Patient support tables and monitors

Abstract: An operating table used with electrosurgery equipment 10 has a high impedance static discharge resistor 6 connected between the table and earth. A monitor 20, 20' is connected across the resistor 6 and is responsive to change in resistance between the table and earth which is indicative of bypassing or failure of the resistor. The monitor 20, 20' provides an output to an alarm 21 and also disables the electrosurgery equipment 10 in response to such a change in resistance.

Description: This invention relates to systems of the kind including a patient support table which is normally electrically-isolated from earth except for a high impedance static discharge resistor connected in series between the table and earth. Surgical operating tables and the like are often used in conjunction with electrical equipment such as electrosurgery equipment. In order to reduce the risk of electrocution of the patient, the table is usually electrically isolated from ground, apart from a static discharge path provided through a high impedance resistor of about 100KΩ connected in series between the table and earth. If, however, an earthed object is inadvertently placed in contact with the table or the patient, this provides a low impedance earth leakage path which bypasses the resistor. This can put the patient at risk if he should come into contact with a live conductor. It is the object of the present invention to provide a system including a table and monitor which can be used to reduce the risk of danger to the patient. According to the present invention there is provided a system of the above-specified kind, characterised in that the system includes a monitor connected to permanently monitor the resistance during patient use across the resistor and that the monitor is responsive to change in resistance between the table and earth and provides an output in accordance therewith. The monitor may include a capacitor, a voltage source for charging the capacitor, a switch for periodically connecting the capacitor across the resistor and a unit for detecting the decay rate of the capacitor through the resistor, the detecting unit being responsive to a change in the decay rate and providing an output in accordance therewith. Alternatively, the monitor may include a transformer, the primary winding of the transformer being connected to a source of alternating voltage and the secondary winding being connected across the resistor, the monitor including a unit for detecting change in voltage across the primary winding indicative of change in resistance between the table and earth and for providing an output in accordance therewith. The system may include an alarm unit, the output being provided to an alarm unit to provide an audible or visual warning. The system may include an electrosurgery unit, the output being provided to the electrosurgery unit to disable the electrosurgery unit. The monitor unit may be incorporated in the table or connected to the table by a cable. A system including an operating table and a monitor, in accordance with the present invention, will now be described, by way of example, with reference to the accompanying drawings, in which: Figure 1 illustrates the system in use with electrosurgery apparatus; Figure 2 is a circuit diagram of one form of monitor; and Figure 3 is a circuit diagram of an alternative form of monitor. With reference to Figure 1, the table includes a conventional patient support platform 1 at the upper end of an adjustable column 2 which is mounted on a base 3. The base 3 stands on the floor 4 on four electrically-insulative feet 5, such as of rubber, or on insulating castors or wheels (not shown). The table is made mainly of metal and it is, therefore, electrically conductive but is isolated from earth by the feet 5. In order to discharge static that may build up on the table, a high impedance resistor 6 is connected between earth and the base 3 or some other part of the table. The impedance of the resistor 6 is 100KΩ such that current flow through it is limited to values that are safe to the patient in the event that a live conductor should come into contact with the patient or table. The table may be used in conjunction with electrosurgery equipment 10 forming a part of the system. This equipment comprises an electrosurgery generator 11 that produces RF current to a hand-held electrode 12 which is applied by the surgeon to the patient's tissue so as to effect cutting, coagulation or a combination of cutting and coagulation. A return current path to the generator 11 is provided via a large area plate electrode 13 attached to a part of the patient. As so far described, the system is conventional. The system differs from previous systems, however, in that it includes a monitor 20 connected across the resistor 6 which detects change in resistance between the table and earth; the monitor responds to such a change by providing an output on line 14 to an alarm unit 15, and, or alternatively, to the electrosurgery generator 11. The alarm unit 15 provides a visual or audible alarm, or both, to alert the user that isolation of the table has been compromised. The output provided to the electrosurgery generator 11 could be arranged to disable the generator and terminate supply of power. It is preferable that an alarm is also given since there are often other electrical equipment in the vicinity of the table, such as electrically-operated instruments, monitors, lights and the like which, if faulty, could lead to a live conductor coming into contact with the patient or table. The monitor 20 may take several different forms, two of which are illustrated in Figures 2 and 3 respectively. With reference first to Figure 2, the monitor 20 has a capacitor 21 the electrodes of which are connected to respective change-over switches 22 and 23. Although illustrated as being mechanical devices, the switches 22 and 23 would in practice be conventional electronic switches controlled by a switching unit 24. With the switches 22 and 23 in the position illustrated, the capacitor 21 is connected across a battery 25, or other dc voltage source, so that the capacitor is charged. When the position of the switches 22 and 23 changes to that shown in broken lines, the capacitor 21 is instead connected across the resistor 6. In this position, the charge on the capacitor 21 is dissipated through the resistor 6. The monitor 20 also includes a voltage measuring unit 26 connected across the capacitor 21 which measures the rate of decay of charge from the capacitor when it is connected across the resistor 6. If there is an alternative path to earth in addition to the resistor 6, there will be a reduction in the time taken for the charge on the capacitor to decay to a set voltage. This reduction in decay time is detected by the unit 26 which provides an output on line 14 to the alarm unit 15 and, or alternatively, to the electrosurgery generator 11. Similarly, if the resistor 6 should fail open circuit, this would lead to an increase in the decay time which is also detected by the unit 26 and signalled to the user by an appropriate alarm. Figure 3 shows an alternative monitor 20' which includes a transformer 30 having a secondary winding 31 connected across the resistor 6 via an isolating capacitor 32. The primary winding 33 of the transformer 30 is connected to a source 34 of high frequency voltage. A voltage measuring unit 35 is connected across the primary winding 33. If an alternative path to earth exists, apart from the resistor 6, the secondary winding will be short circuited, thereby causing the voltage on the primary winding to reduce. The voltage measuring unit 35 detects any such change in voltage on the primary and provides an output on line 14 accordingly to the alarm unit 15 and, or alternatively, to the electrosurgery generator 11. The resistor 6 and monitor 20 may, for example, be incorporated within the base 3 of the table or they may be connected to the table by a cable. It will be appreciated that various other forms of monitor could be used which are capable of responding to change in resistance between the table and earth.

Claims: A system including a patient support table which is normally electrically-isolated from earth except for a high impedance static discharge resistor (6) connected in series between the table and earth, characterised in that the system includes a monitor (20, 20') connected to permanently monitor the resistance during patient use across the resistor (6), and that the monitor (20, 20') is responsive to change in resistance between the table and earth and provides an output in accordance therewith. A system according to Claim 1, characterised in that the monitor (20) includes a capacitor (21), a voltage source (25) for charging the capacitor (21), a switch (22, 23) for periodically connecting the capacitor (21) across the resistor (6) and a unit (26) for detecting the decay rate of the capacitor (21) through the resistor (6), and characterised in that the detecting unit (26) is responsive to a change in the decay rate and provides an output in accordance therewith. A system according to Claim 1, characterised in that the monitor (20') includes a transformer (30), that the primary winding (31) of the transformer (30) is connected to a source of alternating voltage (34) and the secondary winding (33) is connected across the resistor (6), and characterised in that the monitor (20') includes a unit (35) for detecting change in voltage across the primary winding (31) indicative of change in resistance between the table and earth and for providing an output in accordance therewith. A system according to Claim 1 or 2 including an alarm unit (15), and characterised in that the output is provided to the alarm unit (15) to provide an audible or visual warning. A system according to any one of the preceding claims, including an electrosurgery unit (10), and characterised in that the output is provided to the electrosurgery unit (10) to disable the electrosurgery unit. A system according to any one of the preceding claims, characterised in that the monitor unit (20, 20') is incorporated in the table. A system according to any one of Claims 1 to 5, characterised in that the monitor unit (20, 20') is connected to the table by a cable.

Inventors: GARDNER JOHN ADRIAN, HANNANT KEITH, GARDNER, JOHN ADRIAN, HANNANT, KEITH
Applicants: SMITHS INDUSTRIES PLC, SMITHS INDUSTRIES PUBLIC LIMITED COMPANY
Application year: 1992
Main classification: A61B17
IPC codes: A61B18