Skip to main content    

An official website of the United States government

Here’s how you know

**Official websites use .gov**              A **.gov** website belongs to an official government organization in the United States.            

**Secure .gov websites use HTTPS**              A **lock** (  LockA locked padlock) or **https://** means you’ve safely connected to the .gov website. Share sensitive information only on official, secure websites.            

https://www.nist.gov/laboratories/tools-instruments/si-euv-photodiode-transfer-standard

            -  Topics  -  All Topics 
-  Advanced communications 
-  Artificial intelligence 
-  Bioscience 
-  Buildings and construction 
-  Chemistry 
-  Climate 
-  Cybersecurity 
-  Electronics 
-  Energy 
-  Environment 
-  Fire 
-  Forensic science 
-  Health 
-  Information technology 
-  Infrastructure 
-  Manufacturing 
-  Materials 
-  Mathematics and statistics 
-  Metrology 
-  Nanotechnology 
-  Neutron research 
-  Performance excellence 
-  Physics 
-  Public safety 
-  Resilience 
-  Standards 
-  Transportation 
 
            -  Publications 
            -  Labs & Major Programs  -  Laboratories    -  Communications Technology Laboratory 
  -  Engineering Laboratory 
  -  Information Technology Laboratory 
  -  Material Measurement Laboratory 
  -  Physical Measurement Laboratory 
-  User Facilities    -  NIST Center for Neutron Research 
  -  CNST NanoFab 
-  Research Test Beds 
-  Research Projects 
-  Tools & Instruments 
-  Major Programs    -  Baldrige Performance Excellence Program 
  -  CHIPS for America Initiative 
  -  Manufacturing Extension Partnership (MEP) 
  -  Office of Advanced Manufacturing 
  -  Special Programs Office 
  -  Technology Partnerships Office 
 
            -  Services & Resources  -  Standards and Measurements    -  Calibration Services 
  -  Laboratory Accreditation (NVLAP) 
  -  Quality System 
  -  Standard Reference Materials (SRMs) 
  -  Standards.gov 
  -  Time Services 
  -  Office of Weights and Measures 
-  Software 
-  Data    -  Chemistry WebBook 
  -  National Vulnerability Database 
  -  Physical Reference Data 
  -  Standard Reference Data (SRD) 
-  Storefront 
-  License & Patents 
-  Computer Security Resource Center (CSRC) 
-  NIST Research Library 
 
            -  News & Events  -  News 
-  Events 
-  Blogs 
-  Feature Stories 
-  Awards 
-  Video Gallery 
-  Image Gallery 
-  Media Contacts 
 
            -  About NIST  -  About Us    -  Leadership 
  -  Organization Structure 
  -  Budget & Planning 
-  Contact Us 
-  Visit 
-  Careers    -  Student programs 
-  Work with NIST 
-  History    -  NIST Digital Archives 
  -  NIST Museum 
  -  NIST and the Nobel 
-  Educational Resources 
 

TOOLS & INSTRUMENTS

# Si EUV Photodiode Transfer Standard

Silicon (Si) photodiodes are issued by NIST as transfer standard detectors throughout the extreme ultraviolet spectral region (5 nm to 254 nm). Each device is an n-on-p junction with a thin (typically 5 nm) oxide layer grown on the front surface and specially processed to improve the radiation hardness. Once the chip is manufactured, it is mounted in a ceramic case. Redundant leads are wire bonded to the chip to make an electrical connection to two pins where user electronics may be easily attached. Incident radiation generates electron-hole pairs in the Si which cause an electric current to flow across the junction.

The responsivity of a Si photodiode is the externally detected electric current per unit of incident power. Typically, this is expressed in units of A / W. For wavelengths below about 60 nm, the responsivity is nearly independent of wavelength and has a value of about 0.28 A / W. The exact value depends on the transmission through the silicon dioxide layer and any internal losses (*e.g.* surface recombination) in the photodiode, both of which have some wavelength dependence and may vary considerable between different manufacturing lots. At longer wavelengths, the oxide transmission, charge injection from oxide absorption, and internal losses become much more significant. Additionally, the mean electron-hole pair creation energy is no longer constant with wavelength as the indirect bandgap absorption becomes significant. This manifests as a reduction on the responsivity at wavelengths longer than about 60 nm.

Si photodiodes have much higher responsivity than aluminum oxide photoemissive detectors, require no bias voltage, and are small and lightweight. However, they are noisier and exhibit higher dark currents than photoemissive standards and have broadband response from the infrared to the x-ray region.

## Specifications/Capabilities

EUV Calibration Range: 5 nm to 254 nm

Useful Range: x-ray (5 keV to 10 keV) to infrared (1,100 nm)

Typical Responsivity: 0.07 A / W to 0.27 A / W

Dynamic Range: 10 pW to 10 mW (wavelength dependent)

Active Area: 10 mm by 10 mm

## Scientific Opportunities/Applications

The Si transfer standard photodiodes are general-purpose radiometric detectors optimized for the extreme ultraviolet spectral region and are suitable for almost any application requiring the measurement of radiant power. Detector lots are selected for high radiation hardness and low-noise characteristics. These photodiodes are space-qualified and have significant flight heritage.

Optical / photometry / laser metrologyOptical physics and communicationsCalibration services

## Organizations

                          - NIST Headquarters
                            - Laboratory Programs
                              - Physical Measurement Laboratory
                                - Sensor Science Division
                                  - Optical Radiation Group

## Details

## Contact

                    -  Robert E Vest  robert.vest@nist.gov      (301) 975-3992   

## Related Projects

Extreme Ultraviolet Detector Calibration Service

## Related Publications

Stable Silicon Photodiodes for Absolute Intensity Measurements in the VUV and Soft X-Ray Regions

Present Status of Radiometric Quality Silicon Photodiodes

Response of a Silicon Photodiode to Pulsed Radiation

### HEADQUARTERS

301-975-2000

Webmaster | Contact Us | Our Other Offices

TwitterFacebookLinkedInInstagramYouTubeGiphyRSS FeedMailing List

Feedback

                      -  Site Privacy 
                      -  Accessibility 
                      -  Privacy Program 
                      -  Copyrights 
                      -  Vulnerability Disclosure 
                      -  No Fear Act Policy 
                      -  FOIA 
                      -  Environmental Policy 
                      -  Scientific Integrity 
                      -  Information Quality Standards 
                      -  Commerce.gov 
                      -  Science.gov 
                      -  USA.gov 
                      -  Vote.gov