PRIME UNDER COUNTER BI SERIES FREEZERS

LABHP‐5‐UFMB, LABHP‐5‐UFMBSS

OWNER’S INSTRUCTIONS

This manual describes how to operate and care for your appliance to get the best, most efficient, performance.

READ THIS BOOK!

Note to Customer:

This merchandise was carefully packed and thoroughly inspected before leaving our plant.  Responsibility for its

safe delivery was assumed by the carrier upon acceptance of the shipment.    As directed on the side of your

packing carton, claims for loss or damage sustained in transit must be made on the carrier as follows:

A.) Visible Loss, Damage, Shortage External Evidence of Loss or Damage: This type of damage must be noted on

the freight bill and acknowledged by the carrier’s agent (driver) at time of delivery. Make sure you get a signed

copy. Send a written request for an inspection to the carrier.  

B.) Concealed Damage: This type of damage may not be discovered until the unit is being unpacked. When

concealed damage is discovered, stop unpacking immediately and contact the carrier immediately to report

the claim and request an inspection. This should be done as soon as possible and, in any case, must be done

within 15 days or receiving the merchandise. If at all possible, do not move the item and save all packaging

material for carrier’s inspection.  

C.) FAILURE TO FOLLOW THESE INSTRUCTIONS MAY RESULT IN THE CARRIER REFUSING TO HONOR YOUR

COMPANY’S CLAIM. UNDER NO CIRCUMSTANCES SHOULD THE MERCHANDISE BE RETURNED TO THE

MANUFACTURER. NO RETURNS WILL BE ACCEPTED WITHOUT PRIOR AUTHORIZATION.
2

TABLE OF CONTENTS

RECEIVING.......................................................................................... 3

INSTALLATION.................................................................................... 4

TEMPERATURE CONTROL ................ERROR! BOOKMARK NOT DEFINED.

REFRIGERATOR OPERATION ............................................................... 6

QUICK TROUBLESHOOTING GUIDE ................................................... 10

MAINTENANCE AND CLEANING.......ERROR! BOOKMARK NOT DEFINED.

PRODUCT WARRANTY INFORMATION.............................................. 15

BEFORE CALLING THE MANUFACTURER’S TECHNICAL SERVICE, please have the unit’s model

and serial number ready as well as the problem description.  The model and serial number is

located on the serial tag which can be found on the interior left OR right upper wall of the unit.

For convenience, you may want to record the following information here for easy access in the

future.

Model number: __________________________________

Serial number: __________________________________

Date of delivery: _________________________________

   

Horizon Scientific 

125 Varnfield Drive 

Summerville, SC 29483 

Phone: 800-648-4041 

Fax: 843-821-8051 

Warranty: 

Two-Year Parts and Labor 

5 Years Compressor Parts 

Revision Date 

03/09/2012 
3

RECEIVING

Your unit was built, packaged, and inspected with extreme care. We shipped it to you using

carriers we trust with a proven track record of careful handling, good customer service, and on

time delivery.  Unfortunately, regardless of all of these efforts sometimes accidents happen and

occasionally those accidents result in shipping damage. When the carrier picked up the

merchandise from us, they assumed responsibility for its condition en route to you. Thus, any

claims for shipping damage must be filed with the carrier. Like anybody else, carriers don’t like

to pay out on insurance claims, so their claims procedures and requirements are very

restrictive. You should consult the carrier’s website for their specific claims procedures. You

should also know that time is of the essence.  

There are two general types of shipping damage. The first is visible damage. This type of

damage includes visible loss, damage, shortage or any external evidence of loss or damage that

is visible at time of delivery. This type of damage must be noted in detail on your delivery

receipt. Make sure the driver signs and dates the delivery receipt, acknowledging the

damage. This has to happen at the time of delivery or it won’t happen at all. Keep a copy for

your records and send another to the carrier’s damage claims department along with a formal

request for an inspection report. Follow up with a phone call. Their contact information can be

found on the carrier’s web site.  

The second type of shipping damage is concealed damage. This type of damage will probably

not be apparent at time of delivery and may not be discovered until unpacking and inspecting

the unit. Remember, time is of the essence here. You should unpack and inspect the unit as

soon as possible. Each day that passes reduces the likelihood that the carrier will pay the claim.

As soon as the concealed damage is discovered, stop unpacking and retain all packing

materials. Contact the carrier by phone to report the claim. Note the date and time and

person you spoke with. Get a claim number. Follow up with a written letter referencing the

claim number and including a formal request for an inspection. Again, consult the carrier’s

website for specific claim instructions and follow them precisely.  

AS STATED ABOVE, THE CARRIER IS YOUR SOLE SOURCE FOR SATISFACTION OF A DAMAGE

CLAIM. UNDER NO CIRCUMSTANCES SHOULD THE MERCHANDISE BE RETURNED TO THE

MANUFACTURER. NO RETURNS WILL BE ACCEPTED WITHOUT PRIOR AUTHORIZATION.

   
4

INSTALLATION  

UNCRATING – Move your refrigerator as

close to the final location as possible

before unpacking. Remove the carton

and the styrofoam strips.    The

location should be as close as possible

to the power outlet so no extension

cord is needed. This unit requires a

minimum of 3 inches on the sides

and top.  Do not store material on the

top of this unit.  

LEVELING  You can level your unit with

the screw‐type adjustable leveling

legs on the front corners of the unit.    Turn counterclockwise to raise the corner; turn

clockwise to lower it. Leveling legs are required in front only.   Unit may have a slight tilt

from front to back after legs are installed.  This is acceptable and is recommended on this

unit.

POWER SOURCE – The supply circuit to this cabinet

must conform to all National and Local Electrical

Codes.    Consult the cabinet Serial‐Data plate for

voltage, cycle, phase, and amp requirements before

making connection.    VOLTAGE SHOULD NOT VARY

MORE THAN 5% FROM SERIAL PLATE RATINGS.    A

separate circuit is recommended to prevent possible

loss of product due to over‐loading or failure of other equipment on the same circuit.  

PROTECT THE CIRCUIT WITH A 20 AMP DELAY‐TYPE

FUSE OR CIRCUIT BREAKER.    Do not use an extension

cord.  Be sure your unit is properly grounded.  Use the

3‐prong plug provided into a 3‐prong grounded outlet.

(Only this method complies with national electrical

codes, local codes and ordinances.)   Unless the above

grounding method is followed, you are not protected

against severe or lethal shock in the event of a short

circuit of an electrical component or wiring of the unit.

STARTING – There are no compressor shipping bolts to

loosen or valves to open.  All that is necessary after the

unit has been properly leveled is to plug the service

cord into an electrical outlet    When starting this new
5

appliance, allow the cabinet to operate a minimum of four hours or until it has started

cycling normally before placing product in the cabinet.   The motor compressor may start

and stop several times when the unit is first started or after defrosting, especially if the

weather is very hot.  This is only normal functioning of the motor overload protector.  The

motor compressor will cycle normally as soon as the excess heat has been removed.    

Shelf Installation  

Hook shelf clips onto pilaster strips (see Illustration). Position all four shelf clips equal in

distance from the floor for flat shelves. Wire shelves are oriented so that cross support bars are

facing down.

Place shelves on shelf clips making sure all corners are seated properly.
6

Freezer Operation

After the unit is properly installed and power is applied, it will take some time before the

system is cooled down to temperature and cycling normally. You should wait 8 hours on the

first startup before beginning to add product to the unit. This ensures that the unit is installed

and operating properly before being put into service   After this wait time, the unit should be

cycling in the design temperature range. The units are calibrated before leaving the factory, so

no adjustment should be necessary.  

Loading the units will again cause temperature to rise as the warmer product is introduced into

the compartment. If a large amount of product is to be introduced, it is a good idea to do it in

stages, allowing several hours between stages to allow temperature to stabilize again before

introducing additional warm product. This will minimize the temperature transient while

loading. Other tips for successful loading include:  

‐ Leave about 2‐3 inches of free space along the back and sides of the unit to allow for proper

air flow and, therefore, more even cooling of the product.  

‐ When loading the top shelf, avoid blocking the evaporator fan(s) (if installed). There should

be at least 4 inches of clearance around the fan to allow proper air flow.  

‐ Do not overload the unit. A full load is about 70% full. Additionally, the load should be

distributed evenly from top to bottom and side to side for best results.  

‐ Minimize the time the door is open. On top of letting the cold air out, you are also letting

warm, moist air in which can result in more condensation and/or frost in the unit.  

Remember that the units are calibrated to the design temperature range before leaving the

factory. We also do extensive testing to ensure that these temperatures will result in product

temperatures in the desired band. There should be no need to adjust the temperature control

on these units, but if it is necessary, they can be adjusted using the electronic temperature

control on the front of the unit.

It is STRONGLY RECOMMENDED that you contact the manufacturer’s Technical Support

Department prior to performing any temperature adjustments to ensure the adjustment is

necessary and, if so, it is performed correctly.  

   
7

TempTech­2000 Electronic Temperature Control

Description and Functions

The T‐1000 Electronic Temperature Control is a digital microprocessor based temperature

control, indication, and alarm system. It continuously monitors the temperature of the

refrigerator/freezer through its sensor assembly.  When the temperature warms to the point

where it reaches the upper temperature limit, the controller sends a signal to turn on the

compressor, which cools the unit. The snowflake icon will illuminate on the display. The

compressor is kept running until the temperature reaches the cold setting.   When this lower

temperature set point is reached the controller turns off the compressor and the snowflake

icon extinguishes.  The refrigerator/freezer warms over time and the process continues.  

The T‐1000 also provides the following functions:

 Red LED for visual indication of an alarm condition

 Beeper for audible indication of an alarm condition

 A preprogrammed 5 minute delay from the time a setpoint is attained until an alarm

condition is declared. This prevents spurious alarms for things like door openings.  

 A choice of remote alarm contacts, one normally open and one normally closed for

connecting remote alarms or annunciation systems.

5

Operating Mode

The TempTech‐2000 normally displays the current interior temperature in degrees Celsius. To

display the compressor off setpoint, press the SET button and release. The display will

momentarily display the setpoint and then revert to the current temperature.

The control also records the warmest and the coldest temperatures seen by the controller since

the memory was last reset.  To access these temperatures, press the up arrow for the warmest

temperature or press the down arrow for the coldest temperature.  To clear out the memory,

press the corresponding arrow button. Immediately press and hold the SET button until the

“rSt” message is displayed. The corresponding temperature memory is now erased.
8

Programming Mode

Note: The TempTech‐2000 programming mode gives access to several

functions that are not intended for user adjustment. The procedures

below provide direction to the functions that you can adjust. Please do

not attempt to change any of the other functions in the programming

area or your unit may be damaged or become unstable.    

Locking and Unlocking the Control

The control keys can be locked out to prevent inadvertent operation.  

‐ To lock the control keys, simultaneously press and hold the up arrow and

the down arrow. The display will show “POF”.  At this point you can only

use keys to view the SET temp and the minimum and maximum stored

temperatures.  

‐ To unlock the control keys, simultaneously press and hold the up arrow

and the down arrow. The display will show “Pon”. The control keys are

now unlocked and you can perform the programming functions below.  

Adjusting the temperature control setpoints:

NOTE: The temperature control setpoints are preset at the factory to

provide the desired air and product temperatures inside of the unit and

require no further adjustment. Please contact the manufacturer’s

Technical Service Department before making any adjustments to

determine if adjustment is necessary and, if so, to make sure it is

performed properly.

‐ To set the compressor off setpoint, press and hold the SET key for 2 seconds. The C or F icon

will flash. Use the up or down keys to change the value. Press Set again to lock in the value.

‐ To set the compressor on setpoint, you will need to adjust the differential, or Hysteresis

setting. This is the number of degrees the controller will allow temperature to drift up until

it restarts the compressor. For example, if the SET point is 3 degrees Celsius and the

Hysteresis is set at 3, then the compressor will cycle from 3 to 6 degrees Celsius. To change

this value, “Hy” will display meaning Hysteresis. Press Set again and the current value will

be displayed. Use the UP and/or DOWN arrow keys to adjust the value and press SET again

to lock it in. Press SET and UP to return to Operating Mode.
9

Adjusting other User Adjustable settings:

‐ To change between Celsius and Fahrenheit scales, press SET and DOWN keys for 3 seconds

to enter programming mode. “Hy” will display. Press the UP or DOWN keys several times to

cycle through the parameters until “CF” is displayed. Press SET to display the current setting

(either C for Celsius or F for Fahrenheit). Use the UP or DOWN key to choose the desired

setting. Press SET again to lock it in. Press SET and UP to return to Operating Mode.

‐ To change the Hi or Low Alarm setpoints. , press SET and DOWN keys for 3 seconds to enter

programming mode. “Hy” will display. Press the UP or DOWN keys several times to cycle

through the parameters until “ALU” for Alarm Upper (High Temperature Alarm) or ALL for

Alarm Lower (Low Temperature Alarm) is displayed. Press SET to display the current value.

Use the UP and/or DOWN key to choose the desired value. Press SET again to lock it in.

Press SET and UP to return to Operating Mode.

Remote Alarm Contacts

The Remote Alarm Contracts are located on top of the unit directly above the controller. See

the picture below for connection information.  
10

Quick Troubleshooting Guide

Check these items before calling for service

PROBLEM: POSSIBLE CAUSE / SOULTIONS:

Unit does not run

 Electrical circuit is not 110‐120V 60Hz.

 The power cord is not plugged in.

 No power at electrical outlet. Check to make sure

breaker is not tripped or fuse is not blown. Additionally,

make sure unit is not plugged into a Ground Fault Circuit

Interrupter (GFCI) type of outlet.  

Unit does not maintain at the

proper temperature

 Check the room temperature.  We recommend the

refrigerator or freezer should be placed in the air

conditioned room between 65°F to 85°F.  If the room

temp is too warm, the refrigerator or freezer may not be

able to maintain the interior temp at proper range.   

 Door is not closed properly.

 Amount of stored product is overloaded.

 Product replacements are pushed against rear wall or

interrupted the proper refrigerator air circulation.  For

the proper air circulation, place the products evenly on

each shelf.  Do not push against the refrigerator’s rear or

side walls.

 Evaporator is blocked by frost or ice.  Remove the

products, unplug the refrigerator or freezer power, and

allow the unit to defrost.  If the problem still exists, call

for service.  

Appliance runs too long

 Prolong door openings.

 Control set too cold.

 Room temperature is high which will make the unit work

harder to keep cool.  

Temperature of external wall

surface is warm

 The exterior walls can be as much as 30 degrees warmer

than room temperature due to the embedded

condenser coils. This is normal when the unit is

operating.   

Compressor noises

 Check the unit level. Also ensure none of the pipes in

the compressor area are touching each other or against

the side.  

 Compressor may be overheated.  Please check the room

temp and ensure the range is within 65°F to 85°F.   

 A buzzing noise may indicate improper voltage to the

compressor. Check supply voltage.  

 If the problem still exists, call for service.
11

PROBLEM: POSSIBLE CAUSE / SOULTIONS:

Moisture collects inside

 Door gasket is not sealing properly. Check for debris,

cracks, and items passing through door at the gasket.         

 The refrigerator or freezer is facing a doorway or is

underneath of air conditioning vent. Relocate the unit or

redirect air vent.

 Too many door openings. Minimize time door is open.

 Hot, humid weather increases condensation.

 Make sure there is a water trap (U‐shaped loop) in the

drain tube near the compressor. This will “trap” a small

amount of water in the loop and prevent air from

entering the chamber through the tube.  

Moisture collects on outside

surface  Hot, humid weather increases condensation.

 As humidity decreases, moisture will disappear.

The display shows “SO” and

the compressor does not run  The temperature sensor is open. Call for service.                  

The display shows “SC” and

the compressor does not run  The temperature sensor is shorted. Call for service.  

The display shows “HH” and

the compressor does not run  The ambient temperature is above 40C/104F. Stop using

immediately.

The display shows “LL” and the

compressor does not run  The ambient temperature is below ‐40C/‐40F. Stop using

immediately.  

MOISTURE DURING THE SUMMER SEASON

The amount of moisture, condensation, or high humidity related issues increase during the summer

and, in most cases, will self‐resolve when the weather cools down.  Please note a refrigeration

system does NOT generate moisture or water but simply condenses the moisture that is already in

the chamber. Keeping the unit in an air conditioned, low humidity space will resolve many issues.

Other things you should check

1. Location of the refrigerator (See Quick Troubleshooting Guide above)

2. Door sealing and frequency of door opening event (See Quick Troubleshooting Guide above)

3. Make sure there is a water trap (U‐shaped loop) in the drain tube near the end. This will

“trap” a small amount of water in the loop and prevent air from entering the chamber

through the tube.  

BEFORE CALLINGTHE MANUFACTURER’S TECHNICAL SUPPORT DEPARTMENT, please have the

unit’s model and serial number ready as well as the problem description.  The model and serial

number is located on the serial tag which can be found on the interior left upper wall of the

unit.
12

Maintenance and Cleaning

These units require very little maintenance. They should be cleaned periodically to keep them

running efficiently. Use the cleaning agents and suggestions below for best results.  

If 1⁄4 to 1⁄2 inch of frost accumulates or if the unit will not maintain proper temperature, it should

be defrosted per the procedure on page 14.  

CLEANING

PART CLEANING AGENTS TIPS AND PRECAUTIONS

Interior and Door

Liners

Soap and water

Baking soda and water

Use 2 tablespoons of baking soda in 1 quart of

warm water

Be sure to wring excess water out of sponge or

cloth before cleaning around controls, light bulb

or any electrical parts.

Glass Doors (if

applicable)

Soap and water Clean the glass with a mild detergent and water

on a soft cloth or sponge. Rinse with water and

wipe dry

Door Gaskets Soap and water Wipe gaskets and their seating surfaces with a

clean soft cloth

Shelves Soap and water Do not wash removable shelves in dishwasher

Exterior and

Handles

Soap and water

Non Abrasive Glass

Cleaner

Do not use commercial household cleaners,

ammonia, or alcohol to clean handles

Use a soft cloth to clean smooth handles

Do not use a dry cloth to clean smooth handles

Stainless Steel Care and Cleaning

Use the proper tools‐  When cleaning your stainless steel surfaces, take care to use non‐

abrasive tools.    Soft cloths and plastic scouring pads will not harm the steel’s passive layer.  

Stainless steel pads can also be used, but the scrubbing motion must be in the directing of the

polishing lines.

Clean with the polish lines‐ Some stainless steels come with visible polishing lines or “grain”.  

When visible lines are present, you should always scrub in a motion that is parallel to them.  

When the grain cannot be seen, play it safe and use a soft cloth or plastic scouring pad.

Use alkaline, alkaline chlorinated or non‐chloride containing cleaners‐  While many

traditional cleaners are loaded with chlorides, there is now an ever increasing choice of non‐

chloride cleaners.   If you are not sure of your cleaner chloride content, contact your cleaner

supplier.    If they tell you that your present cleaner contains chlorides, ask if they have an

alternative.  Avoid cleaners containing quaternary salts as they also can attach stainless steel

and cause pitting and rusting.
13

Treat your water‐ Though this is not always practical, softening hard water can do much to

reduce deposits.  There are certain filters that can be installed to remove corrosive elements.  

If you are not sure of the proper water treatment, call a treatment specialist.

Keep your equipment clean‐ Use proper cleaners at recommended strength.  Clean frequently

to avoid build‐up of hard stubborn stains.

Rinse, Rinse, Rinse‐ If chlorinated cleaners are used, you must rinse and wipe dry immediately.  

The sooner you wipe off standing water, the better.  After wiping the unit down, allow it to air

dry. The oxygen helps maintain the stainless steels passivity film.

Never use hydrochloric acid (muriatic acid) on stainless steel.

Regularly restore /  passivate stainless steel

Job Cleaning Agent Comments

Routine Cleaning Soap, ammonia,

detergent medallion

Apply with cloth or

sponge

Fingerprints & smears Arcal 20, Lac‐O‐Nu

Ecoshine

Provides barrier firm

Stubborn stains &

discoloration

Cameo, Talc, Zud, First

Impression

Rub in direction of polish

lines

Grease & fatty acids, blood Easy‐off, DeGrease It

Over Aid

Excellent removal on all

finishes

Grease & oil Any good commercial

detergent

Apply with a sponge or

cloth

Restoration – Passivation Benefit, Super Sheen   

Passive Film Breakdown

 Corrosion on metals is everywhere.    We recognize it quickly on iron and steel as

unsightly yellow/ orange rust.    Such metals are called “active” because they actively

corrode in the natural environment.

 Contrary to popular belief, Stainless Steels are susceptible to rusting. The first signs are

on the microscopic level.  If you were to look at them under a microscope or through a

magnifying glass, you would see small pits and cracks will grow and deepen while all the

time exuding unsightly, red‐orange rust. More severe and visible cracking can also take

place.

 Stainless steels are protected by a thin layer of chromium oxide, and it important to

keep that film of chromium oxide intact. If the passive film of your stainless steel has

been broken, you are subject to corrosion and rust. The integrity of the chromium oxide

can be compromised by:

‐ Mechanical abrasion from steel pads, brushes, and scrapers.

‐ Hard Water leaves mineral deposits behind that, if left to dry will break down the

passive layer and rust your stainless steel.

‐ Chlorides are in water, food, and table salt, but most prominently in cleaning agents.   

   
14

DEFROSTING

It is important to defrost and clean the unit when 1⁄4 to 1⁄2 inch of frost has accumulated or when

the unit will not maintain proper temperature. When defrosting becomes necessary, press the

DEF key for 2 seconds. The compressor will stop. Remove the product and leave the door open

while defrosting the evaporator. The frost should melt and drain through the drain tube to the

drain pan under the unit.  Once the unit is defrosted, clean it using the cleaning suggestions in

the table above. Wipe away any excess water. After 25 minutes, the unit will automatically

return to normal operation. If you wish to end the defrost cycle early, simply press the DEF key

again for 2 seconds. The defrost icon will extinguish on the display and the compressor will

return to normal cycling operation.  

   
15

PRODUCT WARRANTY INFORMATION

HORIZON SCIENTIFIC, INC. PRODUCT WARRANTY

Horizon Scientific, Inc. warrants to the original purchaser every new Horizon Scientific, Inc. refrigerated unit, the

cabinet and all parts thereof, to be free from defects in material or workmanship, when such unit is installed, used,

and maintained in accordance with provided instructions, for a period of two (2) years.  The warranty period starts

two weeks from the date of shipment from Horizon Scientific, Inc.  This two week period allows ample shipping

time so that the warranty will go into effect at approximately the same time your equipment is delivered.  Unless

subject to prior written agreement with Horizon Scientific, Inc., this warranty does not allow for any warranty start

deferment greater than two weeks from date of shipment due to a delayed installation and/or start‐up.  By

purchasing any product from Horizon Scientific, Inc., you and any entity for which you are purchasing acknowledge

and agree to each and every provision contained herein, and all other Notices and Terms provided to Purchaser by

Horizon Scientific, Inc., which are hereby incorporated.

Under this warranty, Horizon Scientific, Inc., through its authorized service organizations, will repair, or at its

option, replace any part found to contain a manufacturing defect in material or workmanship without charge to

the owner for parts, service labor or any shipping or cartage costs.  Replacement or repaired parts will be

warranted for only the unexpired portion of the original warranty.

ADDITIONAL THREE YEAR COMPRESSOR WARRANTY

In addition to the two (2) year warranty stated above, Horizon Scientific, Inc. warrants its hermetically and semi‐

hermetically sealed compressors to be free from defects in both material and workmanship under normal use and

service for a period of three (3) additional years from the end of the initial two (2) year warranty period, but not to

exceed five (5) years after shipment from Horizon Scientific, Inc.

Compressors determined by Horizon Scientific, Inc. to have been defective within this extended time period will, at

Horizon Scientific, Inc.'s option, be either repaired or replaced with a compressor or compressor parts of similar

design and capacity.   

The three (3) year extended compressor warranty applies only to hermetically and semi‐hermetically sealed parts

of the compressor and does not apply to any other parts or components, including, but not limited to, cabinet,

paint finish, temperature control, refrigerant, metering device, driers, motor starting equipment, fan assembly or

any other electrical components.

Horizon Scientific, Inc.’s sole obligation under this warranty is limited to either repair or replacement of parts,

subject to the additional limitations below.  This warranty neither assumes nor authorizes any person to assume

obligations other than expressly covered by this warranty.

NO CONSEQUENTIAL DAMAGES.  Horizon Scientific, Inc. is not responsible for economic loss; profit loss; or special,

indirect or consequential damages, including without limitation, losses or damages arising from contents spoilage

claims whether or not on account of refrigeration failure, electrical failure, power failure, or compressor failure.  
16

HORIZON SCIENTIFIC, INC.’S MAXIMUM CUMULATIVE LIABILITY RELATIVE TO ALL CLAIMS AND LIABILITIES,

INCLUDING OBLIGATIONS UNDER ANY INDEMNITY, WHETHER OR NOT INSURED, SHALL NOT EXCEED THE COST OF

THE PRODUCT(S) GIVING RISE TO THE CLAIM OR LIABILITY.

WARRANTY IS NOT TRANSFERABLE.  This warranty is not assignable and applies only in favor of the original

purchaser/user to whom delivered.  Any such assignment or transfer shall void the warranties herein made and

shall void all warranties, express or implied, including any warranty of merchantability or fitness for a particular

purpose.

NO IMPLIED WARRANTY OF MERCHANTABILITY OR FITNESS FOR A PARTICULAR PURPOSE.  There are no other

warranties, express, implied, or statutory, except the two (2) year warranty and the additional three (3) year

compressor warranty as described above.  These warranties are exclusive and in lieu of all other warranties,

including implied warranty and merchantability or fitness for a particular purpose.  There are no warranties which

extend beyond the description on the face hereof, whether based on contract, warranty, tort (including

negligence), strict liability, indemnity, or any other legal theory, and whether arising out of warranties,

representations, instructions, installations, or non‐conformities from any cause.  Purchaser further acknowledges

that the purchase price of the Product reflects these warranty terms and remedies.

ALTERATION, NEGLECT, ABUSE, MISUSE, ACCIDENT, DAMAGE DURING TRANSIT OR INSTALLATION, FIRE, FLOOD

OR OTHER EXTERNAL CAUSES.   

Horizon Scientific, Inc. is not responsible for the repair or replacement of any parts that Horizon Scientific, Inc.

determines have been subjected after the date of manufacture to alteration, neglect, abuse, misuse, accident,

damage during transit or installation, fire, flood or other external causes.  It does not apply to defects resulting

from failure to properly install, operate or maintain the product in accordance with the printed instructions

provided, or damage caused by the storage of any corrosive material that comes in contact with the interior or

exterior portions of the cabinet, or the use of spark producing equipment or containers (such as galvanized or

carbonized steel containers) that come in contact with any interior portion of the cabinet.

OUTSIDE U.S./CANADA.  This warranty does not apply to, and Horizon Scientific, Inc. is not responsible for, any

warranty claims made on products sold or used outside the United States and Canada.

CHOICE OF LAW/VENUE.  The laws of the State of South Carolina shall govern the validity, interpretation and

enforcement of this warranty, regardless of conflicts of law principles.  Purchaser agrees that proper venue for any

action to enforce the terms of this warranty shall be the Dorchester County District Courts, South Carolina.  

Purchaser submits the jurisdiction of such courts over the Purchaser and the subject matter of any such action.  

Any action for breach of these warranty provisions must be commenced within one (1) year after that cause of

action has accrued.

WARRANTY CLAIMS.  To obtain prompt warranty service, simply contact the manufacturer at 800‐648‐4041.  

Horizon Scientific, Inc.’s shipping records showing date of shipment shall be conclusive in establishing the warranty

period.  All claims should include:  model number of the refrigerator, the serial number of the cabinet, proof of

purchase, date of installation, and all pertinent information supporting the existence of the alleged defect.  Any

repairs must be authorized by Horizon in order for the warranty to be honored.