Skip to main content    

An official website of the United States government

Here’s how you know

**Official websites use .gov**              A **.gov** website belongs to an official government organization in the United States.            

**Secure .gov websites use HTTPS**              A **lock** (  LockA locked padlock) or **https://** means you’ve safely connected to the .gov website. Share sensitive information only on official, secure websites.            

https://www.nist.gov/pml/sensor-science/fluid-metrology/long-wavelength-acoustic-flowmeter

            -  Topics  -  All Topics 
-  Advanced communications 
-  Artificial intelligence 
-  Bioscience 
-  Buildings and construction 
-  Chemistry 
-  Climate 
-  Cybersecurity 
-  Electronics 
-  Energy 
-  Environment 
-  Fire 
-  Forensic science 
-  Health 
-  Information technology 
-  Infrastructure 
-  Manufacturing 
-  Materials 
-  Mathematics and statistics 
-  Metrology 
-  Nanotechnology 
-  Neutron research 
-  Performance excellence 
-  Physics 
-  Public safety 
-  Resilience 
-  Standards 
-  Transportation 
 
            -  Publications 
            -  Labs & Major Programs  -  Laboratories    -  Communications Technology Laboratory 
  -  Engineering Laboratory 
  -  Information Technology Laboratory 
  -  Material Measurement Laboratory 
  -  Physical Measurement Laboratory 
-  User Facilities    -  NIST Center for Neutron Research 
  -  CNST NanoFab 
-  Research Test Beds 
-  Research Projects 
-  Tools & Instruments 
-  Major Programs    -  Baldrige Performance Excellence Program 
  -  CHIPS for America Initiative 
  -  Manufacturing Extension Partnership (MEP) 
  -  Office of Advanced Manufacturing 
  -  Special Programs Office 
  -  Technology Partnerships Office 
 
            -  Services & Resources  -  Standards and Measurements    -  Calibration Services 
  -  Laboratory Accreditation (NVLAP) 
  -  Quality System 
  -  Standard Reference Materials (SRMs) 
  -  Standards.gov 
  -  Time Services 
  -  Office of Weights and Measures 
-  Software 
-  Data    -  Chemistry WebBook 
  -  National Vulnerability Database 
  -  Physical Reference Data 
  -  Standard Reference Data (SRD) 
-  Storefront 
-  License & Patents 
-  Computer Security Resource Center (CSRC) 
-  NIST Research Library 
 
            -  News & Events  -  News 
-  Events 
-  Blogs 
-  Feature Stories 
-  Awards 
-  Video Gallery 
-  Image Gallery 
-  Media Contacts 
 
            -  About NIST  -  About Us    -  Leadership 
  -  Organization Structure 
  -  Budget & Planning 
-  Contact Us 
-  Visit 
-  Careers    -  Student programs 
-  Work with NIST 
-  History    -  NIST Digital Archives 
  -  NIST Museum 
  -  NIST and the Nobel 
-  Educational Resources 
 

Physical Measurement LaboratorySensor Science Division

## Fluid Metrology Group

-  Laboratory Homepage 
-  Division Homepage 
-  Groups  Expand or Collapse    -  Dimensional Metrology 
  -  Fluid Metrology 
  -  Optical Radiation 
  -  Remote Sensing 
  -  Surface and Interface Metrology 
  -  Thermodynamic Metrology 
-  Collaboration Opportunities 
-  Employment and Internship Opportunities  Expand or Collapse    -  Employment at NIST 
  -  NIST Professional Research Experience Program (PREP) 
  -  NIST/NRC Postdoctoral Research Associateships Program 
-  Services and Standards 
-  Staff Directory 

# Long-wavelength acoustic flowmeter

## Share

Facebook

Linkedin

Twitter

Email

Figure 1. 1:100 scale (10 cm OD pipe) LWAF facility. The fan forces air through a calibrated flow standard (0.2 % uncertainty with a 95 % confidence level), then through a swirl-generating turn-around section, and finally through the LWAF.

Conventional gas flow measurements conducted in large ducts (such as the smoke stack of a coal-burning power plant) have uncertainties of 5 % to 10 %. Consequently, the quantity of pollutants (sulfur dioxide, mercury, nitrogen oxides, and carbon dioxide) emitted by such ducts have equally large uncertainties. As part of NIST's Greenhouse Gas Measurements Program, we are testing long-wavelength acoustic flow meters (LWAFs) to reduce this uncertainty. LWAFs measure the average volume flow of flue gases in smoke stacks. The measured volume flow can be combined with measurements of the flue gas's pressure, temperature, and composition to determine the mass of pollutants emitted by the stack.

To test LWAFs, we constructed the calibrated, 1:100 scale model, test facility shown in Fig. 1. The model LWAF determined the speed of sound in ambient air with a standard uncertainty of 0.05 %. Within this uncertainty, the speed of sound agrees with the value calculated from NIST's REFPROP database.

The same LWAF determined the average flow velocity. It agreed, within ±1 %, with the velocity determined from a NIST-calibrated flow standard upstream from the LWAF. This good agreement was maintained for flows up to 24 m/s, even after a swirl-inducing turn-around section was inserted between the flow standard and the LWAF. (See Fig. 3.) Similar uncertainties were obtained with highly distorted flows generated by placing obstructions upstream of the LWAF.

To verify the scalability of the method, we will test LWAFs in NIST's 1:10 scale smoke stack simulator.

EnvironmentFlow metrology and rheology

## Contacts

                -  Keith A Gillis  keith.gillis@nist.gov      (301) 975-2468   

### HEADQUARTERS

301-975-2000

Webmaster | Contact Us | Our Other Offices

TwitterFacebookLinkedInInstagramYouTubeGiphyRSS FeedMailing List

Feedback

            -  Site Privacy 
            -  Accessibility 
            -  Privacy Program 
            -  Copyrights 
            -  Vulnerability Disclosure 
            -  No Fear Act Policy 
            -  FOIA 
            -  Environmental Policy 
            -  Scientific Integrity 
            -  Information Quality Standards 
            -  Commerce.gov 
            -  Science.gov 
            -  USA.gov 
            -  Vote.gov