Skip to main content    

An official website of the United States government

Here’s how you know

**Official websites use .gov**              A **.gov** website belongs to an official government organization in the United States.            

**Secure .gov websites use HTTPS**              A **lock** (  LockA locked padlock) or **https://** means you’ve safely connected to the .gov website. Share sensitive information only on official, secure websites.            

https://www.nist.gov/pml/quantum-measurement/atomic-spectroscopy/electron-beam-ion-trap-ebit/trapped-ion-dynamics

            -  Topics  -  All Topics 
-  Advanced communications 
-  Artificial intelligence 
-  Bioscience 
-  Buildings and construction 
-  Chemistry 
-  Climate 
-  Cybersecurity 
-  Electronics 
-  Energy 
-  Environment 
-  Fire 
-  Forensic science 
-  Health 
-  Information technology 
-  Infrastructure 
-  Manufacturing 
-  Materials 
-  Mathematics and statistics 
-  Metrology 
-  Nanotechnology 
-  Neutron research 
-  Performance excellence 
-  Physics 
-  Public safety 
-  Resilience 
-  Standards 
-  Transportation 
 
            -  Publications 
            -  Labs & Major Programs  -  Laboratories    -  Communications Technology Laboratory 
  -  Engineering Laboratory 
  -  Information Technology Laboratory 
  -  Material Measurement Laboratory 
  -  Physical Measurement Laboratory 
-  User Facilities    -  NIST Center for Neutron Research 
  -  CNST NanoFab 
-  Research Test Beds 
-  Research Projects 
-  Tools & Instruments 
-  Major Programs    -  Baldrige Performance Excellence Program 
  -  CHIPS for America Initiative 
  -  Manufacturing Extension Partnership (MEP) 
  -  Office of Advanced Manufacturing 
  -  Special Programs Office 
  -  Technology Partnerships Office 
 
            -  Services & Resources  -  Standards and Measurements    -  Calibration Services 
  -  Laboratory Accreditation (NVLAP) 
  -  Quality System 
  -  Standard Reference Materials (SRMs) 
  -  Standards.gov 
  -  Time Services 
  -  Office of Weights and Measures 
-  Software 
-  Data    -  Chemistry WebBook 
  -  National Vulnerability Database 
  -  Physical Reference Data 
  -  Standard Reference Data (SRD) 
-  Storefront 
-  License & Patents 
-  Computer Security Resource Center (CSRC) 
-  NIST Research Library 
 
            -  News & Events  -  News 
-  Events 
-  Blogs 
-  Feature Stories 
-  Awards 
-  Video Gallery 
-  Image Gallery 
-  Media Contacts 
 
            -  About NIST  -  About Us    -  Leadership 
  -  Organization Structure 
  -  Budget & Planning 
-  Contact Us 
-  Visit 
-  Careers    -  Student programs 
-  Work with NIST 
-  History    -  NIST Digital Archives 
  -  NIST Museum 
  -  NIST and the Nobel 
-  Educational Resources 
 

Physical Measurement LaboratoryQuantum Measurement Division

## Atomic Spectroscopy Group

-  Electron Beam Ion Trap (EBIT)  Expand or Collapse    -  EBIT Introduction 
  -  EBIT Operating Principles 
  -  EBIT History  Expand or Collapse      -  EBIT Milestones 
    -  The History of NRL's Involvement in NIST's EBIT Facility 
  -  Other EBITs Around the Globe 
  -  Typical Operating Parameters of the NIST EBIT 
  -  EBIT Design Tutorial 
  -  Guide to Reported NIST EBIT Data 
  -  Current and Future EBIT Research 
  -  X-ray Polarization 
  -  Visible and UV Spectroscopy 
  -  Electron-Ion Collisions  Expand or Collapse      -  Radiative and Dielectronic Recombination 
  -  Trapped Ion Dynamics  Expand or Collapse      -  Charge state simulation 
  -  Atomic Lifetimes 
  -  Ion-Surface Interactions 
  -  EBIT Bibliography 
  -  People Using the EBIT Facility  Expand or Collapse      -  2004 EBIT Hike 
    -  EBIT Group Photo Montage 
    -  EBIT Second Group Photo Montage 
  -  Fellowships for Research at the NIST EBIT 

# Trapped Ion Dynamics

## Share

Facebook

Linkedin

Twitter

Email

#### Ion Orbits

In order to have a better understanding of the motion of the highly charged ions inside the trap a computer simulation was developed by Eric Meyer. The program accounts for the presence of the magnetic field of the superconducting coils, the electron beam, and the neutral background gas atoms. It can also simulate the effect of changing electron beam currents which has importance in time resolved (e.g., lifetime) measurements. Here are a few examples of different orbits.

#### Charge state simulation

The charge state balance (e.g., the population of different charge states) inside the EBIT is determined by the balance between the different ionization and recombination processes. In order to predict the charge states of the ions inside the trap and to help to determine the EBIT parameters for the optimalization for the production of a certain charge state these processes should be taken into account. To this end a computer program which was originally developed by Penetrante in Livermore and modified by Margolis at Oxford was installed and further modified. The predictions of the simulation can be checked by analyzing the x-rays emitted from the trap. An alternative way to look into the same problem is the extraction and charge to mass ratio analysis of the highly charged ions. Experimental efforts in this direction are under way.

### Atomic Lifetimes

The measurement of excited-state lifetimes is complementary to measuring transition wavelengths as a way of studying atomic structure. Although the lifetimes are determined by the same wavefunctions as the energy levels the measurements of the atomic decays carry different information since they are sensitive to the long-range behavior of the wavefunctions. The knowledge of the lifetimes also has important practical applications. They are critical in the density diagnostics of laboratory and astrophysical plasmas.

The principle of measuring lifetimes with an EBIT lies in the periodic fast switching of different voltages in the machine. Since the ions are created and excited with the same beam of electrons, by changing the electron beam energy one can selectively exclude certain levels from being excited. This can simply be done by setting the electron beam energy below the excitation threshold of the level to be excluded. Without further excitation the time dependence of the emitted photon signal carries the information about the lifetime of the level. After a certain period of time (determined by the lifetime of the level) the electron beam energy is set to be above the excitation threshold to repopulate the level and repeat the sequence. An alternative method for measuring lifetimes with an EBIT is to switch off the electron beam completely, take data, and turn the beam back again to re-excite the ions in the trap. While the electron beam is off, the ions remain trapped by the magnetic field. The lifetime range that can be measured with an EBIT is determined by the capabilities for the fast switching of voltages. In principle the 10 ns to 10 ms lifetime range can be addressed by this method. Since this lifetime range is only partially covered by other methods the EBIT is a unique tool for measuring the lifetime of long living metastable levels.

In a recent experiment we have measured the lifetime of a visible light emitting metastable level **[18]**. The transition takes places within the ground state configuration of titanium-like ions. The measured lifetimes fall into the millisecond range.

### Ion-Surface Interactions

The EBIT was originally developed for *in situ* spectroscopic measurements of highly charged ions. It soon turned out, however, that the machine is easily capable of producing the highest charge states not accessible even for sophisticated ion sources. To use the EBIT as a source for ions an extraction system has to be built which removes the ions from the trap region and transports them outside the machine. The construction of the extraction system on the NIST EBIT was completed in the first half of 1995 **[12,20]**. When the ions are extracted from the EBIT they pass several ion beam steering and shaping ion-optical elements. A bending magnet is available as a part of the system to charge-to-mass select the ions and direct them into experimental chambers. Recent experiments with the extracted ions involve the study of nanometer scale damages of insulator surfaces caused by the impact of highly charged ions. X-ray **[16]** and Electron spectroscopic investigations **[15]** of metal-surface -- highly-charged-ion interactions are under way.

Both the atomic spectroscopy and ion-surface interactions component of our work is proceeding in connection with International SEMATECH.

PhysicsAtomic / molecular / quantum

-  **‹** Radiative and Dielectronic Recombination 
-  Charge state simulation **›** 

### HEADQUARTERS

301-975-2000

Webmaster | Contact Us | Our Other Offices

TwitterFacebookLinkedInInstagramYouTubeGiphyRSS FeedMailing List

Feedback

            -  Site Privacy 
            -  Accessibility 
            -  Privacy Program 
            -  Copyrights 
            -  Vulnerability Disclosure 
            -  No Fear Act Policy 
            -  FOIA 
            -  Environmental Policy 
            -  Scientific Integrity 
            -  Information Quality Standards 
            -  Commerce.gov 
            -  Science.gov 
            -  USA.gov 
            -  Vote.gov