X

X

5 Committee Draft OIML/5CD

Date: 21 October, 2016

Reference number: 

TC_9_p_1_N001

Supersedes document: R 60-1/4 CD

OIML TC 9 : 

Instruments for Measuring Mass and Density

Circulated to P- and O-members and 

liaison international bodies and external 

organizations for:

P-members votes and 

comments

O-members and liaisons

comments

by: 27 January, 2017

Title: 

Metrological Regulation for Load Cells

Part 1: Metrological and Technical 

Requirements

Clean version

Convener: 

John Barton

National Institute of Standards and 

Technology, 

USA

TITLE OF THE CD (English):

OIML R 60- Parts 1 

Metrological Regulation for Load Cells

Part 1: Metrological and Technical Requirements

TITRLE DU CD (French):

OIML R 60-1

Réglementation métrologique des celleules de pesée

Partie 1: Exigences métrologiques et techniques, 

Original version in: English

October 2016 OIML R 60-1

1

Contents

PART 1 METROLOGICAL AND TECHNICAL REQUIREMENTS 2

1. INTRODUCTION 2

2. SCOPE 3

3. TERMINOLOGY (TERMS AND DEFINITIONS) 4

3.1. GENERAL DEFINITIONS 4

3.2. CATEGORIES OF LOAD CELLS 6

3.3. CONSTRUCTION OF LOAD CELLS 6

3.4. METROLOGICAL CHARACTERISTICS OF A LOAD CELL 7

3.5. RANGE, CAPACITY AND OUTPUT TERMS 7

3.6. ILLUSTRATION OF CERTAIN DEFINITIONS 9

3.7. MEASUREMENT AND ERROR TERMS 10

3.8. INFLUENCES AND REFERENCE CONDITIONS 13

3.9. ABBREVIATIONS 14

4. DESCRIPTION OF LOAD CELLS 14

5. METROLOGICAL REQUIREMENTS 15

5.1. PRINCIPLE OF LOAD CELL CLASSIFICATION 15

5.2. MEASURING RANGES 17

5.3. MAXIMUM PERMISSIBLE MEASUREMENT ERRORS 18

5.4. REPEATABILITY ERROR 19

5.5. PERMISSIBLE VARIATION OF RESULTS UNDER REFERENCE CONDITIONS 19

5.6. INFLUENCE QUANTITIES (RATED OPERATING CONDITIONS) 20

5.7. REQUIREMENTS FOR LOAD CELLS EQUIPPED WITH ELECTRONICS 21

6. TECHNICAL REQUIREMENTS 24

6.1. SOFTWARE 24

6.2. INSCRIPTIONS AND PRESENTATION OF LOAD CELL INFORMATION 25

ANNEX B (MANDATORY) DEFINITIONS FROM OTHER APPLICABLE INTERNATIONAL

PUBLICATIONS A-1

ANNEX C (MANDATORY) OIML CERTIFICATE OF CONFORMITY FOR LOAD CELLS -

FORMAT OF CERTIFICATE B-1

ANNEX D (INFORMATIVE) OIML CERTIFICATE OF CONFORMITY FOR LOAD CELLS C-1

ANNEX E (INFORMATIVE) SELECTION OF LOAD CELL(S) FOR TESTING - A PRACTICAL 

EXAMPLE D-1

ANNEX F (INFORMATIVE) LOAD TRANSMISSION TO THE LOAD CELL E-1

BIBLIOGRAPHY I
OIML R 60-1 October 2016

2

Part 1 Metrological and technical requirements

1. Introduction

The subject of this Recommendation, load cells comprise a distinct element or module within 

other complex instruments. Load cells do not produce distinct quantitative values that are 

inherently identified or associated with denominations or units. The data that can be extracted 

from a load cell is simply a measurement of change in the output of the load cell in relation to the 

input. This relative change must be converted by other elements or modules within an 

instrument into values that are meaningful measurements which can then be used to identify a 

quantity.

Although strain gauge technology was a primary focus in the initial development of R60, it is to 

be understood that load cells that operate using other principles may also be evaluated under this 

Recommendation.
October 2016 OIML R 60-1

3

2. Scope 

2.1.

This Recommendation prescribes the principal metrological static requirements and static 

evaluation procedures for load cells used in the determination of conformity to this 

recommendation. It is intended to provide authorities with uniform means for determining the 

metrological characteristics of load cells used in measuring instruments that are subjected to 

metrological controls.

It is acknowledged that test procedures found in Part 2 of this Recommendation (see section 8) 

are useful in the evaluation of load cells that are currently found in service (i.e., primarily strain 

gauge design) however, there may be variations in designs for load cells that will require 

additional or modified test procedures to appropriately evaluate them. These additional test 

procedures may be annexed when necessary.

Except where otherwise specified, these requirements apply regardless of the technology or 

operating principle employed. The requirements and evaluation procedures in this 

Recommendation have been drafted to be non-specific with regard to load cell design and their 

operating principles.

2.2.

This Recommendation utilizes the principle that several measurement errors shall be considered 

together when applying load cell performance characteristics to the permitted error envelope. 

Thus, it is not considered appropriate to specify individual errors for given characteristics (e.g., 

non-linearity, hysteresis, effects of influence factors), but rather to consider the total error 

envelope allowed for a load cell as the limiting factor. The use of an error envelope allows the 

balancing of the individual contributions to the total error of measurement while still achieving 

the intended final result.

Note: the error envelope may be defined as the boundary of the combined individual errors

(see Table 4) as a function of the force introduced by the applied load (expressed in mass units) 

over the measuring range. The combined error determined may be positive or negative and 

include the effects of nonlinearity, hysteresis and temperature.

2.3.

“Weighing modules” as noted in OIML R76 [1], T.2.2.7 (see Annex A, A.2.1), are not covered 

by this recommendation. Weighing instruments that include load cells and which give an 

indication of mass are the subjects of separate Recommendations. While digital load cells may 

be covered under this Recommendation, a load cell that produces an output consisting of more 

than digital “raw counts” will not be covered under R60. In the illustration from OIML R76 

below, the scope of R60 would not extend beyond module #3.
OIML R 60-1 October 2016

4

Analog load cell__________________________2

Digital load cell __________________________2 + 3

Electronic load cell _______________________2 + 3 + (4)*

Load cells within scope of OIML R 60 ________2 + 3

* Numbers in brackets indicate options

From OIML R76:

Definition of typical modules within a weighing system (other combinations are possible)

Figure 1. Typical Components in a Weighing Instrument

3. Terminology (Terms and definitions)

The terms most frequently used in the load cell field and their definitions are given below (see 

3.6 for an illustration of certain definitions). The terminology used in this Recommendation 

conforms to OIML V 1 International Vocabulary of Basic and General Terms in Metrology 

(VIM) [2], to OIML V 2 International Vocabulary of Terms in Legal Metrology (VIML) [3], to 

OIML D 9 Principles of metrological supervision [4], to OIML D 11 General Requirements for 

electronic measuring instruments [5], and to OIML B 3 OIML Certificate System for Measuring 

Instruments [6].

In addition, for the purposes of this Recommendation, the following definitions apply:

3.1. General definitions

3.1.1. durability [VIML 5.15]

Ability of a measuring instrument to maintain its performance characteristics over a 

period of use.

Peripheral devices

Data storage 

device

Secondary Primary display display

Key(s) or 

keyboard to 

operate

Further data 

processing 

(e.g., tare, price 

calculation)

Data processing 

(scaling) Analog load cell ADC

Mechanical and 

electrical 

connecting 

elements

Other peripheral 

devices

1 2 3 4 5

6

7

Net, gross, tare and other 

primary indication (e.g. price)

Weighing 

value 

(in mass units)

Digital “raw values” 

(counts)

Analog 

signal

Printer
October 2016 OIML R 60-1

5

3.1.2. legal metrology [VIML 1.01]

practice and process of applying statutory and regulatory structure and enforcement to 

metrology.

(For notes, please refer to VIML)

3.1.3. load cell 

measuring transducer that will produce an output in response to an applied load.. 

This output may be converted by another device into measurement units such as 

mass.

3.1.3.1. non-electronic load cell

load cell from which the output provides either measureable data or direct 

information representing the measurand value.

Note: The ratio between output and input may be adjustable and this type of load 

cell may utilize:

• passive electronics (e.g., strain gauges); and

• passive temperature compensation elements

3.1.3.2. electronic load cell

a type of digital load cell (3.1.3.3.) which is capable of performing the functions as 

described under “non-electronic” load cell (3.1.3.1.) and which also utilizes active 

electronics and would therefore be equipped with metrologically relevant software.

Note: This type of load cell may utilize the active electronics for:

• gaining an electronic representation of the measurand value;

• active temperature compensation; and

• similar functions being of influence to the measurand value.

3.1.3.3. digital load cell

electronic load cell which includes an analogue to digital conversion device 

providing a representation of the measurand value in some digital format.

3.1.4. marking [VIML 2.19]

affixing of one or more marks.

(For notes, please refer to VIML)

3.1.5. metrological supervision [VIML 2.03]

activity of legal metrological control to check the observance of metrology laws and 

regulations.

(For notes, please refer to VIML)
OIML R 60-1 October 2016

6

3.1.6. measuring transducer [VIM 3.7] 

device, used in measurement, that provides an output quantity having a specified 

relation to the input quantity.

3.1.7. performance test

test to verify whether the load cell under test is capable of performing its intended 

functions.

3.1.8. rated operating condition [VIM 4.9]

operating condition that must be fulfilled during measurement in order that a 

measuring instrument or measuring system perform as designed.

(For notes, please refer to VIM)

3.1.9. sealing [VIML 2.20]

means intended to protect the measuring instrument against any unauthorized 

modification, readjustment, removal of parts, software, etc.

(For notes, please refer to VIML)

3.1.10. type (pattern) evaluation [VIML 2.04]

conformity assessment procedure on one or more specimens of an identified type 

(pattern) of measuring instruments which results in an evaluation report and / or an 

evaluation certificate.

(For notes, please refer to VIML)

3.1.11. type approval [VIML 2.05]

decision of legal relevance, based on the review of the type evaluation report, that the 

type of a measuring instrument complies with the relevant statutory requirements and 

results in the issuance of the type approval certificate.

3.2. Categories of Load Cells

3.2.1. Application of load

3.2.1.1. compression loading

applying a compressive force to the load receptor of a load cell. 

3.2.1.2. tension loading

applying a tension force to the load receptor of a load cell.

3.3. Construction of load cells*

* See also sections 1 (Introduction) and 2.1
October 2016 OIML R 60-1

7

3.3.1. strain gauge

analog resistive element that is attached to a load cell structure and changes resistance 

depending on the deformation of the load cell structure when compression or tension 

forces are applied to the load cell.

3.4. Metrological characteristics of a load cell 

3.4.1. humidity symbol 

symbol assigned to a load cell that indicates the conditions of humidity under which 

the load cell has been tested.

3.4.2. load cell family 

group of load cells, which for the purposes of type evaluation, are considered one a 

family and that are of:

a) the same material or combination of materials (for example, mild steel, stainless 

steel or aluminum); 

b) the same design of the measurement technique (for example, strain gauges bonded 

to metal);

c) when used, the same type attachment of the strain gauge to the load cell (e.g., use 

of the same or similar adhesive);

d) the same method of construction (for example, shape, sealing of strain gauges, 

mounting method, manufacturing method); the same set of specifications (for 

example, output rating, input impedance, supply voltage, cable details); and

e) one or more load cell groups where all load cells within the group possess 

identical metrological characteristics (as listed in 5.1.5 – including: class; nmax; 

temperature rating; etc.). 

Note: The examples provided are not intended to be limiting.

3.5. Range, capacity and output terms

3.5.1. load cell interval 

subdivision of the load cell measuring range. 

3.5.2. load cell measuring range 

range between the maximum load of the measuring range Dmax and minimum load of 

the measuring range Dmin 

Load cell measuring range = (Dmax – Dmin)

3.5.3. load cell output 

measurable quantity into which a load cell converts the measured input quantity.

3.5.4. load cell verification interval (v) 
OIML R 60-1 October 2016

8

load cell interval, expressed in units of mass, used in the test of the load cell for 

accuracy classification.

3.5.5. maximum capacity (Emax)

largest value of a quantity expressed in units of mass, which may be applied to a load 

cell. 

3.5.6. maximum load of the measuring range (Dmax) 

largest value of a quantity expressed in units of mass which can be introduced to a 

load cell under test. 

3.5.7. maximum measuring range 

range of values of the quantity expressed in units of mass that may be applied to a 

load cell.

Note:

maximum measuring range is the range between maximum capacity Emax and 

minimum dead load Emin

[maximum measuring range = (Emax – Emin)]

3.5.8. maximum number of load cell verification intervals (nLC) 

maximum number of load cell verification intervals into which the maximum

measuring range may be divided.

3.5.9. minimum dead load (Emin) 

smallest value of a quantity (expressed in mass units) that may be applied to a load 

cell. 

3.5.10. minimum dead load output return (DR) 

difference of load cell output, expressed in units of mass at the minimum dead load 

(Emin), measured before and after application of a load of Dmax

3.5.11. minimum load cell verification interval (vmin) 

smallest load cell verification interval in units of mass into which the maximum

measuring range (Emax – Emin) can be divided.

3.5.12. minimum load of the measuring range (Dmin) 

smallest value of a quantity expressed in units of mass, applied to a load cell under 

test. 

3.5.13. number of load cell verification intervals (n) 

total of load cell verification intervals into which the maximum measuring range,

Emax – Emin is divided.
October 2016 OIML R 60-1

9

3.5.14. relative minimum dead load output return Z 

ratio of the maximum measuring range, to two times the minimum dead load output 

return, DR. 

Note: This ratio is used to describe multi-interval instruments.

3.5.15. relative minimum load cell verification interval Y 

ratio of the maximum measuring range, to the minimum load cell verification 

interval, vmin. 

Note: This ratio describes the resolution of the load cell independent from the load 

cell capacity

3.5.16. safe load limit (Elim) 

maximum load that can be applied without producing a permanent shift in the 

performance characteristics beyond those specified.

3.5.17. warm-up time 

time between the moment power is applied to a load cell and the moment at which the 

load cell is capable of complying with the requirements. 

3.6. Illustration of certain definitions 

The terms that appear above the central horizontal line (related to parameters Emin and Emax) in 

Figure 2 below are parameters that are fixed by the design of the load cell. The terms that appear 

below that line (related to parameters Dmin and Dmax) are parameters that are variable, dependent 

on the conditions of the test of a load cell (in particular, those load cells used in weighing 

instruments).
OIML R 60-1 October 2016

10

Minimum Dead Load

 Emin

Maximum Capacity

 Emax

No Load

Maximum Measuring Range Safe Load 

Limit

Elim

Load Cell Measuring Range

Minimum Load

Dmin

Maximum Load

Dmax

The following statements apply: (see also 8.7.3.4)

a) (Dmin – Dmax) ≤ (Emin-Emax)

b) Emin ≤ Dmin ≤ (0.1 Emax), and (0.9 Emax) ≤ Dmax ≤ Emax

Figure 2. Illustration of certain definitions

3.7. Measurement and error terms 

3.7.1. creep 

change in load cell output occurring with time while under constant load and with all 

environmental conditions and other variables also remaining constant.

3.7.2. apportioning factor (pLC) 

the value of a dimensionless fraction expressed as a decimal (for example, 0.7) 

representing that portion of an error observed in the (weighing) instrument which is 

attributed to the load cell alone.

Note: This value is used in determining MPE (see 3.7.10) 

3.7.3. durability error [VIML 5.16]

difference between the intrinsic error after a period of use and the initial intrinsic 

error of a measuring instrument 
October 2016 OIML R 60-1

11

3.7.4. expanded uncertainty 

quantity defining an interval about the result of a measurement that may be expected 

to encompass a large fraction of the distribution of values that could reasonably be 

attributed to the measurand. [Guide to the Expression of Uncertainty in 

Measurement, BIPM, IEC, IFCC, ISO, IUPAC, IUPAP, OIML, 2008] [7]

3.7.5. fault [VIML 5.12]

difference between the error of indication and the intrinsic error of a measuring 

instrument.

(For notes, please refer to VIML)

3.7.6. fault detection output 

electrical representation issued by the load cell indicating that a fault condition exists. 

3.7.7. hysteresis error 

difference in load cell output readings for the same applied force between the reading 

obtained by increasing the load from minimum load (Dmin), and the reading obtained 

by decreasing the load from maximum load (Dmax). 

3.7.8. initial intrinsic error [VIML 5.11]

intrinsic error of a measuring instrument as determined prior to performance tests and 

durability evaluations

3.7.9. load cell intrinsic error 

error resulting from a load cell, determined under reference conditions.

3.7.10. maximum permissible error (MPE) [VIM 4.26]

extreme value of measurement error, with respect to a known reference quantity 

value, permitted by specifications or regulations for a given measurement, measuring 

instrument, or measuring system.

(For notes, please refer to VIM)

3.7.11. measurement error [VIM 2.16)]

measured quantity minus a reference quantity value.

Note: The term “measurement error” in this Recommendation refers to load cell 

measurement errors.

(For additional notes, please refer to VIM)

3.7.12. measured quantity (value) [VIM 2.10)]

quantity value representing a measurement result.

(For notes, please refer to VIM)
OIML R 60-1 October 2016

12

3.7.13. non-linearity 

deviation from the average of the values of load cell signals from a straight line 

through zero force applied and maximum force applied.

3.7.14. repeatability error 

difference between load cell output readings taken from consecutive tests under the 

same loading and environmental conditions of measurement.

3.7.15. resolution [VIM 4.14]

smallest change in a quantity being measured that causes a perceptible change in the 

corresponding indication.

(For note, please refer to VIM)

3.7.16. significant durability error [VIML 5.17]

durability error exceeding the value specified in the applicable Recommendation.

(For note, please refer to VIML)

3.7.17. significant fault [VIML 5.14]

fault exceeding the applicable fault limit value.

(For note, please refer to VIML)

3.7.18. span stability 

capability of a load cell to maintain the load cell output of the load cell’s measuring 

range over a period of use within specified limits. 

3.7.19. temperature effect on minimum dead load output

change of the signal output under minimum dead load due to a change in ambient 

temperature.

3.7.20. temperature effect on sensitivity 

change in sensitivity due to a change in ambient temperature.

3.7.21. type approval mark [VIML 3.07]

mark applied to a measuring instrument certifying its conformity to the approved type
October 2016 OIML R 60-1

13

3.8. Influences and reference conditions 

3.8.1. disturbance [VIML 5.19]

influence quantity having a value within the limits specified in the relevant 

Recommendation, but outside the specified rated operating conditions of a measuring 

instrument.

3.8.2. influence factor [VIML 5.17]

influence quantity having a value which ranges within the rated operating conditions 

of a measuring instrument.

(For notes, please refer to VIML)

3.8.3. influence quantity [VIM 2.52]

quantity that, in a direct measurement, does not affect the quantity that is actually 

measured, but affects the relation between the indication and the measurement result.

(For examples and notes, please refer to VIM)

3.8.4. Rated operating condition [VIM 4.9]

operating condition that must be fulfilled during measurement in order that a 

measuring instrument or measuring system perform as designed.

(For note, please refer to VIM)

3.8.5. reference (operating) condition [VIM 4.11]

operating condition prescribed for evaluating the performance of a measuring 

instrument or measuring system or for comparison of measurement results.

(For notes, please refer to VIM)
OIML R 60-1 October 2016

14

3.9. Abbreviations

AC Alternating Current

CH Cyclic Humidity

DC Direct Current

DR Minimum Deadload Output Return

EMC Electro Magnetic Compatibility

EUT Equipment Under Test

IEC International Electrotechnical Committee

ISO International Organization for Standardization 

I/O Input/Output

LC Load Cell

MPE Maximum Permissible Error

NH No Humidity

OIML International Organization of Legal Metrology

SH Steady-State Humidity

VIM International Vocabulary of Metrology – Basic and General Concepts and Associated 

Terms

4. Description of Load Cells

A load cell provides an output proportional to a force resulting from applying a load. Load cells 

may be used as a single transducer or applied together with other load cells in a system where the 

design allows such application. The term “load cell” in this Recommendation is not limited to 

any particular type of technology or design principle.

While many technologies are used in the design of load cells, those used in legal metrology 

applications are commonly designed to to provide an electrical output relative to a mechanical 

input. Both analog and digital outputs are recognized in load cells within that category. 

Although strain gauge technology was a primary focus in the development of R60, it is to be 

understood that load cells that operate using other principles may also be evaluated under this 

Recommendation. Variations of transducers that operate using alternative basis of input/output 

may include, but are not limited to: pressure (e.g., hydraulic, pneumatic); vibratory frequency; 

and magnetic forces.

The term load cell may describe an elemental component/module or a somewhat more complex 

instrument including constituents that perform functions such as signal filtering and analog-todigital conversion.
October 2016 OIML R 60-1

15

5. Metrological requirements

5.1. Principle of load cell classification

The classification of load cells into specific accuracy classes is provided to facilitate their 

application to various measuring systems. In the application of this Recommendation, it should 

be recognized that the effective performance of a particular load cell may be improved by 

compensation means within the measuring system with which it is applied. Therefore, it is not 

the intention of this Recommendation to require that a load cell be of the same accuracy class for 

a load cell as the measuring system in which it may be applied. Nor does it require that a 

measuring instrument, indicating in units of mass for example, use a load cell which has been 

approved during a separate type evaluation. All data/items found in 5.1.1 to 5.1.7 shall be 

specified by the manufacturer

5.1.1. Accuracy classes and their symbols

Load cells shall be ranked, according to their overall performance capabilities, into

one of the four accuracy classes whose designations are as follows: 

Class A; Class B; Class C; Class D.

5.1.2. Maximum number of load cell verification intervals

The maximum number of load cell verification intervals, nLC, into which the 

maximum measuring range Emax – Emin (see 3.5.8) can be divided in a measuring 

system shall be within the limits presented in Table 1. 

Maximum Number of Load Cell Verification Intervals (nLC) according to 

accuracy class.

Class A Class B Class C Class D

Lower Limit 50 000 5 000 500 100

Upper Limit Unlimited 100 000 10 000 1 000

Table 1

5.1.3. Minimum load cell verification interval 

The minimum load cell verification interval, vmin, shall be specified by the 

manufacturer (see 3.5.11 in combination with 3.5.15).
OIML R 60-1 October 2016

16

5.1.4. Supplementary classifications 

Load cells shall also be classified by the intended manner in which a load is applied 

to the load cell wherever there would be a risk of confusing the manner of loading 

(i.e., compression loading, tension loading or, universal). A load cell may bear 

different classifications according to the intended manner in which a load is applied 

to the load cell. The manner of loading for which the classification(s) applies(y) shall 

be specified. For multiple capacity load cells, each capacity shall be classified 

separately. 

5.1.5. Complete load cell classification 

The load cell shall be classified corresponding to the following six parameters: 

a) accuracy class designation (see 5.1.1 and 6.2.4.1); 

b) maximum number of load cell verification intervals (see 5.1.2 and 6.2.4.5); 

c) intended manner of the application of the load, if necessary (see 5.1.4 and 

6.2.4.2); 

d) special limits of working temperature, if applicable (see 6.2.4.3); 

e) humidity symbol, if applicable (see 6.2.4.4); and 

f) additional characterization information, as listed below in Figure 3, 5.1.6, and 

5.1.7.

An example illustrating the six parts of the load cell classification is shown in 

Figure 3. 

For example: C 3 NH ↑ -5/30

Accuracy class

1.

2. SH

3. CH or no

Special

e.g. – 5/30

– 5 °C to + 30 °C

Note: This is only

when the

Maximum number of load cell

intervals, stated in units of 1 000:

e.g. 3 represents 3 000

1.5 represents 1 500

limits are different

From -10°C to + 40°C

- Class 

B - Class B 

C - Class C 

D - Class D

↑

↓

↓

↑

↓ or ↑

↑ ↓

↑ ↑
October 2016 OIML R 60-1

17

Figure 3. Complete load cell classification

5.1.6. Standard classification 

Standard classifications shall be used; examples are shown in Table 2.

Classification 

symbol Description

C2 Class C, 2 000 intervals

C3 ↓ 5/35 Class C, 3 000 intervals, compression, + 5 °C to + 35 °C ↑

C2 NH Class C, 2 000 intervals, not to be subjected to humidity test

Table 2. Examples of load cell classification

5.1.7. Multiple classifications

Load cells that have comprehensive classifications for the manner in which the load is 

applied to the load cell shall be accompanied by the relative information for each 

classification. Example is shown in Table 3. An illustration of the standard 

classification symbols, using an example is shown in Figure 3.

Classification 

Symbol Description

C2 ↑ Class C, 2 000 intervals

C1.5 ↓ Class C, 1 500 intervals

C1 ↓

↑ - 5/30 Class C, 1 000 intervals, compression, – 5 °C to + 30 C

C3 ↑

↓ - 5/30 Class C, 3 000 intervals, tension, – 5°C to + 30 °C

Table 3. Examples of Multiple Classifications

5.2. Measuring ranges

5.2.1. Minimum load of the measuring range (Dmin) (see 3.5.12)

The value of the smallest load applied to a load cell during test which is expressed in 

units of mass shall not be less than Emin (see 3.5.9).

5.2.2. Maximum load of the measuring range (Dmax) (see 3.5.6)

The value of the largest load applied to a load cell during test which is expressed in 
OIML R 60-1 October 2016

18

units of mass shall not be greater than Emax (see 3.5.5). 

5.3. Maximum permissible measurement errors

Under the rated operating conditions in 5.6, the maximum permissible error (MPE) shall not 

exceed the values stated in 5.5

The MPE is applicable after increasing as well as decreasing the force applied (i.e., they include 

hysteresis).

5.3.1. Maximum permissible errors for each accuracy class 

The maximum permissible measurement errors for each accuracy class are related to 

the maximum number of load cell verification intervals (nLC) specified for the load 

cell (see 5.1.2) and to the actual value of the load cell verification interval, v. 

5.3.2. Type evaluation

The MPE (see 3.7.10) on type evaluation shall be the values derived using the 

expressions contained in the left column of Table 4. The apportioning factor, pLC

shall be chosen and declared (if other than 0.7) by the manufacturer and shall be in 

the range of 0.3 to 0.8 

(0.3 ≤ pLC ≤ 0.8)1. 

Where “m” is the value (expressed in mass) representing the force introduced by the 

load applied

MPE

Load, m

Class A Class B Class C Class D

pLC × 0.5 v

pLC × 1.0 v

pLC × 1.5 v

0 ≤ m ≤ 50 000 v

50 000 v < m ≤ 200 000 v

200 000 v < m

0 ≤ m ≤ 5 000 v

5 000 v < m ≤ 20 000 v

20 000 v < m ≤ 100 000v

0 ≤ m ≤ 500 v

500 v < m ≤ 2 000 v

2 000 v < m ≤ 10 000 v

0 ≤ m ≤ 50 v

50 v < m ≤ 200 v

200 v < m ≤ 1 000 v

Table 4. Maximum Permissible Errors (MPE) on Type Evaluation

The value of the apportioning factor, pLC shall appear on the OIML certificate, if the 

value is not equal to 0.7. If the apportioning factor, pLC is not specified on the 

certificate then the value 0.7 shall be assumed. The maximum permissible error may 

be positive or negative and is applicable to both increasing and decreasing loads.

 1

Associated with apportionment of error provisions contained within OIML R 76-1, 3.5.1 and 3.10.2.1 [1]; R 50-1, 2.2.3 [28]; R 51-1, 5.2.3.4 [28]; R 61-1, 

5.2.3.3 [28]; R 106-1, 5.1.3.2 [28]; or R 107-1, 5.1.4.1 [28], when load cell is applied to such instruments.
October 2016 OIML R 60-1

19

The limits of error shown include errors due to nonlinearity, hysteresis and 

temperature effect on sensitivity over certain temperature ranges, specified in 5.6.1.1 

and 5.6.1.2. Further errors, not included in the Table 4 limits of error, are treated 

separately.

5.4. Repeatability error

The maximum difference between the results of five identical load applications for classes A and 

B and of three identical load applications for classes C and D shall not be greater than the 

absolute value of the MPE for that load.

5.5. Permissible variation of results under reference conditions

5.5.1. Creep 

The difference between the reading taken upon the application of a maximum load 

(Dmax) and the reading observed within and after 30 minutes of exposure of 90% to 

100% of Emax shall not exceed 0.7 times the absolute value of MPE for the applied 

load.*

Example: 

Load cell class: C3 (declared by the manufacturer)

Apportioning factor: pLC = 0.7 (declared by manufacturer)

Applied load: Dmax = Emax (test specification)

Maximum difference between the reading = 0.7 × (0.7 × 1.5v) = 0.735

The difference in readings taken after 20 minutes of exposure to 90% to 100% of Emax

and at 30 minutes of exposure to 90% to 100% of Emax shall not exceed 0.15 times the 

absolute value of MPE.

Example: 

Load cell class: C3 (declared by the manufacturer)

Apportioning factor: pLC = 0.7 (declared by manufacturer)

Applied load: Dmax = Emax (test specification)

Maximum difference between the initial reading = 0.15 × (0.7 × 1.5v) = 0.1575

*Regardless of any value declared by the manufacturer for the apportioning factor, pLC, the 

MPE for creep shall be determined from Table 4 using the apportioning factor, pLC = 0.7.

5.5.2. Minimum dead load output return (DR) (see 3.5.10)

The difference between the initial reading of the minimum load output (Dmin) and the reading 

of Dmin at the conclusion of the creep test (5.5.1), shall not exceed half the value of the load 

cell verification interval (0.5 v).
OIML R 60-1 October 2016

20

Note:

It should be noted that DR is the minimum dead load output return expressed in mass units 

(g, kg, t). DR has to be adjusted in a value expressed in load cell verification intervals v.

5.6. Influence quantities (Rated operating conditions)

Load cells are to be evaluated under the conditions specified in 5.6.1 - 5.6.3. An evaluation may 

include additional special testing performed under conditions that vary from those specified in 

5.6.1 - 5.6.3 if requested and specified by the applicant submitting the load cell for evaluation. 

This special testing may be performed in addition to, but not instead of testing under the 

specified conditions in 5.6.1 – 5.6.3. 

Load cells that are equipped with functions typically performed by complete instruments (may 

be required to be evaluated against additional requirements contained in other OIML 

Recommendations for those complete instruments. These additional evaluations are outside the 

scope of this Recommendation (see 2.3. and Figure 1).

5.6.1. Temperature

5.6.1.1. Temperature limits 

Excluding temperature effects on minimum dead load output, the load cell shall perform 

within the limits of error in 5.3.2 over the temperature range of – 10 °C to + 40 °C, unless 

otherwise specified as in 5.6.1.2 below. 

Note: National legislation may prescribe alternate temperature limits outside of the range 

specified above as appropriate for local climatic conditions and the environmental 

conditions that can be anticipated.

5.6.1.2. Special limits 

Load cells for which particular limits of working temperature are specified shall satisfy, 

within those ranges, the conditions defined in 5.3.2. The span of these ranges shall be at 

least: 

5 °C for load cells of class A;

15 °C for load cells of class B;

30 °C for load cells of classes C and D.

5.6.1.3. Temperature effect on minimum dead load output 

The minimum dead load output of the load cell over the temperature range, as specified 

in 5.6.1.1 or 5.6.1.2, shall not vary by an amount greater than the apportioning factor, pLC, 

times the minimum load cell verification interval, vmin, for any change in ambient 

temperature of:

2 °C for load cells of class A; 

5 °C for load cells of class B, C and D.

5.6.2. Barometric pressure 

The output of the load cell shall not vary by an amount greater than the minimum load cell 
October 2016 OIML R 60-1

21

verification interval, vmin, for any incremental change in barometric pressure equivalent to 1 

kPa.

5.6.3. Humidity

With respect to humidity conditions, this Recommendation defines 3 humidity classes: CH

(cyclic humidity - as standard), NH (no humidity), and SH (steady-state humidity). In case of 

class NH, or SH, the class designation shall be marked on the load cell. In the case of class 

CH, class designation marking of the load cell is not mandatory.

5.6.3.1. Humidity error – CH or unmarked load cells

This requirement is only applicable to load cells marked CH or with no humidity symbol 

marking and not applicable to load cells marked NH or SH.

The influence of exposure to temperature cycles specified in 8.10.5.12 on the load cell 

output for minimum load shall not be greater than 4 % of the difference between the 

output on the maximum capacity, Emax, and that at the minimum dead load Emin. 

The influence of exposure to temperature cycles specified in 8.10.5.12 on the load cell 

output for the maximum load shall not be greater than the load cell verification interval 

v.

5.6.3.2. Humidity error – SH marked load cells

This requirement is only applicable to load cells marked SH and not applicable to load 

cells marked NH or CH or with no humidity symbol marking.

A load cell shall meet the MPE applicable to the load applied as specified in Table 4,

when exposed to conditions of relative humidity variations as specified in 8.10.6.11

5.7. Requirements for digital load cells

5.7.1. General requirements

In addition to the other requirements of this Recommendation, a digital load cell shall 

comply with the following requirements. The MPE shall be determined using an 

apportioning factor, pLC greater than or equal to 0.7 and lesser than or equal to 0.9 (0.7 ≤ pLC

≤ 0.9) substituted for the apportioning factor, pLC, that is declared by the manufacturer and 

applied to the other requirements.

If a load cell is configured with substantial additional electronic functions (e.g., display of 

indications, frequency counter) that are typical of an electronic weighing instrument, it may 

be considered outside the scope of this Recommendation and need to undergo additional 

evaluation using requirements contained in other OIML Recommendations which are 

applicable to weighing instruments.
OIML R 60-1 October 2016

22

5.7.1.1. Faults

A digital load cell shall be designed and manufactured such that when it is exposed to 

electrical disturbances either:

a) significant faults do not occur; or

b) significant faults are detected and acted upon.

If significant faults do occur, and the load cell is equipped with the intelligence to detect 

and act upon significant faults through the instrument that the load cell is installed in, the 

reporting of and acting upon significant faults would then be evaluated under the 

appropriate Recommendation for the complete instrument.

Messages of significant faults should not be confused with other messages presented.

Note: A fault, in value that is equal to or smaller than the load cell verification interval, v, 

is allowed. 

5.7.1.2. Acting upon significant faults 

When a significant fault has been detected, either the load cell shall be made inoperative 

automatically or a fault detection output shall be issued automatically. This fault 

detection output shall continue until fault has been resolved. 

5.7.1.3. Durability 

The load cell shall be suitably durable so that the requirements of this Recommendation 

may be met in accordance with the intended use of the load cell. 

5.7.1.4. Compliance with requirements 

A digital load cell is presumed to comply with the requirements in 5.7.1.1 and 5.7.1.3, if 

it passes the examinations specified in 5.7.2 and 8.10.7 

5.7.1.5. Application of the requirements in 5.7.1.1

The requirements in 5.7.1.1 may be applied separately to each individual cause or 

significant fault. The choice of whether 5.7.1.1 a) or 5.7.1.1 b) is applied is left to the 

manufacturer. 

5.7.2. Functional requirements

5.7.2.1. Warm-up time 

During the design warm-up time of a digital load cell there shall be no transmission of measurement results. 
October 2016 OIML R 60-1

23

5.7.2.2. Mains power supply (AC) 

A digital load cell that operates from a mains power supply shall be designed to comply with 

the metrological requirements if the mains power supply varies in voltage from – 15 % to + 

10 % of the supply voltage.

5.7.2.3. Battery power supply (DC) 

A digital load cell that operates from a battery power supply shall either continue to function 

correctly or not provide a measurement result whenever the voltage is below the value 

specified by the manufacturer.

5.7.2.4. Maximum allowable variations during voltage variations:

All functions shall operate as designed.

All measurement results shall be within maximum permissible errors.

5.7.2.5. Disturbances 

When a digital load cell is subjected to the disturbances specified in 8.10.7.5 to 8.10.7.10 

(also summarized in Table 5), the difference between the load cell output due to a 

disturbance and the load cell output without disturbance (fault) shall satisfy the 

conditions in 5.7.1.1. 

Table 5. 

Performance and Stability Tests for a Digital Load Cell

Test Section 8.10 test 

procedure pLC

Characteristic 

under test

Warm-up time 8.10.7.3

1.0

Influence factor

Power voltage variations 8.10.7.4 Influence factor

Short-time power reductions 8.10.7.5 Disturbance

Bursts (electrical fast transients) 8.10.7.6 Disturbance

Surge 8.10.7.7 Disturbance

Electrostatic discharge 8.10.7.8 Disturbance

Electromagnetic susceptibility 8.10.7.9 Disturbance

Immunity to conducted 

electromagnetic fields 8.10.7.10 Disturbance

Span stability 8.10.7.11 Influence factor
OIML R 60-1 October 2016

24

5.7.2.6. Span stability: maximum allowable variation requirements (not applicable to 

class A load cells) 

When a digital load cell is subjected to the span stability test specified in 8.10.7.11, the 

variation in the load cell span measurement results shall not exceed the greater of: half 

the load cell verification interval; or half the absolute value of the MPE for the applied 

test load Dmax. 

6. Technical Requirements

6.1. Software

Provision shall be made for appropriate sealing by mechanical, electronic and/or cryptographic 

means, making any change that affects the metrological integrity of the device impossible or 

evident.

Any embedded programming (i.e., firmware) that influences the raw count output of the load cell 

will be evaluated under the terms of this Recommendation. In addition, if the software modifies 

load cell performance, not exceeding the functions of analog to digital conversion and the 

linearization of the load cell output, then that software shall be evaluated under the terms in this 

Recommendation and in accordance with OIML D31 Edition 2008(E) [8] Any weighing 

instrument function shall be evaluated under other appropriate Recommendations for weighing 

instruments. 

Functionality of any software which is not covered by this Recommendation, e. g. functionalities 

of weighing instruments, is outside the scope of this Recommendation and not evaluated. It may 

be required to undergo additional evaluations against other requirements contained in the 

applicable OIML Recommendations for weighing instruments.

The requirements which are relevant to the evaluation of load cells and provided in OIML D 31 

Edition 2008 (E) [8] have to be fulfilled for the load cell by taking into account the following 

aspects.

a). In general, for load cells, the severity level I, examined with validation procedure A, is 

required.

b). For legally relevant software of digital load cells the following statements according to 

OIML D31 shall be applied.

1) The exception described in D 31, 5.1.1 [8] for an imprint of the software 

identification is allowed.

2) The level of conformity of manufactured devices to the approved type is according to 

D 31, 5.2.5 (clause a) [8]. 

3) Updating the legally relevant software of a load cell in the field is possible via 

verified or traced update according to D31, 5.2.6.2 and 5.2.6.3 [8]
October 2016 OIML R 60-1

25

4) The software documentation shall include descriptions according to the applicable 

requirements of D 31, 6.1.1 [8]

The validation procedures are described in D 31, 6.4 [8]

6.2. Inscriptions and presentation of load cell information

Technical information markings including load cell classifications as indicated in 5.1.5 must be 

specified for the load cell(s).

6.2.1. Mandatory markings on the load cell 

The following mandatory markings shall be clearly an indelibly marked on the load cell:

a. Manufacturer’s name or trade mark

b. Manufacturer’s type designation or load cell model

c. Serial number

d. Maximum capacity as: Emax = (in units g, kg, t, )

e. Year of production

f. OIML certificate number (if applicable)

g. Type approval mark

If due to the limitation of the size of the load cell, it is impossible to apply all mandatory 

markings, the manufacturer’s name or trade mark, the load cell type designation, the serial 

number, and the maximum capacity shall be provided as a minimum on the load cell itself. 

All other mandatory information shall be provided in an accompanying document supplied

by the manufacturer and submitted to the user. Where such a document is provided, the 

information required in 6.2.2 shall also be given therein. 

6.2.2. Mandatory additional information 

The following mandatory information shall be provided in a document accompanying the 

load cell supplied by the manufacturer and submitted to the user (or, if space permits, they 

may be marked on the load cell). Where the information provided is associated with a 

specific unit of measure, the unit (g, kg, t,) shall also be specified.

a. Manufacturer’s name or trade mark

b. Type designation

c. Accuracy class(es); see 6.2. 4.1

d. Type of load; see 6.2.4.2

e. Working temperature when required; see 6.2.4.3

f. Humidity symbol when required; see 6.2.4.4

g. Maximum capacity as: Emax =

h. Minimum dead load as: Emin =
OIML R 60-1 October 2016

26

i. Safe load limit as: Elim =

j. Minimum load cell verification interval as vmin (or relative minimum dead load 

verification interval Y) =

k. Value of the apportioning factor, pLC, if not equal to 0.7; and

l. Other pertinent conditions that must be observed to obtain the specified performance 

(for example, electrical characteristics of the load cell such as output rating, input 

impedance, supply voltage, cable details, mounting torque, etc.)

6.2.3. Non-mandatory additional information 

In addition to the information required in 6.2.2, the following information may optionally be 

specified: 

a. for a weighing instrument (for example a multiple range instrument according to 

OIML R 76) [1], the relative vmin, Y, where Y = Emax / vmin (see 3.5.15);

b. for a weighing instrument (for example a multi-interval instrument according to 

OIML R 76) [1], the relative DR, Z, where Z= Emax / (2 × DR) (see 3.5.14) and the 

value of DR (see 3.5.10) is set at the maximum permissible minimum dead load 

output return according to 8.10.1.

c. other information considered necessary or useful by the manufacturer.
October 2016 OIML R 60-1

27

6.2.4. Specific markings

6.2.4.1. Accuracy class designation 

Class A load cells shall be designated by the character “A”, class B by “B”, class C by 

“C” and class D by the character “D”. 

6.2.4.2. Designation of the type of load applied to the load cell 

The designation of the type of load 

applied to the load cell shall be specified 

when it is not clearly apparent from the 

load cell construction, using the symbols 

shown in Table 6.Tension

↑

↓

Compression ↓

↑

Beam ↑ or ↓

Universal ↑ ↓

↓ ↑

Table 6. 

Symbols for Different Types of Load Transmission Principles

6.2.4.3. Working temperature designation 

The special limits of working temperature, as referred to in 5.6.1.2, shall be specified 

when the load cell cannot perform within the limits of error in 5.3.to 5.6 over the 

temperature range specified in 5.6.1.1. In such cases, the limits of temperature shall be 

designated in degrees Celsius (°C).

6.2.4.4. Humidity symbols

a). A load cell not designed to meet performance criteria evaluated under 8.10.5 or 

8.10.6 shall be marked by the symbol NH. 

b). A load cell submitted for evaluation and designed to meet performance criteria 

evaluated under 8.10.5 shall be marked by the symbol CH or not be marked with 

any humidity classification.

c). A load cell submitted for evaluation and manufactured to meet performance 

criteria evaluated under 8.10.6 shall be marked by the symbol SH. 

6.2.4.5. Maximum number of load cell verification intervals 

The maximum number of load cell verification intervals for which the accuracy class 

applies shall be designated in actual units (e.g., 3 000) or, when combined with the 
OIML R 60-1 October 2016

28

accuracy class designation (see 6.2.4.1 above) to produce a classification symbol (see 

5.1.6), it shall be designated in units of 1 000. 
OIML R 60-1 Annex A

A-1

Annex A (Mandatory) Definitions from other applicable international 

publications

A.1 Definitions from OIML D 11 [4]

A.1.1 Electronic measuring instrument (OIML D 11, 3.1)

instrument intended to measure an electrical or non-electrical quantity using electronic means 

and/or equipped with electronic devices .

A.1.2 Module (OIML D11, 3.2)

device performing a specific function or functions and (usually) manufactured and constructed 

such that it can be separately evaluated according to prescribed metrological and technical 

performance requirements.

A.1.3 Device (OIML D 11, 3.3)

identifiable instrument or part of an instrument or of a family of instruments that performs a 

specific function or functions

A.1.4 Checking facility (OIML D11, 3.19)

facility incorporated in a measuring instrument which enables significant faults to be detected 

and acted upon

A.1.5 Automatic checking facility (OIML D 11, 3.19.1)

checking facility that operates without the intervention of an operator.

A.1.6 Permanent automatic checking facility (type P) (OIML D 11, 3.19.1.1)

automatic checking facility that operates at each measurement cycle.

A.1.7 Intermittent automatic checking facility (type I) (OIML D 11, 3.19.1.2)

automatic checking facility that operates at certain time intervals or per fixed number of 

measurement cycles.

A.1.8 Non-automatic checking facility (type N) (OIML D 11, 3.19.2)

checking facility that requires the intervention of an operator.
Annex A OIML R 60-1

A-2

A.1.9 Durability protection facility (OIML D 11, 3.20)

facility incorporated in a measuring instrument that enables significant durability errors to be 

detected and acted upon.

A.1.10 Test (OIML D 11, 3.21)

series of operations intended to verify the compliance of the equipment under test (EUT) with 

specified requirements.

A.1.11 Test procedure (OIML D 11, 3.21.1)

detailed description of the test operations.

A.1.12 Performance test (OIML D 11, 3.21.4)

test intended to verify whether the EUT is able to accomplish its intended functions

A.1.13 Mains power (OIML D 11, 3.22)

primary external source of electrical power for an instrument, including all sub-assemblies. 

(Examples: public or local power grid (AC or DC) or external generator.

A.1.14 Power converter (power supply device) (OIML D 11, 3.23)

sub-assembly converting the voltage from the mains power to a voltage suitable for other subassemblies.

A.1.15 Auxiliary battery (OIML D 11, 3.25)

battery that is:

• mounted in, or connected to, an instrument that can be powered by the mains power as 

well; and

• capable of supplying power to the complete instrument for a reasonable period of time.

A.1.16 Back-up battery (OIML D 11, 3.26)

battery that is intended to maintain power supply for specific functions of an instrument in the 

absence of the primary power supply 

Example: To preserve stored data. 

A.2 Definitions from OIML R76 [1]
OIML R 60-1 Annex A

A-3

A.2.1 Weighing Module [T.2.2.7]

Part of the weighing instrument that comprises all mechanical and electronic devices (i.e. load 

receptor, load-transmitting device, load cell, and analog data processing device or digital data 

processing device) but not having the means to display the weighing result. It may optionally 

have devices for further processing (digital) data and operating the instrument.

OIML R 60-1 Annex B

B-1

Annex B (Mandatory) OIML Certificate of conformity for load 

cells - Format of certificate

Member State OIML CERTIFICATE OF CONFORMITY OIML certificate no. 

Issuing Authority 

Name: 

...............................................................................................................................................................................

Address: 

....................................................................................................................................................................... Person respon

....................................................................................................................................................................................

Applicant 

Name: 

............................................................................................................................................................................... 

Address: 

............................................................................................................................................................................ 

Manufacturer of the certified type (if the manufacturer is not the applicant) ...............................................................

Identification of the certified type: Load cell (construction principle, e.g., strain gauge, compression).....................

.............................................................................................................................................................................................................

Model designation 

Maximum capacity, Emax 

Accuracy class 

Maximum number of load cell 

verification intervals, nmax 

Minimum verification interval, vmin 

Apportioning factor, pLC

(Additional characteristics and identification, as applicable according to R 60, 3.4.2 and 5.1.5, continued overleaf or on 

addendum if necessary) 
Annex B OIML R 60-1

B-2

This certificate attests the conformity of the above-mentioned type (represented by the samples identified in the associated test 

report(s) with the requirements of the following Recommendation of the International Organization of Legal Metrology -

OIML): 

R 60 Metrological regulation for load cells Edition ....................... for accuracy class ...............................................

This certificate relates only to the metrological and technical characteristics of the type of instrument

concerned, as covered by the relevant OIML International Recommendation.

This certificate does not bestow any form of legal international approval.

The conformity was established by tests described in the associated test report no. ......, which includes ....... pages.

Identification(s) and signature(s) or stamp(s), of (as applicable): 

Issuing Authority: CIML Member: 

Date: ....................................... Date: .......................................
OIML R 60-1 Annex B

B-3

Page 1. This certificate includes ....... pages.

OIML certificate no. 

* * * 

Additional characteristics1 and identification, as applicable to R 60, 3.4.2 and 5.1.5 (continued)

Model designation 

(Additional characteristics, per 3.4.2 

and 5.1.5) 

Special conditions: 

..................................................................................................................................................

..................................................................................................................................................

....................................................................

................................................................................................................................................

................................................................................................................................................

................................................................................................................................................

................................................................................................................................................

................................................................................................................................................

................................................................................................................................................

................................................................................................................................................

................................................................................................................................................

Important note:

Apart from the mention of the certificate’s reference number and the name of the OIML Member State in which 

the certificate was issued, partial quotation of the certificate or the associated test report is not permitted, though 

they may be reproduced in full. 

1 

The table with the essential technical data may, upon request by the manufacturer, be placed on the certificate or on an addendum. 
Annex B OIML R 60-1

B-4

Page 2. This certificate includes ....... pages.

B.1. Contents of addendum to certificate (Informative) 

Addendum to certificate no. ............................................................................................................................

(Name and type of the load cell) ...........................................................................................................................................

B.2. Technical data 

The essential technical data for the OIML certificates of conformity are listed on the certificate (at the request of the 

manufacturer) alternatively, in the case of limited space on the certificate the following information may be provided: 

Table B.1 Technical data

Model designation Designation Example Units

Classification C4

Additional markings –

Maximum number of load cell 

verification intervals nLC 4 000

Maximum capacity Emax 30 000 kg

Minimum dead load, relative Emin / Emax 0 %

Relative vmin (ratio to minimum load cell 

verification interval) Y = Emax / vmin 24 000

Relative DR (ratio to minimum dead 

load output return) Z = Emax / (2 × DR) 7 500

Rated output* 2.5 mV/V*

Maximum excitation voltage 30 V

Input impedance (for strain gauge load 

cells) RLC 4 000 Ω

Temperature rating – 10/+ 40 °C

Safe overload, relative Elim / Emax 150 %

Cable length 3 m

Additional characteristics per 3.4.2 and 

5.1.5** –

* Note: For load cells with digital output this refers to the number of counts for Emax 

** Note: For load cells with digital output this is not required 
OIML R 60-1 Annex B

B-5

Page 1. The Annex to the certificate includes ....... pages.

OIML R 60-1 Annex C

C-1

Annex C (Informative) OIML Certificate of conformity for load 

cells

This Annex is provided as an example of supplemental information that may be included in the OIML 

Certificate format and is intended to compliment the OIML Certificate format found in Annex B.

Certificate history

Certificate release Date Essential changes

xxx xxx primary certificate

1. Technical Data

The metrological characteristics of the load cells type xxx are listed in Table C1. Further technical data are 

listed in the data sheet of the manufacturer at page 5 to 6 of this annex.

Table C1: Essential data

Accuracy class C3

Maximum number of load cell 

intervals 

nLC 3000

Rated output mV/V 2

Maximum capacity Emax kg 150 / 200 / 250 / 300 / 500 / 750

Minimum load cell verification 

interval

vmin = (Emax / Y) kg Emax / 15000

Minimum dead load output 

return

DR = (1⁄2 Emax / Z) kg 1⁄2 Emax / 5000

Dead load: xxx%·Emax; Safe overload: xxx%·Emax; Input impedance: xxx Ω

2. Tests

The determination of the measurement error, the stability of the dead load output, repeatability and creep in the 

temperature range of –10°C to +40°C as well as the tests of barometric pressure effects and the determination 

of the effects of static damp heat have been performed according to OIML R60 (2015) as shown in Table C2 

on the load cell nominated in the test report with the reference No. xxx, dated xxx.
Annex C OIML R 60-1

C-2

Table C2: Tests performed 

Test R60 (2015) tested 

samples

result

Temperature test and repeatability at 

(20 / 40 / -10 / 20°C)

5.3.1.1; 

5.4

8.10.1 150 kg +

Temp. effect on minimum dead load output at 

(20 / 40 / -10 / 20°C)

5.6.1.3 8.10.1.

16

150 kg +

Creep test at 

(20 / 40 / -10 / 20°C)

5.5.1 8.10.2 150 kg +

Minimum dead load output return at 

(20 / 40 / -10 / 20°C)

5.5.2 8.10.3 150 kg +

Barometric pressure effects at ambient temperature 5.6.2 8.10.4 150 kg +

Damp heat test , static, marked SH 5.6.3.2 8.10.6 150 kg +

3. Description of the load cell 

{Example}

The load cells (LC) of the series xxx are double bending beam load cells. They are made of aluminium, the 

strain gauge application is hermetically sealed. Further essential characteristics are given in the data sheet, 

see chapter 6 of this annex. 

Picture of load cell

Figure 1: Load cell type xxx

The complete type designation is indicated as follows in the example on the name plate:

Picture of name plate

Figure 2: Name plate

4. Documentation

{Example}

− Test Report No. xxx; C3; Y=xxx; Z=xxx; Emax=xxx kg; SN: xxx

− Datasheet No. Xxx

− Technical Drawing No. Xxx

5. Further information

The manufacturing process, material and sealing (i.e., environmental protection) of the produced load cells 

have to be in accordance with the tested patterns; essential changes must be identified and communicated to 

the issuing authority and are only allowed with the permission of the issuing authority based on the impact of 

those changes on the certification process.
OIML R 60-1 Annex C

C-3

Sufficient information shall be included to describe the patent design.

The typical errors related to linearity, hysteresis and temperature coefficient as indicated in the data sheet 

point out possible single errors of a pattern; however the overall error of each pattern is determined by the 

maximum permissible error according OIML R60 No 5.3.1.1.

The technical data, the dimensions of the load cell and the principle of load transmission are given in chapter 

6 of this annex, have to be complied with. 

6. Data sheet and dimensions 

Specifications of the Load Cell Family

Accuracy class acc. to 

OIML R60

C3

Rated output C mV/V 2,0 ± 0,2

Maximum capacity Emax kg 150 / 200 / 250 / 300 / 500 / 750

Max. number of load cell 

intervals

nLC 3000

Min. load cell verification 

interval

vmin kg Emax / 15000

Minimum dead load output 

return (MDLOR)

DR kg 1⁄2·Emax / 5000

Minimum dead load %·Emax 0

Safe load limit %·Emax 150

Ultimate load %·Emax 300

Excitation voltage, 

recommended

UEXE V 10 – 12 DC

Excitation voltage, maximum V 15 DC

Input resistance RLC Ω 404 ± 10

Output resistance Rout Ω 350 ± 3

Insulation resistance RISO MΩ ≥ 2000

Compensated temperature 

range

T °C - 10 ... + 40

Load cell material Aluminium

Cable length L m 2

Coating Silicone rubber
Annex C OIML R 60-1

C-4

Wiring

The load cell is provided with a shielded 4 or 6 

conductor cable. The cable length is indicated in the 

accompanying document. The shield will be 

connected or not connected to the load cell 

according to customers preference.

Connections

Connections 4-wires 6-wires

 Excitation + red red

 Excitation − black black

 Signal + green green

 Signal − white white

 Sense + -- blue

 Sense − -- yellow

 Shield purple purple

 Cable length 2 m

Picture of the load cell dimensions
OIML R 60-1 Annex D

D-1

Annex D (Informative) Selection of load cell(s) for testing - a 

practical example

D.1. This Annex describes a practical example showing the complete procedure for the selection of test 

samples out of a load cell family. 

D.2. Assume a family consisting of three groups of load cells, differing in class, maximum number of load 

cell verification intervals, nmax, and maximum capacities, Emax. The capacities, Emax, overlap between the 

groups according to the following example: 

Group 1: Class C, nmax = 6 000, Y = 18 000, Z = 6 000 

Emax: 50 kg, 100 kg, 300 kg and 500 kg 

Group 2: Class C, nmax = 3 000, Y = 12 000, Z = 4 000 

Emax: 100 kg, 300 kg, 500 kg, 5 000 kg, 10 t, 30 t and 50 t 

Group 3: Class B, nmax = 10 000, Y = 25 000, Z = 10 000 

Emax: 500 kg, 1 000 kg and 4 000 kg 

D.2.1. Summarize and sort the load cells with respect to Emax and accuracy as follows: 
Annex D OIML R 60-1

D-2

Class Y <--- lowest Emax, kg ---> highest 

nmax

Group Z vmin, kg 

C3 12 000 100 300 500 5 000 10 000 30 000 50 000 

3 000 

2 4 000 0.0083 0.025 0.042 0.42 0.83 2.5 4.17 

C6 18 000 50 100 300 500 

6 000 

1 6 000 0.0028 0.0055 0.0167 0.028 

B10 25 000 500 1 000 4 000 

10 000 

3 10 000 0.020 0.040 0.16 

D.2.2. Identify the smallest capacity load cells in each group to be tested, according to 9.4.4:

Class Y <--- lowest Emax, kg ---> highest 

nmax

Group Z vmin, kg 

C3 12 000 100 300 500 5 000 10 000 30 000 50 000 

3 000 

2 4 000 0.0083 0.025 0.042 0.42 0.83 2.5 4.17 

C6 18 000 50 100 300 500 

6 000 

1 6 000 0.0028 0.0055 0.0167 0.028 

B10 25 000 500 1 000 4 000 

10 000 

3 10 000 0.020 0.040 0.16 
OIML R 60-1 Annex D

D-3

In this example, select and identify:

C6 - 50 kg (full evaluation test required)

B10 - 500 kg (full evaluation test required)

Although load cell C3 - 100 kg is the smallest capacity in its group, its capacity falls within the range of 

other selected load cells having better metrological characteristics. Therefore, it is not selected.

D.2.3. Begin with the group with the best metrological characteristics (in this example, B10) and in accordance 

with 8.4.2, select the next largest capacity between 5 and 10 times that of the nearest smaller capacity 

load cell which has been selected. When no capacity meets this criterion, the selected load cell shall be 

that having the smallest capacity exceeding 10 times that of the nearest smaller capacity load cell which 

has been selected. Continue this process until all load cell capacities in the group have been considered. 

Class Y <--- lowest Emax, kg ---> highest 

nmax

Group Z vmin, kg 

C3 12 000 100 300 500 5 000 10 000 30 000 50 000 

3 000 

2 4 000 0.0083 0.025 0.042 0.42 0.83 2.5 4.17 

C6 18 000 50 100 300 500 

6 000 

1 6 000 0.0028 0.0055 0.0167 0.028 

B10 25 000 500 1 000 4 000 

10 000 

3 10 000 0.020 0.040 0.16 

In this example, select and identify: 

B10 - 4 000 kg (full evaluation test required)

D.2.4. Move to the group with the next best characteristics (in this example, C6) and, in accordance with 8.4.2

select the next largest capacity between 5 and 10 times that of the nearest smaller capacity load cell 

which has been selected. When no capacity meets this criterion, the selected load cell shall be that 
Annex D OIML R 60-1

D-4

having the smallest capacity exceeding 10 times that of the nearest smaller capacity load cell which has 

been selected. Continue this process until all load cell capacities in the group have been considered. 

Class Y <--- lowest Emax, kg ---> highest 

nmax

Group Z vmin, kg 

C3 12 000 100 300 500 5 000 10 000 30 000 50 000 

3 000 

2 4 000 0.0083 0.025 0.042 0.42 0.83 2.5 4.17 

C6 18 000 50 100 300 500 

6 000 

1 6 000 0.0028 0.0055 0.0167 0.028 

B10 25 000 500 1 000 4 000 

10 000 

3 10 000 0.020 0.040 0.16 

In this example, there is no change to the load cells selected. The capacities of the load cells C6 

- 300 kg and C6 - 500 kg exceed the capacity of the load cell C6 - 50 kg by greater than 5 

times but not greater than 10 times. However, a 500 kg load cell of better metrological 

characteristics (from group B10) has already been selected. Therefore, in order to minimize the 

number of load cells to be tested according to 8.3.1, neither cell is selected. 

D.2.5. Again, and repeating this process until all groups have been considered, move to the group with the next 

best characteristics (in this example, C3) and in accordance with 8.4.4, select the next largest capacity 

between 5 and 10 times that of the nearest smaller capacity load cell which has been selected. When no 

capacity meets this criterion, the selected load cell shall be that having the smallest capacity exceeding 

10 times that of the nearest smaller capacity load cell which has been selected. Continue this process 

until all load cell capacities in the group and all groups have been considered. 
OIML R 60-1 Annex D

D-5

Class Y <--- lowest Emax, kg ---> highest 

nmax

Group Z vmin, kg 

C3 12 000 100 300 500 5 000 10 000 30 000 50 000 

3 000 

2 4 000 0.0083 0.025 0.042 0.42 0.83 2.5 4.17 

C6 18 000 50 100 300 500 

6 000 

1 6 000 0.0028 0.0055 0.0167 0.028 

B10 25 000 500 1 000 4 000 

10 000 

3 10 000 0.020 0.040 0.16 

In this example, select and identify:

C3 - 30 000 kg (full evaluation test required) Proceeding from smallest to largest capacity, the only 

capacity of load cell which is greater than 5 times the capacity of an already selected load cell but less 

than 10 times that capacity is the C3 - 30 000 kg load cell. Since the capacity of the C3 - 50 000 kg load 

cell does not exceed 5 times the capacity of the next smaller selected load cell, which is C3 - 30 000 kg, 

according to 8.4.3 it is presumed to comply the requirements of this Recommendation. 

D.2.6. After completing steps D.2.2 to D.2.5 and identifying the load cells, compare load cells of the same 

capacity from different groups. Identify the load cells with the highest accuracy class and highest nmax

in each group (see shaded portion of table below). For those load cells of the same capacity but from 

different groups, identify only the one with the highest accuracy class and nmax and lowest vmin. 
Annex D OIML R 60-1

D-6

Class Y <--- lowest Emax, kg ---> highest 

nmax

Group Z vmin, kg 

C3 12 000 100 300 500 5 000 10 000 30 000 50 000 

3 000 

2 4 000 0.0083 0.025 0.042 0.42 0.83 2.5 4.17 

C6 18 000 50 100 300 500 

6 000 

1 6 000 0.0028 0.0055 0.0167 0.028 

B10 25 000 500 1 000 4 000 

10 000 

3 10 000 0.020 0.040 0.16 

Inspect the values of vmin, Y, and Z for all cells of the same capacity. 

If any load cell of the same capacity has a lower vmin or higher Y than the identified load cell, 

that load cell (or load cells) is also liable for partial evaluation testing, specifically the conduct of 

additional temperature effect on minimum dead load, Emin and barometric pressure effect tests. 

If any load cell of the same capacity has a higher Y than the selected load cell, that load cell (or 

load cells) is also liable for partial evaluation testing, specifically the conduct of additional creep 

and DR tests. 

In this example, the load cells identified above also have the best characteristics of lowest vmin, 

highest Y and highest Z. This is normally the case, but not always. 

D.2.7. If applicable, select the load cell for humidity testing in accordance with 8.4.5, that being the load cell 

with the most severe characteristics, for example the greatest value of nmax or the lowest value of vmin. 

In this example, the load cell with the greatest value of nmax or the lowest value of vmin is the same 

load cell, therefore select: 

B10 - 500 kg (humidity test required) 

Note: The other B10 load cells also possess the same qualifications and are possible choices. The 500 

kg load cell was chosen because it is the smallest of the applicable B10 capacities. Although the C6 - 50 
OIML R 60-1 Annex D

D-7

kg load cell has the lowest vmin of 0.0028, the B10 load cells have the highest nmax, highest accuracy 

class, and the highest Y and Z. 

D.2.8. If applicable, select the load cell for the additional tests to be performed on digital load cells in 

accordance with 8.4.6, that being the load cell with the most severe characteristics, for example the 

greatest value of nmax or the lowest value of vmin. 

D.2.9. Summarizing, the load cells selected for test are: 

In this example, no load cell in the family is equipped with electronics. 

Summary Selected cells

C6 - 50 kg

Load cells requiring full evaluation test B10 - 500 kg

B10 - 4 000 kg

C3 - 30 000 kg

Load cells requiring partial evaluation test None 

Load cell to be tested for humidity B10 - 500 kg 

Digital load cells for additional tests None 

OIML R 60-1 Annex E

E-1

Annex E (Informative) Load transmission to the load cell

This Annex is taken from the WELMEC 2.4 (European cooperation in legal metrology) Guide for Load 

Cells (Issue 2, published in August, 2001). With permission from WELMEC, the following portion of 

that document is reprinted here to provide guidelines for load cell evaluators, during load cell 

performance evaluations. Recognizing the critical role that load cell receptors and load transmission 

devices play in accurate measurements, this Annex is intended to provide information regarding the 

effect of load transmission and recommendations for test design and procedure. The annex is 

informational and not to be considered required practice.

For some types of load cells the kind of load transmission to the load cell has influence on the 

measurements and with this on the test results.

In this annex the standard load transmission devices are listed.

The manufacturer should define whether the load cell works with all standard load transmission devices 

for the type of load cell or with selected standard load transmission devices or with a load cell specific 

load transmission devices.

This information may be considered for the load cell tests and may be marked on the certificate.

Standard load transmission devices

Tables 1 and 2 identify different types of LCs, (compression, tension, ...) and typical load cell mounting 

devices suitable for them. The symbols below classify the mobility between one point of contact on the 

load cell and its counterpart on the load receptor or mounting base.

Symbol Description

Movement possible normal to load axis 

Note: allows for temperature dilatation

Movement possible normal to load axis, with reversing force 

(spring-back effect)

Note: allows for temperature dilatation, also used for damping 

of lateral shock

Inclination possible

Note: allows for tilt of load cell or deflection of load receptor, 

no movement normal to load axis possible

Indicates auto-centering effect of the complete mounting 

assembly of one load cell
Annex E OIML R 60-1

E-2

Remarks on the standard load transmission devices presented in Tables 1 and 2:

All combinations of load cell and transmitting device shown in Tables 1 and 2 can also be utilised in a 

completely reversed manner. 

The load transmission device is independent of the encapsulation, potting or housing which are shown in 

the examples.

(a) Compression LCs (Table 1, upper part)

- The load transmissions 1 to 8 are presented for canister type LCs. Instead, all load 

transmissions may be constructed for S-type or ring type load cells.

- 6a shows a pendulum construction build as a complete unit. 

- 6b and 6c show external pendulum rocker pins combined with ring-type LCs.

- The bearings for all compression load cells may be installed either below or above the LC.

(b) Tension LCs (Table 1, lower part)

- The load transmissions 1 and 2 are presented for canister type LCs. Alternatively, both load 

transmissions may be used for S-type LCs.

(c) Beam LCs (table 2, upper part)

- The drawings present double bending and shear beams, as well as plastic 

potted and encapsulated constructions; all these constructions may be 

combined with ether of the load transmissions 1 to 10.

- The direction of loading, which is given by the manufacturer, has to be 

observed. 

(d) Single point LCs (Table 2, middle part)

- The load transmissions 1 to 10 for the beam LCs may be applied to all single point LCs. 

- The direction of loading, which is given by the manufacturer, has to be observed. 

(e) Double bending beam LCs (Table 2, lower part)

- The table shows examples of common constructions. Variations are possible provided the 

constructions allow enough horizontal flexibility between both ends. 

- The direction of loading, provided by the manufacturer, has to be observed. 

The single bending beams had been exempted for general acceptance, because very small displacements 

of the “force transducing point” may lead to a change of span and linearity. 
OIML R 60-1 Annex E

E-3

Table 1: Schematic drawings for compression and tension LCs
Annex E OIML R 60-1

E-4

Table 2: Schematic drawings for beam LCs
OIML R 60-1 Bibliography

i

Bibliography

[1] "OIML R76 Non-automatic weighing instruments," 2006.

[2] "OIML V 2-200 International Vocabulary of Metrology - Basic and General Concepts and Associated 

Terms (VIM)," 2012.

[3] "OIML V2 International Vocabulary of Terms in Legal Metrology (VIML)," 2013.

[4] "OIML D9 Principles of metrological supervision," 2004.

[5] "OIML D11 General requirements for electronic measuring instruments.," 2013.

[6] "OIML B3 OIML Basic Certificate System for OIML Type Evaluation of Measuring Instruments," 2011.

[7] "Guide to the Expression of Uncertainty in Measurement, BIPM, IEC, IFCC, ISO, IUPAC, IUPAP, OIML, 

JCGM 100," 2008.

[8] "OIML D31 General requirements for software controlled measuring instruments," 2008.

[9] "ISO 8601 Data elements and interchange formats - Information interchange - Representation of dates and 

times," 2004.

[10] "IEC Publication 60068-2-30," 2005-08. 

[11] "IEC Publication 60068-3-4," 2001- 08.

[12] "IEC Publication 60068-2-78," 2012.

[13] "IEC Publication 61000-2-1," 1990-05. 

[14] "IEC Publication 61000-4-1," 2006-10. 

[15] "IEC Publication 61000-4-29," 2000 - 08. 

[16] "IEC Publication 61000-4-11," 2004- 03. 

[17] "IEC Publication 61000-6-1," 2005 - 03. 

[18] "IEC Publication 61000-6-2," 2005-01. 

[19] "IEC Publication 61000-4-4," 2012 - 04. 

[20] "IEC Publication 61000-4-5," 2014 - 05. 

[21] "IEC Publication 61000-4-2," 2008 - 12. 

[22] "IEC Publication 61000-4-3," 2010 - 04. 

[23] "IEC Publication 61000-4-6," 2013 - 10. 

[24] "OIML R61-1 Automatic gravimetric filling instruments," 2004.

[25] "OIML R50-1 Continuous totalizing automatic weighing instruments (belt weighers)," 1997.

[26] "OIML R51-1 Automatic catchweighing instruments," 2006.

[27] "OIML R107-1 Discontinuous totalizing automatic weighing instruments (totalizing hopper weighers)," 

2007.

[28] "OIML R106-1 Automatic rail-weighbridges," 2011.