memory of a microcontroller and transmitted at a lower rate directly after the measurement. Transmitting signals at a rate that is ten times lower would allow sampling of 1 signal of at least 500 Hz or of several signals at a lower rate. As described, the GMR sensor has to be adjusted carefully, and its signal is filtered not to measure the 4-kHz powering field but to measure only the magnetic pulses produced by the implant. Currently the signal-receiving circuit is changed such that the signal will only be received at time points when the powering field is close to the zero crossing, which will ensure that the quality of References 1. Streicher RM, Semlitsch M, Schon R, Weber H, Rieker C (1996) Metal-on- metal articulation for artificial hip joints: laboratory study and clinical results. Proc Inst Mech Eng H 210: 223–232. 2. Drewniak EI, Jay GD, Fleming BC, Crisco JJ (2009) Comparison of two methods for calculating the frictional properties of articular cartilage using a simple pendulum and intact mouse knee joints. J Biomech 42: 1996–1999. 3. Bergmann G, Graichen F, Rohlmann A, Verdonschot N, van Lenthe GH (2001) Frictional heating of total hip implants. Part 1: measurements in patients. J Biomech 34: 421–428. 4. Fialho JC, Fernandes PR, Eca L, Folgado J (2007) Computational hip joint simulator for wear and heat generation. J Biomech 40: 2358–2366. 5. Brockett C, Williams S, Jin Z, Isaac G, Fisher J (2007) Friction of total hip replacements with different bearings and loading conditions. J Biomed Mater Res B Appl Biomater 81: 508–515. 6. Moss SG, Schweitzer ME, Jacobson JA, Brossmann J, Lombardi JV, et al. (1998) Hip joint fluid: detection and distribution at MR imaging and US with cadaveric correlation. Radiology 208: 43–48. 7. Liao YS, Benya PD, McKellop HA (1999) Effect of protein lubrication on the wear properties of materials for prosthetic joints. J Biomed Mater Res 48: 465– 473. 8. Rainer F, Ribitsch V, Ulreich A (1980) Viscosity of synovial fluid and possible artificial lubricants (author’s transl). Acta Med Austriaca 7: 92–95. 9. Fam H, Bryant JT, Kontopoulou M (2007) Rheological properties of synovial fluids. Biorheology 44: 59–74. 10. Chikama H (1985) [The role of protein and hyaluronic acid in the synovial fluid in animal joint lubrication]. Nippon Seikeigeka Gakkai Zasshi 59: 559–572. 11. Swann DA, Bloch KJ, Swindell D, Shore E (1984) The lubricating activity of human synovial fluids. Arthritis Rheum 27: 552–556. 12. Williams S, Leslie I, Isaac G, Jin Z, Ingham E, et al. (2008) Tribology and wear of metal-on-metal hip prostheses: influence of cup angle and head position. J Bone Joint Surg Am 90 Suppl 3: 111–117. 13. Smith SL, Unsworth A (2001) A five-station hip joint simulator. Proc Inst Mech Eng H 215: 61–64. 14. Tepic S, Macirowski T, Mann RW (1985) Experimental temperature rise in human hip joint in vitro in simulated walking. Journal of Orthopaedic Research 3: 516–520. 15. Pritchett JW (2006) Heat generated by knee prostheses. Clin Orthop Relat Res 442: 195–198. 16. Rocchi M, Affatato S, Falasca G, Viceconti M (2007) Thermomechanical analysis of ultra-high molecular weight polyethylene-metal hip prostheses. Proc Inst Mech Eng H 221: 561–568. 17. Bergmann G, Graichen F, Rohlmann A, Verdonschot N, van Lenthe GH (2001) Frictional heating of total hip implants. Part 2: finite element study. J Biomech 34: 429–435. 18. Lu Z, McKellop H (1997) Frictional heating of bearing materials tested in a hip joint wear simulator. Proc Inst Mech Eng H 211: 101–108. 19. Leon SA, Asbell SO, Arastu HH, Edelstein G, Packel AJ, et al. (1993) Effects of hyperthermia on bone. II. Heating of bone in vivo and stimulation of bone growth. Int J Hyperthermia 9: 77–87. 20. Moritz AR, Henriques FC (1947) Studies of Thermal Injury: II. The Relative Importance of Time and Surface Temperature in the Causation of Cutaneous Burns. Am J Pathol 23: 695–720. Hip Implant for Temperature Measurements the signals is significantly less dependent on the exact adjustment of the power coil and the GMR sensor. Author Contributions Conceived and designed the experiments: GND GB FG AR. Performed the experiments: FG PD JD GB GND AR. Analyzed the data: FG PD JD GB GND AR. Contributed reagents/materials/analysis tools: GND AR GB JD. Wrote the paper: GB FG PD JD AR GND. 21. Li S, Chien S, Branemark PI (1999) Heat shock-induced necrosis and apoptosis in osteoblasts. J Orthop Res 17: 891–899. 22. Yoshida K, Uoshima K, Oda K, Maeda T (2009) Influence of heat stress to matrix on bone formation. Clin Oral Implants Res 20: 782–790. 23. Gronkiewicz K, Majewski P, Wisniewska G, Pihut M, Loster BW, et al. (2009) Experimental research on the possibilities of maintaining thermal conditions within the limits of the physiological conditions during intraoral preparation of dental implants. J Physiol Pharmacol 60 Suppl 8: 123–127. 24. Schneider E, Michel MC, Genge M, Zuber K, Ganz R, et al. (2001) Loads acting in an intramedullary nail during fracture healing in the human femur. J Biomech 34: 849–857. 25. Holland HJ, Gra¨tz H, Braunschweig M, Kuntz M (2012) INHUEPRO: sensorsystem for use in implants. Proc MikroSytemTechnik Kongress 2011 Darmstadt, VDE-Verlag: 437–440. 26. Taylor SJ, Perry JS, Meswania JM, Donaldson N, Walker PS, et al. (1997) Telemetry of forces from proximal femoral replacements and relevance to fixation. J Biomech 30: 225–234. 27. Puers R, Catrysse M, Vandevoorde G, Collier RJ, Louridas E, et al. (2000) A telemetry system for the detection of hip prosthesis loosening by vibration analysis. Sensors and Actuators A 85: 42–47. 28. Bergmann G, Graichen F, Siraky J, Jendrzynski H, Rohlmann A (1988) Multichannel strain gauge telemetry for orthopaedic implants. J Biomech 21: 169–176. 29. Graichen F, Arnold R, Rohlmann A, Bergmann G (2007) Implantable 9- channel telemetry system for in vivo load measurements with orthopedic implants. IEEE Trans Biomed Eng 54: 253–261. 30. Graichen F, Bergmann G (1991) Four-channel telemetry system for in vivo measurement of hip joint forces. J Biomed Eng 13: 370–374. 31. Graichen F, Bergmann G, Rohlmann A (1999) Hip endoprosthesis for in vivo measurement of joint force and temperature. J Biomech 32: 1113–1117. 32. Moss C, Weinrich N, Sass W, Mueller J (2010) Integration of a telemetric system within an intramedullary nail for monitoring of the fracture healing progress. Proc 3rd Int Symp on Applied Sciences in Biomedical and Communication Technologies: 1–5. 33. Marschner U, Gra¨tz H, Jettkan B, Ruwisch D, Woldt G, et al. (2009) Integration of a wireless lock-in measurement of hip prosthesis vibrations for loosening detection. Sensors and Actuators A 156 145–154. 34. Moss Ch, Sass W, Weinrich N, Seide K, Mu¨ller J (2009) Low Frequency RFID – Strain Measurement on Passive Implants. Proc SOMSED ‘09 1: 97–100. 35. Brown RH, Burstein AH, Frankel VH (1982) Telemetering in vivo loads from nail plate implants. J Biomech 15: 815–823. 36. Karrholm J (2010) The Swedish Hip Arthroplasty Register (www.shpr.se). Acta Orthop 81: 3–4. 37. ISO7206–4 (2010) Implants for surgery – Partial and total hip joint prostheses – Part 4: Determination of endurance properties and performance of stemmed femoral components. 38. ISO7206–6 (1992) Implants for surgery – Partial and total hip joint prostheses – Part 6: Determination of endurance properties of head and neck region of stemmed femoral components. 39. ISO7206–8 (1995) Implants for surgery – Partial and total hip joint prostheses – Part 8: Endurance performance of stemmed femoral components with application of torsion. PLOS ONE | www.plosone.org 7 August 2012 | Volume 7 | Issue 8 | e43489