Document headings vary by document type but may contain
                    the following: the agency or agencies that issued and signed a document
the number of the CFR title and the number of each part
                      the document amends, proposes to amend, or is directly
                      related to
the agency docket number / agency internal file number
the RIN which identifies each regulatory action listed in
                      the Unified Agenda of Federal Regulatory and Deregulatory
                      Actions See the
                    Document Drafting Handbook
                    for more details.
 Department of Health and Human Services Food and Drug Administration 21 CFR Part 866
[Docket No. FDA-2017-N-5719]
 AGENCY: Food and Drug Administration, HHS.
 ACTION: Final order.
 SUMMARY: The Food and Drug Administration (FDA or we) is classifying the 
                        *Streptococcus spp.*
                         nucleic acid-based assay into class II (special controls). The special controls that apply to the device type are identified in this order and will be part of the codified language for the 
                        *Streptococcus spp.*
                         nucleic acid-based assay's classification. We are taking this action because we have determined that classifying the device into class II (special controls) will provide a reasonable assurance of safety and effectiveness of the device. We believe this action will also enhance patients' access to beneficial innovative devices, in part by reducing regulatory burdens.
 DATES: This order is effective October 30, 2017. The classification was applicable on April 16, 2014.
 FOR FURTHER INFORMATION CONTACT: Steven Tjoe, Center for Devices and Radiological Health, Food and Drug Administration, 10903 New Hampshire Ave., Bldg. 66, Rm. 4550, Silver Spring, MD 20993-0002, 301-796-5866, 
                        *steven.tjoe@fda.hhs.gov.*


SUPPLEMENTARY INFORMATION:  I. Background Upon request, FDA has classified the 
                    *Streptococcus spp.*
                     nucleic acid-based assay as class II (special controls), which we have determined will provide a reasonable assurance of safety and effectiveness. In addition, we believe this action will enhance patients' access to beneficial innovation, in part by reducing regulatory burdens by placing the device into a lower device class than the automatic class III assignment. The automatic assignment of class III occurs by operation of law and without any action by FDA, regardless of the level of risk posed by the new device. Any device that was not in commercial distribution before May 28, 1976, is automatically classified as, and remains within, class III and requires premarket approval unless and until FDA takes an action to classify or reclassify the device (see 21 U.S.C. 360c(f)(1)). We refer to these devices as “postamendments devices” because they were not in commercial distribution prior to the date of enactment of the Medical Device Amendments of 1976, which amended the Federal Food, Drug, and Cosmetic Act (FD&C Act). FDA may take a variety of actions in appropriate circumstances to classify or reclassify a device into class I or II. We may issue an order finding a new device to be substantially equivalent under section 513(i) of the FD&C Act to a predicate device that does not require premarket approval (see 21 U.S.C. 360c(i)). We determine whether a new device is substantially equivalent to a predicate by means of the procedures for premarket notification under section 510(k) of the FD&C Act and part 807 (21 U.S.C. 360(k) and 21 CFR part 807, respectively). FDA may also classify a device through “De Novo” classification, a common name for the process authorized under section 513(f)(2) of the FD&C Act (21 U.S.C. 360c(f)(2)). Section 207 of the Food and Drug Administration Modernization Act of 1997 established the first procedure for De Novo classification (Pub. L. 105-115). Section 607 of the Food and Drug Administration Safety and Innovation Act modified the De Novo application process by adding a second procedure (Pub. L. 112-144). A device sponsor may utilize either procedure for De Novo classification. Under the first procedure, the person submits a 510(k) for a device that has not previously been classified. After receiving an order from FDA classifying the device into class III under section 513(f)(1) of the FD&C Act, the person then requests a classification under section 513(f)(2). Under the second procedure, rather than first submitting a 510(k) and then a request for classification, if the person determines that there is no legally marketed device upon which to base a determination of substantial equivalence, that person requests a classification under section 513(f)(2) of the FD&C Act. Under either procedure for De Novo classification, FDA is required to classify the device by written order within 120 days. The classification will be according to the criteria under section 513(a)(1) of the FD&C Act (21 U.S.C. 360c(a)(1)). Although the device was automatically within class III, the De Novo classification is considered to be the initial classification of the device. We believe this De Novo classification will enhance patients' access to beneficial innovation, in part by reducing regulatory burdens. When FDA classifies a device into class I or II via the De Novo process, the device can serve as a predicate for future devices of that type, including for 510(k)s (see 21 U.S.C. 360c(f)(2)(B)(i)). As a result, other device sponsors do not have to submit a De Novo request or PMA in order to market a substantially equivalent device (see 21 U.S.C. 360c(i), defining “substantial equivalence”). Instead, sponsors can use the less-burdensome 510(k) process, when necessary, to market their device. II. De Novo Classification For this device, FDA issued an order on March 20, 2014, finding the Lyra Direct Strep Assay not substantially equivalent to a predicate not subject to a premarket application approval (PMA). Thus, the device remained in class III in accordance with section 513(f)(1) of the FD&C Act when we issued the order. On March 28, 2014, Quidel Corp. submitted a request for De Novo classification of the Lyra Direct Strep Assay. FDA reviewed the request in order to classify the device under the criteria for classification set forth in section 513(a)(1) of the FD&C Act. We classify devices into class II if general controls by themselves are insufficient to provide reasonable assurance of safety and effectiveness, but there is sufficient information to establish special controls that, in combination with the general controls, provide reasonable assurance of the safety and effectiveness of the device for its intended use (see 21 U.S.C. 360c(a)(1)(B)). After review of the information submitted in the request, we determined that the device can be classified into class II with the establishment of special controls. FDA has determined that these special controls, in addition to general controls, will provide reasonable assurance of the safety and effectiveness of the device. Therefore, on April 16, 2014, FDA issued an order to the requestor classifying the device into class II. FDA is codifying the classification of the device by adding 21 CFR 866.2680. We have named the generic type of device 
                    *Streptococcus spp.*
                     nucleic acid-based assay, and it is identified as a qualitative in vitro diagnostic device that is intended to simultaneously detect and 
                    ( print page 50074)
                    identify various 
                    *Streptococcus spp.*
                     nucleic acids extracted directly from clinical specimens. The device detects specific nucleic acid sequences for organism identification. The identification aids in the diagnosis of diseases caused by bacteria belonging to the genus 
                    *Streptococcus*
                     and provides epidemiological information on these diseases. Pathogenic streptococci are associated with infections, such as sore throat, impetigo (an infection characterized by small pustules on the skin), urinary tract infections, rheumatic fever, and kidney disease. FDA has identified the following risks to health associated specifically with this type of device and the measures required to mitigate these risks in table 1. Table 1—Streptococcus SPP. Nucleic Acid-Based Assay Risks and Mitigation Measures
Identified risks
Mitigation measuresIncorrect identification of a pathogenic microorganism by the device can lead to improper patient management
Special controls (1), (2), (3), (4), (5) and (6) (21 CFR 866.2680(b)(1); 21 CFR 866.2680(b)(2); 21 CFR 866.2680(b)(3); 21 CFR 866.2680(b)(4); 21 CFR 866.2680(b)(5); and 21 CFR 866.2680(b)(6)).Failure to correctly interpret test results
Special control (7) (21 CFR 866.2680(b)(7)).Failure to correctly operate the instrument
Special control (8) (21 CFR 866.2680(b)(8)).
 FDA has determined that special controls, in combination with the general controls, address these risks to health and provide reasonable assurance of safety and effectiveness. In order for a device to fall within this classification, and thus avoid automatic classification in class III, it would have to comply with the special controls named in this final order. The necessary special controls appear in the regulation codified by this order. This device is subject to premarket notification requirements under section 510(k). III. Analysis of Environmental Impact The Agency has determined under 21 CFR 25.34(b) that this action is of a type that does not individually or cumulatively have a significant effect on the human environment. Therefore, neither an environmental assessment nor an environmental impact statement is required. IV. Paperwork Reduction Act of 1995 This final order establishes special controls that refer to previously approved collections of information found in other FDA regulations. These collections of information are subject to review by the Office of Management and Budget (OMB) under the Paperwork Reduction Act of 1995 (44 U.S.C. 3501-3520). The collections of information in part 807, subpart E, regarding premarket notification submissions have been approved under OMB control number 0910-0120, and the collections of information in 21 CFR parts 801 and 809, regarding labeling have been approved under OMB control number 0910-0485. List of Subjects in 21 CFR Part 866 Biologics
Laboratories
Medical devices
 Therefore, under the Federal Food, Drug, and Cosmetic Act and under authority delegated to the Commissioner of Food and Drugs, 21 CFR part 866 is amended as follows: PART 866—IMMUNOLOGY AND MICROBIOLOGY DEVICES
 1.The authority citation for part 866 continues to read as follows: Authority:
                        21 U.S.C. 351, 360, 360c, 360e, 360j, 360
                            *l,*
                             371. 2.Add § 866.2680 to subpart C to read as follows: § 866.2680
  (a) 
                            *Identification.*
                             A 
                            *Streptococcus spp.*
                             nucleic acid-based assay is a qualitative in vitro diagnostic device intended to simultaneously detect and identify various 
                            *Streptococcus spp.*
                             nucleic acids extracted directly from clinical specimens. The device detects specific nucleic acid sequences for organism identification. The identification aids in the diagnosis of diseases caused by bacteria belonging to the genus 
                            *Streptococcus*
                             and provides epidemiological information on these diseases. Pathogenic streptococci are associated with infections, such as sore throat, impetigo (an infection characterized by small pustules on the skin), urinary tract infections, rheumatic fever, and kidney disease. (b) 
                            *Classification.*
                             Class II (special controls). The special controls for this device are: (1) Premarket notification submissions must include detailed device description documentation, including the device components, ancillary reagents required but not provided, and a detailed explanation of the methodology including primer/probe sequence, design, and rationale for sequence selection. (2) Premarket notification submissions must include detailed documentation from the following analytical and clinical performance studies: Analytical sensitivity (Limit of Detection), reactivity, inclusivity, precision, reproducibility, interference, cross reactivity, carry-over, and cross contamination. (3) Premarket notification submissions must include detailed documentation from a clinical study. The study, performed on a study population consistent with the intended use population, must compare the device performance to results obtained from well-accepted reference methods. (4) Premarket notification submissions must include detailed documentation for device software, including, but not limited to, software applications and hardware-based devices that incorporate software. (5) Premarket notification submissions must include database implementation methodology, construction parameters, and quality assurance protocols, as appropriate. (6) The device labeling must include limitations regarding the need for culture confirmation of negative specimens, as appropriate. (7) A detailed explanation of the interpretation of results and acceptance criteria must be included in the device's 21 CFR 809.10(b)(9) compliant labeling. (8) Premarket notification submissions must include details on an end user device training program that will be offered while marketing the device, as appropriate.
 Dated: October 25, 2017. Lauren Silvis, Chief of Staff.


[FR Doc. 2017-23513 Filed 10-27-17; 8:45 am] BILLING CODE 4164-01-P