INSULATION ANALYZER APPARATUS AND METHOD OF USE

Abstract: An insulation specimen (10) may be analyzed by removing any absorbed charge from the specimen (10). Test voltage E is applied through system resistance R. ITOT is sensed at time intervals occurring at selected times multiplied by any three points in a power series while maintaining E constant and RIR is calculated from the formula: <IMAGE> and displayed. Apparatus consists of a high voltage power supply (12), current sensing means (14) and specimen terminals for connecting an insulating specimen (10) across the power supply (12) in series with the current sensing means (14). A voltage comparator (16) is connected across the specimen terminals to sense effective voltage across the specimen (10). Computation means (18) receives inputs from timing means (18), the current sensing means (14) and the voltage comparator (16) and is used to calculate insulating current using a formula involving elapsed time measurements of total current through the insulation specimen to calculate insulation current and the insulation resistance of the insulation specimens. Means (18, 2a) is employed to adjust the high voltage direct current power supply (12) and display means (22) permits display of insulation current and insulation resistance readings.

Description: INSULATION ANALYZER APPARATUS AND METHOD OF USE The present invention relates to measurement of insulation resistance, and other related properties, of a specimen using a method and preferably employing a novel insulation analyzer apparatus in accordance with the present invention. Background of the Invention Since close to the beginning of the century, insulation measurements have been made using techniques which have been highly developed and instruments built specifically for that purpose such as the Megger ® sold by James G. Biddle Company. Such instrumentation requires a high degree of knowledge and skill on the part of the operator in order to obtain readings which are reliable and, by following a painstaking procedure, can achieve a high degree of accuracy in measuring : insulation resistance. Persons with the skill and patience to practice this method are becoming relatively unavailable. In 1958, E. B. Curdts made a technical analysis of insulation testing entitled "Insulation Testing By D-C Methods" which he revised and reprinted in 1964 in Biddle Technical Publication 22T1. In that publication, Curdts showed that when d.c. voltage is applied, the current existing in the insulation of a capacitive specimen is always made up of three components, to wit: geometric capacitance current, i g ; absorption current, i a ; conduction or leakage current, i z . These currents will be explained more fully hereafter, but it will be understood that the current which was measured by all conventional insulation testers was i^OT' where Ϊ T QT = i g + i a + ^c* 5 Because the i TO τ is measured by all conventional. insulation testers, various elaborate techniques have been evolved to estimate the value of i c in the presence of i and i a . Typical examples are time- resistance tests, step-voltage tests, polarization 10 index and dielectric absorption ratio tests. These tests have been developed to a considerable degree of sophistication and achieve excellent results when applied by skilled technicians. However, they- suffer from the disadvantages of taking comparatively long 15 periods of time and requiring great care and skill in the measurements and their interpretation. In 22T1, previously referenced E. B. Curdts has described a method of calculating i c - He shows that since the absorption current is a power function of 20 time, constant n (with a value between 0 and 1.0) is the slope of the straight-line current-time curve plotted on a log-log basis. The leakage current i c will represent a. deviation from this curve, and may be calculated from 25 il x O " is . l β 2 ι„ - where i j , i3.- j g a nd i 10 are three values of i TOT measured at different times, based on a 30. constant unit of time multiplied by their subscripts, i.e., 1, 3.16 and 10 minutes. OMPI This relationship is true provided that: a. i q has fallen to a negligible value compared to i a and i c ; which therefore also assumes that b. i_ is negligible at that voltage level when compared to i a and i c « Nature of the Present Invention The present invention involves apparatus for automatically accomplishing these tests. In particular, " the present invention employs an insulation analyzer apparatus which employs a high voltage direct current power supply. Current sensing means is employed and specimen terminals are provided for connecting the insulation specimen across the power supply in series with the current sensing means. A voltage comparator is connected across the specimen • terminals to sense effective voltage across the specimen. Timing means is provided. Computation means for recovering input from the timing means, current sensing means and voltage comparator in calculating insulation current as a function of elapsed time measurements of total current through the insulation specimen to calculate insulation current, and the insulation resistance of the specimen's insulation. Means are provided to adjust the high voltage direct - current power supply. Display means are also provided permitting reading of various insulation and other properties of the specimen. In accordance with the present invention, the following method of analyzing an insulation specimen is provided which may be practiced using the analyzer apparatus. First, remove any absorbed charge from the specimen. Next, apply test voltage E through a fixed - D RE 7 OMPI resistance, R, in series with E. Then, time the interval from V = 0 until V = 0.632E across the specimen. As is well known, this is equal to one time constant for a combination of resistance and capacitance arranged as shown. Next, using formula t = C x R, where t is time in seconds, calculate C and store C in CR. Next, as an optional step, calculate and display of value for 100CR, start timing and count display from 100CR to zero. In the course of this countdown, maintain E constant and measure in time increments related in the sequence (tτ_, 2 t t 3 ) 1, 3.16, 10, for example at 10CR, 31.6CR and 100CR. Calculate time insulation resistance '(RJR) from and that is where C For a better understanding of the present invention, .reference is made to the drawings in which; Fig. 1 is a plot of current in microamperes against time in seconds of i g , i a , ' i c and the total of these three currents ? Fig. 2 is a equivalent circuit representing the insulation specimen broken down into the respective current paths for the three current components, a possible additional current i q / defined below, and showing the power supply and system resistance; and Fig. 3 is a block diagram of an insulation analyzer in accordance with the present invention. Referring to Fig. 1, plots of the four currents are shown. These currents may be described as follows: Geometric Capacitance Current, i,, This is the current due to the capacitance created by the geometric arrangement of electrodes and the dielectric constant(s) of the insulation(s) . -t CR ig ' ε R Where E = open circuit voltage of the dc supply; R = the total internal series resistance; t = time; and C = capacitance of the specimen. Absorption Current, i a This current results from absorption within the imperfect dielectric, caused by polarizations within the molecular chains of the dielectric. i a = Δ VCD t "n = At~ n where V = the increment of applied voltage C = capacitance of the specimen D = a proportionality factor on a per unit basis of applied voltage and the capacitance of the specimen. This depends on the type of insulation and its condition and temperature. A = Δ V CD n = a constant t = time Conduction or Leakage Current, i This current is the resistive current either - y zEA " . OMPI, through the bulk of the insulation or over surfaces such as terminal insulator. E S 1 c R S where E s = the applied direct voltage (at the terminals of the specimen) R s = the insulation resistance of the specimen under test Partial Discharge Current, i_ As shown by R. J. Densley in ASTM-STP 660 Chapter 11, an additional component will occur when the value of E s is above the partial discharge inception voltage for the specimen. i=n iq " . Σ f i<3i ι=l where f^ = discharge repetition rate of the discharges of magnitude, qi, occurring at a discharge site, i. For most purposes i g , the partial discharge current can be ignored and the plot i t is equivalent of i«jo * Ifc should be noted that Figure 1 is plotted on a logarithmic scale. Referring to Fig. 2, the currents may be visualized as dividing through the specimen. As shown, iq is treated as negligible for most purposes and in most cases this is acceptable. In test instrument terms, a typical workable range would be to provide a test supply E with an internal series resistance of 1 M Ω . The capacitive load C of the specimen might range from 10 pF minimum to 10 uF maximum. Thus, ignoring the effect of insulation resistance, the OMPI system provides a range of time constants from RC - 10~ seconds minimum to 10 seconds maximum. These assumptions are illustrative and are not critical to the operation of the instrument but permit practical values to be used with the final specifications of an instrument designed in each individual case based upon need. The method of the present invention in its simple mode of operation can be specified in the following steps: Step 1. Conditioning of the specimen. Remove any absorbed charge remaining by one of three methods: apply reverse dc, reducing " in amplitude to zero; or apply ac reducing to zero; or short circuit the specimen for a significant time. The first two methods are obviously more desirable because the time involved is short. Step 2. Apply the test voltage E via R. Measure the time interval from 0 until - V = 0.632E This time interval gives - t (sees) = C x R (10 6 Ω ) Calculate C, store in memory CR and C. Step 3. Start Timing. Calculate and display 100CR. Count display down to zero (test finished) . Step 4. Maintaining E constant, measure iγoτ i n time increments related in the sequence (t j , t 2 , t 3 ) 1, 3.16, 10, for example 10CR, 31.6CR and 100CR. Calculate true Insulation Resistance ( JR) -£!J E 7J- OMPI from and that is where E R IR = : .- - R r, where R is the resistance shown in Fig. 2, 12b Step. 5. Display and/or print E, RJR/ C, or any other form that can be calculated from these values, e.g., zero frequency power factor, tan δ , etc. It is also possible to calculate the slope r of the absorption current, which may be of interest with some types of insulation. Operation in a complex mode of operation is also possible wherein measurements in the simple mode are , repeated at a voltage E, selected to be below partial discharge inception, and the results stored. The voltage is then increased in steps and the value of the R IR at each voltage compared with the previous value. The point where a decrease, is detected corresponds to the partial discharge inception. At the conclusion of the test the instrument can read out: C, j R , partial discharge inception, partial discharge at a predetermined voltage. The calculation of partial discharge magnitude is based on measuring Δ ic , since Δ £ c (representing partial discharge) is based on one coulomb/sec = 1 ampere (by definition) . The method of detection of Δ -ξlJ £4 OMPI - 9, - I c requires the use of techniques already developed in conventional partial discharge measurement, for example, the James G. Biddle Co. Catalog 665700-00 partial discharge bridge. Discussion of the Measurement Most measurements would be of such a short duration that the delay would be negligible. For example, with a 0.1 vF capacitive specimen, a total measurement time of about 10 sees, is required for the simple mode and for a full analysis of 10 voltage steps, only 1 min. 40 sec. The worst case would be a 10 μF specimen where a time of 16.6 minutes is required for a simple measurement. This is probably acceptable, because the test implies a very unusual measurement such as the largest size of generator or a very long length of cable. No operator intervention would be required during the test. It is quite simple to reduce the timing, either by reducing the initial reading from 10CR to say SCR (at some loss of accuracy) or by reducing R. In the case of portable field apparatus, the value of R should also be selected to permit measurements at a current of less than 5 A for reasons of safety. A number of variations of the basic concept are possible. a. Instead of a constant voltage source for E, it could be replaced by a constant current generator. This somewhat modifies the calculations but otherwise does not change the concept. The decision therefore depends on practical and economic considerations. b. If the initial value of CR is too short for fUREr OMPI accurate timing with a practical microprocessor clock rate (say 4 MHz) it may be desirable to go back to step 1 and repeat the sequence with the higher value of R. Similarly, if CR is very large, measurements could be automatically changed to 50CR total for shorter times at slightly reduced accuracies. c. If high precision in the measurement of C is required, the initial calculation which ignores the effect of R R can be recalculated taking R jR into account. (Note that the inaccuracy of the initial determination of CR will not affect the value of R R since it is only a proportional error.) d. In the presence of noise it may be - advantageous to automatically perform iterative measurements and average the result. Description of the Measuring Apparatus Reference is now made to Fig. 3 which illustrates the insulation analyzer system in schematic form. Typical embodiments of the instrument are a field portable lightweight instrument with a 5 kV 5 mA maximum supply for use in the simple mode only. A more elaborate transportable unit for complex mode of operation may be rated at 30kV 30mA. The system consists of a programmable regulated high voltage d.c. power supply 12 made up of a regulated adjustable voltage source 12a in series with a resistor 12b. The voltage" of the power supply is regulated to a close tolerance by an internal feedback loop. The voltage level is set by reference voltage controlled by a digital signal originating from the microprocessor via connection 18a. A specimen 10, which is to be measured to determine its insulation characteristics and capacitance, is connected across the power supply 12 and in series with the current sensor 14. The current sensor is typically a precision resistor with a value which is small compared with resistance R12b. An electronic voltage comparator 16 is connected in parallel with the specimen 10 and the current sensor 14. This compares the voltage across the specimen 10 and current sensor 14 with a reference voltage. When the voltage reaches 0.632 E (where E is the present' value in 12a), a digital signal is transmitted by connection 16a. The ouput of the current sensor 14, in the form of an analog voltage directly proportional to the sensed current ( i n F 9- 2 ) ' i s connected to an analog-to-digital converter 21. The output of this converter, in digital form, is connected to the microprocessor 18 by connection 21a. The microprocessor system 18 and associated memory 20 perform the control, computation and timing function necessary to the measurement. Stored in the read-only portion of the memory 20 is a program which executes the actions listed as steps 1 through step 5 in the preceeding description. The microprocess and associated integrated circuits and read only and random access memories can be any of a wide range of currently available microprocessor equipment, such as the Radio Corporation of America's COSMAC Microprocessor Type 1802 Handbook MPG 180C "COSMAC Microprocessor Product Guide". A second example is the Zilog Z80 series of OMPI - 12 - microprocessor described in the books "Z80 Microprocessor Programming and Interfacing", books 1 and 2 by E. A. Nichols, J. C. Nichols and P. R. Rony and the comparion book "Microcomputer-Analog Converter Software and Hardware Interfacing" by J. A. Titus, C. A. Titus, P. R. Rony, D. G. Larsen, all published by Howard . Sams & Co., Inc., Indianapolis, IN. The general principles used are similar to those described in "Microcomputers/Microprocessors: Hardware, Software and Applications". J. R. Hilborn, P. M. Tulich, published by Prentice-Hall Inc., Englewood Clifts, NJ. The program to implement steps 1 through 5 given above can be in any form suitable for the microprocessor used. Alternatives to the use of the microprocessor 18 described above would be a system of hard wired logic or a method of analog measurement with suitable timing means and a recorder, both of which could be made to achieve the same results by implementing the program Steps 1 through 5 in the previous description. The new instrument has the primary advantage that it can measure with a single set-up, requiring no special operator skill, the quantity which gives a true indication of insulation quality. At present the same information can only be obtained by a complex and time consuming series of measurements. The new instrument also provides a means of measuring loss due to partial discharge and establishing partial discharge inception. These advantages can be obtained with an instrument which is smaller, lighter and costs less than existing techniques. ^^T RE

Claims: - 13 - ' I CLAIM : 1. Insulation analyzer apparatus comprising: a high voltage direct current power supply, current sensing means, specimen terminals for connecting an insulation specimen across said power supply in series with the current sensing means, a voltage comparator connected across the specimen terminals to sense effective voltage across the specimen, timing means, computation means for receiving inputs from the timing means, current sensing means and voltage comparator and calculating insulation current' using a formula involving elapsed time measurements of total current through the insulation specimen to calculate insulation current and the insulation resistance of the insulation specimens; and means to adjust the high voltage direct current power supply and display means permitting insulation current and insulation resistance readings. 2. The insulation analyzer apparatus of claim 1 in which the computation means is a programmable computer. 3. The insulation analyzer apparatus of claim 1 in which the computation means is logic circuitry. 4. The insulation analyzer apparatus of claim 1 in which E is held constant in the high voltage direct current power supply. - 14 - 5. The insulation analyzer -apparatus of claim 1 in which the output current of the high voltage direct current power supply is held constant. 6. The insulation analyzer apparatus of claim 1 in which the timing means is set for a predetermined period, and the computation means has preestablished criteria for adequacy of the timing period results, whereby if the initial values of CR is too short for convenience in use, the scale is automatically changed by varying the value of R. 7. The insulation analyzer apparatus of claim 1 in which the timing -means is set for a predetermined period, and the computation means has preestablished criteria for adequacy of the timing period results, whereby if the initial values of CR is too long for convenience in use, the scale is automatically changed by varying the value of R. 8. The method of detection of the inception of partial discharge comprising: increasing the magnitude of E; calculating R IR at each value of E; and observing when partial discharge inception occurs by a decrease in the value of R IR s a function of voltage. 9. A method of analyzing an insulation specimen comprising: removing any absorbed charge from the specimen; applying test voltage E through system resistance R; sensing Iιo at time intervals occuring at selected times multiplied by any three points in a power series, while maintaining E constant; and calculating and displaying insulation resistance R jR from formula: E R IR = — ~ R where 10. The method of claim 9 in which the three points in a power series are selected in the ratio of 1, 3.16 and 10. 11. A method of analyzing an insulation specimen comprising: removing any absorbed charge from the specimen; applying test voltage E through system resistance R; timing an interval from one specified voltage level to another across the specimen and using formula t = C x R (where t is time in seconds, C is calculated) ; displaying values of C; starting timing and counting display from nCR to zero; maintaining E constant, while measuring i< j >oτ at each of the timing intervals at nCR, O.lnCR and 0.316CR; and calculating and displaying resistance R jR from formula: where 12. The method of claim 11 in which n = 100. 13. The method of claim 12 in which 100CR is also displayed and incremented by -ICR to zero to indicate the timing of the test. 1 . T e method of claim 9 in which the calculation of C is modified to take into consideration the value of R jR - 15. The method of claim 9 in which a number of alternative measurements are made and the results averaged to reduce the effect of electrical noise.

Inventors: REYNOLDS PETER H, REYNOLDS, PETER, H.
Applicants: BIDDLE CO JAMES G, JAMES G. BIDDLE COMPANY
Application year: 1981
Main classification: G01R31
IPC codes: G01R27, G01R31