MAGNET APPARATUS FOR USE IN MAGNETIC RESONANCE IMAGING SYSTEM

Abstract: A magnet apparatus (6) for use in magnetic reso­ nance imaging systems comprises a superconducting coil (31) which produces a main magnetic field in a working volume (18) defined in the bore (7) in which a patient (1) is accommodated and a cryostat (32) accommodating the superconducting coil (31) and keeping the supercon­ ducting coil (31) in the superconducting state. This cryostat (32) includes only one thermal radiation shield (42) and is configured with a smaller volume than that of conventional cryostats with no reduction in the heat insulation effects. Therefore, the dimensions of the bore (7) are determined to comply with 1.0 ≦ L/B < 1.9 where L is the length along the Z axis of the bore (7) and B is its diameter. Thus, the length L along the Z axis of the bore (7) is determined to be relatively short while the dimensions of the working volume (18) necessary for diagnosis is secured in the bore (7), with the result that the oppressive sensation that the patient (1) has in the bore (7) is lessened.

Description: This invention relates to a magnet apparatus for use in a magnetic resonance imaging system (MRI system) and more specifically to a magnet apparatus having a cryostat to keep a superconducting coil, which produces a static magnetic field, in the superconducting state. The MRI system has formed therein a bore to accommodate a patient. A working volume where the patient's part subject to diagnosis is placed is defined in the bore. A magnet apparatus is disposed surrounding the bore and comprises a superconducting coil to produce a static magnetic field in the working volume and a cryostat to hold the superconducting coil in the superconducting state. A gradient magnetic field is superposed on the static magnetic field and high frequency signals are applied to the patient's part subject to diagnosis. Magnetic resonance signals from that part of the patient are detected and used to form a tomographic image of the part for diagnosis. Incidentally, the static magnetic field formed in the working volume is required to be high homogeneity. This has caused the superconducting coil to be made with a large volume, which led to the increased volume of the cryostat. Thus, the production costs of superconducting coils and cryostats have been pushed up substantially. As superconducting coils and cryostats are great in size and weight it is difficult to transport MRI systems and install these systems in the existing hospitals or the like. Further, the whole body of the patient is accomodated in the long and narrow bore. Therefore, the patient has an oppressive sensation and often suffers from claustrophobia. While MRI diagnosis is in progress, the doctor cannot directly observe the patient. If, under such a situation, the patient's condition takes a sudden turn for the worse, the doctor is unable to take an emergency measure. Prior art document EP-A-0 116 364 discloses a magnet apparatus for use in NMR imaging systems. This magnet apparatus comprises a superconducting coil disposed radially outside a bore a cryostat which accomodates the superconducting coil, a helium chamber, keeping the superconducting coil in the superconducting state, a vacuum container, enclosing the helium chamber, a thermal radiation shield, disposed in the internal space of the vacuum chamber and surrounding the helium chamber and, cooling means for cooling the thermal radiation heat from entering the helium chamber from outside. Further, prior art document GB-A-2 149 901 describes a low temperature container describes a container which comprises a heat shield and a refrigerating heat exchanger contacting the shield. The shield covers a container for a cryogen such as liquid helium. The heat exchanger is connected to a working gas feed pipe and a return pipe to and from a low temperature generating heat exchanger, a preliminary heat exchanger and a compressor of a refrigerator. The heat exchanger are in a vacuum container, the pipes in a vacuum pipe and the cryogen container in a vacuum casing. The cryogen container houses a superconductive electromagnet for an NMR computed tomograph. Further, prior art document EP-A-0 122 498 describes a cryostat including a horizontal hollow space section of room temperature in which a liquid helium vessel is supported by a plurality of support structures each having a multiple cylinder of small thickness and a liquid nitrogen vessel is supported by a plurality of tension rods. Finally, prior art document PATENT ABSTRACTS OF JAPAN, vol. 10, no. 340 (E-455)(2396), 18th. November 1986, (JP-A-61 144 802) discloses a superconductive electromagnet device in which two sets of superconductive coils whereon a superconductive wire is wound around, are arranged and housed in a liquid-helium-filled ocntainer in such a manner that the condition of uniform magnetic field is satisfied. The container is enveloped by a radiant shield, and it is placed in a vacuum container through the intermediary of a heat-insulating vacuum layer. Subsequently, a current supply part corresponding to coils is inserted into the container while the container is being airtightly sealed. The coils are connected to an exciting power source, and the values of current to be applied to the coils are made independent and allow them to make a change. Through these procedures, a highly precise uniform magnetic field is obtained, and when a quenching is generated by some cause, the current on the coil couple is cut-off, and the quantity of emitted energy is reduced. A permanent current switch may be provided on the coil in this case. It is an object of this invention to provide a magnetic apparatus for use in magnet resonance imaging systems, which has been reduced in size and weight and therefore can be transported and installed easily and can be manufactured at reduced cost and which has lessened the patient's sense of oppression. To solve this object the present invention provides a magnet apparatus as specified in claim 1. The volume of the cryostat has been made smaller than before without reducing the heat-insulation effects of the helium chamber. To take an example, the volume of the cryostat is 50% of that of conventional cryostats. As a result, the bore can be designed to comply with the condition that 1.0 ≦ L/B < 1.9 where L is the length of the bore in parallel with the Z axis and B is the diameter of the bore. The reason why the range of the value of L/B has been determined will be mentioned in the following description of the preferred embodiment. The length L parallel with the Z axis of the bore has been defined to be relatively short while the dimensions of the working volume necessary for diagnosis are secured inside the bore. Consequently, the oppressive sensation of the patient in the bore is lessened. Further, the cryostat has been reduced in size and weight. Therefore, MRI systems can be easily transported and installed in the existing hospitals or the like. In addition, the production cost of the cryostat can be decreased. This invention can be more fully understood from the following detailed description when taken in conjunction with the accompanying drawings, in which: Fig. 1 is a perspective view and a block diagram of an MRI system; and Fig. 2 is a vertical sectional view of the magnet apparatus according to this invention; Fig. 1 shows a magnetic resonance imaging system (MRI system) in which a magnet apparatus according to this invention is used. Patient 1 is accommodated in the bore lying on couch 2. The part subject to diagnosis of patient 1 is located in the working volume 18. A magnet apparatus 6 produces a static or a main magnetic field along the Z axis in working space 18. The magnet apparatus 6 is connected to excitation power source 8. X, Y and Z gradient magnetic field coils 5 are arranged radially inside magnet apparatus 6. These X, Y and Z gradient magnetic field coils 5 are respectively connected to excitation power sources 11, 12 and 13. RF coil 3 is disposed surrounding the part subject to diagnosis of patient 1. RF coil 3 is connected to RF oscillator 15 and RF receiver 16. RF oscillator/receiver 15, 16 and excitation power sources 11, 12 and 13 are connected to central control unit 14. This control unit 14 is connected to display/operation console 17. When excited, the magnet apparatus 6 produces a uniform magnetostatic field in the working volume. Under this condition, the X, Y and Z gradient magnetic fields produced by X, Y and Z gradient magnetic coils 5 are superposed on the static magnetic field. At the same the time RF oscillator 15 is driven according to a pulse sequence, pulse signals are sent to RF coil 3. Thus, magnetic resonance signals are induced in the patient's part under diagnosis. These magnetic resonance signals are detected by RF coil 3 and sent through RF receiver 16 to central control unit 14, which performs the image reproduction process and provides image information. This image information is converted into video signals which are used to display a tomographic image on display console 17. Referring to Fig. 2, the magnet apparatus according to this invention will now be described. Magnet apparatus 6 comprises superconducting coil 31 for producing a static or a main magnetic field in working volume 18 and cryostat 32 to keep magnet coil 31 in the superconducting state. Cryostat 32 includes helium chamber 41 containing liquid helium at a temperature of 4.2K. In this helium chamber 41, superconducting coil 31 is immersed with liquid helium and kept in superconducting state. Helium chamber 41 is covered by single thermal radiation shield 42. Cryostat 32 also further includes vacuum container 43 having an internal space held in a vacuum. Both helium chamber 41 and thermal radiation shield 42 are accommodated in the internal space. Multiple layer heat insulating material 44 is arranged also in the internal space. Vacuum container 43 comprises a cylindrical wall which defines bore 7. Cryostat 32 comprises means for cooling thermal radiation shield 42. The cooling means is constituted of a refrigerator having a single refrigeration stage. The cooling means comprises refrigerator 45 provided at the top of vacuum container 43, cylinder 46 having one end connected to refrigerator 45 and another end, and single refrigeration stage 48 connected to another end of cylinder 46 and connected through heat transfer plate 47 to thermal radiation shield 42. The refrigerant cooled by refrigerator 45 is adiabatically expanded in cylinder 46. Thus, refrigeration stage 48 is cooled to the temperature of 30K. Refrigeration stage 48 takes the heat through heat transfer plate 47 from radiation shield 42. Consequently, radiation shield 42 is cooled to 30K and kept at this temperature. In this manner, thermal radiation shield 42, the internal space and heat insulating material 44 of vacuum container 43, cooperating with one another, prevent the radiant heat from entering helium chamber 41. Thus, helium chamber 41 is insulated from the radiant heat and the temperature in helium chamber 41 is kept at 4.2K. The conventional cryostat will now be described briefly. The conventional cryostat comprises an inner thermal radiation shield enclosing a helium chamber and connected to a refrigeration stage of 20K and an outer thermal radiation shield enclosing the inner shield and connected to a refrigeration stage of 80K. The inner and outer shields are cooled continuously at 20K and 80K, respectively, and cooperate with the internal space of the vacuum chamber to cut off the radiant heat. However, the cryostat, having two shields, is relatively large in volume. This results in the relatively long bore and the relatively great sense of oppression that the patient has. In contrast, cryostat 32 includes single thermal radiation shield 42. Therefore, cryostat 32 can be designed to have a small volume. The surface area of the cryostat can be made smaller than before, accordingly. As a result, the amount of radiant heat that may enter the helium chamber through the surface of the cryostat can be made smaller than in conventional cryostats. Therefore, the cryostat, though with only one thermal radiation shield, exerts sufficient heat insulation effects to insulate the helium chamber from the heat. In other words, the cryostat can be decreased in volume without reducing the heat insulation effects of the helium chamber. For example, the volume of the cryostat of this invention is half that of the conventional cryostat. Further, the thermal radiation shield 42 is cooled by refrigerator 45 with a single refrigeration stage. For this reason, the refrigerant such as liquid nitrogen which is conventionally used to cool the thermal radiation shield is not required. Thus, the container to store the refrigerant can be done away with and therefore the volume of the cryostat can be further decreased. The cryostat has a reduced volume. Therefore, the bore dimensions may be determined to conform to the condition that 1.0 ≦ L/B < 1.9 where L is the length along the Z axis of bore 7 and B is its diameter. The L/B is determined for the reasons described below. It is considered that the required diameter of the working volume for diagnosis is 100 cm. Therefore, the L needs to be more than or equal to 100 cm. The B may be about 100 cm. Meanwhile, in order to lessen the sense of oppression that the patient has in bore 7, the L needs to be less than 190 cm. From the foregoing, we have: 100 cm ≦ L < 190 cm. Therefore, (100 cm/100 cm) ≦ L/B < (190 cm/100 cm). That is, 1.0 ≦ L/B < 1.9. Thus, the length L along the Z axis of the bore is designed to be relatively short while the working volume necessary for diagnosis is secured in the bore. As a result, the patient's oppressive sensation is lessened. Further, the cryostat is reduced both in volume and weight. This makes it easy to transport MRI systems and install them in the existing hospitals or the like. The production cost of cryostats is also decreased. As described above, the volume of the cryostat is made smaller than before with no reduction in the heat insulating effects of the helium chamber. Therefore, the bore can be designed to have dimensions that comply with 1.0 ≦ L/B < 1.9. Thus, the length L along the Z axis of the bore is determined to be relatively short. In consequence, the patient's trepidation in bore 7 is eased. The magnet apparatus further comprises a film sheet 65 serving as a magnetic shield and consisting of a superconducting material (e.g. NbTi). This film sheet 65 is bonded to the external circumferential surface of the cylindrical part of helium chamber 41 and the outer faces of the disc-like end parts thereof. Film sheet 65 may be bonded to the internal cylindrical surface of helium chamber 41. That is, film sheet 65 has only to be cooled by helium chamber 41 and kept in the superconducting state. When the leakage flux of the static magnetic field permeates film sheet 65, film sheet 65 produces a magnetic flux to offset the leakage flux by the Meissner effect. Thus, the magnetic fringe field is reduced. This film sheet is extremely light compared with a yoke magnetic shield. This facilitates the transportation and installation of the MRI system. In addition, film sheet 65 is positioned closer than the yoke magnetic shield to superconducting magnet coil 31. For this reason, film sheet 65 can isolate the magnetic fringe field more efficiently than the yoke magnetic field.

Claims: A magnet apparatus for use in magnetic resonance imaging systems, said magnet apparatus having an axis and a bore (7) extending coaxially with the axis (Z) and adapted to accommodate an examined body (1), said bore (7) having the length L parallel with the axis and the diameter B , said magnet apparatus comprising: a superconducting coil (31) disposed coaxially with the axis and radially outside the bore (7) and capable of producing a main magnetic field inside the bore (7); and a cryostat (32) having a cylindrical wall which defines the bore (7), accommodating said superconducting coil (31) and keeping said coil (31) in the superconducting state, said cryostat including: a helium chamber (41) containing liquid helium, said helium chamber (41) accommodating said superconducting coil (31) immersed in the liquid helium, thereby keeping said superconducting coil (31) in the superconducting state; a vacuum container (43) having a vacuum internal space in which said helium chamber (41) is accomodated; a thermal radiation shield (42) disposed in the internal space of said vacuum container (43) and surrounding said helium chamber (41); and means for cooling said thermal radiation shield to a specified temperature, whereby said thermal radiation shield (42) cooperates with the internal space of said vacuum container (43) to prevent the greater part of the radiant heat from entering said helium chamber (41) from outside said vacuum container (43), thus said helium chamber (41) being insulated from the radiant heat, said magnet apparatus being characterized in that: the dimensions of the bore (7) comply with 1.0 ≦ L/B < 1.9 between length L and diameter B of the bore (7), a magnetic shield is provided to isolate the flux of the main magnetic field leaking from inside the bore (7) to its outside, and said magnetic shield includes a film magnetic shield (65) formed of a superconducting material which is fitted closely to said helium chamber (41), and produces a magnetic flux to offset the leakage flux of the main magnetic flux penetrating into the inside of the film magnetic shield (65). A magnet apparatus according to claim 1, characterized in that said cooling means includes a refrigerator (45) having a refrigeration stage (48) connected to said thermal radiation shield, whereby said refrigeration stage takes heat from said thermal radiation shield (42) and cools said radiation shield (42). A magnet apparatus according to claim 1, characterized in that said cooling means includes a tube (61) fitted closely to said thermal radiation shield (42), said tube (61) having one end (62) open in said helium chamber (41) and the opposite end (63) open at the outside of said vacuum container (43), whereby the helium evaporated in said helium chamber (41) passes through said tube (61) and is vented to the outside of said vacuum container (43), and as it is vented, the evaporated helium cools said thermal radiation shield (42). A magnet apparatus according to claim 1, characterized in that said vacuum container (43) has a heat-insulating material (44) filled in its internal space. A magnet apparatus according to claim 1, characterized in that said magnetic shield includes a yoke magnetic shield (55) which is made of a ferromagnetic material and is provided outside the cryostat (32) to absorb the leakage flux of the main magnetic field.

Inventors: MATSUTANI, KINYA
Applicants: KABUSHIKI KAISHA TOSHIBA
Application year: 1989
Main classification: A61B5
IPC codes: H01F6, A61B5, G01R33, F17C3, A61B10, H01F7