The scale operator presses the LBSJKG. key 175 to alternate between die two measurement systems, if desired. The weight is recorded and the weighing process is complete. The module will automatically shut off. In addition, the scale operator can press die HOLD button if it is desired to add or remove items from the bed. In the USER mode, as described above, die HOLD key 174 is pressed when weight is desired to be added or subtracted from the mattress, such as a pillow, a blanket, and the like. The HOLD key is a special tare, i.e. zero, function key that allows for changes in the initial zero weight setting widiout effecting the patient's weight reading. This holds the patient's weight in memory, and automatically subtracts or adds the weight diat is removed or added. When the module is in the HOLD mode the LCD 180 reads "hold." Normally, the HOLD key is pressed again to release the module from the HOLD state, once e items have been removed or have been added to die bed. Also, die HOLD function will automatically release after approximately ten minutes if the HOLD key is not pressed to release it manually. The patient's true and accurate weight will still be displayed after die HOLD key is released and die objects have been placed upon or removed from die bed while the HOLD key was locked. The ZERO key 176 is used to zero the scale to 0 pounds when no patient is residing on the mattress. This ZERO function is primarily used when the scale is initially set up, prior to use by the patient. The ZERO key 176 is a regular tare or zero function key. It is used to tare out any weight on the scale before weighing die patient. For example, is zeros out the scale with die mattress and blankets diat are provided before the patient is placed upon the bed. By pressing the ZERO key 176 once, the LCD display will show the "-0-" string. In order to prevent the mistake of pressing the ZERO button accidentally, die user needs to continue to press the ZERO button for at least five seconds to confirm the zeroing operation. During is time, the display will continue to show the "-0-" string. If the ZERO key is released too soon, die zeroing function will cancel and die display will return to its original reading. However, if die ZERO key is held for die proper length of time, die "-0-" string will begin flashing to indicate the function has taken place. The ZERO key should dien be released and "0.0" will show on the display. The weight should be close to zero after die zeroing operation. The control and display module has three modes of operation. The first mode is described above as the USER mode. The other two are the CALIBRATION mode and the CONTINUOUS mode. The CALIBRATION mode is primarily intended for use by scale technicians and service personnel to adjust the scale's performance. The CALIBRATION mode is used by die service personnel for calibrating and adjusting die control and display module. As described above, die USER mode is used by the typical users to operate the scale for weighing. To jump between the these two modes, a mode selection switch is located inside the module 170. When die system is placed in CONTINUOUS mode by operation of the internal mode selection switch, die display continuously displays the patient's weight. The control and display module 170 can be used as either a hand held device, or can be removably mounted, for example, on the bed frame using a clamping device. FIG. 11 shows a side view of a control and display module 170 that is "hinged" at 151 to a clamping bracket 150. To mount the module 170 upon the bracket 150, the module projection 171 is placed between, and men passed through, die hinge points 151 of die bracket 150. The module housing 179 and die clamping bracket are preferably constructed of a polycarbonate material. The module may be rotated about the hinge points 151 to more easily enable the medical personnel to read the display. Additionally, the module may be easily removed from the bracket by sliding it upwards. This allows medical personnel to operate die module widiout having to leave it on its bracket hinge points. The module bracket 150 is removably attached to the bed frame head board, foot board, or odier appropriate mounting area by clamping the bracket 150 to the mounting surface. The bracket is able to maintain its position by first loosening the knob 154 and sliding die clamp 153 until die mounting surface is forcibly caught between die clamping surface 156 and die bracket surface 155. The knob 154 is then rotated to tighten die clamp. While the presently preferred scale 10 is comprised of four load receiver units, diere are other forms in which a scale system according to this invention can be arranged. One alternate arrangement is to construct the scale load assembly of one large load receiver unit diat covers die entire length of die bed. This could be used, for example, with a more conventional bed frame that does not articulate. The load sensing beams can be placed with dieir orientation either similar to or perpendicular to the orientation (cross-bed) used in die system shown in FIGs. 1 and 2 of die present arrangement. If the beams are oriented similarly to the preferred arrangement, men die load sensing beam ends terminate along the opposite sides of the bed. Alternately, die load sensing beams can run perpendicular, i.e. from the "head" to the "foot" of the bed, if merely one large scale load receiver unit is used. If more man one scale unit is used in a bed, each with this perpendicular beam arrangement, then d e orientation of the load sensing beams can be oriented relatively "head" to "foot" on each scale unit. While a scale system with four load receiver units is the preferred arrangement, it will be understood diat a lesser number as well as a greater number of scale units could be used, as desired. In die presently preferred arrangement, each scale unit 20 A-D is comprised of a substantially rigid structure 25 with lightening holes that form a substantially "grid-like" geometric pattern. This patterned structure provides relatively high strength widi low mass as compared to using a metal structure with a solid, i.e., non "grid- like" pattern. While die current patterned structure is preferred, other arrangements, such as die solid structure mentioned below, are possible. The units can also be constructed with lightening holes diat do not form any particular geometric pattern. Also, while in the preferred arrangement the platforms 21 are constructed of aluminum, the platforms can be constructed of stainless steel, synthetic materials, or composites. Examples include, but are not limited to, constructing the platforms either partially or totally of carbon graphite or fiberglass materials as well as metal, metal matrix or ceramic materials and die like. In another alternate arrangement of the platforms, a combination of aluminum and honeycomb structure may be used. Preferably die honeycomb is sandwiched between one or more sheets of aluminum mat provides a relatively high strengdi-to-weight ratio. However, in the current invention, it is preferable to sandcast die aluminum truss-like platform. While sandcasting is presently preferred, odier ways of forming the aluminum platforms include diecasting, machining, and die like. While in the preferred arrangement die platform is desired to be substantially rigid, by using die one large platform that is substantially die same size as die bed frame as described above, less rigidity is allowable. For example, this can be accomplished by putting more of the load cell sensors along the edges of die platform. The several load sensors, i.e., load sensitive beams, can be placed along the platform edges. This requires less overall rigidity of die one large platform than would be required if merely four sensors are employed, one at each corner of die large single bed-sized platform. Further, die beams can be oriented perpendicular to die current preferred arrangement along die length of the scale units. If one scale unit is used, die beams can all be of the same lengdi. If more than one scale unit is desired, but die units are substantially uniform in lengdi, as in "head" to "foot", then a standardized load sensitive beam can be used in the mounting at is perpendicular relative to die preferred arrangement where the length of a beam is oriented at a right angle across the line formed from the "head" to the "foot" of the bed. As described above, platforms of varying length, as in from "head" to "foot" of the bed, are preferable. The beams can be oriented along a line diat is formed from the "head to the "foot" of the bed, even though the scale units are of varying length. However, in diat arrangement, the load sensitive beams may be required to be of varying length, with each beam length adapted to its particular scale unit length. Instead, in the presently preferred form of the invention, by mounting the load sensitive beams as shown in FIG. 2, all load sensitive beams for all platforms can be made of the same standardized length. This makes every load sensitive beam fully interchangeable on any platform of any bed scale of diat style. The scale system 70 of FIG. 7 includes a load sensitive beam 72 that is attached to the load bearing platform 74 by screws 71. The load sensitive beam 72 has a portion 77 that is rigidly attached to the platform structure 74 by screws 71 or the like. A load sensor 73 is placed suitably on a relatively flexible portion 78 of the load sensitive beam which is unconnected to and deflectable relative to the platform. The sensor 73 is placed between die screws 71 and a dowel pin 75 which passes dirough the center of die beam end 79. The dowel pin is attached on both ends to die walls of a channel 76 that is parallel to die beam and spans the space between the opposite sides of a bed frame. The beam is mounted and centered widiin the walls of this channel. Other than for the dowel pin, the relatively flexible portion 78 of die beam is not supported between me screws 71 and die end of die beam which is substantially collocated with the dowel pin. Functionally, the load bearing platform 72 of scale system 70 supports the patient's weight. The platform is carried via d e load sensitive beams by die channels 76. The ends of die channels are supported on the bed frame members 60 by feet carried by the channel at its end. The load is dius transmitted dirough the beam flexure zones 78. The sensors 73 detect d e amount of beam deflection in each flexure zone caused by the transmitted load. The dowel pin arrangement allows for relatively accurate measurement of that deflection. This accurate measurement occurs because substantially all side loads are excluded, and only d e top loading of die patient on die scale is measured. The control and display module dien converts the sensor measurements into a measurement of weight. Each beam 72 shown in FIG. 7 is supported in a manner approximating, and functionally equivalent to the simply supported beam 125 shown in FIG. 3B. Each beam 72 is supported at its dowel pin end 79 in a manner which corresponds to the way beam 125 is supported at its respective ends 126 by die simple end supports 121. Beam 125 has a relatively large central cross-sectional area 123 that is transverse to the beam. This central cross-sectional area that is an analogy to the portion of beam 72 which is affixed to platform 74. The relatively small cross-sectional area portion 124 adjacent the ends of beam 125 is an analogy to end portions 78 of beam 72. A force 122, representing the mass of the object to be weighed by die scale 10 as shown in FIG. 1, is applied as shown in FIG. 3B to the top surface 127 of d e representational beam. The force 122 is the functional equivalent of die patient to be weighed by scale system 70. The relatively large beam cross-section 123 substantially transmits, without significant flexing, the applied force 122 to the relatively small beam cross-section 124 where flexure of the beam occurs. There are other load sensitive beam arrangements that are functional equivalents but physical variants from the preferred beam arrangement described above. Such variants include those in which material is removed from the cross-section of a load sensitive beam in the vicinity of the sensing device. For example, in the preferred arrangement as shown in FIG. 2, material is removed, or drilled out, from the vicinity of die beam neutral axis between its top and bottom surfaces, dius to provide the zone of reduced cross-section 44. If desired, one may instead remove material from the external surfaces of die beam to impart to die beam a wasp-waist type of configuration. Other arrangement techniques can be used to concentrate beam stresses in the area of a load sensor so diat a highly accurate reading of the patient's weight may be obtained. As shown in FIG. 5, anodier scale system 91 according to this invention includes plural load receivers each comprised of a truss-like lightened platform core 95 having a cover 94. The core 95 can be the same material as the cover 94, or of different materials. The cover and core can be comprised of aluminum, or of a different material such as any of the materials mentioned above for scale units. The platform can be machined out of a solid block of aluminum so mat the core and cover are integrally connected. Anodier example includes constructing die cover out of fiberglass widi die truss-like core made of aluminum, and men suitably bonded to each odier. FIG. 5 also shows that, where a load assembly is comprised of plural load receiver platforms, they can be hinged to each other. When hinged, the scale load receiver units can be folded and carried as a set. The bidirectional hinges 92, depending upon die attachment to die bed frame at is chosen, can facilitate the scale in remaining properly situated upon die bed frame. By having one or more units hinged together widi a flexible type hinge as is known in the art, the electrical cabling need not be disconnected and reconnected between die hinged units each time the scale is moved from bed to bed. As shown in Fig. 5, the hinged units can be latched together via die latch 93 for transport. The arrangement can either have two units which are hinged together, but are separate from the remaining two units which are also hinged to each odier, or all four units can be hinged togetiier if desired. When die respective hinged units are folded togedier, the latch 93 is used to keep the units folded until ready for installation again. FIG. 5 also shows anodier form of load sensitive beam system 80, an enlarged view of which is shown in FIG. 6. A channel or other suitable arrangement is formed during manufacture of the platform core and cover, such that the beam system can be mounted in a recessed manner as shown in FIG. 6. Beam system 80 includes a beam member 82 having an end 88B which is rigidly attached by screws 81 to the truss-like platform core 95 as well as to the cover structure 94. The beam's other end 88A is rigidly attached to a mounting strapping plate 85 structure via screws 84. The strapping plate carries suitable straps or the like by which the plate can be buckled or otherwise securely affixed to die surface of a respective bed frame section. A load sensor 83 is suitably mounted to the load sensitive beam 82 near the center of the beam, which is collocated with the flexible beam portion 87. The strapping plate includes a cut out portion 86 which allows clearance for the beam end 88B to flex into this cut-out region. Similarly, die beam end 88A is allowed to traverse into a cut-out region 89 diat is provided above diat beam end in core 95. This modular load sensitive beam 80 is structurally different from die presently preferred beam arrangement and is functionally represented by FIG. 3C. Load sensitive beam 80 does not extend across the width of the bed. Instead, the beam is located along an edge of die scale load assembly. A plurality of beams 80 are placed along die perimeter, i.e. along each side of die load assembly. Functionally, each beam 80 is supported at its ends in a manner approximating e end supports of beam 135 as shown in FIG. 3C. FIG. 3C shows a supported representational beam 135 at is die functional equivalent of the beam 80 as shown in FIG. 6. The beam 135 is supported at its bed frame end 134 by the fixed support 137. The platform end 133 of die beam is attached to the support 138. The beam has a center portion 136 which is relatively flexible. A force 132, representing the mass of the object to be weighed by die scale is applied to the top surface 131 of the support 138. The force 132 is the functional equivalent of the patient to be weighed by the current invention. The fixed ends 133, 134 of the beam cause me force to be transmitted to the beam's central flexible portion 136, causing stress deformation of d e center of the beam. Suitably mounted sensors are collocated in die vicinity of die center of the beam to sense the deformation of the center of the beam. The strain measurement is men converted by a control and display module 170, as shown in FIG. 11-12 and described below, into a representation of weight. Another scale system 190 of this invention is shown in FIGs. 13 and 14. In is arrangement, the scale is actually a part of a mattress assembly. The scale is inserted into pockets 193 located in die bottom of a mattress cover 192, die pockets being closable via zippers 191. The scale then becomes a part of the mattress, forming a mattress/scale set 190. A mattress of suitable kind rests upon the top of the scale, inside die mattress cover. Pockets 193 are provided to allow e mattress more flexibility in the flexure regions 194 when the mattress is used on an articulated bed. A center zipper 195 provides for assembly and disassembly of die mattress and scale units, so that it can be divided in half, allowing for easier carrying of the scale by die personnel moving it. Suitable clearance holes can be provided, if desired, to allow the scale feet to protrude, for example. Alternately, the entire mattress can be strapped onto the bed frame sections, or any other attachment apparatus may be used to hold the mattress/scale set in place. Other arrangements include having die mattress be collocated upon the scale in a pocketless mattress cover. Several means of attachment can be used to cause the scale to be properly positioned on a bed frame. One such arrangement consists of attachment latches that, for example, are rotatably mounted on a scale unit; such latches can be similar to latches 93 shown in FIG. 5. Each latch can cooperate with a hook, pin or die like diat is installed on die bed frame. Another mechanism effective to cause die scale load receiver units to stay in place on a bed frame is to provide rubber feet similar to die rubber feet shown in FIG. 2, but modified to include a mounting post within die rubber foot. The post can extend dirough die bottom of the foot and into a hole in d e bed frame. This provides that the scale is supported by the rubber foot cushions but is immobilized from lateral movement by the posts. By providing this rubber cushion, die likelihood of damage to the scale is reduced. Without providing some resilient cushioning, the scale can possibly be damaged if someone, for example, a heavy person, jumps on the bed or otherwise causes a significant impact loading to the scale. Odier causes of impact loading can include performing vigorous cardiopulmonary resuscitation ("CPR"). In order to help prevent damage from impact loading, the scale preferably is constructed widi support feet diat are made of a resilient rubber. Another mediod of mounting the scale load receiver unit on a bed frame is by using angle brackets, or die like, in place of the rubber feet. FIG. 8A shows a bracket-type support device 63 diat functions to resist sliding of die scale unit off of the bed. An angle-type bracket 63 is placed upon the bed frame section 64. Movement of the scale unit is restricted by using at least two brackets for each scale unit, one on each of the opposite sides of die bed. FIG. 8B shows another mechanism for mounting a load assembly to resist sliding of die scale off of the bed frame. FIG. 8B shows a clamping-type support device 65 mat functions to cause the scale load receiver units to be more rigidly affixed to die underlying bed frame structure 67. A thumb screw 66 is threadably engaged with the device 65. The device operates to cooperate with a bottom surface of a bed frame rail to clamp a scale support post 68 to the rail. As noted above, the presently preferred scale 10 has two connectors on each beam, one male and one female. As previously described, this male and female connector arrangement allows for the present scale arrangement to be used without need for any odier external cabling in order for die several scale load receivers to be connected to each other and to the control and display unit. However, if enhanced weighing accuracy is desired, dien an additional auxiliary cable, as shown in FIG. 15, can be used. The auxiliary bus cable 185 has preferably three connectors 186, 187, 188. Connector 186 is connected to die "foot" scale unit's unused connector. Connector 187 is connected to the "head" scale unit's unused connector. Finally, connector 188 is connected to the control and display module. By using mis external auxiliary bus cable 185, the scale's measurement accuracy can be enhanced. Alternate wiring arrangements of the scale system include eight connection version comprised of one cable emanating from each load sensitive beam and connectable to a bus cable extending along the scale load assembly to the control and display module. Modular phone jack connectors can be used, as opposed to the commonly available DIN type male and female connectors used in die eight connection scale system described above. Any connector style which does not interfere widi die operation of the scale can be used. It is preferred diat a connector's width or height not exceed the clearance height of a scale unit. The clearance height of the scale unit is the smallest distance measured from the top of the bed frame on which die scale unit is mounted, to the bottom of the scale unit's truss-like platform, load sensitive beam, or other load supporting structure to which die load sensitive beam is attached. By having connectors diat exceed die clearance height of die scale unit, die transmission of the applied load dirough a beam and its load sensors onto the scale feet can be interrupted. Such interruption may cause the connector to support some or all of the load applied to die adjacent load receiver unit, diereby by-passing one or more load sensors and producing a weight measurement error. The present invention provides a reliable, low cost, and retrofittable in-bed scale system which can work with any conventional mattress. The scale also can work with the mattress described in PCT International Publication No. WO 92/07541 and most other fluid mattresses. Other mattresses include, but are not limited to, low air loss and air flow mattresses, for example. The total weight of the presently preferred scale system is approximately 45 pounds, wi an overall height of approximately 1-1/4 inch. The preferred scale system consists of four individual load receiver panels that are separately placed on die bed frame to allow for bed articulation. The connections between panels, and to the readout unit, are made via flexible wiring cabling. The approximate calibrated distributed load capacity of die scale is between 0 and 600 pounds (272 kilograms), with a sensitivity of approximately one-tenth pound, and an overall system accuracy of approximately one-half pound. However, die scale can weigh or support subjects up to approximately 1 ,200 pounds without damage to the scale. A scale according to die present invention is not "built-in" to die bed frame. Therefore, it avoids requiring die user to purchase an entire new unique bed just to have a scale that is usable while the patient lies in the bed. The in-bed scale is easy to transport from bed to bed and does not need re-calibration each time it is moved from bed to bed. The scale system does not require any specially trained personnel to install or set up the scale when moved from bed to bed or when placed on a bed for the first time. To install e scale system, its load assembly is first placed between die mattress and die bed frame. Then approximately four electrical connections are made by merely plugging together standard type connectors. This electrically connects the scale units to each odier and to die control and display module. To measure a patient's weight, health care personnel merely examine the readout unit after first pressing a button to cause the unit to display the patient's weight. Medical personnel can then record the weight on the patient's chart or use the information as required. The present scale system avoids die problems of other types of scales mat often take many minutes, and a great deal of effort, to move the patient or to cause die patient to get up and cooperate to have his/her weight measured. Instead, widi die present scale system, die only intrusion to die patient, if any, is that it is preferable to put the patient's bed in a non- articulated state prior to reading die weight. Upon reading die weight, die bed can be re-articulated to die patient's comfort, as desired. This entire operation takes significantly less time than die traditional methods of weighing a patient. This also causes little or no patient discomfort nor does it cause any significant disturbance of the patient. By using die present invention, medical personnel will not be required to physically move a patient, either manually or with a stretcher or hammock, for example. Moving the patient can be extremely difficult, as the patient may be critically ill. Also, it is often required diat more than one medical staff member be available to move a patient, as usually a patient weighs more than one person can lift. With the present invention, only one person is required and need not lift or move the patient. This prevents injury both to die patient and to die medical personnel. The present scale is able to determine a patient's weight within the range diat is likely to change due to nutritional tissue gains and losses, as opposed to weight changes that reflect changes in total body water. This provides the medical staff with the ability to monitor a patient's weight and to modify therapeutic treatment based upon the weight measurement, if desired. The scale is very reliable and has no mechanical moving parts, unlike most other scales. Because the present scale is located between die bed frame and the mattress, the patient's weight is constantly available at any time the patient is in the bed. Even when the scale is in use, the patient remains relatively undisturbed and die bed remains mobile and ready for movement. Workers skilled in the art to which this invention pertains will understand that the foregoing description of the presently preferred and other structural and procedural aspects of die invention has been presented for purposes of illustration and example, and mat such description is not an exhaustive catalog of all of the ways in which die invention can be embodied or practiced. Those workers will understand diat variations and modifications of die described structures and procedures can be made while using die principles and advances of die invention and without departing from the fair scope of die invention as set forth in the following claims.

Claims: WHAT IS CLAIMED IS: 1. A scale system for measuring the weight of a person supported on a bed frame, e scale system comprising: a) a load assembly mountable on a bed frame between the frame and a mattress having length and width dimensions related to length and width dimensions of the frame, the load assembly comprising at least one substantially planar and substantially rigid load receiver, the load assembly having a height dimension which is small relative to its wid and lengdi dimensions whereby the presence of die load assembly between the bed frame and a mattress does not appreciably raise the mattress above the frame, b) a plurality of load sensors disposed at selected locations in die load assembly each having an output signal representative of e load carried by the assembly in die vicinity of the sensor, and c) a system output device operable for receiving the several sensor output signals and for producing a system output signal which is representative of the total load carried by die load assembly. ' 2. A scale as defined in claim 1 wherein the scale is comprised of a single load receiver with its length dimension defined with reference to the length of a bed frame. 3. A scale as defined in claim 1 wherein the bed is comprised of a plurality of sections and die load assembly is comprised of a corresponding plurality of load receivers having length and width dimensions related to the corresponding dimensions of the corresponding sections of die bed frame. 4. A scale as defined in claim 3 wherein the bed frame sections are articulated and e load receivers are mountable to the frame sections to accommodate and not substantially interfere with articulation of the bed frame sections. 5. A scale as defined in claim 1 including means for releasably mounting me load assembly to the bed frame. 6. A scale as defined in claim 5 wherein the mounting means includes a means for resiliently supporting the load assembly on the bed frame. 7. A scale as defined in claim 6 wherein die mounting means includes a beam rigidly attached to the load receiver at selected locations along the beam, and means cooperable with at least one other location on the beam for releasably coupling the beam to d e bed frame. 8. A scale as defined in claim 7 wherein a load sensor is associated with each odier beam location. 9. A scale as defined in claim 7 in which the mounting means is arranged for releasably connecting the load receiver to the bed frame. 10. A scale as defined in claim 6 wherein the mounting means comprise a plurality of load bearing feet depending from the load receiver at locations selected for engagement of the several feet on the bed frame upon placement of die load receiver upon die frame, and a plurality of feet retainers affixable to the frame adjacent die places of engagement of die feet wi e frame for securing the feet from lateral movement relative to die frame. 11. A scale as defined in claim 10 wherein the feet retainers are adapted for adhesive attachment to the bed frame. 12. A scale according to claim 7 wherein the load assembly is comprised of at least one load receiver, each load receiver having a substantially rectangular platform configuration, there being a beam end adjacent each corner of each load receiver via which the receiver is mountable to a bed frame, and a load sensor disposed proximate each receiver corner adjacent the corresponding beam end. 13. A scale according to claim 12 wherein, adjacent each said beam end, a beam has a flexure zone of reduced cross-sectional area, and die corresponding load sensor comprises a strain gage coupled to the beam flexure zone. 14. A scale according to claim 13 wherein each strain gage comprises at least one strain-sensitive variable resistor, a plurality of the resistors associated with each load receiver being interconnected to define a bridge circuit. 15. A scale according to claim 12 wherein each load receiver comprises a pair of beams disposed substantially parallel to each other adjacent respective opposite edges of the receiver, whereby there is a beam end adjacent each corner of the receiver, each beam being securely affixed to die receiver along a central portion of its lengdi and having opposite end portions which are unconnected to the receiver to deflect under load as cantilever beams, the beam end portions comprising components of corresponding load sensors which further include a strain gage coupled to each beam end portion. 16. A scale according to claim 15 wherein each strain gage defines a pair of strain sensitive resistors. 17. A scale according to claim 16 wherein the strain gages associated with each beam are interconnected so that the several resistors thereof cooperate as a variable resistance. 18. A scale according to claim 17 wherein each beam and strain gage combination in die load assembly is substantially identical to each odier such combination in terms of its sensitivity to load, me no-load value of die variable resistance, and d e change in resistance as a function of load. 19. A scale according to claim 1 wherein the load sensors are disposed at selected locations adjacent each longitudinal edge of the load assembly in an array in which the sensors are disposed in pairs with each sensor adjacent one edge being opposite a corresponding sensor adjacent the opposite load assembly edge. 20. A scale according to claim 19 including a beam associated with each pair of sensors, each beam being secured along a central portion of its length to the adjacent load receiver, being unsecured to the load receiver along opposite end portions diereof, a load sensor coupled to each beam end portion, and means for mounting e load assembly to a bed frame via the ends of each beam, whereby upon placement of the load assembly on a bed frame and upon application of a load to the load assembly the beams can deflect in me end portions thereof. 21. A scale according to claim 20 wherein each beam end portion defines a flexure zone in which die beam is of reduced cross-sectional area, and die load sensors comprise strain gages coupled to the beams in the flexure zones thereof. 22. A scale according to claim 21 wherein each beam and load sensor combination is substantially identical to each other such combination in terms of its sensitivity to load and its no-load characteristics. 23. A scale according to claim 21 wherein each strain gage defines a pair of strain sensitive resistors, the resistors associated with each beam being interconnected to respond as a single strain sensitive variable resistance to load applied to die load assembly. 24. A scale according to any one of claims 19, 20, 21, 22 and 23 in which die load assembly is comprised of a plurality of load receivers and there are two pairs of load sensors associated widi each load receiver. 25. A scale according to claim 19 including a control and display module operable for displaying in selected units of measure the magnitude of a load carried by die load assembly.

Inventors: BUMBALOUGH TIMOTHY RAY, BUMBALOUGH, TIMOTHY, RAY
Applicants: BIO CLINIC CORP, BUMBALOUGH TIMOTHY RAY, BIO CLINIC CORPORATION, BUMBALOUGH, TIMOTHY, RAY
Application year: 1993
Main classification: G01G
IPC codes: G01G19