MAGNETIC STRUCTURE

Abstract: Magnetic shielding of a magnet such as a first set of superconducting coils (20, 22, 24) as used in nuclear magnetic resonance (NMR) applications includes a second set of coils (26, 28, 30) around and spaced from the first set of coils. The magnetic flux of the second set of coils opposes and offsets the magnetic flux of the first set of coils in the region outside of the second set of coils but is less than the magnetic flux of the first set of coils in the region within the first set of coils. The uniformity of magnetic flux within the first set of coils is not adversely affected by the magnetic flux produced by the second set of coils. In a preferred embodi­ ment the coils of the first set are independently energized to establish a uniform field within the coils having suppressed lower order derivatives of flux with respect to position within the uniform field. The external field of the first set of coils comprises a dipole which is offset by the field of the second set of coils. Higher order multipoles beyond the dipole are also suppressed by the geometries of the two sets of coils.

Description: MAGNETIC STRUCTURE This invention relates generally to magnetic structures, and more particularly the invention relates to magnetic structures for use in nuclear magnetic resonance (#R) applications and the like. NMR, sometimes referred to as magnetic resonance imaging (MRI), is a new technology which can be used to produce two-dimensional slices of the human body on the video screen. The technology offers an advantage over the commercially available computed tomography (CT) scanners which require exposure of the patient to ionizing radiation. Further, NMR can be employed to observe many different types of atoms and identify the chemical environment of each atom. NMR uses a strong magnetic field to line up atoms whose nuclei have an odd number of protons. A second RF magnetic field, usually applied at right angles to the first, is used to pump energy into these nuclei, flipping them over. When the applied magnetic field is removed, the nuclei flip back and give up the energy in the form of weak but detectable electromagnetic waves. These signals are used by a computer to produce images. NMR images can be formed with protons in magnetic fields of only a few kilogauss. However, NME information from other nuclei, such as fluourine and phosphorous requires fields of a few Tesla. Indeed, even with protons there are many advantages to using magnetic fields of one to two Tesla. On the other hand, strong magnetic fields outside of the viewing area can present a danger to humans not only from possible deleterious biological effects but also from ferromagnetic objects which are subject to powerful magnetic forces in the field gradient. Standard practice has been to produce the necessary uniform fields with either end-corrected solenoids or with a number (typically four) of coils distributed along the axis of symmetry of the system. The field uniformity is determined by the geometry of the system and the relative currents in the various coils or windings, while the field magnitude is determined by the magnitude of the currents. In these systems, all of the different components produce magnetic fields in the same direction, and far from the windings the magnetic field necessarily looks as though it was produced by a dipole. The field thus decreases as the cube of the distance from the magnet. Heretofore, shielding of the magnetic field outside the region of interest has been handled in two ways. Access to the region where the field is above a nominal five gauss limit can be restricted by various means, or the field can be contained through the use of ferromagnetic shielding. For a two Tesla whole body magnet the field drops to S gauss only at a distance of 30 or 40 feet from the magnet. Access to this region must be restricted from all sides, and from the top and bottom. The use of iron to shield the outside world from the magnet is difficult and expensive because of the amount of iron necessary. For a reasonably sized room the weight of the iron amounts to many tons. Further, in addition to the building structure having to be able to support the weight of the iron, the forces between the iron shielding and the magnet have to be carefully considered. The use of a second magnetic field to offset a first magnetic field is heretofore known. See US-A-3493304, US-A-3671902, US-A-3466499. However, these applications are unrelated to NMR applications and do not encounter the constraints found in NMR applications. US-A-3493304 discloses a structure for generating a uniform magnetic field within a spherical volume by employing a single surface winding having contiguous turns over the spherical surface except at the axis of generation. Small access channels are provided at the vicinity of the axis. A shield winding is provided on a concentric spherical surface with the two windings connected in series and in magnetic opposition. US-A-3671902 discloses a shield for a single superconductive coil which comprises a plurality of conductive coils which surround the superconductive coil both laterally and longitudinally. US-A-3466499 discloses a shield for an elongate cylindrical conductor comprising a second, elongate cylindrical conductor coaxial with the first conductor. The second conductor accommodates a current sheet having a direction and magnitude to produce a field which squashes the field produced by the inner current sheet. None of these disclosures is relevant to NMR applications. According to the present invention there is provided a magnetic structure, for use in NMR applications, characterised by: a first set of coils provided about a volume and independently energizable to establish a field with suppressed lower order derivatives of flux through said volume and a dipole with higher order magnetic moments outside of said first set of coils, and a second set of coils provided about the first set of coils and independently energizable for producing a magnetic field which opposes said dipole of said first set of coils, while not substantially affecting the uniformity of said field through said volume. With structures in accordance with the present invention reduced external fields are realized while maintaining a uniformity of magnetic field in the NMR inspection area. The coil geometry is such that when the currents in the coils are set to eliminate various orders of derivatives of flux with respect to position in the center of the volume, corresponding multipole moments of the field outside of the coil region are simultaneously eliminated. A uniform field inside the coils corresponds to an external field which decreases as the cube of the distance from the coils (a dipole field), while in general a component of the field in the center which varies as the nth power of the position at the center corresponds to a field which decreases as the (n+3) power of the distance from the center. Thus, when the currents in the first plurality of coils are set to produce a very uniform field in the center, the external field is that of a very pure dipole. The properties of the second set of coils are similar to the first set in that there is a correspondence between the derivatives of the field produced by the second set in their center and the external multipoles produced by this set, The second set of coils may differ in number and angular placement from the first set of coils. Accordingly, the second set of coils can be used to produce a very uniform field in their center and a very pure dipole outside their volume. The dipole field produced by the second set of coils can be adjustable to cancel or to reduce substantially the dipole field produced by the first set of coils. The remaining external fields are greatly reduced in magnitude and decrease with a very high power of the distance from the coils. The total magnetic field within the first set of coils remains highly homogeneous, and is only slightly reduced by the field of the second set of coils. Preferably, the first set of coils and the second set of coils comprise superconducting coils. Means can be provided for generating currents in the first and second sets of coils of magnitude and directions whereby the dipole magnetic field outside of the second set of coils is cancelled or substantially reduced. By providing effectively shielding as in magnetic structure in accordance with the invention, the volume to which access must be restricted is reduced enormously as is the amount and weight of iron required to shield the remaining field, if desired. Moreover, the uniform magnetic field through the volume is enhanced by eliminating the lower order derivatives of flux. This invention will now be described by way of example with reference to the drawings, in which: Figure 1 is a perspective view of a magnetic structure in accordance with the invention; Figure 2 is a section view of the magnetic structure of Figure 1 taken along the line 2-2; Figure 3 illustrates the magnetic shielding available with the magnetic structure of Figure 2; Figure 4 illustrates the external field of a structure in accordance with the invention and the external field of a conventional magnet; and Figure 5 is a section view of sets of coils showing coil locations for different coil cross-sections. Referring now to the drawing, Figure 1 is a perspective view of the outside of a magnetic structure for use in NMR imaging. Conventionally, the magnetic structure comprises a generally spherical vacuum housing 10 having a cylindrical opening 12 therethrough for receiving a patient and couch for imaging purposes. As above described, the magnet must be of sufficient strength to generate a uniform magnetic field within the opening 12 of the order of one to two Tesla. The magnetic structure appears as a dipole outside of the housing, and because of the field strength dangerously high magnetic fields exist up to distances many feet from the housing 10. Shielding required to make the magnetic structure useful in an ordinary hospital space, for example, would be prohibitive in weight. Figure 2 is a section view of the magnetic structure of Figure 1 taken along the line 2-2. A first coils means comprising a first set of superconducting coils 20, 22, and 24 surround the cylindrical opening 12 and define a first strong and uniform magnetic field necessary within the cylindrical opening 12, the coils being supported by suitable supports as shown. Positioned about the first set of coils 20-24 and spaced therefrom is a second set of superconducting coils 26, 28 and 30. The second set of coils is energised by a current having a direction to generate a second magnetic field in opposition to the magnetic field generated by the first set of coils 20-24. The magnitude of the second magnetic field is established to cancel the dipole moment of the first magnetic field outside of the magnetic structure. However, the second field is substantially less than the first magnetic field within the opening 12, and the field within the opening remains uniform. The magnitude of the field within the opening is established by the currents through the two sets of coils. To further enhance the magnetic shielding a ferromagnetic shield 31 can be provided around the magnetic structure. This arrangement of coil sets is the so-called Maxwell three-coil configuration. The inner three coils are on a sphere of 80 cm radius and the outer three are on a sphere of 125 cm radius. The current in the outer coils is adjusted to exactly cancel the external dipole (about 408 of the current in the inner coils) and the resultant central field is thereby reduced to about 74% of that produced by the inner coils alone. This central field varies as the sixth power and the external field as the inverse ninth power of the normalized radius. The following table illustrates the central field uniformity: CENTRAL FIELD UNIFORMITY Field Uniformity Diameter of Volume 1 x 10-7 9 cm 1 x 10 6 13 cm 1 x 10 5 20 cm 1 x 10 4 29 cm The structure of Figure 1 uses forced-low cooling from a dedicated refrigerator 35 that is part of the magnet installation. This refrigerator provides a stream of saturated liquid helium which flows in tubes or channels 32 in thermal contact with the superconducting windings. These channels may be integrated into the winding itself if necessary, and they may be run to areas of high local heat leak, such as support points. After passing through the magnet, the stream still saturated, returns to the refrigerator. The refrigerator also provides a flow of helium gas at 20 K to cool shielding that intercepts the thermal radiation and conductions from room temperature. The heat load on this shielding is controlled by means of multi-layer insulation shown at 34. However, this insulation does not have to be of the highest performance. The shield cooling gas also returns to the refrigerator, near 60 i, The structure is within vacuum vessel 34. Structure 36 supports the coils along with fiberglas reinforced straps 38, thermal radiation shield 40 is positioned between the coils and the vessel 34. A non-conducting bore tube and coil-foil thermal radiation shield 42 is provided near the central opening of the magnet structure. The closed-cycle, a forced-flow cryogenic system allows enormous simplifica- tion of the magnet construction over systems employing pool-boiling. Figure 3 illustrates the total magnetic field outside of the magnetic structure in the presence of the opposing magnetic field. The magnetic field in the central region is 1.5 Tesla. From this drawing it is seen that the five gauss contour is confined to a distance within approximately 1-2 meters of the magnetic structure. Thus, the magnetic structure can be used with NMR imaging equipment in a typical hospital laboratory room. Figure 4 compares the external field of the present structure with that of a conventional NMR imaging magnet. The internal field produced by either magnet is 1.5 Tesla. The invention can be mathematically described with long solenoids although it must be emphasized that the use of solenoids does not offer the advantage of other systems to be discussed shortly. The field on the axis of a solenoid can be written as EMI9.1 where z is the distance along the axis from the center of the solenoid, L is the length of the solenoid, a is its radius, and Bo is the field at the center. It is straightforward to expand this expression in powers of llz to show that when z is large compared with L and a, EMI9.2 Now, suppose we have two concentric solenoids, forming a symmetric system. For simplicity assume that the length of the solenoids are the same, and assume that they are long compared with their radii so that ("a2+L2/4)/L L/2. Then if we adjust the currents in the solenoids so that bo1 a21 +B02 a22 = 0. we will have a system with no dipole field and the field at large z will fall off as l/z5. The field in the center of the system will be EMI10.1 The penalty which has to be paid for the reduced external field is an increase in the current in the coils in order to produce the same central fields as without the cancelling coils. With normal conducting coils this penalty may be quite severe, but with a superconducting system the gains are worth the increased winding weight and complexity. Systems of coils located on the surface of spherical shells are even more interesting than the solenoidal example above. It is well known that the magnetic field produced by current flowing on a spherical shell is absolutely uniform inside the shell and is a perfect dipole outside the shell, if the current density varies as sink, where Ok is the angle between the internal field and the vector to the shell. The fact that the external field is a perfect dipole guarantees that the field outside the larger of the two concentric spherical shells is identically zero, as long as the ratio of the current densities on the two shells is adjusted to cancel the dipoles. The field inside the smaller shell will be somewhat reduced, but it will still be absolutely uniform. To be specific, the 'z' component of the field inside a shell is proportional to I/a, while the outside is proportional to I/a (a/z)'. If there are two shells, then h 2 ' o a1 4 12 sets the external field to zero. The field inside the smaller EMI11.1 where B1 is the field produced by the inner shell alone. While the field produced by the two concentric shells is ideal from some point of view, the configuration usually won't be very practical. The problem, of course, is access to the uniform field region. While access could be attained by arranging for the shells to be opened temporarily along JA constant planes, fields produced by discrete planar coils providing inherent accessibility should be superior for many purposes. To this end, consider a single spherical shell with a finite number of coils, wound at discrete angles i on the surface of the shell. This, of course, sets the planes of the various coils parallel to each other. Considering the direction perpendicular to these planes as the 'z' axis, it can be shown that the magnetic field generated by the 'i'th coil along the axis is given by: EMI11.2 or EMI12.1 In these equations, Ii, and ai refer to the current and the coordinate of the ith coil located on the spherical shell of radius a. Pn (x) is the associated Legendre polynominal in standard notation, and z is the distance along the axis from the center of the sphere. Note that the diameter of the coil is given by a sin ai. The total field produced by a set of coils can be obtained by adding the contribution of the individual coils as given in equations 7 and 8. It can now be seen that if the ai and Ii of a set of coils is chosen so that the coefficient of any of the zm 1 term is zero for the internal field, then the coefficient of the z#(m+2) term will also be zero. To produce the most uniform field inside the sphere, the obvious procedure is to choose the ai and Ii to eliminate as many of the zm-1 terms (for m Ire 1) as possible. (Depending on whether the desire is for maximally uniform field at z r o, or for maximum average uniform ity over some specific region, or some other particular requirement, the details of the procedure will differ but the idea will not.) The presence of the mtl term will give a uniform field inside and a dipole outside. Every zm-l term eliminated inside will automatically eliminate a #-(m+2) multipole term outside. Thus as the internal field becomes more uniform, the external field becomes a more perfect dipole. As examples, consider a two coil system, and a three coil system. If the coils are placed and excited symmetrically about the i-0 plane ¯rr/3) then half the terms in the field expansion will automatically vanish. This symmetry will be assumed. For the two coil system we can writes EMI13.1 EMI13.2 sin a (5 cos2 a -1). if ¯ is chosen then the z2 term vanishes. The fields (on axis) inside and outside the sphere can then be written: EMI13.3 As the radius of each coil is given by a sin a2 and the coil separation is given by 2a cos a, the choice cos2a=i/S is just that for the coil radius to be the same as the separation. This geometry is familiar as the Helmholtz configuration. For the three-coil system we write: EMI13.4 From the symmetry condition discussed earlier, a a #12. We now choose Io/Il and al so that both the z and the z term vanish. The two conditions can be written (P1 (cos EMI13.5 sin a (21 cos4 a - 14 cos2 a+1) ) as EMI13.6 These are satisfied by cos2 al r 3/7, EMI14.1 With these choices, B: can be written as: EMI14.2 This three coil configuration, not as well known as the Helmholtz coils, is called the Xaxwell geometry. The purpose of these examples which are well known is to illustrate that the multipole components (other than the dipole) outside the sphere are reduced to zero at the same time that the spatial derivatives of B are reduced to zero inside the sphere. In the usual coil arrangement the fact that the dipole field outside is becoming more and more perfect is rather irrelevant. The dipole field is large and carries a substantial amount of flux. The small perturbations in the dipole field at modes values of (a/z) are completely irrelevant to the field and shielding problem. However, if two sets of coil geometries are used, but with different shell radii a, then the picture changes substantially. As discussed earlier, one set of coils can be used to cancel the dipole field of the other. In this case, the only fields outside the larger sphere are the higher order multipoles which decrease very rapidly as 2 increases. As an illustration, consider two sets of the three coil geometry. Let the smaller set carry current lo in its center coil and have a radius of aO. Let the larger coil set carry current Il and have radius a1 The coils are concentric and coaxial, so the resulting field on axis can be written as: EMI15.1 2 Now choose 11/10 (a1/a0) a -l so that the dipole term is zero. The expressions become EMI15.2 It is seen that the inner field is reduced by the factor 1 - (aO/al)3 from its value without the field due to the outer sphere. However, the field outside now changes very rapidly1 as intended. As a specific example, assume that the outer coils are on a radius of 125 cm and the inner coils are on a radius of 80 cm. The inner field is then approximately 74% of its uncompensated value. However, if the inner field is maintained at a value of 1.5 Telsa, the external field has dropped to 5 gauss at a distance of 2.8 meters. If the dipole term weren't cancelled, this distance would have to be 11.5 meters. The equations also show that the uniformity of the field in the central region is excellent, even though the field falls off so rapidly outside. In the example, the field varies by only a part in ID7 over a volume 9 cm in diameter or by a part in 106 over a volume of 13 cm in diameter. The excellent field uniformity of the magnets will not be preserved in the presence of nearby ferromagnetic material or other sources of magnetic fields. This problem is, of course, not unique to the particular winding geometry discussed here. It is typically solved through the use of correction winding. to null the offending fields and gradients. It is also necessary that the external sources be constant in time and position, and that they not be too close. Since many such sources are not easily controlled (cars, trucks, elevators, tool carts, etc.), it is frequently necessary to surround the room containing the uniform field magnet with ferromagnetic shielding. In the normal winding configuration this also reduces the field outside the shield to allowed levels. The problem with the use of the ferromagnetic shield with conventional magnets is that the weight of the material required can be enormous. The self shielding magnets described here require much less material. The following simple model shows that the required material differs by a factor of 500 or more between the two magnets. Assume that the field outside some magnet falls off as The The field can then be written as B B5 (a5/r)n where B8 is the average field over the surface of a sphere of radius ds outside the magnet. The flux which has to return outside the radius as is EMI16.1 This flux must be returned inside the shielding material, and through the requirement that it not be saturated sets its minimum thickness. To illustrate the requirements, compare the 6 coil self shielded magnet discussed earlier with the 3 coil conventional magnet Assume a spherical shield with 3 meter radius and assume that the central field is 1.5 elsa. Interestingly, in the model being used, the weight of the shield for the conventional magnet is independent of the shield radius. The field at the shield radius will be approximately 3.8 gauss for one magnet and 280 gauss for the other, leading to a return flux of 31 gauss meter2 and 16 kilogauss m2 respectively. If the shielding material is assumed to saturate at 10 kg, then 31 cm2, or 1.6 m2 is required, for a thickness of .16 mm or 8.5 cm. If this is translated into the weight of an iron sphere of 3 meter radius and the indicated thickness, assuming a density of 7.5 gms/cm3, the conventional magnet requires 80 tons, while the self shielding magnet requires only 300 lbs. In addition to the weight of the shield, the forces between the shield and magnet need to be taken into account. These forces are proportional to the product of the magnitude of the field and the field gradient at the location of the shield. Although the gradient of the self shielded magnet's fields is larger than that of the conventional magnet, the magnitude of the fields is such that the forces in the self shielded case are only 1/2000 those of the conventional case, at the 3 meter point. The calculations presented are based upon idealized coils consisting of filamentary current loops as approximations in some sense to an equally idealized current sheet on the surface of a sphere. Real systems will have to be made with coils of finite dimensions, and will require some modifications in some of the details of the preceding equations. Enough work has been done on numerical calculations using dimensions of realistic coils to show that the advantage of these ideas can be realized in practical designs for central fields of at least 2 Tesla. Further, in the described embodiment the coils are assumed to be on a spherical surface. However, the coils could be arranged on any generally spheroidal surface including an approximation thereof, the approximation being dependent on a number of factors including coil cross-section. This is illustrated in the section views of several sets of internal coils having different coil cross-sections shown in Figure 5. The coils 50 form a first set in which the aspect ratio (height-to-width, h/w) is 1; the coils 52 form a second set in which the aspect ratio is 0.1; and the coils 54 form a set in which the aspect ratio is 0.03. The placement of the coils can be compared with the placement of ideal filament coils 56. The coil sets 50-54 were generated by computer simulation of seal coils based on the integration of the expression for magnetic field with respect to i for the real coil configurations and assuming a uniform current density in the coils. The discussion of the technique for cancellation of the external fields has concentrated on examples in which the desired field in the interior region of the coils is uniform. It should be clear, however, that the technique is perfectly adaptable to the production of rather arbitrary field configurations within the central region, while retaining the property of very low external fields. An example is the production of the uniform gradient fields required in NMR imaging. These fields are typically pulsed and have to be superimposed on the large uniform field. The induction of eddy currents in nearby conductors can be a serious problem. The use of the external field cancelling method being presented here can allow the gradient fields to be produced with greatly reduced problems due to the eddy currents.

Claims: CLAIMS 1. A magnetic structure, for use in NMR applications, characterised by a first set of coils (20, 22, 24) provided about a volume (10) and independently energizable to establish a field with suppressed lower order derivatives of flux through said volume (10) and a dipole with higher order magnetic moments outside of said first set of coils (20, 22, 24), and a second set of coils (26, 28, 30) provided about the first set of coils (20, 22, 24) and independently energizable for producing a magnetic field which opposes said dipole of said first set of coils (20, 22, 24) while not substantially affecting the uniformity of said field through said volume (10). 2. A magnetic structure as claimed in claim 1, characterised by an opening (12) for receiving a patient said first set of coils (20, 22, 24) and said second set of coils (26, 28, 30) being arranged on generally spheroidal surfaces about the opening (12). 3. A magnetic structure as claimed in Claim 1 or Claim 2, characterised in that said first and second sets of coils (20, 22, 24 and 26, 28, 30) are configured and energized to suppress lower orders of derivatives of flux within said first and second sets of coils and to suppress corresponding multipole moments of the magnetic flux outside of said second set of coils (26, 28, 30), said first and second sets of coils suppressing multipole moments outside of the structure. 4. A magnetic structure as claimed in any preceding claim, characterised in that said first set of coils (20, 22, 24) and said second set of coils (26, 28, 30) comprise superconducting coils. 5. A method of shielding a magnetic structure for use in NMR imaging characterised by the steps of providing a first set of coils (20, 22, 24) which define a first spheroidal surface, energising each of said first set of coils to establish a uniform magnetic field within said first spheroidal surface and a dipole outside of said first spheroidal surface, providing a second set of coils (26, 28, 30) which define a second spheroidal surface concentric with and outside of said first spheroidal surface, and energizing each of said second set of coils to establish a magnetic field outside of said second spheroidal surface which offsets said dipole while not significantly affecting the uniformity of said magnetic field within said first spheroidal surface. 6. A method as claimed in Claim 5, characterised in that each coil of said first set of coils is independently energized to suppress lower order derivatives of flux with respect to position within said uniform magnetic field and simultaneously suppress lower order magnetic moments outside of said first spheroidal surface.

Inventors: FAIRBANK, WILLIAM MARTIN, MCASHAN, MICHAEL SANFORD, SMITH, TODD IVERSEN
Applicants: THE BOARD OF TRUSTEES OF THE LELAND STANFORD JUNIOR UNIVERSITY
Application year: 1985
Main classification: G01N24
IPC codes: H01L39, G01R33, A61B10, H01F6, A61B5