Skip to main content    

An official website of the United States government

Here’s how you know

**Official websites use .gov**              A **.gov** website belongs to an official government organization in the United States.            

**Secure .gov websites use HTTPS**              A **lock** (  LockA locked padlock) or **https://** means you’ve safely connected to the .gov website. Share sensitive information only on official, secure websites.            

https://www.nist.gov/laboratories/tools-instruments/lunar-spectro-radiometer

            -  Topics  -  All Topics 
-  Advanced communications 
-  Artificial intelligence 
-  Bioscience 
-  Buildings and construction 
-  Chemistry 
-  Climate 
-  Cybersecurity 
-  Electronics 
-  Energy 
-  Environment 
-  Fire 
-  Forensic science 
-  Health 
-  Information technology 
-  Infrastructure 
-  Manufacturing 
-  Materials 
-  Mathematics and statistics 
-  Metrology 
-  Nanotechnology 
-  Neutron research 
-  Performance excellence 
-  Physics 
-  Public safety 
-  Resilience 
-  Standards 
-  Transportation 
 
            -  Publications 
            -  Labs & Major Programs  -  Laboratories    -  Communications Technology Laboratory 
  -  Engineering Laboratory 
  -  Information Technology Laboratory 
  -  Material Measurement Laboratory 
  -  Physical Measurement Laboratory 
-  User Facilities    -  NIST Center for Neutron Research 
  -  CNST NanoFab 
-  Research Test Beds 
-  Research Projects 
-  Tools & Instruments 
-  Major Programs    -  Baldrige Performance Excellence Program 
  -  CHIPS for America Initiative 
  -  Manufacturing Extension Partnership (MEP) 
  -  Office of Advanced Manufacturing 
  -  Special Programs Office 
  -  Technology Partnerships Office 
 
            -  Services & Resources  -  Standards and Measurements    -  Calibration Services 
  -  Laboratory Accreditation (NVLAP) 
  -  Quality System 
  -  Standard Reference Materials (SRMs) 
  -  Standards.gov 
  -  Time Services 
  -  Office of Weights and Measures 
-  Software 
-  Data    -  Chemistry WebBook 
  -  National Vulnerability Database 
  -  Physical Reference Data 
  -  Standard Reference Data (SRD) 
-  Storefront 
-  License & Patents 
-  Computer Security Resource Center (CSRC) 
-  NIST Research Library 
 
            -  News & Events  -  News 
-  Events 
-  Blogs 
-  Feature Stories 
-  Awards 
-  Video Gallery 
-  Image Gallery 
-  Media Contacts 
 
            -  About NIST  -  About Us    -  Leadership 
  -  Organization Structure 
  -  Budget & Planning 
-  Contact Us 
-  Visit 
-  Careers    -  Student programs 
-  Work with NIST 
-  History    -  NIST Digital Archives 
  -  NIST Museum 
  -  NIST and the Nobel 
-  Educational Resources 
 

TOOLS & INSTRUMENTS

# Lunar spectro-radiometer

Design of the lunar irradiance instrument is based on commercially available spectrographs. These spectrographs can be stable and sub-components are readily available to make a custom instrument. To account for their non-uniform response and the non-uniformity of the lunar image, the spectrographs are illuminated by an integrating sphere at the focus of a conventional on-axis, 25-cm diameter telescope. Below is a simplified optical schematic showing one possible configuration.

Moonlight is collected with a 25-cm (10 inch) diameter, Ritchey-Chrétien f/4 telescope. The moonlight passes through one of the positions of a filter wheel before being focused to form an image at the plane of an aperture wheel. Following the aperture wheel is a lens that re-images at f/1.5 onto the entrance aperture of an integrating sphere. To minimize sensitivity of the response to small pointing errors when collecting the entire lunar disk, the lunar image under fills the aperture. Diffused light from the sphere provides uniform illumination of three fiber bundles, each of which is connected to a spectrograph covering a different spectral range. Some of the filters will measure linear polarization. Other filters along with reference sources and detectors will measure changes in the spectrograph wavelength scale and changes in optical through-put at various stages.

A similar simpler instrument that observes stars is to provide atmospheric corrections. In this instrument starlight is directly coupled into the spectrographs.

Both instruments together are compact enough to fit on a single mount and to fly on a high-altitude balloon.

Finally, there is a provision to measure the solar irradiance that will allow LUSI to provide the lunar reflectance as a secondary data product. A small diameter, non-imaging solar collector will fiber-couple sunlight into the integrating sphere. Stability of the collector throughput will be monitored using a small, detector-stabilized LED external source.

### Calibration and Characterization

Calibration of the instrument will be performed at the facility for Spectral Irradiance and Radiance responsivity Calibrations with Uniform Sources (SIRCUS) at NIST. This detector-based facility has its fundamental radiometric scale is set by an absolute cryogenic radiometer, the Primary Optical Watt Radiometer (POWR), whose power responsivity is known and ultimately tied to electrical standards. Calibrated detectors and precision apertures set the irradiance scale. However, a stable and relatively uniform source is needed to transfer this scale to the lunar instruments.

To simulate the Moon, a collimator coupled with a laser-fed integrating sphere source will be used to provide a large diameter beam that diverges by the angular extent of the Moon. At the focus of the collimator the integrating sphere exit port will be fitted with apertures of various shapes and orientations to test sensitivity of the instruments to different lunar phases. A similarly positioned linear polarizer will allow testing of sensitivity to polarization. The source feeding the sphere will be either one of several existing lasers at SIRCUS that provide continuous wavelength coverage from 210 nm to 2500 nm or a monochromator fed by a supercontinuum source.

The best uncertainties associated with transferring the irradiance and radiance scales to a radiometer are expected to be 0.1 % (k=1) or better from 350 nm to 2500 nm. However, the overall measurement uncertainty when deducing the lunar irradiance will depend on additional factors, including the stability of the lunar instrument, expected to be at the 0.1 % level, and the uncertainty in deducing the transmittance of the atmosphere, which will vary strongly with wavelength, estimated to be approximately 0.5 % through visible atmospheric window bands and less through the longer wavelength window bands. Knowledge of the atmospheric transmittance is expected to limit the overall uncertainty to approximately 0.5 %.

Optical / photometry / laser metrologyCalibration services

## Organizations

                          - NIST Headquarters
                            - Laboratory Programs
                              - Physical Measurement Laboratory
                                - Sensor Science Division
                                  - Remote Sensing Group

## Details

## Contact

                    -  Stephen Maxwell  stephen.maxwell@nist.gov      (301) 975-4950   

## Related Projects

Lunar Spectral Irradiance (LUSI)

Spectral irradiance and radiance responsivity calibrations using uniform sources (SIRCUS)

Primary optical watt radiometer (POWR)

NIST Stars

### HEADQUARTERS

301-975-2000

Webmaster | Contact Us | Our Other Offices

TwitterFacebookLinkedInInstagramYouTubeGiphyRSS FeedMailing List

Feedback

                      -  Site Privacy 
                      -  Accessibility 
                      -  Privacy Program 
                      -  Copyrights 
                      -  Vulnerability Disclosure 
                      -  No Fear Act Policy 
                      -  FOIA 
                      -  Environmental Policy 
                      -  Scientific Integrity 
                      -  Information Quality Standards 
                      -  Commerce.gov 
                      -  Science.gov 
                      -  USA.gov 
                      -  Vote.gov