Skip to main content    

An official website of the United States government

Here’s how you know

**Official websites use .gov**              A **.gov** website belongs to an official government organization in the United States.            

**Secure .gov websites use HTTPS**              A **lock** (  LockA locked padlock) or **https://** means you’ve safely connected to the .gov website. Share sensitive information only on official, secure websites.            

https://www.nist.gov/laboratories/tools-instruments/thermal-infrared-transfer-radiometer-txr

            -  Topics  -  All Topics 
-  Advanced communications 
-  Artificial intelligence 
-  Bioscience 
-  Buildings and construction 
-  Chemistry 
-  Climate 
-  Cybersecurity 
-  Electronics 
-  Energy 
-  Environment 
-  Fire 
-  Forensic science 
-  Health 
-  Information technology 
-  Infrastructure 
-  Manufacturing 
-  Materials 
-  Mathematics and statistics 
-  Metrology 
-  Nanotechnology 
-  Neutron research 
-  Performance excellence 
-  Physics 
-  Public safety 
-  Resilience 
-  Standards 
-  Transportation 
 
            -  Publications 
            -  Labs & Major Programs  -  Laboratories    -  Communications Technology Laboratory 
  -  Engineering Laboratory 
  -  Information Technology Laboratory 
  -  Material Measurement Laboratory 
  -  Physical Measurement Laboratory 
-  User Facilities    -  NIST Center for Neutron Research 
  -  CNST NanoFab 
-  Research Test Beds 
-  Research Projects 
-  Tools & Instruments 
-  Major Programs    -  Baldrige Performance Excellence Program 
  -  CHIPS for America Initiative 
  -  Manufacturing Extension Partnership (MEP) 
  -  Office of Advanced Manufacturing 
  -  Special Programs Office 
  -  Technology Partnerships Office 
 
            -  Services & Resources  -  Standards and Measurements    -  Calibration Services 
  -  Laboratory Accreditation (NVLAP) 
  -  Quality System 
  -  Standard Reference Materials (SRMs) 
  -  Standards.gov 
  -  Time Services 
  -  Office of Weights and Measures 
-  Software 
-  Data    -  Chemistry WebBook 
  -  National Vulnerability Database 
  -  Physical Reference Data 
  -  Standard Reference Data (SRD) 
-  Storefront 
-  License & Patents 
-  Computer Security Resource Center (CSRC) 
-  NIST Research Library 
 
            -  News & Events  -  News 
-  Events 
-  Blogs 
-  Feature Stories 
-  Awards 
-  Video Gallery 
-  Image Gallery 
-  Media Contacts 
 
            -  About NIST  -  About Us    -  Leadership 
  -  Organization Structure 
  -  Budget & Planning 
-  Contact Us 
-  Visit 
-  Careers    -  Student programs 
-  Work with NIST 
-  History    -  NIST Digital Archives 
  -  NIST Museum 
  -  NIST and the Nobel 
-  Educational Resources 
 

TOOLS & INSTRUMENTS

# Thermal infrared transfer radiometer (TXR)

The NIST Thermal-infrared Transfer Radiometer (TXR) is used for measuring the radiance temperature in the thermal-infrared spectral region of extended-area blackbody sources, over the range of about 190 K to 350 K, either in vacuum chambers or ambient condition laboratories. The TXR is a portable radiometer with two channels, one at 5 μm and the other at 10 μm. It uses a photovoltaic InSb detector for the 5-μm channel, and a photovoltaic mercury cadmium telluride (MCT) detector for the 10-μm channel. The detectors, filters, and reflective optics are built into a liquid-nitrogen cryostat, and the entire radiometer is vacuum/cryogenic compatible and designed to be deployed inside of the typical space-simulating vacuum chambers used for space-flight instrument calibrations. The TXR has a self-contained vacuum jacket and liquid-nitrogen reservoir, and so can be also used in a typical laboratory environment with the outer case of the cryostat at ambient pressure and room temperature. This allows it to be characterized and calibrated by a full array of infrared facilities in the Sensor Science Division at NIST, much of it in an ambient laboratory environment. The standard uncertainty for radiance measurements using the TXR is of the order of 0.2 % at 10 μm, which corresponds to about 60 mK in radiance temperature near 300 K, with absolute calibration traceable to the kelvin through a NIST water-bath blackbody.

The TXR has so far been successfully deployed 13 different times (between 1999 and 2015) to several different aerospace calibration facilities, where it performed in-situ measurements of various sources in a space-simulating chambers. These measurements were used to verify the infrared radiance scales in currently used by several NASA, NOAA, DOE, and DOD satellite programs. The TXR is maintained at NIST when it is not on a deployment.

## Specifications/Capabilities

PhysicsAerospace

## Organizations

                          - NIST Headquarters
                            - Laboratory Programs
                              - Physical Measurement Laboratory
                                - Sensor Science Division
                                  - Remote Sensing Group

## Details

## Contact

                    -  Joseph P Rice  joe.rice@nist.gov      (301) 975-2133   

## Related Projects

Transfer radiometers and reference detectors

## Related Publications

The NIST EOS Thermal-Infrared Transfer Radiometer

### HEADQUARTERS

301-975-2000

Webmaster | Contact Us | Our Other Offices

TwitterFacebookLinkedInInstagramYouTubeGiphyRSS FeedMailing List

Feedback

                      -  Site Privacy 
                      -  Accessibility 
                      -  Privacy Program 
                      -  Copyrights 
                      -  Vulnerability Disclosure 
                      -  No Fear Act Policy 
                      -  FOIA 
                      -  Environmental Policy 
                      -  Scientific Integrity 
                      -  Information Quality Standards 
                      -  Commerce.gov 
                      -  Science.gov 
                      -  USA.gov 
                      -  Vote.gov