Skip to main content    

An official website of the United States government

Here’s how you know

**Official websites use .gov**              A **.gov** website belongs to an official government organization in the United States.            

**Secure .gov websites use HTTPS**              A **lock** (  LockA locked padlock) or **https://** means you’ve safely connected to the .gov website. Share sensitive information only on official, secure websites.            

https://www.nist.gov/pml/sensor-science/fluid-metrology/calibration-pitot-tubes-turbulence

            -  Topics  -  All Topics 
-  Advanced communications 
-  Artificial intelligence 
-  Bioscience 
-  Buildings and construction 
-  Chemistry 
-  Climate 
-  Cybersecurity 
-  Electronics 
-  Energy 
-  Environment 
-  Fire 
-  Forensic science 
-  Health 
-  Information technology 
-  Infrastructure 
-  Manufacturing 
-  Materials 
-  Mathematics and statistics 
-  Metrology 
-  Nanotechnology 
-  Neutron research 
-  Performance excellence 
-  Physics 
-  Public safety 
-  Resilience 
-  Standards 
-  Transportation 
 
            -  Publications 
            -  Labs & Major Programs  -  Laboratories    -  Communications Technology Laboratory 
  -  Engineering Laboratory 
  -  Information Technology Laboratory 
  -  Material Measurement Laboratory 
  -  Physical Measurement Laboratory 
-  User Facilities    -  NIST Center for Neutron Research 
  -  CNST NanoFab 
-  Research Test Beds 
-  Research Projects 
-  Tools & Instruments 
-  Major Programs    -  Baldrige Performance Excellence Program 
  -  CHIPS for America Initiative 
  -  Manufacturing Extension Partnership (MEP) 
  -  Office of Advanced Manufacturing 
  -  Special Programs Office 
  -  Technology Partnerships Office 
 
            -  Services & Resources  -  Standards and Measurements    -  Calibration Services 
  -  Laboratory Accreditation (NVLAP) 
  -  Quality System 
  -  Standard Reference Materials (SRMs) 
  -  Standards.gov 
  -  Time Services 
  -  Office of Weights and Measures 
-  Software 
-  Data    -  Chemistry WebBook 
  -  National Vulnerability Database 
  -  Physical Reference Data 
  -  Standard Reference Data (SRD) 
-  Storefront 
-  License & Patents 
-  Computer Security Resource Center (CSRC) 
-  NIST Research Library 
 
            -  News & Events  -  News 
-  Events 
-  Blogs 
-  Feature Stories 
-  Awards 
-  Video Gallery 
-  Image Gallery 
-  Media Contacts 
 
            -  About NIST  -  About Us    -  Leadership 
  -  Organization Structure 
  -  Budget & Planning 
-  Contact Us 
-  Visit 
-  Careers    -  Student programs 
-  Work with NIST 
-  History    -  NIST Digital Archives 
  -  NIST Museum 
  -  NIST and the Nobel 
-  Educational Resources 
 

Physical Measurement LaboratorySensor Science Division

## Fluid Metrology Group

-  Laboratory Homepage 
-  Division Homepage 
-  Groups  Expand or Collapse    -  Dimensional Metrology 
  -  Fluid Metrology 
  -  Optical Radiation 
  -  Remote Sensing 
  -  Surface and Interface Metrology 
  -  Thermodynamic Metrology 
-  Collaboration Opportunities 
-  Employment and Internship Opportunities  Expand or Collapse    -  Employment at NIST 
  -  NIST Professional Research Experience Program (PREP) 
  -  NIST/NRC Postdoctoral Research Associateships Program 
-  Services and Standards 
-  Staff Directory 

# Calibration of pitot tubes in turbulence

## Share

Facebook

Linkedin

Twitter

Email

Today, the carbon dioxide emitted from the flue-gas stacks of coal-burning power plants is measured to control the emissions of the pollutants sulfur dioxide, nitrogen oxides (nitric oxide, nitrogen dioxide), and mercury. However, present-day carbon dioxide emission measurements may have uncertainties in the range 5% to 10%. NIST is conducting research to reduce the uncertainties to 1% or 2%. These lower uncertainties will be needed if the US Environmental Protection Agency (EPA) puts a price on carbon dioxide emissions, for example by a carbon tax or by cap-and-trade regulations.

**Calibration factor of a multi-hole Pitot tube as a function of air speed at turbulence intensities of 0.27% to 2.0%. (pitch = 20°; yaw = 0°) The arrows show that, at low turbulence, the calibration factor is smaller with increasing air speed than with decreasing air speed.**

Carbon dioxide emissions are determined by mapping the flow in stacks using 1-, 2- and 3-dimensional differential pressure devices, such as standard Pitot static tubes, S-probes and multi-hole Pitot tubes. NIST has outfitted its low-turbulence wind tunnel with an automated stage to calibrate such differential pressure devices as a function of air-speed, pitch angle, and yaw angle. During preliminary measurements, Crowley, et al. discovered that the calibration can show hysteresis that gradually disappears as the turbulence in the flow increases. Thus calibrations must account for the turbulence in the flow to achieve the target accuracies of 1% or 2%.

**The recirculation zone near a multi-hole Pitot tube is associated with the hysteresis in the calibration factor of the Pitot tube.**

In addition to improving the calibration of differential pressure devices, NIST is developing a scale-model smoke stack test-bed to study calibration protocols. NIST is also studying unconventional techniques for measuring carbon dioxide emissions such as a long-wavelength acoustic flow-meter.

Flow metrology and rheologyCalibration services

## Contacts

                -  Iosif Isaakovich Shinder  iosif.shinder@nist.gov      (301) 975-5943   

### HEADQUARTERS

301-975-2000

Webmaster | Contact Us | Our Other Offices

TwitterFacebookLinkedInInstagramYouTubeGiphyRSS FeedMailing List

Feedback

            -  Site Privacy 
            -  Accessibility 
            -  Privacy Program 
            -  Copyrights 
            -  Vulnerability Disclosure 
            -  No Fear Act Policy 
            -  FOIA 
            -  Environmental Policy 
            -  Scientific Integrity 
            -  Information Quality Standards 
            -  Commerce.gov 
            -  Science.gov 
            -  USA.gov 
            -  Vote.gov