Skip to main content    

An official website of the United States government

Here’s how you know

**Official websites use .gov**              A **.gov** website belongs to an official government organization in the United States.            

**Secure .gov websites use HTTPS**              A **lock** (  LockA locked padlock) or **https://** means you’ve safely connected to the .gov website. Share sensitive information only on official, secure websites.            

https://www.nist.gov/laboratories/tools-instruments/low-pressure-gauges

            -  Topics  -  All Topics 
-  Advanced communications 
-  Artificial intelligence 
-  Bioscience 
-  Buildings and construction 
-  Chemistry 
-  Climate 
-  Cybersecurity 
-  Electronics 
-  Energy 
-  Environment 
-  Fire 
-  Forensic science 
-  Health 
-  Information technology 
-  Infrastructure 
-  Manufacturing 
-  Materials 
-  Mathematics and statistics 
-  Metrology 
-  Nanotechnology 
-  Neutron research 
-  Performance excellence 
-  Physics 
-  Public safety 
-  Resilience 
-  Standards 
-  Transportation 
 
            -  Publications 
            -  Labs & Major Programs  -  Laboratories    -  Communications Technology Laboratory 
  -  Engineering Laboratory 
  -  Information Technology Laboratory 
  -  Material Measurement Laboratory 
  -  Physical Measurement Laboratory 
-  User Facilities    -  NIST Center for Neutron Research 
  -  CNST NanoFab 
-  Research Test Beds 
-  Research Projects 
-  Tools & Instruments 
-  Major Programs    -  Baldrige Performance Excellence Program 
  -  CHIPS for America Initiative 
  -  Manufacturing Extension Partnership (MEP) 
  -  Office of Advanced Manufacturing 
  -  Special Programs Office 
  -  Technology Partnerships Office 
 
            -  Services & Resources  -  Standards and Measurements    -  Calibration Services 
  -  Laboratory Accreditation (NVLAP) 
  -  Quality System 
  -  Standard Reference Materials (SRMs) 
  -  Standards.gov 
  -  Time Services 
  -  Office of Weights and Measures 
-  Software 
-  Data    -  Chemistry WebBook 
  -  National Vulnerability Database 
  -  Physical Reference Data 
  -  Standard Reference Data (SRD) 
-  Storefront 
-  License & Patents 
-  Computer Security Resource Center (CSRC) 
-  NIST Research Library 
 
            -  News & Events  -  News 
-  Events 
-  Blogs 
-  Feature Stories 
-  Awards 
-  Video Gallery 
-  Image Gallery 
-  Media Contacts 
 
            -  About NIST  -  About Us    -  Leadership 
  -  Organization Structure 
  -  Budget & Planning 
-  Contact Us 
-  Visit 
-  Careers    -  Student programs 
-  Work with NIST 
-  History    -  NIST Digital Archives 
  -  NIST Museum 
  -  NIST and the Nobel 
-  Educational Resources 
 

TOOLS & INSTRUMENTS

# Low Pressure Gauges

NIST has developed best-in-the-world primary standard pressure measurement capabilities for pressures spanning the range of 0.01 Pa to 360 kPa through the operation of four liquid-column manometers. While the principle behind the primary standard is simple, the manometers are operationally complex, requiring accurate determination of temperature, density, gravity, speed of sound, and ultimately column height. The NIST-designed manometers are unique in that they use an ultrasonic interferometric technique that enables operational column height resolution of 10 nm to 20 nm. Custom electronics and software implement a pulse-echo technique that uses the change in phase between sent and received radio-frequency acoustic signals to determine the length of a mercury or oil column. A diffraction correction model further refines the technique.
                    - 360 kPa Mercury Manometer.  The 360 kPa Ultrasonic Interferometer Manometer is three meters tall and provides the nation with best-in-the-world capabilities for measuring absolute and differential pressure from 1 Pa to 360 kPa with an expanded uncertainty (k=2) of [(6 x 10  - 3  Pa) 2  + (5.2 x 10 -6  * P) 2 ] 1/2 , where P is the pressure being measured. It is primarily operated for calibration of piston gauges (absolute or differential) between the pressures of 10 kPa and 360 kPa.
                    	
                    - 160 kPa Mercury Manometer.  The 160 kPa Ultrasonic Interferometer Manometer, or 1 meter manometer, provides the nation with best-in-the-world capabilities for measuring absolute and differential pressure from 1 Pa to 160 kPa with an expanded uncertainty (k=2) of [(6 x 10  - 3  Pa) 2  + (5.2 x 10 -6  * P) 2 ] 1/2 , where P is the pressure being measured. It is primarily operated for calibration of capacitance diaphragm gauges, resonance silicon gauges, or quartz Bourdon gauges between the pressures of 1 Pa and 360 kPa.
                    	
                    - 140 Pa Oil Manometer.  The 140 Pa Ultrasonic Interferometer Manometer is the only manometer at NIST that it uses oil instead of mercury as the working fluid. Oil provides for a lower vapor pressure and lower density, enabling measurements to pressures as low as 10 -2  Pa. This manometer provides measurements of absolute and differential pressure from 3 Pa to 140 Pa with an expanded uncertainty (k=2) of [(3x10 -3  Pa) 2  + (36 x 10  - 6  * P) 2 ] 1/2  and from 0.01 Pa to 3 Pa of [(7 x 10 -4  Pa) 2  + (1 x 10  - 3  * P) 2 ] 1/2 , where P is the pressure being measured. It is primarily operated for calibration of capacitance diaphragm gauges between the pressures of 0.01 Pa and 140 Pa.
                    

**How to Arrange a Calibration**
                    - Contact the staff member in charge of the calibration service for your artifact with any technical or calibration cost questions.
                    	
                    - Calibrations are performed in batches. Contact staff member for scheduling.
                    	
                    - Ship your artifact with a copy of the purchase order. Use a sturdy box in which we can return the calibrated instrument.
                    	
                    - The purchase order may be sent to contact by e-mail, fax, or mail. Please include the following information:		                       - service ID for the calibration
                      			
                      - model and serial number of your artifact
                      			
                      - customer contact name, address, phone, and e-mail
                      			
                      - return shipping method (e.g., shipping account number)
                      
                    

For contact information, shipping address, or cost of calibration, see Low pressure, vacuum, and leak measurements.

Thermodynamics

## Organizations

                          - NIST Headquarters
                            - Laboratory Programs
                              - Physical Measurement Laboratory
                                - Sensor Science Division
                                  - Thermodynamic Metrology Group

## Details

## Contact

                    -  Jacob Edmond Ricker  jacob.ricker@nist.gov      (301) 975-4475   
                    -  Jay H. Hendricks  jay.hendricks@nist.gov      (301) 975-4836   

### HEADQUARTERS

301-975-2000

Webmaster | Contact Us | Our Other Offices

TwitterFacebookLinkedInInstagramYouTubeGiphyRSS FeedMailing List

Feedback

            -  Site Privacy 
            -  Accessibility 
            -  Privacy Program 
            -  Copyrights 
            -  Vulnerability Disclosure 
            -  No Fear Act Policy 
            -  FOIA 
            -  Environmental Policy 
            -  Scientific Integrity 
            -  Information Quality Standards 
            -  Commerce.gov 
            -  Science.gov 
            -  USA.gov 
            -  Vote.gov