Installation and Operation Manual 

Thermo Scientific

General Purpose, Flammable Storage, 

and Explosion Proof 

Laboratory Refrigerators and Freezers 
© 2015 Thermo Fisher Scientific. All rights reserved
Refrigerators,

Refrigerator-Freezers and Freezers

Cool-Lab, General Purpose

05LCEETSA 

05LCEETSV 10LCEETSA

Explosion-Proof

10ECEETSA

10ECEETSV

05EREETSA

05EFEETSA

Flammable Material Storage 

10FCEETSA

10FCEETSV

05FREETSA

05FFEETSA

 05LREETSA 12LCEETSA

05LREETSV 

 04LFEETSA

02LFEETSA

10LCEETSV
2 

Table of Contents

Safety Information .............................................................................................................................................................. 4

AlertSignals .................................................................................................................................................................... 4

Explosion-Proof Refrigerators and Freezers...................................................................................................................... 5

Overview......................................................................................................................................................................... 5

MODEL 10ECEETSA, 10ECEETSV...............................................................................................................................6 

MODEL 05EREETSA ......................................................................................................................................................7 

MODEL 05EFEETSA......................................................................................................................................................7 

Flammable Materials Storage Refrigerators and Freezers...................................................................................................8 

Overview......................................................................................................................................................................... 8

MODEL 10FCEETSA, 10FCEETSV................................................................................................................................9 

MODEL 05FREETSA.................................................................................................................................................... 10 

MODEL 05FFEETSA.....................................................................................................................................................10 

Cool Lab, General-Purpose Laboratory Refrigerators/Freezers .......................................................................................11 

MODEL 05LCEETSA, 05LCEETSV .............................................................................................................................11 

Overview.......................................................................................................................................................................11

MODEL 05LREETSA, 05LREETSV..............................................................................................................................11 

MODEL 04LFEETSA.................................................................................................................................................... 12 

MODEL 02LFEETSA ....................................................................................................................................................13

MODEL 10LCEETSA, 10LCEETSV.............................................................................................................................14

MODEL 12LCEETSA ....................................................................................................................................................15

Unpacking and Installation Shipping .................................................................................................................................16

Carton ...........................................................................................................................................................................16

Unpacking.....................................................................................................................................................................16

Location ........................................................................................................................................................................16

Clearance......................................................................................................................................................................16

Electrical .......................................................................................................................................................................17

Be Advised....................................................................................................................................................................17

How to Seal Killark® Box Conduit with Fiber and Sealing Compound to Help Protect Against Explosions 

(Explosion-Proof Units Only) ......................................................................................................................................18

Operation ..........................................................................................................................................................................21

EnvironmentalOperating Conditions ............................................................................................................................21

Start-UpProcedure .......................................................................................................................................................21

RestartProcedure .........................................................................................................................................................21

Control Panel Lights .....................................................................................................................................................22

How to Save Energy ....................................................................................................................................................23

Safety Tips ....................................................................................................................................................................23

Troubleshooting................................................................................................................................................................ 24

Maintenance .....................................................................................................................................................................25

Cleaning of Units ...........................................................................................................................................................25

Interior/Exterior and Door Gaskets ...............................................................................................................................25

Condenser .....................................................................................................................................................................25

ManualDefrost Procedure (Models 05FFEETSA,05EFEETSA,

05LCEETSA, and 05LCEETSV)................................................................................................................................... 26

ManualDefrost Procedure (Model 12LCEETSA Only) .................................................................................................27

Reversing the Front Door (Model 10ECEETSA)...........................................................................................................28
3 

Replacement Parts...........................................................................................................................................................32

Ordering Procedures ........................................................................................................................................................33

OneYear Limited Warranty...............................................................................................................................................34

Reversing the Front Door (Model 05FREETSA, 05EREETSA and 05LREETSA) .................................................... 29

Reversing the Front Door (Model12LCEETSA)..................................................................................... 30
4

Safety Information

Your satisfaction and safety are important to Thermo Scientific and a complete understanding of this unit 

is necessary to attain these objectives.

As the ultimate user of this apparatus, it is your responsibility to understand its proper function and 

operational characteristics. This instruction manual should be thoroughly read and all operators given 

adequate training before attempting to place this unit in service. Awareness of the stated cautions and 

warnings, and compliance with recommended operating parameters – together with maintenance 

requirements – are important for safe and satisfactory operation. The unit should be used for its intended 

application; alterations or modifications will void the Warranty.

This product is not intended, nor can it be used, as a sterile or patient connected device. In addition, this 

apparatus is not designed for use in Class I, II or III locations as defined by the US National Electrical 

Code, unless otherwise noted.

Alert Signals

DANGER RISK OF CHILD ENTRAPMENT 

Before you throw away your old refrigerator or freezer:

Take off doors 

 

 Leave the shelves in the place so that children may not easily climb inside.

If the equipment is used in a manner not specified by the manufacturer, the 

protection provided by the equipment may be impaired. 

ON 

OFF 

Safety Alert 

Important operating instructions. To reduce the risk of injury 

or poor performance of the unit. Read the user manual before 

putting the equipment into operation.

WARNING Indicates an immediately hazardous situation, which if 

not avoided, will result in death or serious injury. 

CAUTION Indicates an immediately hazardous situation, which if 

not avoided, may result in minor to moderate injury.

(No symbol) CAUTION (Without Safety Alert Symbol) indicates a situation that 

may result in property damage.

Shock Hazard 

Use of this equipment involves power supplies which convert 

line voltage to low voltage power. Do not modify or use power 

supplies other than OEM equipment. Connection of the 

power supply may require a properly grounded receptacle. 

Potential for electrical shock or equipment damage exists if 

precautions are not followed.

Frost bite/ 

Low 

Temperature 

Avoid contact with cold freezer surfaces potential for 

cold burns or skin sticking to cold surfaces.
5 

Explosion-Proof Refrigerators and

Freezers 

Overview 

Conventional refrigerators and freezers are not suitable for storing 

flammable materials. Such units have components in their 

electrical and refrigeration systems that can trigger explosions of 

flammable air-vapor mixtures inside the unit and/or in the 

immediate surrounding area. 

The Authority Having Jurisdiction (AHJ) determines if work areas 

are designated as a hazardous location with respect to the presence 

of flammable gases or vapors. Such locations are defined in 

(National Fire Protection Agency) NFPA 70 Articles 500- 501 and 

OSHA 29 CFR1910.307. Some of these classified areas are 

expected to experience concentrations of flammable gases and/or 

vapors at or above their lower flammability limits for extended 

periods of time. 

The construction of our explosion-proof units has been evaluated by 

Underwriters Laboratories (UL) and are suitable for use in classified 

areas requiring Class I, Groups C and D* protected equipment. The 

electrical components such as thermostats, wiring, splices, relays 

and compressor motors on explosion-proof units are safely housed 

within explosion-proof enclosures and conduit. Compressor surface 

temperatures have been evaluated and determined to remain below 

the flash point of materials found in Class I, Groups C and D. 

All models have heavy-gauge, rigid, steel construction with a 

durable enamel finish. Interiors have epoxy enamel or ABS plastic 

construction. Each unit is insulated throughout for energy-efficient 

operation. 

These units are ideal for storing cyclopropane, ethyl ether, 

ethylene, acetone, alcohol, benzene, butane, gasoline, hexane, 

lacquer solvent vapors, naphtha, natural gas or propane along with 

many other potentially hazardous materials. 

*The notation Class 1, Groups C and D is an accepted abbreviation

for Class I, Div 1, Groups C and D; Class I Zone 1 Group IIB.
6 

EXPLOSION-PROOF UNITS

MODEL 10ECEETSA, 10ECEETSV 

23.5” Wide, 

Two-door Refrigerator/Freezer 

Features: 

• Overall 10.1 cu. ft.

• Three adjustable shelves

• Four door shelves

• Hydraulic thermostat

• Adjustable natural air-flow vent

• Manual defrost

• ABS plastic interior

• White color

Model Refrigerator Chamber Dimensions

H x W x D inches (cm) Freezer Chamber Dimensions

H x W x D inches (cm)

Total Vol. 

Cu. Ft. 

Exterior Dimensions

H x W x D inches

(cm)

10ECEETSA

38.5" x 19.5" x 13.5" 

(98 x 50 x 34 cm)

11.5" x 17.5" x 15" 

(29 x 44 x 38 cm) 10.1 60" x 23.5" x 31.5" 

(152 x 60 x 80 cm) 

10ECEETSV 10.1 60" x 23.5" x 31.5" 

(152 x 60 x 80 cm) 

Model Electrical Characteristics 

Volts/Hz, Watts, Amps 

Refrig.Temp. Range 

° C (° F)

Freezer Temp. Range 

° C (° F)

Net Wt. 

Lbs. (kg)

Ship Wt. 

Lbs. (kg)

10ECEETSA 120/60, 230, 1.9 1° to 12° 

(33.8° to 53.6°)

-12° to -20° 

(10.4° to -4°) 120 (54.5) 150 (68) 

10ECEETSV 230/50, 360, 1.5 -12° to -20° 

(10.4° to -4°)

120 (54.5) 150 (68) 

NOTE: Amps listed are at normal run mode, starting amps may be higher.

38.5" x 19.5" x 13.5"

 (98 x 50 x 34 cm)

11.5" x 17.5" x 15" 

(29 x 44 x 38 cm) 

1° to 12° 

(33.8° to 53.6°)
7 

EXPLOSION-PROOF UNITS

MODEL 05EREETSA 

23.5” Wide, 

Under counter refrigerator 

Features: 

• Cold-wall design

• Ample door storage

• Adjustable thermostat

• Rear rollers

• Front legs level-adjust

• Magnetic gasket

• Sturdy metal hinges

• Seamless one-piece

• polypropylene liner

• White color

• One glass shelf

Note (05EREETSA)

THIS UNIT IS DESIGNED FOR FREE 

STANDING INSTALLATION ONLY. IT 

MAY BE INSTALLED UNDER A 

COUNTER ONLY IF SUFFICIENT SPACE 

FOR AIR FLOW IS PROVIDED. IT IS 

SUGGESTED THE UNIT BE RECESSED 

ONLY 21” TO PERMIT EASY ACCESS 

TO THE LOCK. 

MODEL 05EFEETSA 

21.5” Wide, 

Under-counter freezer 

Features: 

• Adjustable thermostat

• Magnetic gasket

• Hermetically-sealed

compressor

• Manual defrost

• White color

Model Refrigerator Chamber Dimensions 

H x W x D inches (cm)

Freezer Chamber Dimensions 

H x W x D inches (cm)

Total Vol. 

Cu. Ft. 

Exterior Dimensions 

H x W x D inches (cm)

05EREETSA

27.5” x 20” x 16” 

(70 x 51 x 40.64 cm) N/A 5.5

3.5” x 23.5” x 28” 

(85 x 59 x 71cm)

05EFEETSA N/A

28.5” x 18” x 15” 

(72 x 46 x 38 cm) 5.0 33.5”x 21.5” 31” 

(85 x 55 x 79 cm) 

Model Electrical Characteristics 

Volts/Hz, Watts, Amps

Refrig.Temp. Range

 °C (°F) 

Freezer Temp. Range 

°C (°F) 

Net Wt. 

Lbs. (kg) 

Ship Wt. 

Lbs. (kg) 

120/60, 160, 1.3 1° to 12° 

(33.8° to 53.6°) 

N/A 100 (45.3) 110 (50) 

120/60, 132, 1.1 N/A 

-12° to -20° 

(10.4° to -4°) 

110 (50) 

NOTE: Amps listed are at normal run mode, starting amps may be higher.

05EREETSA

05EFEETSA 100 (45.3) 
8 

Flammable Materials Storage 

Refrigerators and Freezers 

Overview 

Conventional refrigerators and freezers are not suitable for 

storing flammable materials. Such units have components in 

their electrical and refrigeration systems that can trigger 

explosions of flammable air-vapor mixtures inside the unit. 

Flammable Materials Storage (FMS) refrigerators/freezers are 

designed for use in locations, which are not classified by the 

Authority Having Jurisdiction (AHJ) as hazardous. Under normal 

operating conditions the build up or presence of flammable 

vapors will not occur in the environment external to the unit. 

(Commonly known as “Ordinary Locations”) 

FMS units are NOT designed for use in Class I, Groups 

C and D environments, which require an ExplosionProof Refrigerator/Freezer. 

FMS units are designed and evaluated by Underwriters 

Laboratories (UL) to meet the requirements of the National Fire 

Protection Association Standards Nos. 45, 70 and 99. These 

units have no internal electrical components that could trigger an 

explosion or fire of hazardous materials inside the unit. 

These units are ideal for storing cyclopropane, ethyl ether, 

ethylene, acetone, alcohol, benzene, butane, gasoline, hexane, 

lacquer solvent vapors, naphtha, natural gas or propane along 

with many other potentially hazardous materials. 

All models have heavy-gauge, rigid, steel construction with a 

durable enamel finish. Interiors have epoxy enamel or ABS 

plastic construction. Each unit is insulated throughout for 

energy-efficient operation. 
9 

FLAMMABLE MATERIALS STORAGE UNITS

Note

In order for freezer to reach -20°C, vent on 

inside upper rear of refrigerator must be 

closed. 

MODEL 10FCEETSA, 10FCEETSV

23.5” Wide 

Two-door refrigerator/freezer 

Features: 

• Overall 10.1 cu. ft.

• Adjustable thermostat

• ABS plastic interior

• Manual defrost

• White color

Model Refrigerator Chamber Dimensions 

H x W x D inches (cm)

Freezer Chamber Dimensions

H x W x D inches (cm)

Total Vol. 

Cu. Ft.

Exterior Dimensions 

H x W x D inches (cm)

10FCEETSA

39” x 19.5” x 18” 

(140 x 69 x 51 cm)

12” x 18” x 16.25” 

(30 x 46 x 41 cm) 10.1

60” x 23.5” x30.5” 

(152 x 60 x 77.47 cm)

10FCEETSV 39” x 19.5” x 18” 

(140 x 69 x 51 cm)

12” x 18” x 16.25” 

(30 x 46 x 41 cm) 10.1

60” x 23.5” x 30.5” 

(152 x 60 x 77.47 cm)

Model Electrical Characteristics

Volts/Hz, Watts, Amps

Refrig. Temp. Range 

°C (°F)

Freezer Temp. Range

°C (°F)

Net Wt. 

Lbs. (kg)

Ship Wt. 

Lbs. (kg)

10FCEETSA 120/60, 230, 1.9

1° to 12° 

(33.8° to 53.6°)

-12° to -20° 

(10.4° to -4°)

120 (54.5) 150 (68)

10FCEETSV 230/50, 360, 1.5 120 (54.5) 150 (68)

NOTE: Amps listed are at normal run mode, starting amps may be higher. 

l

1° to 12° 

(33.8° to 53.6°)

-12° to -20° 

(10.4° to -4°)
10 

FLAMMABLE MATERIALS STORAGE UNITS

MODEL 05FREETSA

23.5” Wide, no-frost 

under-counter all refrigerator with 

key lock 

Features: 

• Cold-wall design

• Ample door storage

• Adjustable thermostat

• Rear rollers

• Front legs level-adjust

• Magnetic gasket

• Sturdy metal hinges

• Seamless one-piece

polypropylene liner

• White color

• One glass shelf

NOTE (05FREETSA): THESE UNITS 

ARE DESIGNED FOR FREE 

STANDING INSTALLATION ONLY. 

THEY MAY BE INSTALLED UNDER 

A COUNTER ONLY IF SUFFICIENT 

SPACE FOR AIRFLOW IS 

PROVIDED. 1” ON THE SIDES, TOP 

AND REAR ARE RECOMMENDED. 

MODEL 05FFEETSA

21” Wide 

Front-opening freezer 

Features: 

• Adjustable thermostat

• Magnetic gasket

• Hermetically-sealed

compressor

• Manual defrost

• White color

Model

Refrigerator Chamber Dimensions 

H x W x D inches (cm)

Freezer Chamber Dimensions 

H x W x D inches (cm)

Total Vol. 

Cu. Ft.

Exterior Dimensions 

H x W x D inches (cm)

05FREETSA

27.5” x 20” x 16” 

(70 x 51x 41cm)

N/A 5.5

33.5” x 23.5” x 29.5”

(85 x 59 x 75 cm)

05FFEETSA N/A

28.5” x 18” x 15”

(72 x 46 x 38 cm) 5.0 33.5” x 21” x 29.5” 

(85 x 53 x 75 cm)

 NOTE: Amps listed are at normal run mode, starting amps may be higher. 

Model

Electrical Characteristics 

Volts/Hz, Watts, Amps

Refrig. Temp.Range 

°C (°F) 

Freezer Temp. Range

°C (°F)

Net Wt. 

Lbs. (kg)

Ship Wt. Lbs. 

(kg) 

05FREETSA 120/60, 160, 1.3 1° to 12° 

(33.8° to 53.6°) N/A 100 (45.3) 120 (54.5) 

05FFEETSA 120/60, 132,1.1 N/A -12° to -20° 

(10.4° to -4°) 73 (53) 85 (38.5)
11 

Cool Lab, General-Purpose Laboratory 

Refrigerators/Freezers 

MODEL 05LCEETSA, 05LCEETSV

23.63” Wide, 

Refrigerator/freezer 

Features: 

• One-piece seamless interior

• Three Glass, slide-out shelves

• Three in-door shelves

• Adjustable foot

• Drip tray

• Manual defrost

• White color

NOTE (05LCEETSA and 05LREETSA): 

THESE UNITS ARE DESIGNED FOR FREE 

STANDING INSTALLATION ONLY. IT MAY 

BE INSTALLED UNDER A COUNTER ONLY IF 

SUFFICIENT SPACE FOR AIRFLOW IS 

PROVIDED. 

Overview 

These are general-purpose units available in all refrigerator, 

all freezer and combination refrigerator/freezer models. 

MODEL 05LREETSA, 05LREETSV 

23.5” Wide, 

Under-counter Refrigerator

Features: 

• ABS plastic interior

• Four adjustable shelves

• Adjustable thermostat

• Magnetic door gasket

• White color

• Key lock

• One glass shelf

Model

Refrigerator Chamber Dimensions 

H x W x D inches (cm)

Freezer Chamber Dimensions 

H x W x D inches (cm)

Total Vol. 

Cu. Ft.

Exterior Dimensions 

H x W x D inches (cm)

05LCEETSA 5.1

33.5” x 23.63” x 23.5” 21” x 15” x 18”

(53x 38 x 46 cm)

5” x 15” x 11”

(13 x 38 x 28 cm) (85 x 60 x 60 cm)

05LCEETSV 5.1

33.5” x 23.63” x 23.5”

(85 x 60 x 60 cm)

05LREETSA 5.5

27.5” x 19” x 13”

(70x 48 x 33 cm)

33.3” x 23.5” x 24.5”

(85 x 59 x 62 cm)

05LREETSV 5.5

27.5” x 19” x 13”

(70x 48 x 33 cm)

33.3” x 23.5” x 24.5”

(85 x 59 x 62 cm)

NOTE: Amps listed are at normal run mode, starting amps may be higher.

Model 

Electrical Characteristics 

Volts/Hz,Watts, Amps

Refrig. Temp.Range 

°C (°F) 

Freezer Temp. Range 

°C (°F) 

Net Wt. Lbs. 

(kg) 

Ship Wt. 

Lbs. (kg) 

05LCEETSA 100 (45.4) 120 (54.5) 

05LCEETSV 230/50, 160, 0.66 100 (45.4) 120 (54.5) 

05LREETSA 115/60, 160, 1.3 N/A 100 (45.4) 120 (54.5) 

05LREETSV 230/50, 200, 0.8 N/A 100 (45.4) 120 (54.5) 

21” x 15” x 18”

(53x 38 x 46 cm)

5” x 15” x 11”

(13 x 38 x 28 cm)

N/A 

N/A 

1° to 12° (33.8° to 53.6°) 

1° to 12° (33.8° to 53.6°) 

1° to 12° (33.8° to 53.6°) 

1° to 12° (33.8° to 53.6°) 

-12° to -20° (10.4° to -4°) 

-12° to -20° (10.4° to -4°)

115/60, 160, 1.3 
12

COOL LAB GENERAL-PURPOSE LABORATORY REFRIGERATORS/FREEZERS

MODEL 04LFEETSA

23.5” Wide, 

Under-counter freezer 

Features: 

• Exterior thermostat

• High temperature alarm

• Reversible door

• 3 slide out drawers

• Key lock

• White color

Model

Refrigerator Chamber Dimensions 

H x W x D inches (cm)

Freezer Chamber Dimensions 

H x W x D inches (cm)

Total Vol. 

Cu. Ft.

Exterior Dimensions 

H x W x D inches (cm)

04LFEETSA N/A

24.25” x 17.75” x 16.5” 

(62 x 45 x 42 cm) 4.0 33.5” x 23.5” x 23.5” 

(85 x 60 x 60 cm)

Model

Electrical Characteristics

Volts/Hz, Watts, Amps

Refrig. Temp. Range 

°C (°F)

Freezer Temp. Range 

°C (°F)

Net Wt.

Lbs. (kg)

Ship Wt. 

Lbs. (kg)

04LFEETSA 115/60, 144, 1.2 N/A

-12° to -20° 

(10.4° to -4°)

100 (45.4) 120 (54.5)

NOTE: Amps listed are at normal run mode, starting amps may be higher.
13

COOL LAB GENERAL-PURPOSE LABORATORY REFRIGERATORS/FREEZERS

MODEL 02LFEETSA

19.75” Wide, 

Cube Size Freezer 

Reversible Door Swing 

Features: 

• Adjustable thermostat

• Manual defrost

• Front-mounted lock

• 100% CFC Free

Model

Refrigerator Chamber Dimensions

H x W x D inches(cm)

Freezer Chamber Dimensions

H x W x D inches (cm)

Total Vol. 

Cu. Ft.

Exterior Dimensions

H x W x D inches(cm)

02LFEETSA N/A

14" x 15.5" x 13.5"

(36 x 39 x 34 cm) 1.6

20.75" x 19.75" x 21" 

(53 x 50 x 53 cm)

Model

Electrical Characteristics 

Volts/Hz, Watts, Amps

Refrig. Temp. Range 

° C (° F)

Freezer Temp. Range 

° C (° F)

Net Wt. 

Lbs. (kg)

Ship Wt. 

Lbs. (kg)

02LFEETSA 115/60, 83, 1.2 N/A

-12° to -20°

(10.4° to 4°)

49 (22) 55 (29)

NOTE: Amps listed are at normal run mode, starting amps may be higher. 
14

COOL LAB GENERAL-PURPOSE LABORATORY REFRIGERATORS/FREEZERS

MODEL 10LCEETSA, 10LCEETSV

23.5” Wide, 

Two-Door Refrigerator/Freezer 

Features: 

• 3 adjustable shelves

• Manual defrost

• White color

Model

Refrigerator Chamber Dimensions

H x W x D inches (cm)

Freezer Chamber Dimensions

H x W x D inches (cm)

Total Vol. 

Cu. Ft.

Exterior Dimensions H 

x W x D inches (cm)

10LCEETSA

38.5" x 19.5" x 13.5" 

(98 x 50 x 34.29 cm)

11.5" x 17.5" x 15" 

(29.21 x 44.45 x 38 cm) 10.1

60" x 23.5" x 25.5" 

(152.4 x 60 x 64.77 cm)

10LCEETSV 10.1

Model

Electrical Characteristics 

Volts/Hz, Watts, Amps

Refrig. Temp. Range 

° C (° F)

Freezer Temp. Range 

° C (° F)

Net Wt. 

Lbs. (kg)

Ship Wt. 

Lbs. (kg)

10LCEETSA 115/60, 360, 3.0

1° to 12° 

(34° to 53.6°)

-12° to -20° 

(10.4° to -4°)

120 (54.5) 150 (68.24)

10LCEETSV 230/50, 360, 1.5

1° to 12° 

(34° to 53.6°)

-12° to -20° 

(10.4° to -4°)

120 (54.5) 150 (68.24)

NOTE: Amps listed are at normal run mode, starting amps may be higher. 

38.5" x 19.5" x 13.5" 

(98 x 50 x 34.29 cm)

11.5" x 17.5" x 15" 

(29.21 x 44.45 x 38 cm)

60" x 23.5" x 25.5" 

(152.4 x 60 x 64.77 cm)
15

COOL LAB GENERAL-PURPOSE LABORATORY REFRIGERATORS/FREEZERS

MODEL 12LCEETSA

23.75” Wide, 

Combination Refrigerator/Freezer 

Features: 

• 4 adjustable refrigerator shelves

• 3 fixed freezer shelves

• Automatic defrost refrigerator

• Manual defrost freezer

• Reversible doors

• 2 independent adjustable thermostat

• 2 independent compressors

• White color

Model

Refrigerator Chamber Dimensions

H x W x D inches (cm)

Freezer Chamber Dimensions

H x W x D inches (cm)

Total Vol. 

Cu. Ft.

Exterior Dimensions H 

x W x D inches (cm)

12LCEETSA

35" x 19" x 13.5" 

(89 x 48 x 34 cm)

28" x 17" x 13.5" 

(71 x 43 x 34 cm) 11.8

79" x 23.75" x 23.75" 

(199 x 60 x 60 cm)

Model

Electrical Characteristics 

Volts/Hz, Watts, Amps

Refrig.Temp.Range 

° C (° F)

Freezer Temp. Range 

° C (° F)

Net Wt. 

Lbs. (kg)

Ship Wt. 

Lbs. (kg)

12LCEETSA 115/60, 648, 5.4

1° to 12° 

(34° to 53.6°)

-12° to -20° 

(10.4° to -4°)

192 (87) 251(114)

NOTE: Amps listed are at normal run mode, starting amps may be higher. 
16

Unpacking and Installation 

Shipping Carton 

This should be inspected upon delivery. When received, 

carefully examine for any shipping damage before 

unpacking. If damage is discovered, the delivering carrier 

should both specify and sign for the damage on your copy 

of the delivery receipt. 

Open the carton carefully making certain that all parts are 

accounted for before packaging materials are discarded. 

After unpacking, if damage is found, promptly report it to 

the carrier and request a damage inspection promptly. 

IMPORTANT: Failure to request an inspection of 

damage within a few days after receipt of shipment 

absolves the carrier from any liability for damage. You 

must call for a damage inspection promptly. 

Unpacking 

Use the list below when unpacking to verify that the 

complete unit has been received. Do not discard packing 

materials until all is accounted for. 

The following items are included in the shipment: 

Refrigerator/Freezer 

Operation Manual 057-196-00 

528-609-00 

528-022-00 

Volt Warning Tag - 

240V Registration Card 

Inspection Tag 528-028-00 

Location 

Locate your unit in the most convenient place and near a 

grounded electric outlet. If units are placed on a countertop, the front should be 3" or more back from the edge to 

avoid accidental tipping of the unit. 

If possible, locate your unit out of direct sunlight and away 

from heat sources such as a radiator, stove or heat duct. 

Clearance 

Position the unit so that there is at least 12” clearance on 

the top, 3” at the back, and 2” on each side. It is 

recommended that even more space be left at the back of 

the unit in case maintenance is required. 
17

INSTALLATION

Caution

DO NOT REMOVE, under any 

circumstance, the grounding 

prongs from the 3-prong power 

cord supplied with all units. 

Caution

DO NOT USE electrical 

extension cords that may 

result in voltage loss and 

possible hazardous operation. 

Warning

Explosion-proof units do not 

come with line cords. They 

require rigid conduit to be run 

directly in order to seal off the 

fitting on thermostat housing. 

This should be done by a 

licensed electrician and follow 

all local and electrician and 

follow all local electrical codes. 

If any questions pertaining to 

electrical safety arises, please 

refer to article 501 of the 

National Electrical Code. 

Electrical 

Units must be connected to a grounded outlet matching the 

nameplate and/or the information furnished in this manual. 

If you are not sure about the outlet, you should contact a 

qualified electrician for assistance. 

The unit should always be connected to its own individual outlet. 

Be Advised 

WARNING: UNLESS UNIT IS SPECIFICALLY 

DESIGNED FOR COMBUSTIBLE OR FLAMMABLE 

ATMOSPHERES DO NOT USE IN THE PRESENCE OF 

FLAMMABLE OR COMBUSTIBLE MATERIALS OR 

EXPLOSIVE GASES. DO NOT USE IN THE PRESENCE OF 

PRESSURIZED OR SEALED CONTAINERS— FIRE OR 

EXPLOSION MAY RESULT CAUSING DEATH. 

CAUTION: BEFORE CONNECTING THE FINAL POWER 

SUPPLY, CHECK THE ELECTRICAL CHARACTERISTICS 

OF THE UNIT NAMEPLATE TO SEE THAT IT IS IN 

AGREEMENT WITH THE POWER SUPPLIED. IN ADDITION, 

POWER SHOULD BE WIRED TO THE UNIT ACCORDING 

TO THE ELECTRICAL SCHEMATIC AND ALL APPLICABLE 

CODES. ONLY QUALIFIED ELECTRICIANS SHOULD WORK 

ON THE ELECTRICAL PORTION OF ANY UNIT 

INSTALLATION. 

CAUTION: STORAGE BY USER OF ANY MATERIALS IN THE 

PRODUCT THAT MAY CAUSE A DETERIORATION OF THE 

PRODUCT SHALL BE DEEMED TO CONSTITUTE 

ABNORMAL AND IMPROPER USAGE OF THE PRODUCT 

FOR PURPOSES OF THIS WARRANTY. 

WARNING: RISK OF CHILD ENTRAPMENT! Before you 

discard your old refrigerator or freezer: 

• Remove door(s),

• Leave the shelves in place so that children may not

easily climb inside.
18

Unscrew and Remove Cover 

INSTALLATION

Figure 1: Sealing the Killark Box

How to Seal Killark® Box Conduit with Fiber and 

Sealing Compound to Help Protect Against 

Explosions (Explosion-Proof Units Only):

The purpose of the procedure that follows is to build fiber rope

dams on the left and right hubs of the horizontal conduit. The

fiber rope damns will surround conduit wiring that is housed

inside the horizontal conduit.

When both the left and right fiber rope dams have been 

pressed into place, sealing compound is poured between the 

two and forms into an airtight plug.

All of this is done in order to prevent the very real threat of 

gas entering the Killark box and a resulting serious explosion.

After the unit wires have been pulled through the horizontal 

conduit the following procedure is required:

• Turn power off at the circuit breaker before proceeding.

• Place a small amount of sealing compound granules,

enclosed, into a clean mixing vessel. Add small amounts of

water while stirring until a thick paste is formed, then

carefully continue adding smaller amounts of water until a

thick gravy consistency is achieved—NOT WATERY.

Discard any material that becomes too stiff to use. Never

attempt to restore workability by stirring in more water.

• Locate silver Killark box, back/top-center of unit.

• Unscrew conduit domed-cover.

Conduit Domed-Cover

Conduit Opening

Horizontal Conduit 

Left Hub Right Hub 
19

INSTALLATION

Figure 2: Horizontal Conduit, Cutaway 

• Insert fiber rope material down into horizontal conduit opening. Pressing down firmly, work the

material into the left hub and—most importantly—being sure the material COMPLETELY

SURROUNDS THE WIRING, from the top to the bottom, completely blocking this end of the

horizontal conduit.

• Insert fiber rope material down into horizontal conduit opening. Pressing down firmly, work the material

into the right hub and—most importantly—being sure the material COMPLETELY SURROUNDS THE

WIRING, from the top to the bottom, completely blocking this end of the horizontal conduit.
20

INSTALLATION

• Pour sealing compound down in between the two fiber rope dams filling the remaining space. Pour slowly,

being careful not to trap air bubbles. Immediately wipe off any spilled sealing compound.

• Screw conduit domed-cover back onto conduit opening.

NOTE: INITIAL SETUP OF SEALING COMPOUND WILL OCCUR IN APPROXIMATELY 30 

MINUTES HOWEVER, THE SEALING COMPOUND REQUIRES A MINIMUM OF 8 HOURS ABOVE 

32oF TO DEVELOP SUFFICIENT STRENGTH TO WITHSTAND EXPLOSIONS. 
21

Operation 

Environmental Operating Conditions 

POLLUTION DEGREE*: 

INSTALLATION 

CATEGORY*: ALTITUDE: 

HUMIDITY: 

ELECTRICAL SUPPLY: 

VOLTAGE TOLERANCE: 

TEMPERATURE: PRODUCT 

USAGE: 

2 

II 

2000 Meters MSL (Mean Sea Level) 

80% maximum, non-condensing 

120VAC or 230VAC 

±10% of normal rated line 

15oC to 40oC 

This product is intended for use indoors only 

*Refer to IEC 664-1

 Start-Up Procedure

Rotate the control knob clockwise to lower the

temperature and counterclockwise raise it.

To check chamber temperature, place a dial

thermometer on a shelf in the center of the chamber. 

Initially, rotate the temperature control knob to an

arbitrary setting. Allow approximately 2 hours for the

temperature to initially stabilize. Check the temperature

and compare with the dial setting.

Adjust dial further to reach the desired operating

temperature. After chamber initially stabilizes, allow 1/2

to 1 hour for the chamber temperature to stabilize after

subsequent temperature adjustments.

Because the markings on the dial do not indicate 

specific temperatures, use them AS REFERENCE

POINTS ONLY for any future setting of the temperature.

If the room thermostat is turned below 60oF at night,

consider setting the temperature control one step

colder. It should be left at this setting for the night time

period; return temperature control to original setting

when the room thermostat is returned to its normal

setting.

 Restart Procedure

If unit is unplugged or turned off, allow 3 minutes before

restarting or plugging it back in.
22

OPERATION

Control Panel Lights

(Model 04LFEETSA)

The freezer is controlled with the thermostat knob. 

1. Red control light

2. Yellow control light

3. Thermostat knob

Modes of operation 

Continuous operation – the yellow light is on. 

The cooling system operates continuously. This mode is 

selected when you want to freeze large amounts. 

Automatic operation – the yellow light is off. 

The interior temperature is controlled by the thermostat 

which turns the cooling system on and off. 

The frequency of switching the cooling system on and off 

depends on: 

• The thermostat knob position (thermostat setting),

• How often you open the door: and

• Ambient temperature.

Temperature Selection 

When the freezer operates automatically, the interior 

temperature is controlled by the thermostat. The most 

suitable thermostat positions are in the middle between 

max and min. 

We recommend setting the thermostat on positions 

towards max only in case you want to accelerate cooling 

or when the operation mode shall correspond to the 

ambient temperature. 

Thermostat setting towards min help you save energy 

provided that the freezer is loaded with smaller amount. 

Change of the ambient temperature affects temperature in 

the interior of the appliance. Choose correct setting of the 

thermostat knob. 

Red Control Light 

The red light is on when there is something wrong with the 

temperature in the freezer. 

1 1 3 
23

OPERATION

How to Save Energy

• Be sure to follow location suggestions as mentioned in the

previous INSTALLATION section.

• Wipe moisture from glassware or other materials before

placing them in a unit.

• Don't overcrowd the unit. Too many items can increase

electrical energy demand in order to keep everything

cool.

• Close the door as soon as possible in hot, humid

weather.

• Make certain that the door is closed tightly.

• As soon as frost has accumulated to 1/4”, defrost.

• Keep containers covered, when possible, to reduce

moisture buildup.

• Set operating temperature no colder than necessary for

the items being refrigerated.

Safety Tips

• After a unit is in operation, do not touch the cold surfaces,

particularly when hands are damp. Skin may adhere to the

extremely cold surfaces.

• Never disconnect your unit by pulling on the power cord.

Always grip the plug securely and pull straight out from the

outlet.

• Do not use a power cord that shows cracks or abrasions.

Have a qualified electrician repair or replace damaged

cords immediately.
24

Troubleshooting 

In the event that your unit is not operating properly, check the following before calling for service 

assistance this may save you the cost of unnecessary service calls. 

Symptom Possible Cause of Problem

Unit not operating. Make certain that the unit is connected to a grounded outlet. 

Make certain that temperature control knob is ON. 

Check that circuit breaker is not tripped or fuse is not blown.

Unit runs continuously. Make certain that there is no heavy frost accumulation. If there is, defrost 

unit. 

Look at condenser to see if there is layer of dust or lint. Clean if required 

with a dry brush or vacuum. 

A leak around the door gasket will allow cold air to escape. This causes unit 

to work harder than necessary to maintain cold temperatures. Re-seat or 

replace the gasket if worn. 

If you have temperature set too cold, this may cause unit to run 

continuously. Check optimum running temperature. 

Is the ambient air over 43oC (109oF), or the units located close to 

heat sources?If possible, move to a different location. 

An unusually high frequency of door openings and closings can increase 

operating load. Unit will stabilize as these are decreased.

Noise problems. This can be caused by contents of unit being set too close and rattling 

against each other. Rearrange contents as needed. 

Hissing or gurgling noise is caused by refrigerating fluid circulating and is 

normal. 

Noise can result if unit is not level on floor. Check with level. 

Fan noise: normal airflow can cause this, not a problem.
25

Maintenance 

Note

Make no attempt to service or repair a 

Thermo Scientific product under 

warranty before consulting your 

Thermo Scientific dealer. After the 

warranty period, such consultation is 

still advised, especially when the 

repair may be technically 

sophisticated or difficult. If assistance 

is needed beyond what the distributor 

can provide, please call Customer 

Service at 800-438-4851. No 

merchandise should be returned 

directly to the factory without 

obtaining a Return Materials 

Authorization (RMA) number from 

Customer Service. 

Warning 

Disconnect plug from electrical outlet 

before attempting any maintenance 

or repair of this unit. 

Any internal adjustments or repairs 

must be performed by a qualified 

service representative. 

Cleaning of Units 

• Disconnect power cord from its outlet.

• Set the temperature control to the OFF position.

• The unit designs permit easy and rapid cleaning and

should not take more than a few minutes. Remember to

wear protective gloves to prevent frost bite, especially

when removing items from freezer units

• Do not use abrasive scouring powders, waxes, solvents,

furniture polish, undiluted detergents or cleansers

containing petroleum products on the surfaces of units.

Interior/Exterior and Door Gaskets 

A solution of mild soap and water can be used for cleaning the 

interior, exterior and door gaskets with a soft, clean cloth. Rinse 

with clean water and dry thoroughly before reconnecting and 

turning on the unit. 

Condenser 

With forced-fan vented units, remove the screws that mount the 

grill to the unit. Pull temperature control knobs straight out. This 

will expose condenser for cleaning
26

MAINTENANCE

Caution

Do not use any sharp instrument, 

blade or scraper to remove ice 

and frost on refrigerator surfaces 

because of the very real danger of 

puncturing the cooling coil. 

Caution

Do not use any electrical device 

to defrost the unit. 

When frost accumulates to 1/4" or 

more, the operating efficiency of 

the unit will be affected. 

Manual Defrost Procedure 

(Models 05FFEETSA, 05EFEETSA, 05LCEETSA,

and 05LCEETSV)

Use the following procedure to defrost manually: 

• Rotate the temperature control knob to the OFF

setting.

• Disconnect the power cord from its outlet.

• Remove contents of unit. If practical, wrap contents in

paper and then in a heavy blanket to maintain

temperature of items, especially those removed from a

freezer. Wear protective gloves to prevent frostbite

when handling cold items.

• Open door and allow free circulation of ambient air.

• To speed the process, place pan of warm water inside

the chamber.

• Wipe out the interior.

• Replace contents.

• Reconnect power cord to outlet and set temperature

control to desired operating temperature.
27

MAINTENANCE

Caution

Do not use any sharp instrument, 

blade or scraper to remove ice and 

frost on refrigerator surfaces 

because of the very real danger of 

puncturing the cooling coil. 

Caution

Do not use any electrical device to 

defrost the unit. 

When frost accumulates to 1/4" or 

more, the operating efficiency of the 

unit will be affected. 

Manual Defrost Procedure 

(Model 12LCEETSA Only) 

Use the following procedure to defrost manually: 

1. Turn the temperature control knob to the 0 setting.

2. Disconnect the power cord from its outlet.

3. Remove contents of unit. If practical, wrap contents in

paper and then in a heavy blanket to maintain

temperature of items, especially those removed from a

freezer. Wear protective gloves to prevent frostbite

when handling cold items.

4. Open door and allow free circulation of ambient air.

5. Fold out the draining spout, mount the enclosed

extension and place a bowl under it.

6. To speed the process, place a pan of warm water

inside the chamber.

7. Wipe out the interior when all frozen condensate has

thawed.

8. Replace contents.

9. Reconnect power cord to outlet and set temperature

control to desired setting.

Model 12LCEETSA Illustrating drip pan position 
28

MAINTENANCE

Reversing the Front Door 

(Model 10ECEETSA) 

1. Switch off at the socket outlet and pull out the mains

plug.

2. Remove all items from inside the appliance.

3. Using a cross-headed screwdriver, remove the

top hinge screws.

4. Remove the top hinge while holding the door.

5. Lift the door carefully from the lower hinge.

6. Remove the lower hinge.

7. Reposition the upper hinge.

8. Remove the lower hinge plugs. Remove plugs

with tip of small a screwdriver.

9. Reposition the door hinge bushing and plugs on

the opposite side.

10. Reposition the lower hinge on the opposite side

and fit the hinge washer.

11. Refit the door.

12. Locate the top hinge in the door and loosely

attach to the cabinet.

13. Align the door and then tighten screws firmly on

the top hinge. Check that the door open and

closes easily. Make sure there are no gaps that

would allow air into the cabinet, adjust the top

hinge if necessary.
29

MAINTENANCE

Reversing the Front Door

(Model 05FREETSA, 05EREETSA and 05LREETSA) 

1. Remove bolts and plugs using a screwdriver.

2. Unscrew the bolts connecting the upper hinges to the

upper platform. Remove the upper platform by first pulling

it forward and then lifting it upwards.

3. Remove the door by unscrewing the bolts in the upper

hinges.

4. Remove the buffer by unscrewing the bolts on both of its

sides.

5. Break off the part behind the buffer through which the

lower hinges pass using a knife or your hands.

6. Remove the lower hinge and mount it on the other side

tightening its bolts.

7. Remove the hole plugs of the door handle with a knife.

8. Fit the hole plugs of the door handle to the other side.

9. Fit the buffer to its place.

10. Mount the door to its place.

11. Fit the upper hinges into their places—do not tighten the

bolts, but leave then loose.

12. Adjust the vertical and parallel position of the door by

eyesight then tighten the bolts of the upper hinges.

13. Fit the upper platform into its place.

14. Fit the bolts connecting the upper hinges and the upper

platform but, do not tighten them.

15. Adjust the upper platform until it is level, then tighten the

bolts in operations

16. Fit the plugs of the upper platform.
30

MAINTENANCE

Reversing the Front Door 

(Model 12LCEETSA) 

1. Turn thermostats to 0 and disconnect power cords

2. Remove all items from inside the appliance

3. Using the proper hex screwdriver, remove the

top hinge screws

4. Remove the top hinge while holding the door

5. Lift the door carefully from the middle hinge

6. Remove the middle hinge and lift the lower door

off of pivot

7. Remove the kick panel. Unscrew nuts and

pivot, remount on the opposite side of the angle

iron.
31

MAINTENANCE

8. Remove plastic caps on the new middle hinge side.

Remove screw and mount lower door and fasten middle

hinge

9. Remove the door handle by removing mounting screws

and remount on opposite side

10. Mount the upper door on pivot and fasten the upper hinge

on opposite side then tighten screws

11. For adjustment of doors, loosen screws and reposition door

into place and tighten screws

12. Move handle to opposite side of door
32

Replacement Parts 

To obtain replacement parts information and 

pricing, please call the Customer Service 

Department at 1-800-438-4851 and have the 

unit’s model, serial and code numbers 

available. This information is located on data 

plates on the rear of the unit. 
33

Ordering Procedures 

Please refer to the Specification Plate for the complete 

model number, serial number, and series number 

when requesting service, replacement parts or in any 

correspondence concerning this unit. 

All parts listed herein may be ordered from the Thermo 

Scientific dealer from whom you purchased this unit or 

can be obtained promptly from the factory. When service 

or replacement parts are needed we ask that you check 

first with your dealer. If the dealer cannot handle your 

request, then contact our Customer Service Department 

at 800-438-4851. 

Prior to returning any materials, please contact our 

Customer Service Department for a “Return Materials 

Authorization” number (RMA). Material returned without 

an RMA number will be refused. 
34

One Year Limited Warranty 

This Thermo Scientific product is warranted to be free of defects in materials and workmanship for one 

(1) year from the first to occur of (i) the date the product is sold by the manufacturer or (ii) the date the 

product is purchased by the original retail customer (the “Commencement Date”). Except as expressly 

stated above, the MANUFACTURER MAKES NO OTHER WARRANTY, EXPRESSED OR IMPLIED, 

WITH RESPECT TO THE PRODUCTS AND EXPRESSLY DISCLAIMS ANY AND ALL WARRANTIES, 

INCLUDING BUT NOT LIMITED TO, WARRANTIES OF DESIGN, MERCHANT ABILITY AND FITNESS 

FOR A PARTICULAR PURPOSE. 

An authorized representative of the manufacturer must perform all warranty inspections. In the event of a 

defect covered by the warranty, we shall, as our sole obligation and exclusive remedy, provide free 

replacement parts to remedy the defective product. In addition, for products sold within the continental 

United States or Canada, the manufacturer shall provide free labor to repair the products with the 

replacement parts, but only for a period of ninety (90) days from the Commencement Date. 

The warranty provided hereunder shall be null and void and without further force or effect if there is any (i) 

repair made to the product by a party other than the manufacturer or its duly authorized service 

representative, (ii) misuse (including use inconsistent with written operating instructions for the product), 

mishandling, contamination, overheating, modification or alteration of the product by any customer or third 

party or (iii) use of replacement parts that are obtained from a party who is not an authorized dealer of 

Thermo Scientific products. 

Heating elements, because of their susceptibility to overheating and contamination, must be returned to 

the factory and if, upon inspection, it is concluded that failure is due to factors other than excessive high 

temperature or contamination, the manufacturer will provide warranty replacement. As a condition to the 

return of any product, or any constituent part thereof, to the factory, it shall be sent prepaid and a prior 

written authorization from the manufacturer assigning a Return Materials Number to the product or part 

shall be obtained. 

IN NO EVENT SHALL THE MANUFACTURER BE LIABLE TO ANY PARTY FOR ANY DIRECT, 

INDIRECT, SPECIAL, INCIDENTAL, OR CONSEQUENTIAL DAMAGES, OR FOR ANY DAMAGES 

RESULTING FROM LOSS OF USE OR PROFITS, ANTICIPATED OR OTHERWISE, ARISING OUT OF 

OR IN CONNECTION WITH THE SALE, USE OR PERFORMANCE OF ANY PRODUCTS, WHETHER 

SUCH CLAIM IS BASED ON CONTRACT, TORT (INCLUDING NEGLIGENCE), ANY THEORY OF 

STRICT LIABILITY OR REGULATORY ACTION. 

E-mail: mkt@thermofisher.com 

Web: www.thermo.com 
WEEE Compliance

WEEE Compliance. This product is required to comply with the European 

Union’s Waste Electrical & Electronic Equipment (WEEE) Directive 2002/96EC. 

It is marked with the following symbol. Thermo Fisher Scientific has contracted 

with one or more recycling/disposal companies in each EU Member State, and this 

product should be disposed of or recycled through them. Further information on 

Thermo Fisher Scientific compliance with these Directives, the recyclers in your 

country, and information on Thermo Scientific products which may assist the 

detection of substances subject to the RoHS Directive are available at 

www.thermo.com/

WEEE Konformittät. Dieses Produkt muss die EU Waste Electrical & Electronic 

Equipment (WEEE) Richtlinie 2002/96EC erfüllen. Das Produkt ist durch 

folgendes Symbol gekennzeichnet. Thermo Fisher Scientific hat Vereinbarungen 

getroffen mit Verwertungs/Entsorgungsanlagen in allen EU-Mitgliederstaaten und 

dieses Produkt muss durch diese Firmen widerverwetet oder entsorgt werden. 

Mehr Informationen über die Einhaltung dieser Anweisungen durch Thermo 

Fisher Scientific, die Verwerter und Hinweise die Ihnen nützlich sein können, die 

Thermo Scientific Produkte zu identifizieren, die unter diese RoHS. 

Anweisungfallen, finden Sie unter www.thermo.com/

Conformità WEEE. Questo prodotto deve rispondere alla direttiva dell’ Unione 

Europea 2002/96EC in merito ai Rifiuti degli Apparecchi Elettrici ed Elettronici 

(WEEE). È marcato col seguente simbolo.Thermo Fisher Scientific ha stipulato 

contratti con una o diverse società di riciclaggio/smaltimento in ognuno degli Stati 

Membri Europei. Questo prodotto verrà smaltito o riciclato tramite queste 

medesime. Ulteriori informazioni sulla conformità di Thermo Fisher Scientific con 

queste Direttive, l’elenco delle ditte di riciclaggio nel Vostro paese e informazioni 

sui prodotti Thermo Scientific che possono essere utili alla rilevazione di sostanze 

soggette alla Direttiva RoHS sono disponibili sul sito www.thermo.com/

Conformité WEEE. Ce produit doit être conforme à la directive européenne 

(2002/96EC) des Déchets d’Equipements Electriques et Electroniques (DEEE). Il 

est marqué par le symbole suivant. Thermo Fisher Scientific s’est associé avec une 

ou plusieurs compagnies de recyclage dans chaque état membre de l’union 

européenne et ce produit devraitêtre collecté ou recyclé par celles-ci. Davantage 

d’informations sur laconformité de Thermo Fisher Scientific à ces directives, les 

recycleurs dans votre pays et les informations sur les produits Thermo Scientific 

qui peuvent aider le détection des substances sujettes à la directive RoHS sont 

disponibles sur www.thermo.com/
38

Important

For your future reference and when contacting the factory, please have the following 

information readily available:

Model Number: 

Serial Number: 

Date Purchased: 

The above information can be found on the dataplate attached to the equipment. If

available, please provide the date purchased, the source of purchase (manufacturer 

or specific agent/rep organization), and purchase order number. 

IF YOU NEED ASSISTANCE:

SALES DIVISION

Phone: 828/658-2711 

 800/252-7100

FAX: 828/645-3368

LABORATORY PARTS and SERVICE

Phone: 800/438-4851

FAX: 828/658-2576

TECHNICAL SUPPORT

Phone: 800/438-4851
Thermo Fisher Scientific Inc.

275 Aiken Road

Asheville, NC 28804

United States

www.thermofisher.com

057-196-00 Rev. D