Skip to main content    

An official website of the United States government

Here’s how you know

**Official websites use .gov**              A **.gov** website belongs to an official government organization in the United States.            

**Secure .gov websites use HTTPS**              A **lock** (  LockA locked padlock) or **https://** means you’ve safely connected to the .gov website. Share sensitive information only on official, secure websites.            

https://www.nist.gov/laboratories/tools-instruments/sircus-facility-and-uncertainties

            -  Topics  -  All Topics 
-  Advanced communications 
-  Artificial intelligence 
-  Bioscience 
-  Buildings and construction 
-  Chemistry 
-  Climate 
-  Cybersecurity 
-  Electronics 
-  Energy 
-  Environment 
-  Fire 
-  Forensic science 
-  Health 
-  Information technology 
-  Infrastructure 
-  Manufacturing 
-  Materials 
-  Mathematics and statistics 
-  Metrology 
-  Nanotechnology 
-  Neutron research 
-  Performance excellence 
-  Physics 
-  Public safety 
-  Resilience 
-  Standards 
-  Transportation 
 
            -  Publications 
            -  Labs & Major Programs  -  Laboratories    -  Communications Technology Laboratory 
  -  Engineering Laboratory 
  -  Information Technology Laboratory 
  -  Material Measurement Laboratory 
  -  Physical Measurement Laboratory 
-  User Facilities    -  NIST Center for Neutron Research 
  -  CNST NanoFab 
-  Research Test Beds 
-  Research Projects 
-  Tools & Instruments 
-  Major Programs    -  Baldrige Performance Excellence Program 
  -  CHIPS for America Initiative 
  -  Manufacturing Extension Partnership (MEP) 
  -  Office of Advanced Manufacturing 
  -  Special Programs Office 
  -  Technology Partnerships Office 
 
            -  Services & Resources  -  Standards and Measurements    -  Calibration Services 
  -  Laboratory Accreditation (NVLAP) 
  -  Quality System 
  -  Standard Reference Materials (SRMs) 
  -  Standards.gov 
  -  Time Services 
  -  Office of Weights and Measures 
-  Software 
-  Data    -  Chemistry WebBook 
  -  National Vulnerability Database 
  -  Physical Reference Data 
  -  Standard Reference Data (SRD) 
-  Storefront 
-  License & Patents 
-  Computer Security Resource Center (CSRC) 
-  NIST Research Library 
 
            -  News & Events  -  News 
-  Events 
-  Blogs 
-  Feature Stories 
-  Awards 
-  Video Gallery 
-  Image Gallery 
-  Media Contacts 
 
            -  About NIST  -  About Us    -  Leadership 
  -  Organization Structure 
  -  Budget & Planning 
-  Contact Us 
-  Visit 
-  Careers    -  Student programs 
-  Work with NIST 
-  History    -  NIST Digital Archives 
  -  NIST Museum 
  -  NIST and the Nobel 
-  Educational Resources 
 

TOOLS & INSTRUMENTS

# SIRCUS facility and uncertainties

The basic SIRCUS calibration setup is shown at the right. The output from a tunable laser is sent through an intensity stabilizer, then through a 'speckle removal system' (typically an ultrasonic bath for fiber-coupled systems like in the UV-Vis-NIR SIRCUS) and into the integrating sphere. Beamsplitters send some of the laser radiation into a spectrum analyzer and a wavemeter to monitor the spectral properties and the wavelength of the light. The irradiance is measured first by the reference instrument and the radiometric quantity is measured by the Radiometer Under Test. A monitor detector on the wall of the integrating sphere accounts for any laser intensity changes between the two measurements. A linear encoder records the position of the radiometer stage and is used to radiometrically determine the distance between the sphere and the irradiance detector reference plane (using the inverse square law). A computer controls the measurement sequence and records the data.

The lasers and output powers used in the UV-Vis-NIR SIRCUS are shown below. Several different tunable lasers are used. The UV region (200 nm to 400 nm) a pulsed laser (Ti-Sapphire) is used, but has been verified to be equivalent to continuous-wave (cw) lasers for standard detector calibrations. We are working to further extend the current range of wavelength coverage.

### SIRCUS uncertainties:

The SIRCUS facility provides the state-of-the-art low uncertainty calibration for various optical radiometers and instruments used in many applications. Currently an expanded uncertainty of 0.08 % (k=2) is achievable for an irradiance responsivity calibration in the visible and near IR region for high-quality instruments. The primary uncertainty components and their estimated magnitude are listed below.

For the IR spectral range, a liquid-helium-cooled Electrical Substitution Bolometer (ESB) is characterized and calibrated against irradiance mode Si trap radiometers and is used as the reference detector. InGaAs and InSb working standards are used for cases where adequate uncertainty can be achieved more conveniently.

The uncertainty for irradiance responsivity measurements that have been done to date is currently at about 3 % (k=2), and current developments are directed at reducing this to a general goal of 0.2 % (k=1) Also, the IR-SIRCUS is an optimum monochromatic uniform source facility that can provide detector-based calibration support for digital imaging systems such as detector arrays, infrared cameras, IR focal plane array radiometers, and spectrometers.

The increased accuracy of transfer and working standard radiometers, calibrated on the IR-SIRCUS facility, will increase the accuracy of radiometric measurements in the infrared wavelength range. The IR-SIRCUS will provide an absolute calibration for InGaAs detectors, Extended InGaAs detectors, InSb detectors, pyroelectric detectors and HgCdTe detectors with high accuracy. The IR-SIRCUS facility is the basis to spread the new high-accuracy detector-based calibrations for military, academia, and industry in the infrared.

Optical / photometry / laser metrologyCalibration services

## Organizations

                          - NIST Headquarters
                            - Laboratory Programs
                              - Physical Measurement Laboratory
                                - Sensor Science Division
                                  - Remote Sensing Group

## Details

## Contact

                    -  John T. Woodward IV  john.woodward@nist.gov      (301) 975-5495   
                    -  Ping-Shine Shaw  ping-shine.shaw@nist.gov      (301) 975-4416   

## Related Projects

Spectral irradiance and radiance responsivity calibrations using uniform sources (SIRCUS)

### HEADQUARTERS

301-975-2000

Webmaster | Contact Us | Our Other Offices

TwitterFacebookLinkedInInstagramYouTubeGiphyRSS FeedMailing List

Feedback

                      -  Site Privacy 
                      -  Accessibility 
                      -  Privacy Program 
                      -  Copyrights 
                      -  Vulnerability Disclosure 
                      -  No Fear Act Policy 
                      -  FOIA 
                      -  Environmental Policy 
                      -  Scientific Integrity 
                      -  Information Quality Standards 
                      -  Commerce.gov 
                      -  Science.gov 
                      -  USA.gov 
                      -  Vote.gov