United States

Department of

Commerce

SPECIFICATIONS,

Technology

Administration

National

Institute of

Standards and

Technology

TOLERANCES, AND

OTHER TECHNICAL

REQUIREMENTS FOR

WEIGHING AND

MEASURING DEVICES

as adopted by the 86th

National Conference on

Weights and Measures 2001

NIST

Handbook44

2002
Specifications,

Tolerances, and 

Other Technical 

Requirements for

Weighing and 

Measuring

as adopted by the 86th National Conference

on Weights and Measures 2001

44 NIST

Handbook 

2002 Edition (issued November 2001)

Supersedes Handbook 44 2001 Edition

Editors:

Tina G. Butcher

Terry L. Grimes

Richard Suiter

Juana Williams

NIST Office of Weights & Measures

Henry V. Oppermann, Office 

Director

Gaithersburg, MD 20899

U.S. Department of Commerce

Donald L. Evans, Secretary

Technology Administration

Phillip J. Bond, Under

Secretary for Technology

National Institute of Standards 

and Technology

Karen Brown, Acting Director
National Institute of Standards and Technology Handbook 44, 2002 Edition 

Natl. Inst. Stand. Technol. Handb. 44, 2002 Ed., 284 pages (Nov. 2001) 

CODEN: NIHAE2 
iii

Foreword

Handbook 44 was first published in 1949, having been preceded by similar handbooks of various designations and in

several forms, beginning in 1918.

Handbook 44 is published in its entirety each year following the Annual Meeting of the National Conference on Weights

and Measures. This 2002 edition was developed by the Committee on Specifications and Tolerances of the National

Conference on Weights and Measures with the assistance of the Office of Weights and Measures of the National

Institute of Standards and Technology (NIST). The handbook includes amendments endorsed by the 86th National

Conference on Weights and Measures during its Annual Meeting in 2001.

The National Institute of Standards and Technology has a statutory responsibility for "cooperation with the States in

securing uniformity of weights and measures laws and methods of inspection." In partial fulfillment of this

responsibility, the Institute is pleased to publish these recommendations of the National Conference.

Karen H. Brown 

Acting Director

National Institute of Standards

 and Technology

This handbook conforms to the concept of primary use of SI (metric) measurements recommended in the Omnibus Trade

and Competitiveness Act of 1988 by citing SI units before inch-pound units where both units appear together and

placing separate sections containing requirements in SI units before corresponding sections containing requirements

in inch-pound units. In some cases, however, trade practice is currently restricted to the use of inch-pound units;

therefore, some requirements in this Handbook will continue to specify only inch-pound units until the Conference

achieves a broad consensus on the permitted SI units. 

In accord with NIST policy, the meter/liter spellings are used in this document. However, the metre/litre spellings are

acceptable, and are preferred by the National Conference on Weights and Measures.

It should be noted that a space has been inserted instead of commas in all numerical values greater than 9999 in this

document, following a growing practice, originating in tabular work, to use spaces to separate large numbers into groups

of three digits. This avoids conflict with the practice in many countries to use the comma as a decimal marker.
PLACE HOLDER PAGE 

THIS PAGE INTENTIONALLY LEFT BLANK 
v

Outline of Contents

Page

Foreword iii

2001 Amendments (2002 Edition) vii

Introduction 1

Section 1.

1.10. General Code 1-1

Section 2.

2.20. Scales 2-3

2.21. Belt-Conveyor Scale Systems 2-37

2.22. Automatic Bulk Weighing Systems 2-45

2.23. Weights 2-51

2.24. Automatic Weighing Systems - Tentative Code 2-55

Section 3.

3.30. Liquid-Measuring Devices 3-3

3.31. Vehicle-Tank Meters 3-19

3.32. Liquefied Petroleum Gas and Anhydrous

 Ammonia Liquid-Measuring Devices 3-27

3.33. Hydrocarbon Gas Vapor-Measuring Devices 3-35

3.34. Cryogenic Liquid-Measuring Devices 3-43

3.35. Milk Meters 3-49

3.36. Water Meters 3-55

3.37. Mass Flow Meters 3-59

3.38. Carbon Dioxide Liquid-Measuring Devices - Tentative Code 3-69

Section 4.

4.40. Vehicle Tanks Used as Measures 4-3

4.41. Liquid Measures 4-7

4.42. Farm Milk Tanks 4-9

4.43. Measure-Containers 4-15

4.44. Graduates 4-17

4.45. Dry Measures 4-21

4.46. Berry Baskets and Boxes 4-23
vi

Outline of Contents (Continued)

Section 5.

5.50. Fabric-Measuring Devices 5-3

5.51. Wire- and Cordage-Measuring Devices 5-7

5.52. Linear Measures 5-9

5.53. Odometers 5-11

5.54. Taximeters 5-13

5.55. Timing Devices 5-19

5.56.(a) Grain Moisture Meters 5-23

5.56.(b) Grain Moisture Meters 5-31

5.57. Near-Infrared Grain Analyzers - Tentative Code 5-37

5.58. Multiple Dimension Measuring Devices - Tentative Code 5-43

Appendices

A. Fundamental Considerations A-1

B. Units and Systems of Measurement B-1

C. General Tables of Units of Measurement C-1

D. Definitions D-1
vii

2001 Amendments

The following table lists the codes, paragraphs, and pages in which amendments were made by the 86th National Conference on

Weights and Measures. In the column headed "Action," changes are noted as "added," "amended," "deleted," or "renumbered." Each

code, section, or paragraph that has been changed will be noted as "Added 2001" or "Amended 2001."

Section Code S&T

Item

No.

Paragraph Action Page

1.10.5 General 310-1B G-S.1.1. Added 1-2

310-2 G.S.1.(g) Added 1-2

310-3 G.S.1.(c) Amended 1-2

2.20. Scales

320-1 Table S.6.3.a. Amended column

headings

2-14

320-1 Table S.6.3.a. Added footnote 1 2-14

320-2 S.6.4. Amended 2-12

320-3 N.1.3.4(a) Amended 2-12

320-5 T.N.3.8. Amended 2-25

2.21. Belt-Conveyor Scale 321-1 UR.2.2. Amended 2-41

2.24. Automatic Weighing

Systems

324-1 T.7.3.1. Amended 2-65

3.30. Liquid Measuring Devices 330-2 T.2.1.3. and

T.2.3.4.

Amended 3-13

3-14

N.4.1.2. Added 3-11

T.2.3.2. Amended 3-14

UR.3.4. Amended 3-16

3.31. Vehicle-Tank Meters 331-2 T.4. Amended 3-23

N.4.1.2. Added 3-23

360-2 S.1.3.3.(a) Amended 3-20

3.32. LPG and Anhydrous

Liquid-Measuring Devices

332-1 T.3. Amended 3-32

N.4.1.2. Added 3-31

360-2 S.1.3.3.(a) Amended 3-28

3.34. Cryogenic Liquid-Measurig

Devices

334-1 T.4. Added 3-47

N.5.1.1. Added 3-46

360-2 S.1.3.3.(a) Amended 3-44

3.35. Milk Meters 360-2 S.1.3.3.(a) Amended 3-50
Section Code S&T

Item

No.

Paragraph Action Page

viii

3.36. Water Meters 360-2 S.1.3.3.(a) Amended 3-56

3.37. Mass Flow Meters 337-1 Table T.2. Amended 3-65

T.3. Amended 3-65

N.6.1.1. Added 3-64

3.38. Carbon Dioxide LiquidMeasuring Devices -

Tentative Code

360-2 S.1.3.3.(a) Amended 3-70

4.42. Farm Milk Tanks 342-1 N.5.1., N.5.2. Added 4-12

5.54. Taximeters 354-1 S.7. Added 5-15

5.56.(a) Grain Moisture Meters (a) 356(a)-1 N.1.1., N.1.2. Deleted footnote 1 &

referemce

5-26

356(a)-2 T. Tolerances Delete footnote 2 &

reference

5-26

N.1., N.1.1. Amended 5-26

N.1.3. Added 5-26

Table T.2.1. Amended 5-27

T.2.2., Table

T.2.2.

Added 5-27

5.56. (b) Grain Moisture Meters (b) 356(b)-1 N.1.1. Deleted footnote 1 &

reference

5-34
Section Code S&T

Item

No.

Paragraph Action Page

ix

5.57. Near-Infrared Grain

Analyzers - Tentative

Code

357-1 A.3.1., A.3.2. Added 5-37

S.1.1. Amended (c) and (e)

Added (g)

5-37

Table S.1.2. Amended 5-38

S.1.3.(b) Amended 5-37

Table N.1.1. Added (Title editorially

renumbered from

S.1.1.(e))

5-41

S.2.5.2. Amended Tab le 5-39

N.1.1., N.1.2. Amended 5-40,

5-41

T.1., Table T.2. Amended 5-41

UR.2.1.,

UR.2.3.(a),

UR.2.3.(b)

Amended 5-41,

5-42

UR.2.5. Amended 5-42

Appendix D. Definitions 310-1A manufactured

device

Amended D-8

310-1A remanufactured

device,

remanufactured

element,

repaired device,

and repaired

element

Added D-13

320-1 electronic link Added D-5

320-2 section capacity Added D-14

321-1 belt-conveyor

scale systems

area

Added D-2
x

2001 Editorial Changes

Section Code Paragraph Action Page

2.20 Scales S.6.3.b. 

Footnote 11

Updated to be consistent with G-S.1. 2-15

T.N.3.8. Added “shown in Table 6” to last sentence

for clarification.

2-25

S.1.11, T able

S.1.11., S.2.1.3.1.

Removed “ and enforceable” for consistency

with other nonretroactive statements.

2-7, 2-8

2.22. Automatic

Bulk

Weighing

Systems

UR.1.1. Removed “and enforceable” for consistency

with other nonretroactive statements.

2-49

2.24. Automatic

Weighing

Systmes

A.1. Clarified that weigh labelers can be static or

dynamic.

2-55

A.2. Inserted exception for Automatic Bulk

Weighing Systems.

2-55

T.3.1.2. Clarified that acceptance and maintenance

tolerance values for dynamic tests are the

same.

2-63

T.3.3. Updated references to NIST Handbook 133. 2-63

3.30. LiquidMeasuring

Devices

S.1.6.5.5., Table

S.2.2.

Deleted “E ffective and” fo r consistency with

other nonretroactive statements.

3-6, 3-8

5.56.(a) Grain

Moisture

Meters

Table S.1.2.,

S.2.4.3.

5-24, 5-

25

5.57. Near Infrared

Grain

Analyzers

A.3.1. Changed “type approval” to type evaluation

for consistency with other referernces.

5-37

Entire Code Deleted “and effective” from nonretroactive

statements for consistency with other

references.

5-37

through

42

Appendix

C

General

Tables of

Units of

Measurement

Italicized “statute mile” and “survey foot” for

consistency and to eliminate ambiguity with

NIST SP 811.

C-15
Introduction

1

 When sitting as a voting body, the National Conference on 1

Weights and Measures (NCWM) is made up of State and

local weights and measures officials from all parts of the

United States. The NCWM normally meets annually.

 Communications to this committee may be addressed as 2

follows: Executive Secretary, National Conference on

Weights and Measures, National Institute of Standards and

Technology, 100 Bureau Drive, Stop 2600, Gaithersburg,

MD 20899-2600.

Introduction

A. Source. - The specifications, tolerances, and other

technical requirements published herein comprise, in their

latest form, all of the current codes as adopted by the

National Conference on Weights and Measures, Inc. The 1

National Conference is supported by the National Institute

of Standards and Technology (NIST), which provides the

NCWM secretariat and publishes NCWM documents.

NIST also develops technical publications for use by

weights and measures agencies; these publications may subsequently be endorsed or adopted by the NCWM.

The Conference Committee on Specifications and

Tolerances, acting at the request of the Conference or upon 2

its own initiative, with the technical assistance of the

National Institute of Standards and Technology, annually

prepares proposed revisions, amendments, or additions to

the material previously adopted by the Conference. (See

paragraph C.) Such revisions, amendments, or additions

are then presented to the Conference as a whole, where they

are discussed by weights and measures officials and

representatives of interested manufacturers and industries.

Eventually the proposals of the Committee are voted upon

only by the weights and measures officials.

All of the specifications, tolerances, and other technical

requirements given herein are recommended by the

National Conference on Weights and Measures for official

promulgation in and use by the States in exercising their

control of commercial weighing and measuring apparatus.

A similar recommendation is made with respect to the local

jurisdictions within a State in the absence of the

promulgation of specifications, tolerances, and other

technical requirements by a State agency.

B. Purpose. - The purpose of these technical requirements

is to eliminate from use, weights and measures and

weighing and measuring devices that give readings that are

false, that are of such construction that they are faulty (that

is, that are not reasonably permanent in their adjustment or

will not repeat their indications correctly), or that facilitate

the perpetration of fraud, without prejudice to apparatus

that conforms as closely as practicable to the official

standards.

C. Amendments. - The Committee on Specifications and

Tolerances of the NCWM provides a mechanism for

consideration of amendments or additions to the

specifications, tolerances, and other technical requirements.

D. Submission of Agenda Items - Preamble. - The

Constitution of the NCWM requires that its officers and

Committees observe the principles of due process for the

protection of the rights and interests of affected parties.

Specifically, it requires that Committees and Officers: (a)

give reasonable advance notice of contemplated studies,

issues to be considered for action, and tentative or definite

recommendations for conference vote, and (b) provide that

all interested parties have an opportunity to be heard. 

E. Submission Process. - Anyone introducing an issue to

the Committee shall use the regional weights and measures

associations to initially consider its merits. Using the

regional associations ensures discussion and evaluation of

issues at the grass-roots level by involving the regional

members in the development, evaluation, and justification

of proposals. The regions include the Central,

Northeastern, Southern, and Western Weights and

Measures Associations. For information on the regional

associations contact NCWM, Inc., 15245 Shady Grove

Road, Suite 130, Rockville, MD 20850 or by telephone at

(240) 632-9454.

F. Procedures

The Committee will consider issues according to the

following procedures:

a. All issues to be considered by the Committee for action

at the upcoming Interim Meeting must be submitted in

writing to the Committee by October 15. Although use of

NCWM Form 15 (see page 6) is not required, it is

recommended for use in submitting proposals to the

NCWM. Proposals shall be sent to the Committee at:
Introduction

2

NIST Office of Weights and Measures (i) Identify the pertinent portion, section, and paragraph of

100 Bureau Drive, Stop 2600 the existing publication that would be changed (e.g., Sec.

Gaithersburg, Maryland 20899-2600 1.10. General Code, paragraph G-A.1. Commercial and

Attention: Specifications and Tolerances Committee Law-Enforcement Equipment.).

b. A copy of the proposal must be sent to the NCWM’s (ii) Provide evidence of consistency with other NCWM

Executive Secretary at the same address. publications such as with other specific device code

G. Criteria for Inclusion in the Committee’s Agenda.

a. Any issue approved by at least one regional association

and received by the November 1 deadline will be

automatically placed on the Committee’s Interim Meeting

Agenda. 

b. Issues that have not been approved by a regional

association, but which are received by November 1, will be H. Exceptions to Policy for Submission of Issues to a

evaluated by the Committee using the criteria in Section H, Committee Agenda; Submission of “Priority” Issues.

Exceptions to Policy and Section I, Committee Agenda.

c. Any proposal received after the November 1 deadline,

but prior to the Interim Meeting, will be evaluated by the

Committee according to Section H, Exceptions to Policy

and Section I, Committee Agenda. Only those issues

determined to be a national “Priority” will be included on

its agenda. 

d. Proposals must be in writing and must include: no Regional Association has Approved the Issues.

(1) a concise statement of the issue or problem outlining

the purpose and national need for its consideration. When

possible, an electronic copy of the background material and

proposed amendment(s) should be submitted in a PC

compatible word processing document format (e.g., Corel

WordPerfect or Microsoft Word) on a high density 8.9 cm

(3.5 inch) disk or by electronic mail using the same format;

(2) background material including test data, analysis of test

data, or other appropriately researched and documented

material from which the Committee will be able to make a

judgment for either a firm recommendation or consideration

of the need for further study; 

(3) proposed solutions to problems stated in specific

language in amendment form to Conference documents; and

(4) practical, realistic, and specific recommendations for

both regulations and test methods to provide for proper

enforcement if a proposal involves a new area of weights

and measures activity. 

When proposals are to modify or add requirements to

existing publications, such as Handbook 44, the proposal

should:

sections.

(iii) Provide evidence of consistency with Federal laws and

regulations (e.g., USDA)

(iv) Relay the positions of businesses, industries, or trade

associations affected by the proposal including supporting

and opposing points of view.

The Committee will use the following criteria to evaluate

issues that have not been approved by a regional

association, but has been received by the November 1

deadline. If an issue is received after the November 1

deadline, it will be included on an agenda if the Committee

determines that it is a national “Priority.” 

Criteria for Inclusion in the Committee’s Agenda when

(1) Issues must have significant legal impact on weights and

measures laws and/or regulations involving:

(a) court cases/attorney general opinions; or

(b) pre-emption by Federal statute or regulation; or

(c) conflict with international standards; or

(d) relationship to laws or regulations of an urgent nature

which could affect health and safety.

(2) The Committee may contact parties that are potentially

affected by a issue (e.g., trade associations, industry, and

consumer groups) for comments. The Committee may

consider these comments and any other information in

determining if the issue should be included on its agenda. 

(3) When the Committee determines that it should consider

an issue as a “Priority” (using the criteria in (1), the issue

will be handled in the following manner:

a. A “Priority” issue received prior to the Interim Meeting

may be added to the Interim Meeting agenda by majority

vote of the Committee.
Introduction

3

b. A “Priority” issue received after the Interim Meeting “V” after the item number in the agenda or prior to the

may be added to the Committee’s Annual Meeting agenda voting session). The Committee may also decide to carry

as: (i) a discussion issue by majority vote of the issues over as “Informational” items for further study,

Committee, or (ii) as a voting item by majority vote of the comment, and development (indicated with an “I” after the

Committee and the NCWM Board of Directors. item number). Items marked “W” in the agenda have been

I. Committee Agenda

a. The Committee will review issues that have been

submitted and selected by majority vote to be included on At the 1998 Annual Meeting, the NCWM established a

its agenda. The Committee will only include those issues process for disseminating information on issues which may

that have been (1) approved by at least one of the regional have merit but are insufficiently developed for Committee

associations; or (2) forwarded by other committees, action. Past practice for handling these issues had been to

subcommittees, NTETC Sectors, task forces, or work either “carry” them forward as informational issues or

groups, or those issues that meet the criteria in Section H, withdraw them. Conference members felt that carrying

Exceptions to Policy. undeveloped information items drained NCWM resources,

b. The Committee will publish an agenda which identifies prematurely discarded despite the valuable work that had

the issues to be discussed during the Interim Meetings. This gone into developing the issue. The NCWM was also

agenda shall be distributed to members approximately 30 interested in providing a mechanism to inform parties about

days prior to the meetings. The agenda will be provided issues that are developing in different localities or in the

upon request to all other interested parties. regional associations. 

(Amended 1998)

J. Interim Meeting

a. The Committee shall hold public hearings at the Interim developed for action at the national level. The NCWM

Meeting for the purpose of discussing and taking comments agreed that “Developing” issues should be submitted by the

on all issues on its agenda. regional associations with a recommendation that they be

b. Upon request, the Committee will provide the Committees will present “D” issues in list format at the end

opportunity for presentations by government officials, of their reports and include a point of contact (including the

industry representatives, consumer groups, or other name and telephone number of the submitter) so that

interested parties during the Interim Meeting. Requests to interested parties can obtain additional information. No

make presentations must be received by the Committee comments will be taken on a “Developing” issue, unless the

Chairman or Technical Advisor at least two weeks prior to Committee agrees to receive the new information in

the start of the meetings. advance of the hearing. In these cases, the Chairman will

K. Interim Meeting Report

a. Issues under consideration by the Committee and upon informed of emerging issues.

which it offers comments or recommendations for

Conference action during the Annual Meetings, will be

included in the Committee’s Interim Reports published in

the Annual Meeting, Program and Committee Reports. a. Weights and measures officials, industry representatives,

b. The Annual Meeting Program and Committee Reports and all other parties are encouraged to submit written

will be prepared and distributed to Conference members comments on issues in the Committees' Interim Reports. 

approximately 3 months prior to the NCWM Annual

Meeting. b. All comments on the Interim Meeting Reports must be

L. Classifications for Agenda Items 

At the Interim Meeting, the Committee can classify items

for voting by the NCWM membership (indicated with a

withdrawn from consideration.

M. Developing Issues

but they also believed that withdrawn items were often

The NCWM established a new “Developing” designation to

allow the Committees to notify the submitter of the issue,

that, while it may have merit, it may not be adequately

presented as “D” issues on the national agenda. The

announce in advance that an issue will be discussed in the

session. The use of this “D” designator is seen as an issue

management tool, as well as a way to keep the membership

N. Comments on Interim Reports

submitted to the Committee, along with a copy to the

Executive Secretary, no later than 1 month preceding the

opening of the Annual Meeting.

O. Annual Meeting
Introduction

4

a. The Committee will hold a public hearing at the Annual

Meeting to discuss issues on its agenda.

b. Those who want to speak on an issue during the public

hearings should request time from the Committee

Chairman. Time limitations on presentations, the discussion of a question, or amendments may be imposed by the

Committee Chairman.

P. Final Committee Reports and Conference Action

a. Following the public hearings, the Committee will

prepare its final report for action by the voting membership

of the Conference. Prior to the session during which it will

be acted on, copies of each final report will be provided for

study.

b. The Chairman of the Committee will present the final

report of the Committee to the Conference body. A vote

will be taken on issues, proposals, or sections in the report

as circumstances require. The Conference will vote on the

entire final report as presented in accordance with

established Conference voting procedures. Parliamentary

procedure according to Roberts Rules of Order as amended

by NCWM Publication No. 1, Bylaws, will be adhered to in

the presentation of and action on Standing Committee

reports.

(Amended 1998)

Q. System of Paragraph Designation. - In order that

technical requirements of a similar nature, or those directed

to a single characteristic, may be grouped together in an

orderly fashion, and to facilitate the location of individual

requirements, the paragraphs of each code are divided into

sections. Each section is designated by a letter and a name,

and each subsection is given a letter-number designation

and a side title.

The letter that appears first in a paragraph designation has

a specific meaning, as follows:

G. The letter G is a prefix and indicates that the

requirement is part of the General Code.

A. Application. These paragraphs pertain to the

application of the requirements of a code.

S. Specification. These paragraphs relate to the

design of equipment. Specification paragraphs are

directed particularly to manufacturers of devices.

N. Note. These paragraphs apply to the official

testing of devices.

T. Tolerance. Tolerances are performance

requirements. They fix the limit of allowable error

or departure from true performance or value.

Sensitivity. The sensitivity requirements, applicable

only to nonautomatic-indicating scales, are

performance requirements and are lettered with a T.

UR. User Requirement. These paragraphs are directed

particularly to the owner and operator of a device.

User requirements apply to the selection,

installation, use, and maintenance of devices.

D. Definitions of Terms. A definitions section

appears in Appendix D to provide the definition of

the terms having a special meaning.

The numerical designation after a letter follows the decimal

system of paragraph identification that fixes both the

relationship and the limitation of the requirements of the

paragraph. For example, in the Scales Code, under

Specifications, the following numerical designations occur:

S. Specifications

S.1. Design of Indicating and Recording Elements and

of Recorded Representations

S.1.1. Zero Indication.

S.1.1.1. Digital Indicating Elements.

S.1.1.2. No-Load Reference Value.

S.1.2. Value of Scale Division Units.

S.1.2.1. Weight Units.

S.1.3. Graduations.

S.1.3.1. Length.

S.1.3.2. Width.

S.1.3.3. Clear Space Between Graduations

In this example, Paragraphs S.1.1., S.1.2., and S.1.3. are

directed and limited to paragraph S.1., which pertains to the

design of indicating and recording elements and of recorded

representations. Paragraphs S.1.1.1. and S.1.1.2. are

directly related to each other, but they are limited to the

design of zero indication. Likewise, paragraphs S.1.3.1.,

S.1.3.2., and S.1.3.3. are directly related to each other, but

they are limited to the design of graduations.

This handbook conforms to the concept of primary use of

SI (metric) measurements recommended in the Omnibus

Trade and Competitiveness Act of 1988 by citing SI metric

units before inch-pound units where both units appear

together and placing separate sections containing

requirements for metric units before corresponding sections

containing requirements for customary units. Occasionally,
Introduction

5

a paragraph or table carries the suffix "M" because the

requirement in SI units is shown as a separate statement,

rather than combined with the inch-pound units. In these

few instances, separate requirements were judged to be

more easily understood than attempting to combine SI and

inch-pound units in a single paragraph or table. In some

cases, however, trade practice is currently restricted to the

use of customary units; therefore, some requirements in this

Handbook will continue to specify only customary units

until the Conference achieves a broad consensus on the

permitted metric units. 

R. Classification of Requirements. - The classification of memory tolerances or tolerance tables, even though these

requirements into "retroactive" and "nonretroactive" status are used frequently. For the Handbook to serve its purpose,

is made in order that the requirements may be put into force it should be available when any of its requirements are to be

and effect without unnecessary hardship and without applied. Direct reference is the only sure way to apply a

wholesale condemnation of apparatus. Retroactive requirement properly and to check whether other

requirements are enforceable with respect to all equipment requirements may be applicable.

and are printed in upright roman type. Nonretroactive

requirements are those that, while clearly desirable, are not This Handbook supplies criteria which enables the

so vital that they should at once be enforced with respect to inspector to determine the suitability, accuracy, and

all apparatus. Nonretroactive requirements are printed

in italic type.

It is not expected that, after their promulgation in a given

jurisdiction, nonretroactive requirements will always remain

nonretroactive. It is entirely proper that a weights and

measures official, following a careful analysis of existing

conditions, fix reasonable periods for the continuance of the

nonretroactive application of particular requirements, after

which such requirements will become retroactive. These

periods should be long enough to avoid undue hardship to

the owners or operators of apparatus and, in the case of

some requirements, should approximate the average useful

life of the apparatus in question. 

In order that all interested parties may have timely and

ample notice of impending changes in the status of

requirements, the following procedure is suggested for the

official who plans to change the classification of

requirements. If sufficient data are available to make such

action feasible, publish in combination with the codes

themselves the date or dates at which nonretroactive

requirements are to become retroactive. In other cases, give

equally effective notice at the earliest practicable date.

A nonretroactive requirement, in italic type, will indicate

the year from which it should be enforced and, in some

cases, the date the requirement shall be changed to retroactive status. For example, [Nonretroactive as of 1978 and

to become retroactive on January 1, 1985]. As a general

rule, each nonretroactive requirement is reviewed after it

has been in effect for 10 years to determine the

appropriateness of its nonretroactive status.

S. Using the Handbook. - Handbook 44 is designed to be

a working tool for the weights and measures official, the

equipment manufacturer, installer, and repairman. The

section on Fundamental Considerations (Appendix A)

should be studied until its contents are well known. The

General Code, with general requirements pertaining to all

devices, obviously must be well known to a user of the

Handbook. The makeup of the specific codes, the order of

paragraph presentation, and particularly paragraph

designation are worthy of careful study.

It is not deemed advisable for a user to attempt to commit to

repetitive consistency of a weighing or measuring device,

both in the laboratory and in the field. However, not all

code sections can be appropriately applied in both settings.

Since some sections are designed to be applied specifically

to tests performed under laboratory conditions, it would be

impractical or unrealistic to apply them to field tests. Not

all tests described in the “Notes” section of the Handbook

are required to be performed in the field as an official test.

An inspector may officially approve or reject a device

which has been tested in accordance with those sections

applicable to the type of test being conducted.

(Paragraph added 1996)
Introduction

6

NCWM Form 15 

 Proposal to a Standing Committee

Committee: ______________________

Date: Regional Association : 

Name/Address of Contact Person Regional Actions: (votes for and against)

Phone:

Fax:

email:

Please Attach Additional Pages and Information as Needed 

Proposal: (Proposed solutions to problems stated in specific language in amendment form to handbooks, if a proposal involves a

new area of weights and measures activity, the recommendations for both regulations and test methods for proper enforcement

must be provided.)

Problem/Justification: (Include a concise statement of the issue or problem outlining the purpose and national need for its consideration.)

Other Contacts: (Provide position statements, comments, etc. from names and addresses of individuals, firms, manufacturers,

and/or trade associations included in developing the proposal.)

Other Reasons For: (If none, please indicate none have emerged.)

Other Reasons Against:

Additional Considerations: (provide cost estimates and state the anticipated benefits for all parties or indicate how the proposal

may affect other requirements, programs, etc.) 

Attachments: (list the accompanying documents, data, studies Suggested Action: (Be specific on what action the committee

etc.) should take on the issue.) 

 Recommend NCWM G Adoption G Withdrawn

G Developing Issue 

G Informational Issue 

G Other (Please describe)
)GPGTCN%QFG



5GE)GPGTCN%QFG

)##RRNKECVKQP

)#%QOOGTEKCNCPF.CY'PHQTEGOGPV

'SWKROGPV6JGUGURGEKHKECVKQPUVQNGTCPEGUCPFQVJGT

VGEJPKECNTGSWKTGOGPVUCRRN[CUHQNNQYU


C 6QEQOOGTEKCNYGKIJKPICPFOGCUWTKPIGSWKROGPV

VJCVKUVQYGKIJVUCPFOGCUWTGUCPFYGKIJKPICPF

OGCUWTKPIFGXKEGUEQOOGTEKCNN[WUGFQTGORNQ[GF

KPGUVCDNKUJKPIVJGUK\GSWCPVKV[GZVGPVCTGCQT

OGCUWTGOGPVQHSWCPVKVKGUVJKPIURTQFWEGQT

CTVKENGUHQTFKUVTKDWVKQPQTEQPUWORVKQPRWTEJCUGF

QHHGTGFQTUWDOKVVGFHQTUCNGJKTGQTCYCTFQTKP

EQORWVKPICP[DCUKEEJCTIGQTRC[OGPVHQTUGTXKEGU

TGPFGTGFQPVJGDCUKUQHYGKIJVQTOGCUWTG


D 6QCP[CEEGUUQT[CVVCEJGFVQQTWUGFKPEQPPGEVKQP

YKVJCEQOOGTEKCNYGKIJKPIQTOGCUWTKPIFGXKEG

YJGPUWEJCEEGUUQT[KUUQFGUKIPGFVJCVKVUQRGTCVKQP

CHHGEVUVJGCEEWTCE[QHVJGFGXKEG


E 6QYGKIJKPICPFOGCUWTKPIGSWKROGPVKPQHHKEKCN

WUGHQTVJGGPHQTEGOGPVQHNCYQTHQTVJGEQNNGEVKQP

QHUVCVKUVKECNKPHQTOCVKQPD[IQXGTPOGPVCIGPEKGU


6JGUGTGSWKTGOGPVUUJQWNFDGWUGFCUCIWKFGD[VJG

YGKIJVUCPFOGCUWTGUQHHKEKCNYJGPWRQPTGSWGUV

EQWTVGU[GZCOKPCVKQPUQHPQPEQOOGTEKCNGSWKROGPVCTG

OCFG

)#%QFG#RRNKECVKQP6JKU)GPGTCN%QFGUJCNN

CRRN[VQCNNENCUUGUQHFGXKEGUCUEQXGTGFKPVJGURGEKHKE

EQFGU6JGURGEKHKEEQFGTGSWKTGOGPVUUWRGTUGFG)GPGTCN

%QFGTGSWKTGOGPVUKPCNNECUGUQHEQPHNKEV


#OGPFGF

)#5RGEKCNCPF7PENCUUKHKGF'SWKROGPV+PUQHCT

CUVJG[CTGENGCTN[CRRTQRTKCVGVJGTGSWKTGOGPVUCPF

RTQXKUKQPUQHVJG)GPGTCN%QFGCPFQHURGEKHKEEQFGU

CRRN[VQGSWKROGPVHCKNKPID[TGCUQPQHURGEKCNFGUKIPQT

QVJGTYKUGVQHCNNENGCTN[YKVJKPQPGQHVJGRCTVKEWNCT

GSWKROGPVENCUUGUHQTYJKEJUGRCTCVGEQFGUJCXGDGGP

GUVCDNKUJGF9KVJTGURGEVVQUWEJGSWKROGPVEQFG

TGSWKTGOGPVUCPFRTQXKUKQPUUJCNNDGCRRNKGFYKVJFWG

TGICTFVQVJGFGUKIPKPVGPFGFRWTRQUGCPFEQPFKVKQPUQH

WUGQHVJGGSWKROGPV

)#/GVTKE'SWKROGPV'ORNQ[OGPVQHVJG

YGKIJVUCPFOGCUWTGUQHVJGOGVTKEU[UVGOKUNCYHWN

VJTQWIJQWVVJG7PKVGF5VCVGU6JGUGURGEKHKECVKQPU

VQNGTCPEGUCPFQVJGTTGSWKTGOGPVUUJCNNPQVDG

WPFGTUVQQFQTEQPUVTWGFCUKPCP[YC[RTQJKDKVKPIVJG

OCPWHCEVWTGUCNGQTWUGQHGSWKROGPVFGUKIPGFVQIKXG

TGUWNVUKPVGTOUQHOGVTKEWPKVU6JGURGEKHKERTQXKUKQPU

QHVJGUGTGSWKTGOGPVUCPFVJGRTKPEKRNGUWRQPYJKEJVJG

TGSWKTGOGPVUCTGDCUGFUJCNNDGCRRNKGFVQOGVTKE

GSWKROGPVKPUQHCTCUCRRTQRTKCVGCPFRTCEVKECDNG6JG

VQNGTCPEGUQPOGVTKEGSWKROGPVYJGPPQVURGEKHKGF

JGTGKPUJCNNDGGSWKXCNGPVVQVJQUGURGEKHKGFHQTUKOKNCT

GSWKROGPVEQPUVTWEVGFQTITCFWCVGFKPVJGKPEJRQWPF

U[UVGO

)#4GVTQCEVKXG4GSWKTGOGPVU4GVTQCEVKXG

TGSWKTGOGPVUCTGGPHQTEGCDNGYKVJTGURGEVVQCNN

GSWKROGPV4GVTQCEVKXGTGSWKTGOGPVUCTGRTKPVGFJGTGKP

KPWRTKIJVTQOCPV[RG

)#0QPTGVTQCEVKXG4GSWKTGOGPVU

0QPTGVTQCEVKXGTGSWKTGOGPVUCTGGPHQTEGCDNGCHVGTVJG

GHHGEVKXGFCVGHQT


C FGXKEGUOCPWHCEVWTGFYKVJKPC5VCVGCHVGTVJG

GHHGEVKXGFCVG


D DQVJPGYCPFWUGFFGXKEGUDTQWIJVKPVQC5VCVGCHVGT

VJGGHHGEVKXGFCVGCPF


E FGXKEGUWUGFKPPQPEQOOGTEKCNCRRNKECVKQPUYJKEJ

CTGRNCEGFKPVQEQOOGTEKCNWUGCHVGTVJGGHHGEVKXG

FCVG

0QPTGVTQCEVKXGTGSWKTGOGPVUCTGPQVGPHQTEGCDNGYKVJ

TGURGEVVQFGXKEGUVJCVCTGKPEQOOGTEKCNUGTXKEGKPVJG

5VCVGCUQHVJGGHHGEVKXGFCVGQTVQPGYGSWKROGPVKPVJG

UVQEMQHCOCPWHCEVWTGTQTCFGCNGTKPVJG5VCVGCUQHVJG

GHHGEVKXGFCVG0QPTGVTQCEVKXGTGSWKTGOGPVUCTGRTKPVGF

KPKVCNKEV[RG


#OGPFGF

)#'HHGEVKXG'PHQTEGOGPV&CVGUQH%QFG

4GSWKTGOGPVU7PNGUUQVJGTYKUGURGEKHKGFGCEJPGY

QTCOGPFGFEQFGTGSWKTGOGPVUJCNNPQVDGUWDLGEVVQ

GPHQTEGOGPVRTKQTVQ,CPWCT[QHVJG[GCTHQNNQYKPIVJG

CFQRVKQPD[VJG0CVKQPCN%QPHGTGPEGQP9GKIJVUCPF

/GCUWTGUCPFRWDNKECVKQPD[VJG0CVKQPCN+PUVKVWVGQH

5VCPFCTFUCPF6GEJPQNQI[

)55RGEKHKECVKQPU
)GPGTCN%QFG



)5+FGPVKHKECVKQP#NNGSWKROGPVGZEGRVYGKIJVU 6JGTGSWKTGFKPHQTOCVKQPUJCNNDGUQNQECVGFVJCVKVKU

CPFUGRCTCVGRCTVUPGEGUUCT[VQVJGOGCUWTGOGPVRTQEGUU TGCFKN[QDUGTXCDNGYKVJQWVVJGPGEGUUKV[QHVJG

DWVPQVJCXKPICP[OGVTQNQIKECNGHHGEVUJCNNDGENGCTN[ FKUCUUGODN[QHCRCTVTGSWKTKPIVJGWUGQHCP[OGCPU

CPFRGTOCPGPVN[OCTMGFHQTVJGRWTRQUGUQHKFGPVKHKECVKQP UGRCTCVGHTQOVJGFGXKEG

YKVJVJGHQNNQYKPIKPHQTOCVKQP 
#OGPFGFCPF


CVJGPCOGKPKVKCNUQTVTCFGOCTMQHVJGOCPWHCEVWTGT )54GOCPWHCEVWTGF&GXKEGUCPF

QTFKUVTKDWVQT 4GOCPWHCEVWTGF/CKP'NGOGPVU#NNTGOCPWHCEVWTGF


DCOQFGNFGUKIPCVKQPVJCVRQUKVKXGN[KFGPVKHKGUVJG ENGCTN[CPFRGTOCPGPVN[OCTMGFHQTVJGRWTRQUGUQH

RCVVGTPQTFGUKIPQHVJGFGXKEG KFGPVKHKECVKQPYKVJVJGHQNNQYKPIKPHQTOCVKQP


EVJGOQFGNFGUKIPCVKQPUJCNNDGRTGHCEGFD[VJGVGTO 
C VJGPCOGKPKVKCNUQTVTCFGOCTMQHVJGNCUV

/QFGN6[RGQT2CVVGTP6JGUGVGTOUOC[DG TGOCPWHCEVWTGTQTFKUVTKDWVQT

HQNNQYGFD[VJGVGTO0WODGTQTCPCDDTGXKCVKQPQHVJCV

YQTF6JGCDDTGXKCVKQPHQTVJGYQTF0WODGTUJCNNCU 
D VJGTGOCPWHCEVWTGT	UQTFKUVTKDWVQT	UOQFGN

COKPKOWODGIKPYKVJVJGNGVVGT0GI0QQT0Q FGUKIPCVKQPKHFKHHGTGPVVJCPVJGQTKIKPCNOQFGN

6JGCDDTGXKCVKQPHQTVJGYQTF/QFGN UJCNNDG/QF FGUKIPCVKQP

QT/QF =0QPTGVTQCEVKXGCUQH,CPWCT[?

=0QPTGVTQCEVKXG,CPWCT[? 
#FFGF


#FFGF 
#OGPFGF

=0QVG2TGHKZNGVVGTKPIOC[DGKPKVKCNECRKVCNUCNNECRKVCNU FGXKEG CPFTGRCKTGFGNGOGPV CTGCNUQKPENWFGFCNQPI

QTCNNNQYGTECUG? YKVJFGHKPKVKQPUHQTTGOCPWHCEVWTGFFGXKEG CPF


FGZEGRVHQTGSWKROGPVYKVJPQOQXKPIQTGNGEVTQPKE

EQORQPGPVRCTVUCPQPTGRGVKVKXGUGTKCNPWODGT)5(CEKNKVCVKQPQH(TCWF#NNGSWKROGPVCPFCNN

=0QPTGVTQCEVKXGCUQH,CPWCT[? OGEJCPKUOUCPFFGXKEGUCVVCEJGFVJGTGVQQTWUGFKP


GVJGUGTKCNPWODGTUJCNNDGRTGHCEGFD[YQTFUCP CPFKPUVCNNGFHQTWUGUWEJVJCVVJG[FQPQVHCEKNKVCVGVJG

CDDTGXKCVKQPQTCU[ODQNVJCVENGCTN[KFGPVKHKGUVJG RGTRGVTCVKQPQHHTCWF

PWODGTCUVJGTGSWKTGFUGTKCNPWODGTCPF

=0QPTGVTQCEVKXGCUQH,CPWCT[? )52GTOCPGPEG#NNGSWKROGPVUJCNNDGQHUWEJ


HVJGUGTKCNPWODGTUJCNNDGRTGHCEGFD[VJGYQTFU VJCVWPFGTPQTOCNUGTXKEGEQPFKVKQPU

5GTKCN0WODGTQTCPCDDTGXKCVKQPQHVJCVVGTO

#DDTGXKCVKQPUHQTVJGYQTF5GTKCNUJCNNCUCOKPKOWO 
C CEEWTCE[YKNNDGOCKPVCKPGF

DGIKPYKVJVJGNGVVGT5CPFCDDTGXKCVKQPUHQTVJGYQTF

0WODGTUJCNNCUCOKPKOWODGIKPYKVJVJGNGVVGT0 
D QRGTCVKPIRCTVUYKNNEQPVKPWGVQHWPEVKQPCU


GI50505GT0QCPF50Q KPVGPFGFCPF

=0QPTGVTQCEVKXGCUQH,CPWCT[?


I (QTFGXKEGUVJCVJCXGCP06'2%GTVKHKECVGQH

%QPHQTOCPEG
%%0WODGTQTCEQTTGURQPFKPI%% 7PFWGUVTGUUGUFGHNGEVKQPUQTFKUVQTVKQPUQHRCTVUUJCNN

CFFGPFWOPWODGTVJG06'2%%UJCNNDGRTGHCEGFD[VJG PQVQEEWTVQVJGGZVGPVVJCVCEEWTCE[QTRGTOCPGPEGKU

VGTOU06'2%%%%QT#RRTQXCN6JGUGVGTOU FGVTKOGPVCNN[CHHGEVGF

OC[DGHQNNQYGFD[VJGVGTO0WODGTQTCPCDDTGXKCVKQP

QHVJCVYQTF6JGCDDTGXKCVKQPHQTVJGYQTF0WODGT )5+PVGTEJCPIGQT4GXGTUCNQH2CTVU2CTVUQHC

UJCNNCUCOKPKOWODGIKPYKVJVJGNGVVGT0GI0Q FGXKEGVJCVOC[TGCFKN[DGKPVGTEJCPIGFQTTGXGTUGFKP

QT0Q VJGEQWTUGQHHKGNFCUUGODN[QTQHPQTOCNWUCIGUJCNNDG

=0QPTGVTQCEVKXGCUQH,CPWCT[?

FGXKEGUCPFTGOCPWHCEVWTGFOCKPGNGOGPVUUJCNNDG

0QVG&GHKPKVKQPUHQT OCPWHCEVWTGFFGXKEG TGRCKTGF

TGOCPWHCEVWTGFGNGOGPV KP#RRGPFKZ&&GHKPKVKQPU

EQPPGEVKQPVJGTGYKVJUJCNNDGUQEQPUVTWEVGFCUUGODNGF

OCVGTKCNUFGUKIPCPFEQPUVTWEVKQPCUVQOCMGKVRTQDCDNG


E CFLWUVOGPVUYKNNTGOCKPTGCUQPCDN[RGTOCPGPV


C UQEQPUVTWEVGFVJCVVJGKTKPVGTEJCPIGQTTGXGTUCNYKNN

PQVCHHGEVVJGRGTHQTOCPEGQHVJGFGXKEGQT
)GPGTCN%QFG




D UQOCTMGFCUVQUJQYVJGKTRTQRGTRQUKVKQPU )58CNWGU+HITCFWCVKQPUKPFKECVKQPU

)5+PFKECVKPICPF4GEQTFKPI'NGOGPVU URGEKHKEXCNWGUVJGUGUJCNNDGCFGSWCVGN[FGHKPGFD[

)5)GPGTCN#NNYGKIJKPICPFOGCUWTKPI EQODKPCVKQPUVJGTGQHWPKHQTON[RNCEGFYKVJ

FGXKEGUUJCNNDGRTQXKFGFYKVJKPFKECVKPIQTTGEQTFKPI TGHGTGPEGVQVJGITCFWCVKQPUKPFKECVKQPUQT

GNGOGPVUCRRTQRTKCVGKPFGUKIPCPFCFGSWCVGKPCOQWPV TGEQTFGFTGRTGUGPVCVKQPUCPFCUENQUGVJGTGVQCU

2TKOCT[KPFKECVKQPUCPFTGEQTFGFTGRTGUGPVCVKQPUUJCNN RTCEVKECDNGDWVPQVUQRQUKVKQPGFCUVQKPVGTHGTG

DGENGCTFGHKPKVGCEEWTCVGCPFGCUKN[TGCFWPFGTCP[ YKVJVJGCEEWTCE[QHTGCFKPI

EQPFKVKQPUQHPQTOCNQRGTCVKQPQHVJGFGXKEG

)5)TCFWCVKQPU+PFKECVKQPUCPF4GEQTFGF KPFKECVKQPUQTTGEQTFGFTGRTGUGPVCVKQPUCPFVJGKT

4GRTGUGPVCVKQPU FGHKPKPIHKIWTGUYQTFUCPFU[ODQNUUJCNNDGQH

)5#PCNQI+PFKECVKQPCPF DGEQOGQDNKVGTCVGFQTKNNGIKDNG

4GRTGUGPVCVKQP)TCFWCVKQPUCPFCUWKVCDNG

KPFKECVQTUJCNNDGRTQXKFGFKPEQPPGEVKQPYKVJ )58CNWGUQH)TCFWCVGF+PVGTXCNUQT

KPFKECVKQPUFGUKIPGFVQCFXCPEGEQPVKPWQWUN[ +PETGOGPVU+PCP[UGTKGUQHITCFWCVKQPU

)5&KIKVCN+PFKECVKQPCPF4GRTGUGPVCVKQP VJGITCFWCVGFKPVGTXCNUQTKPETGOGPVUUJCNNDGWPKHQTO

&KIKVCNGNGOGPVUUJCNNDGUQFGUKIPGFVJCV VJTQWIJQWVVJGUGTKGU


C #NNFKIKVCNXCNWGUQHNKMGXCNWGKPCU[UVGO )51P&GXKEGU6JCV+PFKECVGQT4GEQTF

CITGGYKVJQPGCPQVJGT KP/QTG6JCP1PG7PKV1PFGXKEGUFGUKIPGFVQ


D #FKIKVCNXCNWGEQKPEKFGUYKVJKVUCUUQEKCVGF OGPVVJGXCNWGUKPFKECVGFCPFTGEQTFGFUJCNNDG

CPCNQIXCNWGVQVJGPGCTGUVOKPKOWO KFGPVKHKGFYKVJCPCRRTQRTKCVGYQTFU[ODQNQTCD

ITCFWCVKQP DTGXKCVKQP


E #FKIKVCNXCNWGTQWPFUQHHVQVJGPGCTGUV

OKPKOWOWPKVVJCVECPDGKPFKECVGFQT )54GRGCVCDKNKV[QH+PFKECVKQPU#FGXKEG

TGEQTFGF UJCNNDGECRCDNGQHTGRGCVKPIYKVJKPRTGUETKDGF


F #FKIKVCN\GTQKPFKECVKQPKPENWFGUVJGFKURNC[QH VKQPU6JKUTGSWKTGOGPVUJCNNDGOGVKTTGURGEVKXGQH

C\GTQHQTCNNRNCEGUVJCVCTGFKURNC[GFVQVJG TGRGCVGFOCPKRWNCVKQPQHCP[GNGOGPVQHVJGFGXKEGKP

TKIJVQHVJGFGEKOCNRQKPVCPFCVNGCUVQPG COCPPGTCRRTQZKOCVKPIPQTOCNWUCIG
KPENWFKPI

RNCEGVQVJGNGHV9JGPPQFGEKOCNXCNWGUCTG FKURNCEGOGPVQHVJGKPFKECVKPIGNGOGPVUVQVJGHWNN

FKURNC[GFC\GTQUJCNNDGFKURNC[GFHQTGCEJ GZVGPVCNNQYGFD[VJGEQPUVTWEVKQPQHVJGFGXKEGCPF

RNCEGQHVJGFKURNC[GFUECNGFKXKUKQP TGRGCVGFQRGTCVKQPQHCNQEMKPIQTTGNKGXKPIOGEJC

=0QPTGVTQCEVKXGCUQH,CPWCT[? PKUOCPFQHVJGTGRGCVGFRGTHQTOCPEGQHUVGRUQT


#OGPFGFCPF QRGTCVKQPUVJCVCTGGODTCEGFKPVJGVGUVKPIRTQEGFWTG

)55K\GCPF%JCTCEVGT+PCP[UGTKGUQH )5/QPG[8CNWGU/CVJGOCVKECN#ITGGOGPV

ITCFWCVKQPUKPFKECVKQPUQTTGEQTFGF #P[TGEQTFGFOQPG[XCNWGCPFCP[FKIKVCNOQPG[

TGRTGUGPVCVKQPUEQTTGURQPFKPIITCFWCVKQPUCPFWPKVU XCNWGKPFKECVKQPQPCEQORWVKPIV[RGYGKIJKPIQT

UJCNNDGWPKHQTOKPUK\GCPFEJCTCEVGT)TCFWCVKQPU

KPFKECVKQPUQTTGEQTFGFTGRTGUGPVCVKQPUVJCVCTG

UWDQTFKPCVGVQQTQHCNGUUGTXCNWGVJCPQVJGTUYKVJ

YJKEJVJG[CTGCUUQEKCVGFUJCNNDGCRRTQRTKCVGN[

RQTVTC[GFQTFGUKIPCVGF=/CFGTGVTQCEVKXGCUQH

,CPWCT[?

QTTGEQTFGFTGRTGUGPVCVKQPUCTGKPVGPFGFVQJCXG

CUWHHKEKGPVPWODGTQHHKIWTGUYQTFUU[ODQNUQT

)52GTOCPGPEG)TCFWCVKQPU

UWEJEJCTCEVGTVJCVVJG[YKNNPQVVGPFGCUKN[VQ

KPFKECVKQPUQTTGEQTFGFTGRTGUGPVCVKQPUVJGXCNWGUQH

KPFKECVGQTTGEQTFKPOQTGVJCPQPGWPKVQHOGCUWTG

=/CFGTGVTQCEVKXG?#OGPFGF

VQNGTCPEGUKVUKPFKECVKQPUCPFTGEQTFGFTGRTGUGPVC
)GPGTCN%QFG



6CDNG

4GRTGUGPVCVKQPQH7PKVU

4GRTGUGPVCVKQP

(QTO+ (QTO++

0COGQH7PKV

+PVGTPCVKQPCN

U[ODQNEQOOQP

WUGU[ODQN


FQWDNGECUG 
UKPINGECUGNQYGT 
UKPINGECUGWRRGT

$CUG5+WPKVU

OGVGT O O O /

MKNQITCO MI MI MI -)

>KXGF5+WPKVU

PGYVQP 0 0 P 0

RCUECN 2C 2C RC 2#

YCVV 9 9 Y 9

XQNV 8 8 X 8

FGITGG%GNUKWU E% E% EE E%

1VJGTWPKVU

NKVGT NQT. . N .

ITCO I I I )

OGVTKEVQP V V VPG 60'

DCT DCT DCT DCT $#4

OGCUWTKPIFGXKEGWUGFKPTGVCKNVTCFGUJCNNDGKP )54GEQTFGF4GRTGUGPVCVKQPQH/GVTKE

OCVJGOCVKECNCITGGOGPVYKVJKVUCUUQEKCVGFSWCPVKV[ 7PKVUQP'SWKROGPVYKVJ.KOKVGF%JCTCEVGT

TGRTGUGPVCVKQPQTKPFKECVKQPVQVJGPGCTGUVEGPVQHOQPG[ 5GVU6JGCRRTQRTKCVGFGHKPKPIU[ODQNUCTG

XCNWG6JKUFQGUPQVCRRN[VQCWZKNKCT[FKIKVCNKPFKECVKQPU UJQYPKP6CDNG

KPVGPFGFHQTVJGQRGTCVQT	UWUGQPN[YJGPVJGUG 
#FFGF

KPFKECVKQPUCTGQDVCKPGFHTQOGZKUVKPICPCNQIEWUVQOGT )5/CIPKHKGF)TCFWCVKQPUCPF+PFKECVKQPU

KPFKECVKQPUVJCVOGGVVJKUTGSWKTGOGPV #NNTGSWKTGOGPVUHQTITCFWCVKQPUCPFKPFKECVKQPUCRRN[


#OGPFGF VQCUGTKGUQHITCFWCVKQPUCPFCPKPFKECVQTOCIPKHKGF

)54GEQTFGF4GRTGUGPVCVKQPU+PUQHCTCU CUETGGP

VJG[CTGCRRTQRTKCVGVJGTGSWKTGOGPVUHQTKPFKECVKPI

CPFTGEQTFKPIGNGOGPVUUJCNNCNUQCRRN[VQTGEQTFGF )5/CTMKPI1RGTCVKQPCN%QPVTQNU+PFKECVKQPUCPF

TGRTGUGPVCVKQPU#NNTGEQTFGFXCNWGUUJCNNDGRTKPVGF (GCVWTGU#NNQRGTCVKQPCNEQPVTQNUKPFKECVKQPUCPF

FKIKVCNN[ HGCVWTGUKPENWFKPIUYKVEJGUNKIJVUFKURNC[URWUJ

=/CFGTGVTQCEVKXG?#OGPFGF DWVVQPUCPFQVJGTOGCPUUJCNNDGENGCTN[CPFFGHKPKVGN[

D[CPQRVKECNU[UVGOQTCUOCIPKHKGFCPFRTQLGEVGFQP

KFGPVKHKGF6JGWUGQHCRRTQXGFRKEVQITCOUQTU[ODQNU
)GPGTCN%QFG



UJCNNDGCEEGRVCDNG=0QPTGVTQCEVKXGCUQH,CPWCT[ 
C GSWKROGPVVQDGRWVKPVQEQOOGTEKCNWUGHQTVJG

? HKTUVVKOG


#OGPFGF

)5.GVVGTKPI#NNTGSWKTGFOCTMKPIUCPF UGTXKEGYKVJKPVJGRTGEGFKPIFC[UCPFKUDGKPI

KPUVTWEVKQPUUJCNNDGFKUVKPEVCPFGCUKN[TGCFCDNGCPFUJCNN QHHKEKCNN[VGUVGFHQTVJGHKTUVVKOG

DGQHUWEJEJCTCEVGTVJCVVJG[YKNNPQVVGPFVQDGEQOG

QDNKVGTCVGFQTKNNGIKDNG 
E GSWKROGPVVJCVJCUDGGPTGVWTPGFVQEQOOGTEKCN

)52TQXKUKQPHQT5GCNKPI'NGEVTQPKE#FLWUVCDNG EQPHQTOVQRGTHQTOCPEGTGSWKTGOGPVUCPFKUDGKPI

%QORQPGPVU#FGXKEGUJCNNDGFGUKIPGFYKVJ QHHKEKCNN[VGUVGFHQTVJGHKTUVVKOGYKVJKPFC[U

RTQXKUKQP
UHQTCRRN[KPICUGEWTKV[UGCNVJCVOWUVDG CHVGTEQTTGEVKXGUGTXKEG

DTQMGPQTHQTWUKPIQVJGTCRRTQXGFOGCPUQHRTQXKFKPI

UGEWTKV[
GIFCVCEJCPIGCWFKVVTCKNCXCKNCDNGCVVJG 
F GSWKROGPVVJCVKUDGKPIQHHKEKCNN[VGUVGFHQTVJGHKTUV

VKOGQHKPURGEVKQPDGHQTGCP[EJCPIGVJCVFGVTKOGPVCNN[ VKOGYKVJKPFC[UCHVGTOCLQTTGEQPFKVKQPKPIQT

CHHGEVUVJGOGVTQNQIKECNKPVGITKV[QHVJGFGXKEGECPDG QXGTJCWNCPF

OCFGVQCP[GNGEVTQPKEOGEJCPKUO

=0QPTGVTQCEVKXGCUQH,CPWCT[? 
G GSWKROGPVWPFGTIQKPIV[RGGXCNWCVKQP

#FGXKEGOC[DGHKVVGFYKVJCPCWVQOCVKEQTCUGOK

CWVQOCVKEECNKDTCVKQPOGEJCPKUO6JKUOGEJCPKUOUJCNN )6/CKPVGPCPEG6QNGTCPEGU/CKPVGPCPEG

DGKPEQTRQTCVGFKPUKFGVJGFGXKEG#HVGTUGCNKPIPGKVJGT VQNGTCPEGUUJCNNCRRN[VQGSWKROGPVKPCEVWCNWUGGZEGRV

VJGOGEJCPKUOPQTVJGECNKDTCVKQPRTQEGUUUJCNNHCEKNKVCVG CURTQXKFGFKP)6

HTCWF


#FFGF
#OGPFGFCPF )6#RRNKECVKQP6QNGTCPEGUKPGZEGUUCPF

)00QVGU

)0%QPHNKEVQH.CYUCPF4GIWNCVKQPU+HCP[

RCTVKEWNCTRTQXKUKQPUQHVJGUGURGEKHKECVKQPUVQNGTCPEGU

CPFQVJGTTGSWKTGOGPVUCTGHQWPFVQEQPHNKEVYKVJGZKUVKPI

5VCVGNCYUQTYKVJGZKUVKPITGIWNCVKQPUQTNQECN

QTFKPCPEGUTGNCVKPIVQJGCNVJUCHGV[QTHKTGRTGXGPVKQP

VJGGPHQTEGOGPVQHUWEJRTQXKUKQPUUJCNNDGUWURGPFGF

WPVKNEQPHNKEVKPITGSWKTGOGPVUECPDGJCTOQPK\GF5WEJ

UWURGPUKQPUJCNNPQVCHHGEVVJGXCNKFKV[QTGPHQTEGOGPVQH

VJGTGOCKPKPIRTQXKUKQPUQHVJGUGURGEKHKECVKQPUVQNGTCP

EGUCPFQVJGTTGSWKTGOGPVU

)06GUVKPI9KVJ0QPCUUQEKCVGF'SWKROGPV

6GUVUVQFGVGTOKPGEQPFKVKQPUUWEJCUTCFKQHTGSWGPE[

KPVGTHGTGPEG
4(+VJCVOC[CFXGTUGN[CHHGEVVJG )745GNGEVKQP4GSWKTGOGPVU

RGTHQTOCPEGQHCFGXKEGUJCNNDGEQPFWEVGFYKVJ

GSWKROGPVCPFWPFGTEQPFKVKQPUVJCVCTGWUWCNCPF )745WKVCDKNKV[QH'SWKROGPV%QOOGTEKCN

EWUVQOCT[YKVJTGURGEVVQVJGNQECVKQPCPFWUGQHVJG GSWKROGPVUJCNNDGUWKVCDNGHQTVJGUGTXKEGKPYJKEJKV

FGXKEG KUWUGFYKVJTGURGEVVQGNGOGPVUQHKVUFGUKIP


#FFGF KPENWFKPIDWVPQVNKOKVGFVQKVUYGKIJKPIECRCEKV[HQT

)66QNGTCPEGU

)6#EEGRVCPEG6QNGTCPEGU#EEGRVCPEG

VQNGTCPEGUUJCNNCRRN[VQ


D GSWKROGPVVJCVJCUDGGPRNCEGFKPEQOOGTEKCN

UGTXKEGHQNNQYKPIQHHKEKCNTGLGEVKQPHQTHCKNWTGVQ


#OGPFGF

VQNGTCPEGUKPFGHKEKGPE[UJCNNCRRN[VQGTTQTUKPGZEGUU

CPFVQGTTQTUKPFGHKEKGPE[TGURGEVKXGN[6QNGTCPEGUQP

QXGTTGIKUVTCVKQPCPFVQNGTCPEGUQPWPFGTTGIKUVTCVKQP

UJCNNCRRN[VQGTTQTUKPVJGFKTGEVKQPQHQXGTTGIKUVTCVKQP

CPFQHWPFGTTGIKUVTCVKQPTGURGEVKXGN[5GG#RRGPFKZ&

&GHKPKVKQPU

)6(QT+PVGTOGFKCVG8CNWGU(QTCECRCEKV[

KPFKECVKQPNQCFXCNWGGVEKPVGTOGFKCVGDGVYGGPVYQ

ECRCEKVKGUKPFKECVKQPUNQCFUXCNWGUGVENKUVGFKPC

VCDNGQHVQNGTCPEGUVJGVQNGTCPEGURTGUETKDGFHQTVJG

NQYGTECRCEKV[KPFKECVKQPNQCFXCNWGGVEUJCNNDG

CRRNKGF

)747UGT4GSWKTGOGPVU

YGKIJKPIFGXKEGUKVUEQORWVKPIECRCDKNKV[
HQT

EQORWVKPIFGXKEGUKVUTCVGQHHNQY
HQTNKSWKF

OGCUWTKPIFGXKEGUVJGEJCTCEVGTPWODGTUK\GCPF

NQECVKQPQHKVUKPFKECVKPIQTTGEQTFKPIGNGOGPVUCPF

VJGXCNWGQHKVUUOCNNGUVWPKVCPFWPKVRTKEGU


#OGPFGF
)GPGTCN%QFG



)74'PXKTQPOGPV'SWKROGPVUJCNNDG EWUVQOCT[OGCPUCPFKPVJGCOQWPVCPFUK\G

UWKVCDNGHQTVJGGPXKTQPOGPVKPYJKEJKVKUWUGF FGGOGFPGEGUUCT[D[UWEJQHHKEKCNHQTVJGRTQRGT

KPENWFKPIDWVPQVNKOKVGFVQVJGGHHGEVUQHYKPF EQPFWEVQHVJGVGUV

YGCVJGTCPF4(+


#FFGF 1VJGTYKUGKVUJCNNDGVJGTGURQPUKDKNKV[QHVJGFGXKEG

)74.KSWKF/GCUWTKPI&GXKEGU6QDG KPENWFKPIUWEJNCDQTCUOC[DGPGGFGFVQKPURGEVVGUV

UWKVCDNGHQTKVUCRRNKECVKQPVJGOKPKOWOFGNKXGT[HQT CPFUGCNVJGFGXKEGCPFVQVTCPURQTVVJGVGUVKPI

NKSWKFOGCUWTKPIFGXKEGUUJCNNDGPQNGUUVJCP GSWKROGPVVQCPFHTQOVJGFGXKEGCUTGSWKTGFD[VJG

FKXKUKQPUGZEGRVVJCVVJGOKPKOWOFGNKXGT[HQTTGVCKN YGKIJVUCPFOGCUWTGUQHHKEKCN

CPCNQIFGXKEGUUJCNNDGPQNGUUVJCPFKXKUKQPU 
#OGPFGF

/CZKOWOFKXKUKQPXCNWGUCPFVQNGTCPEGUCTGUVCVGFKP

VJGURGEKHKEEQFGU )747UG4GSWKTGOGPVU


#FFGF

)74+PUVCNNCVKQP4GSWKTGOGPVU DGQRGTCVGFQPN[KPVJGOCPPGTVJCVKUQDXKQWUN[

)74+PUVCNNCVKQP#FGXKEGUJCNNDGKPUVCNNGF KPUVTWEVKQPUQPVJGGSWKROGPV

KPCEEQTFCPEGYKVJVJGOCPWHCEVWTGT	UKPUVTWEVKQPU

KPENWFKPICP[KPUVTWEVKQPUOCTMGFQPVJGFGXKEG# )74#UUQEKCVGFCPF0QPCUUQEKCVGF

FGXKEGKPUVCNNGFKPCHKZGFNQECVKQPUJCNNDGKPUVCNNGFUQ 'SWKROGPV#FGXKEGUJCNNOGGVCNNRGTHQTOCPEG

VJCVPGKVJGTKVUQRGTCVKQPPQTKVURGTHQTOCPEGYKNNDG TGSWKTGOGPVUYJGPCUUQEKCVGFQTPQPCUUQEKCVGF

CFXGTUGN[CHHGEVGFD[CP[EJCTCEVGTKUVKEQHVJGHQWPFC GSWKROGPVKUQRGTCVGFKPKVUWUWCNCPFEWUVQOCT[

VKQPUWRRQTVUQTCP[QVJGTFGVCKNQHVJGKPUVCNNCVKQP OCPPGTCPFNQECVKQP

)748KUKDKNKV[QH+FGPVKHKECVKQP'SWKR

OGPVUJCNNDGKPUVCNNGFKPUWEJCOCPPGTVJCVCNN )742QUKVKQPQH'SWKROGPV#FGXKEGQT

TGSWKTGFOCTMKPIUCTGTGCFKN[QDUGTXCDNG U[UVGOGSWKRRGFYKVJCRTKOCT[KPFKECVKPIGNGOGPVCPF


#FFGF WUGFKPFKTGEVUCNGUGZEGRVHQTRTGUETKRVKQPUECNGU

)74+PUVCNNCVKQPQH+PFKECVKPIQT4GEQTFKPI CEEWTCVGN[TGCFCPFVJGYGKIJKPIQTOGCUWTKPI

'NGOGPV#FGXKEGUJCNNDGUQKPUVCNNGFVJCVVJGTGKU QRGTCVKQPOC[DGQDUGTXGFHTQOUQOGTGCUQPCDNG

PQQDUVTWEVKQPDGVYGGPCRTKOCT[KPFKECVKPIQT EWUVQOGTCPFQRGTCVQT RQUKVKQP6JGRGTOKUUKDNG

TGEQTFKPIGNGOGPVCPFVJGYGKIJKPIQTOGCUWTKPI FKUVCPEGDGVYGGPVJGGSWKROGPVCPFCTGCUQPCDNG

GNGOGPVQVJGTYKUGVJGTGUJCNNDGEQPXGPKGPVCPFRGT EWUVQOGTCPFQRGTCVQTRQUKVKQPUJCNNDGFGVGTOKPGFKP

OCPGPVN[KPUVCNNGFOGCPUHQTFKTGEVEQOOWPKECVKQP GCEJECUGWRQPVJGDCUKUQHVJGKPFKXKFWCNEKT

QTCNQTXKUWCNDGVYGGPCPKPFKXKFWCNNQECVGFCVC EWOUVCPEGURCTVKEWNCTN[VJGUK\GCPFEJCTCEVGTQHVJG

RTKOCT[KPFKECVKPIQTTGEQTFKPIGNGOGPVCPFCP KPFKECVKPIGNGOGPV

KPFKXKFWCNNQECVGFCVVJGYGKIJKPIQTOGCUWTKPIGNG 
#OGPFGFCPF

OGPV=5GGCNUQ)74?

)74#EEGUUKDKNKV[HQT+PURGEVKQP6GUVKPI &GXKEGU/QPG[QRGTCVGFFGXKEGUQVJGTVJCPRCTMKPI

CPF5GCNKPI2WTRQUGU#FGXKEGUJCNNDGNQECVGFQT OGVGTUUJCNNJCXGENGCTN[CPFEQPURKEWQWUN[FKURNC[GF

UWEJHCEKNKVKGUHQTPQTOCNCEEGUUVJGTGVQUJCNNDG VJGTGQPQTKOOGFKCVGN[CFLCEGPVVJGTGVQCFGSWCVG

RTQXKFGFVQRGTOKV KPHQTOCVKQPFGVCKNKPIVJGOGVJQFHQTVJGTGVWTPQH


C KPURGEVKPICPFVGUVKPIVJGFGXKEG QDVCKPGF6JKUKPHQTOCVKQPUJCNNKPENWFGVJGPCOG


D KPURGEVKPICPFCRRN[KPIUGEWTKV[UGCNUVQVJG RCTV[HQTVJGFGXKEG6JKUTGSWKTGOGPVFQGUPQVCRRN[

FGXKEGCPF VQFGXKEGUCVNQECVKQPUYJGTGGORNQ[GGUCTGRTGUGPV


E TGCFKN[DTKPIKPIVJGVGUVKPIGSWKROGPVQHVJG FKUETGRCPEKGUHQTVJGEWUVQOGT

YGKIJVUCPFOGCUWTGUQHHKEKCNVQVJGFGXKEGD[ 
#OGPFGF

QYPGTQTQRGTCVQTVQUWRRN[UWEJURGEKCNHCEKNKVKGU

)74/GVJQFQH1RGTCVKQP'SWKROGPVUJCNN

KPFKECVGFD[KVUEQPUVTWEVKQPQTVJCVKUKPFKECVGFD[


#FFGF

UJCNNDGRQUKVKQPGFUQVJCVKVUKPFKECVKQPUOC[DG

)744GURQPUKDKNKV[/QPG[1RGTCVGF

OQPKGURCKFYJGPVJGRTQFWEVQTUGTXKEGECPPQVDG

CFFTGUUCPFRJQPGPWODGTQHVJGNQECNTGURQPUKDNG

CPFTGURQPUKDNGHQTTGUQNXKPICP[OQPGVCT[
)GPGTCN%QFG



)74/CKPVGPCPEG4GSWKTGOGPVU VJGQHHKEKCNYKVJUVCVWVQT[CWVJQTKV[OC[TGSWKTG

)74/CKPVGPCPEGQH'SWKROGPV#NN HQTVGUVKPI6JGFGCNGTQTQYPGTQHVJGUGFGXKEGU

GSWKROGPVKPUGTXKEGCPFCNNOGEJCPKUOUCPFFGXKEGU UJCNNRTQXKFGVTCPURQTVCVKQPQHVJGFGXKEGUVQCPF

CVVCEJGFVJGTGVQQTWUGFKPEQPPGEVKQPVJGTGYKVJUJCNN HTQOVJGVGUVNQECVKQP

DGEQPVKPWQWUN[OCKPVCKPGFKPRTQRGTQRGTCVKPI

EQPFKVKQPVJTQWIJQWVVJGRGTKQFQHUWEJUGTXKEG 
D 9JGPVJGTGSWGUVHQTTGOQXCNCPFFGNKXGT[VQC

'SWKROGPVKPUGTXKEGCVCUKPINGRNCEGQHDWUKPGUU EGPVTCNVGUVNQECVKQPKPXQNXGUFGXKEGUWUGFKP

HQWPFVQDGKPGTTQTRTGFQOKPCPVN[KPCFKTGEVKQP UWDOGVGTKPIGIGNGEVTKEJ[FTQECTDQPXCRQT

HCXQTCDNGVQVJGFGXKEGWUGTUJCNNPQVDGEQPUKFGTGF QTYCVGTOGVGTUVJGQYPGTQTQRGTCVQTUJCNNPQV

OCKPVCKPGFKPCRTQRGTQRGTCVKPIEQPFKVKQP KPVGTTWRVVJGWVKNKV[UGTXKEGVQVJGEWUVQOGTQT


#OGPFGF VGPCPVGZEGRVHQTVJGTGOQXCNCPFTGRNCEGOGPVQH

)74#DPQTOCN2GTHQTOCPEG7PUVCDNG QYPGTQTQRGTCVQTVQOKPKOK\GKPEQPXGPKGPEGVQ

KPFKECVKQPUQTQVJGTCDPQTOCNGSWKROGPVRGTHQTOCPEG VJGEWUVQOGTQTVGPCPV#NNTGRNCEGOGPVQT

QDUGTXGFFWTKPIQRGTCVKQPUJCNNDGEQTTGEVGFCPFKH VGORQTCT[OGVGTUUJCNNDGVGUVGFCPFUGCNGFD[C

PGEGUUCT[DTQWIJVVQVJGCVVGPVKQPQHEQORGVGPV YGKIJVUCPFOGCUWTGQHHKEKCNQTDGCTCEWTTGPV

UGTXKEGRGTUQPPGN XCNKFCRRTQXCNUGCNRTKQTVQWUG


#FFGF 
#FFGF

)747UGQH#FLWUVOGPVU9GKIJKPIGNGOGPVU

CPFOGCUWTKPIGNGOGPVUVJCVCTGCFLWUVCDNGUJCNNDG

CFLWUVGFQPN[VQEQTTGEVVJQUGEQPFKVKQPUVJCVUWEJ

GNGOGPVUCTGFGUKIPGFVQEQPVTQNCPFUJCNNPQVDG

CFLWUVGFVQEQORGPUCVGHQTFGHGEVKXGQTCDPQTOCN

KPUVCNNCVKQPQTCEEGUUQTKGUQTHQTDCFN[YQTPQT

QVJGTYKUGFGHGEVKXGRCTVUQHVJGCUUGODN[#P[HCWNV[

KPUVCNNCVKQPEQPFKVKQPUUJCNNDGEQTTGEVGFCPFCP[

FGHGEVKXGRCTVUUJCNNDGTGPGYGFQTUWKVCDN[TGRCKTGF

DGHQTGCFLWUVOGPVUCTGWPFGTVCMGP9JGPGXGTGSWKR

OGPVKUCFLWUVGFVJGCFLWUVOGPVUUJCNNDGUQOCFGCU

VQDTKPIRGTHQTOCPEGGTTQTUCUENQUGCURTCEVKECDNGVQ

\GTQXCNWG

)74#UUKUVCPEGKP6GUVKPI1RGTCVKQPU+HVJG

FGUKIPEQPUVTWEVKQPQTNQECVKQPQHCP[FGXKEGKUUWEJ

CUVQTGSWKTGCVGUVKPIRTQEGFWTGKPXQNXKPIURGEKCN

GSWKROGPVQTCEEGUUQTKGUQTCPCDPQTOCNCOQWPVQH

NCDQTUWEJGSWKROGPVCEEGUUQTKGUCPFNCDQTUJCNNDG

UWRRNKGFD[VJGQYPGTQTQRGTCVQTQHVJGFGXKEGCU

TGSWKTGFD[VJGYGKIJVUCPFOGCUWTGUQHHKEKCN

)745GEWTKV[5GCN#UGEWTKV[UGCNUJCNNDG

CRRTQRTKCVGN[CHHKZGFVQCP[CFLWUVOGPVOGEJCPKUO

FGUKIPGFVQDGUGCNGF

)746GUVKPI&GXKEGUCVC%GPVTCN.QECVKQP


C 9JGPFGXKEGUKPEQOOGTEKCNUGTXKEGTGSWKTG

URGEKCNVGUVHCEKNKVKGUQTOWUVDGTGOQXGFHTQO

UGTXKEGHQTVGUVKPIQTCTGTQWVKPGN[VTCPURQTVGF

HQTVJGRWTRQUGQHWUG
GIXGJKENGOQWPVGF

FGXKEGUCPFFGXKEGUWUGFKPOWNVKRNGNQECVKQPU

VJCVVJGFGXKEGUDGDTQWIJVVQCEGPVTCNNQECVKQP

VJGFGXKEG2TQXKUKQPUUJCNNDGOCFGD[VJG


Section 2

Page

2.20. Scales 2-3

2.21. Belt-Conveyor Scale Systems 2-37

2.22. Automatic Bulk Weighing Systems 2-45

2.23. Weights 2-51

2.24. Automatic Weighing Systems - Tentative Code 2-55
2.20. Scales



Sec. 2.20. Scales

A. Application

A.1. General. - This code applies to all types of weighing

devices other than automatic bulk-weighing systems, beltconveyor scales, and automatic weighing systems. The

code comprises requirements that generally apply to all

weighing devices, and specific requirements that are

applicable only to certain types of weighing devices.

(Amended 1972 and 1983)

A.2. Wheel-Load Weighers, Portable Axle-Load

Weighers, and Axle-Load Scales. - The requirements for

wheel-load weighers, portable axle-load weighers, and axleload scales apply only to such scales in official use for the

enforcement of traffic and highway laws or for the

collection of statistical information by government

agencies.

A.3. Also see General Code requirements.

S. Specifications

S.1. Design of Indicating and Recording Elements and

of Recorded Representations.

S.1.1. Zero Indication.

(a) On a scale equipped with indicating or recording

elements, provision shall be made to either indicate

or record a zero-balance condition.

(b) On an automatic-indicating scale or balance

indicator, provision shall be made to indicate or

record an out-of-balance condition on both sides of

zero.

(c) A zero-balance condition may be indicated by other

than a continuous digital zero indication, provided

that an effective automatic means is provided to

inhibit a weighing operation or to return to a

continuous digital indication when the scale is in an

out-of-balance condition.

(Added 1987)

(Amended 1987, 1993)

S.1.1.1. Digital Indicating Elements. -

(a) A digital zero indication shall represent a balance

condition that is within ±1⁄2 the value of the scale

division.

(b) A digital indicating device shall either

automatically maintain a "center-of-zero" condition

to ±1⁄4 scale division or less, or have an auxiliary or

supplemental "center-of-zero" indicator that defines

a zero-balance condition to ±1⁄4 of a scale division or

less.

[Nonretroactive as of January 1, 1993.]

(Amended 1992)

S.1.1.2. No-Load Reference Value. - On a single

draft manually operated receiving hopper scale

installed below grade, used to receive grain, and

utilizing a no-load reference value, provision shall be

made to indicate and record the no-load reference

value prior to the gross load value.

(Added 1983)

S.1.2. Value of Scale Division Units. - Except for

batching scales and weighing systems used exclusively

for weighing in predetermined amounts, the value of a

scale division "d" expressed in a unit of weight shall be

equal to:

 

(a) 1, 2, or 5; or 

(b) a decimal multiple or submultiple of 1, 2, or 5; 

(c) a binary submultiple of a specific unit of weight.

Examples: scale divisions may be 0.01, 0.02, 0.05; 0.1,

0.2, or 0.5; 1, 2, or 5; 10, 20, 50, or 100; or, scale

divisions may be 1/2, 1/4, 1/8, 1/16, etc.

[Nonretroactive as of January 1, 1986.]

S.1.2.1. Weight Units. - Except for postal scales, a

digital-indicating scale shall indicate weight values

using only a single unit of measure. Weight values

shall be presented in a decimal format with the value

of the scale division expressed as 1, 2, or 5, or a

decimal multiple or submultiple of 1, 2, or 5.

[Nonretroactive and enforceable as of January 1,

1989.]

(Added 1987)

S.1.2.2. Verification Scale Interval. -

S.1.2.2.1. Class I and II Scales and Dynamic

Monorail Scales. If eú d, the verification scale

interval "e" shall be determined by the expression:

d < e <10 d
2.20. Scales



If the displayed division (d) is less than the

verification division (e), then the verification division

shall be less than or equal to 10 times the displayed

division.

The value of e must satisfy the relationship, e =10 of k

the unit of measure, where k being a positive or

negative whole number or zero. This requirement

does not apply to a Class I device with d < 1 mg

where e = 1 mg. If e ú d, the value of "d" shall be a

decimal submultiple of "e," and the ratio shall not be

more than 10:1. If e ú d, and both "e" and "d" are

continuously displayed during normal operation, then

"d" shall be differentiated from "e" by size, shape,

color, etc. throughout the range of weights displayed

as "d."

(Added 1999)

S.1.2.2.2. Class III and IIII. The value of "e" is

specified by the manufacturer as marked on the

device. Except for dynamic monorail scales, "e" must

be less than or equal to "d."

(Added 1999)

S.1.3. Graduations.

S.1.3.1. Length. - Graduations shall be so varied in

length that they may be conveniently read.

S.1.3.2. Width. - In any series of graduations, the

width of a graduation shall in no case be greater than

the width of the clear space between graduations. The

width of main graduations shall be not more than 50

percent greater than the width of subordinate

graduations. Graduations shall be not less than

0.2 mm (0.008 in) wide.

S.1.3.3. Clear Space Between Graduations. - The

clear space between graduations shall be not less than

0.5 mm (0.02 in) for graduations representing money

values, and not less than 0.75 mm (0.03 in) for other

graduations. If the graduations are not parallel, the

measurement shall be made:

(a) along the line of relative movement between the

graduations at the end of the indicator, or 

(b) if the indicator is continuous, at the point of

widest separation of the graduations.

 S.1.4. Indicators.

S.1.4.1. Symmetry. - The index of an indicator shall

be of the same shape as the graduations, at least

throughout that portion of its length associated with

the graduations.

S.1.4.2. Length. - The index of an indicator shall

reach to the finest graduations with which it is used,

unless the indicator and the graduations are in the

same plane, in which case, the distance between the

end of the indicator and the ends of the graduations,

measured along the line of the graduations, shall be

not more than 1.0 mm (0.04 in).

S.1.4.3. Width. - The width of the index of an

indicator in relation to the series of graduations with

which it is used shall be not greater than: 

(a) the width of the narrowest graduation,

[Nonretroactive as of January 1, 2002]

(b) the width of the clear space between weight

graduations, and

(c) three-fourths of the width of the clear space

between money value graduations.

When the index of an indicator extends along the

entire length of a graduation, that portion of the index

of the indicator that may be brought into coincidence

with the graduation shall be of the same width

throughout the length of the index that coincides with

the graduation.

S.1.4.4. Clearance. - The clearance between the

index of an indicator and the graduations shall in no

case be more than 1.5 mm (0.06 in).

S.1.4.5. Parallax. - Parallax effects shall be reduced

to the practicable minimum. 

S.1.5. Weighbeams.

 

S.1.5.1. Normal Balance Position. - The normal

balance position of the weighbeam of a beam scale

shall be horizontal.

S.1.5.2. Travel. - The weighbeam of a beam scale

shall have equal travel above and below the

horizontal. The total travel of the weighbeam of a

beam scale in a trig loop or between other limiting

stops near the weighbeam tip shall be not less than the

minimum travel shown in Tables 1M and 1. When
2.20. Scales



such limiting stops are not provided, the total travel

at the weighbeam tip shall be not less than

8 percent of the distance from weighbeam fulcrum

to the weighbeam tip.

Table 1M.

Minimum Travel of Weighbeam of Beam Scale

Between Limiting Stops

Distance from weighbeam Minimum travel between

fulcrum to limiting stops limiting stops

(centimeters) (millimeter)

30 or less 10

30+ to 50, inclusive 13

50+ to 100, inclusive 18

Over 100 23

Table 1.

Minimum Travel of Weighbeam of Beam Scale

Between Limiting Stops

Distance from weighbeam Minimum travel between

fulcrum to limiting stops limiting stops

(inches) (inch)

12 or less 0.4

12+ to 20, inclusive 0.5

20+ to 40, inclusive 0.7

Over 40 0.9

S.1.5.3. Subdivision. - A subdivided weighbeam bar

shall be subdivided by scale division graduations,

notches, or a combination of both. Graduations on a

particular bar shall be of uniform width and perpendicular to the top edge of the bar. Notches on a

particular bar shall be uniform in shape and

dimensions and perpendicular to the face of the bar.

When a combination of graduations and notches is

employed, the graduations shall be positioned in

relation to the notches to indicate notch values clearly

and accurately.

S.1.5.4. Readability. - A subdivided weighbeam bar

shall be so subdivided and marked, and a weighbeam

poise shall be so constructed, that the weight corresponding to any normal poise position can easily and

accurately be read directly from the beam, whether or

not provision is made for the optional recording of

representations of weight.

S.1.5.5. Capacity. - On an automatic-indicating scale

having a nominal capacity of 15 kg (30 lb) or less and

used for direct sales to retail customers:

(a) the capacity of any weighbeam bar shall be a

multiple of the reading-face capacity,

(b) each bar shall be subdivided throughout or shall

be subdivided into notched intervals, each equal

to the reading-face capacity; and

(c) the value of any turnover poise shall be equal to

the reading-face capacity.

 

S.1.5.6. Poise Stop. - Except on a steelyard with no

zero graduation, a shoulder or stop shall be provided

on each weighbeam bar to prevent a poise from

traveling and remaining back of the zero graduation.

S.1.6. Poises.

S.1.6.1. General. - No part of a poise shall be readily

detachable. A locking screw shall be perpendicular to

the longitudinal axis of the weighbeam and shall not

be removable. Except on a steelyard with no zero

graduation, a poise shall not be readily removable

from a weighbeam. The knife edge of a hanging poise

shall be hard and sharp and so constructed as to allow

the poise to swing freely on the bearing surfaces in the

weighbeam notches. 

S.1.6.2. Adjusting Material. - The adjusting material

in a poise shall be securely enclosed and firmly fixed

in position; if softer than brass, it shall not be in

contact with the weighbeam.

S.1.6.3. Pawl. - A poise, other than a hanging poise,

on a notched weighbeam bar shall have a pawl that

will seat the poise in a definite and correct position in

any notch, wherever in the notch the pawl is placed,

and hold it there firmly and without appreciable

movement. The dimension of the tip of the pawl that

is transverse to the longitudinal axis of the weighbeam

shall be at least equal to the corresponding dimension

of the notches.

S.1.6.4. Reading Edge or Indicator. - The reading

edge or indicator of a poise shall be sharply defined,

and a reading edge shall be parallel to the graduations

on the weighbeam.

S.1.7. Capacity Indication, Weight Ranges, and Unit

Weights. -
2.20. Scales



(a) Gross Capacity. An indicating or recording (a) 1 cent at all unit prices of 25 cents per pound

element shall not display nor record any values and less;

when the total platform load (not counting the

initial dead load that has been canceled by an initial (b) 2 cents at unit prices of 26 cents per pound

zero-setting mechanism) is in excess of 105 percent through $1.25 per pound (special graduations

of scale capacity. defining 5-cent intervals may be employed but

(b) Capacity Indication. Electronic computing scales

(excluding postal scales and weight classifiers) (c) 5 cents at unit prices of $1.26 per pound through

shall neither display nor record a gross or net $3.40 per pound; or

weight in excess of scale capacity plus 9d.

The total value of weight ranges and of unit weights in

effect or in place at any time shall automatically be Value figures and graduations shall not be duplicated

accounted for on the reading face and on any recorded in any column or row on the graduated chart. (See

representation. also S.1.8.2.)

This requirement does not apply to: (1) single-revolution

dial scales, (2) multi-revolution dial scales not equipped

with unit weights, (3) scales equipped with two or more

weighbeams, nor (4) devices that indicate mathematically

derived totalized values.

[Nonretroactive as of January 1, 1993.]

(Amended 1990, 1992, and 1995)

S.1.8. Computing Scales.

S.1.8.1.M. Money-Value Graduations, Metric Unit

Prices. - The value of the graduated intervals

representing money values on a computing scale with

analog indications shall not exceed:

(a) 1 cent at all unit prices of 55 cents per kilogram

and less;

(b) 2 cents at unit prices of 56 cents per kilogram

through $2.75 per kilogram (special graduations

defining 5-cent intervals may be employed but

not in the spaces between regular graduations);

(c) 5 cents at unit prices of $2.76 per kilogram

through $7.50 per kilogram; or

(d) 10 cents at unit prices above $7.50 per kilogram.

Value figures and graduations shall not be duplicated

in any column or row on the graduated chart. (See

also S.1.8.2.)

S.1.8.1. Money-Value Graduations, Inch-Pound

Unit Prices. - The value of the graduated intervals

representing money values on a computing scale with

analog indications shall not exceed:

not in the spaces between regular graduations);

(d) 10 cents at unit prices above $3.40 per pound.

S.1.8.2. Money-Value Computation. - A computing

scale with analog quantity indications used in retail

trade may compute and present digital money values

to the nearest quantity graduation when the value of

the minimum graduated interval is 0.005 kg (0.01 lb)

or less. (Also see Sec. 1.10; G-S.5.5.)

S.1.8.3. Customer’s Indications. - Weight

indications shall be shown on the customer’s side of

computing scales when these are used for direct sales

to retail customers. Computing scales equipped on the

operator’s side with digital indications, such as the net

weight, unit price, or total price, shall be similarly

equipped on the customer’s side. Unit price displays

visible to the customer shall be in terms of single

whole units of weight and not in common or decimal

fractions of the unit. Scales indicating in metric units

may indicate price per 100 grams.

(Amended 1985 and 1995)

S.1.8.3.1. Scales that will function as either a

normal round off scale or as a weight classifier

shall be provided with a sealable means for

selecting the mode of operation and shall have a

clear indication (annunciator), adjacent to the

weight display on both the operator’s and

customer’s side whenever the scale is operating as

a weight classifier.

[Nonretroactive as of January 1, 2001.]

(Added 1999)

S.1.8.4. Recorded Representations, Point-of-Sale

Systems. - The sales information recorded by cash

registers when interfaced with a weighing element

shall contain the following information for items

weighed at the checkout stand: 

 
2.20. Scales



 Weight values shall be identified by kilogram, kg, grams, 1

g, ounces, oz, pound, lb, or the sign "#." For devices

interfaced with scales indicating in metric units, the unit

price may be expressed in price per 100 grams. 

(Amended 1995)

(a) the net weight, [Nonretroactive as of January 1, 1990] 1

(b) the unit price,1

(c) the total price, and (c) Except on Class I scales, audit trails shall use

(d) the product class or, in a system equipped with the format set forth in Table S.1.11.

price look-up capability, the product name or [Nonretroactive as of January 1, 1995]

code number.

S.1.9. Prepackaging Scales.

S.1.9.1. Value of the Scale Division. - On a

prepackaging scale, the value of the intervals

representing weight values shall be uniform

throughout the entire reading face. The recorded

weight values shall be identical with those on the

indicator.

S.1.9.2. Label Printer. - A prepackaging scale or a

device that produces a printed ticket to be used as the

label for a package shall print all values digitally and

of such size, style of type, and color as to be clear and

conspicuous on the label.

S.1.10. Adjustable Components. - An adjustable

component such as a pendulum, spring, or potentiometer

shall be held securely in adjustment and, except for a

zero-load balance mechanism, shall be located within the

housing of the element.

(Added 1986)

S.1.11. Provision for Sealing.

(a) Except on Class I scales, provision shall be made

for applying a security seal in a manner that

requires the security seal to be broken before an

adjustment can be made to any component

affecting the performance of an electronic device.

[Nonretroactive as of January 1, 1979.]

(b) Except on Class I scales, a device shall be designed S.2. Design of Balance, Tare, Level, Damping, and

with provision(s) for applying a security seal that Arresting Mechanisms.

must be broken, or for using other approved means

of providing security (e.g., data change audit trail

available at the time of inspection), before any

change that detrimentally affects the metrological

integrity of the device can be made to any

electronic mechanism.

A device may be fitted with an automatic or a semiautomatic calibration mechanism. This mechanism shall

be incorporated inside the device. After sealing, neither

the mechanism nor the calibration process shall facilitate

fraud. (Amended 1989, 1991, 1993)

S.1.12. Manual Gross Weight Entries. - A device shall

accept an entry of a manual gross weight value only

when the scale is at gross load zero and the scale

indication is at zero in the gross weight display mode.

Recorded manual weight entries, except those on labels

generated for packages of standard weights, shall

identify the weight value as a manual weight entry by

one of the following terms: "Manual Weight," "Manual

Wt," or "MAN WT." The use of a symbol to identify

multiple manual weight entries on a single document is

permitted, provided that the symbol is defined on the

same page on which the manual weight entries appear

and the definition of the symbol is automatically printed

by the recording element as part of the document.

[Nonretroactive as of January 1, 1993.]

(Added 1992)

S.1.13. Vehicle On-Board Weighing Systems: Vehicle

in Motion. - When the vehicle is in motion, a vehicle onboard weighing system shall either:

(a) be accurate; or

(b) inhibit the weighing operation.

(Added 1993)

S.2.1. Zero-Load Adjustment.

S.2.1.1. General. - A scale shall be equipped with

means by which the zero-load balance may be

adjusted. Any loose material used for this purpose

shall be enclosed so that it cannot shift in position and

alter the balance condition of the scale.

S.2.1.2. Scales used in Direct Sales. - A manual

zero-setting mechanism (except on a digital scale with

an analog zero-adjustment mechanism with a range of

not greater than one scale division) shall be operable

or accessible only by a tool outside of and entirely
2.20. Scales



Table S.1.11. Categories of Device and Methods of Sealing

Categories of Device Method of Sealing

Category 1: No remote configuration

capability.

Seal by physical seal or two event counters: one for calibration

parameters and one for configuration parameters.

Category 2: Remote configuration capability,

but access is controlled by physical hardware.

Device shall clearly indicate that it is in the

remote configuration mode and record such

message if capable of printing in this mode.

The hardware enabling access for remote communication must

be at the device and sealed using a physical seal or two event

counters: one for calibration parameters and one for

configuration parameters.

Category 3: Remote configuration capability

access may be unlimited or controlled through

a software switch (e.g., password).

An event logger is required in the device; it must include an

event counter (000 to 999), the parameter ID, the date and time

of the change, and the new value of the parameter. A printed

copy of the information must be available through the device or

through another on-site device. The event logger shall have a

capacity to retain records equal to ten times the number of

sealable parameters in the device, but not more than 1000

records are required. (Note: Does not require 1000 changes to

be stored for each parameter.)

[Nonretroactive as of January 1, 1995]

(Table added 1993)

separate from this mechanism, or it shall be (a) for bench, counter, and livestock scales: 0.6

enclosed in a cabinet. Except on Class I or II scale division;

scales, a balance ball shall either meet this

requirement or not itself be rotatable. (b) for vehicle, axle-load, and railway track scales:

A semiautomatic zero-setting mechanism shall be

operable or accessible only by a tool outside of and (c) for all other scales: 1.0 scale division.

separate from this mechanism or it shall be enclosed [Nonretroactive and enforceable as of

in a cabinet, or it shall be operable only when the January 1, 1981.]

indication is stable within:

(a) plus or minus 3 scale divisions for scales of

more than 2000 kg (5000 lb) capacity in service

prior to January 1, 1981, and for all axle load,

railway track, and vehicle scales; or

(b) plus or minus 1 scale division for all other

scales.

S.2.1.3. Scales Equipped with an Automatic ZeroSetting Mechanism. - Under normal operating

conditions the maximum load that can be "rezeroed,"

when either placed on or removed from the platform

all at once, shall be:

3.0 scale divisions; and

S.2.1.3.1. Automatic Zero-Setting Mechanism on

Class III L Devices - Class III L devices equipped

with automatic zero setting mechanisms shall be

designed with a sealable means to allow the

automatic zero setting to be disabled during the

inspection and test of the device. 

[Nonretroactive as of January 1, 2001]

(Added 1999)

S.2.1.4. Monorail Scales. - On a static monorail scale

equipped with digital indications, means shall be

provided for setting the zero-load balance to within

0.02 percent of scale capacity. On a dynamic

monorail weighing system, means shall be provided to

automatically maintain these conditions.
2.20. Scales



(Amended 1999) A device designed to automatically clear any tare value

S.2.1.5. Initial Zero-Setting Mechanism. -

(a) Scales of accuracy classes I, II, and III may be

equipped with an initial zero-setting device. [Note: On a computing scale, this requires the input of

(b) An initial zero-setting mechanism shall not zero positive total price at a readable equilibrium. Other

a load in excess of 20 percent of the maximum devices require a complete weighing operation,

capacity of the scale unless tests show that the including tare, net, and gross weight determination.]*

scale meets all applicable tolerances for any [*Nonretroactive as of January 1, 1983.] 

amount of initial load compensated by this

device within the specified range.

(Added 1990)

5%QODKPGF<GTQ6CTG
 6 -G[

5ECNGUPQVKPVGPFGFVQDGWUGFKPFKTGEVUCNGU percent of the scale capacity to within 0.02 percent of

CRRNKECVKQPUOC[DGGSWKRRGFYKVJCEQODKPGF\GTQ

CPFVCTGHWPEVKQPMG[RTQXKFGFVJCVVJGFGXKEGKU

ENGCTN[OCTMGFCUVQJQYVJGMG[HWPEVKQPU6JG

FGXKEGOWUVCNUQDGENGCTN[OCTMGFQPQTCFLCEGPVVQ

VJGYGKIJVFKURNC[YKVJVJGUVCVGOGPV0QVHQT&KTGEV

5CNGU

(Added 1998)

S.2.2. Balance Indicator. - On a balance indicator

consisting of two indicating edges, lines, or points, the

ends of the indicators shall be sharply defined. When the

scale is in balance, the ends shall be separated by not

more than 1.0 mm (0.04 in).

S.2.2.1. Dairy-Product Test, Grain-Test,

Prescription, and Class I and II Scales. - Except on

digital indicating devices, a dairy-product-test, graintest, prescription, or Class I or II scale shall be

equipped with a balance indicator. If an indicator and

a graduated scale are not in the same plane, the

clearance between the indicator and the graduations

shall be not more than 1.0 mm (0.04 in).

S.2.2.2. Equal-Arm Scale. - An equal-arm scale shall

be equipped with a balance indicator. If the indicator

and balance graduation are not in the same plane, the

clearance between the indicator and the balance

graduation shall be not more than 1.0 mm (0.04 in).

[Nonretroactive as of January 1, 1989.]

(Added 1988)

S.2.3. Tare. - On any scale (except a monorail scale

equipped with digital indications), the value of the tare

division shall be equal to the value of the scale division.*

The tare mechanism shall operate only in a backward

direction (that is, in a direction of underregistration) with

respect to the zero-load balance condition of the scale.

shall also be designed to prevent the automatic clearing

of tare until a complete transaction has been indicated.*

(Amended 1985)

a unit price, the display of the unit price, and a computed

S.2.3.1. Monorail Scales Equipped with Digital

Indications. - On a static monorail weighing system

equipped with digital indications, means shall be

provided for setting any tare value of less than 5

scale capacity. On a dynamic monorail weighing

system, means shall be provided to automatically

maintain this condition. 

(Amended 1999)

S.2.4. Level-Indicating Means. - Except for portable

wheel-load weighers and portable axle-load scales, a

portable scale shall be equipped with level-indicating

means if its weighing performance is changed by an

amount greater than the appropriate acceptance tolerance

when it is moved from a level position and rebalanced in

a position that is out of level in any upright direction by

5 percent (approximately 3 degrees). The level-indicating

means shall be readable without removing any scale parts

requiring a tool.

[This requirement is nonretroactive and enforceable as

of January 1, 1986, for prescription, jewelers’, and

dairy-product-test scales and scales marked I and II.]

[Note: Portable wheel-load weighers and portable axleload scales shall be accurate when placed out of level up

to and including 5 percent (approximately 3 degrees).]

(Amended 1991)

S.2.4.1. Vehicle On-Board Weighing Systems. - A

vehicle on-board weighing system shall operate within

tolerance when the weighing system is out of level up

to 3 degrees or 5 percent. If the accuracy of the

system is affected by out-of-level conditions normal to

the use of the device, the system shall be equipped

with an out-of-level sensor that inhibits the weighing

operation when the system is out of level to the extent

that the accuracy limits are exceeded.

(Added 1992)
2.20. Scales



S.2.5. Damping Means. - An automatic-indicating scale

and a balance indicator shall be equipped with effective

means to damp oscillations and to bring the indicating

elements quickly to rest.

S.2.5.1. Digital Indicating Elements. - Digital

indicating elements equipped with recording elements

shall be equipped with effective means to permit the

recording of weight values only when the indication is

stable within: 

(a) plus or minus 3 scale divisions for scales of

more than 2000 kg (5000 lb) capacity in service

prior to January 1, 1981, hopper (other than

grain hopper) scales with a capacity exceeding

22 000 kg (50 000 lb), and for all vehicle, axle

load, livestock, and railway track scales;

(b) plus or minus 1 scale division for all other

scales.

The values recorded shall be within applicable

tolerances.

(Amended 1995)

S.2.5.2. Jewelers’, Prescription, and Class I and

Class II Scales. - A jewelers’, prescription, Class I, or

Class II scale shall be equipped with appropriate

means for arresting the oscillation of the mechanism.

S.3. Design of Load-Receiving Elements.

S.3.1. Travel of Pans of Equal-Arm Scale. - The travel

between limiting stops of the pans of a nonautomaticindicating equal-arm scale not equipped with a balance

indicator shall be not less than the minimum travel

shown in Table 2M and 2.

S.3.2. Drainage. - A load-receiving element intended to

receive wet commodities shall be so constructed as to

drain effectively.

Table 2M.

Minimum Travel of Pans of

Nonautomatic Indicating Equal-Arm Scale

Without Balance Indicator

Nominal capacity

(kilograms)

Minimum

travel of pans

(millimeters)

2 or less 9

2+ to 5 inclusive 13

5+ to 12, inclusive 19

Over 12 25

Table 2.

Minimum Travel of Pans of

Nonautomatic Indicating Equal-Arm Scale

Without Balance Indicator

Nominal capacity Minimum

(pounds) (inch)

travel of pans

4 or less 0.35

4+ to 12, inclusive 0.5

12+ to 26, inclusive 0.75

Over 26 1.0

S.3.3. Scoop Counterbalance. - A scoop on a scale

used for direct sales to retail customers shall not be

counterbalanced by a removable weight. A permanently

attached scoop-counterbalance shall indicate clearly on

both the operator’s and customer’s sides of the scale

whether it is positioned for the scoop to be on or off the

scale.

S.4. Design of Weighing Elements.

S.4.1. Antifriction Means. - Frictional effects shall be

reduced to a minimum by suitable antifriction elements.

Opposing surfaces and points shall be properly shaped,

finished, and hardened. A platform scale having a frame

around the platform shall be equipped with means to

prevent interference between platform and frame. 

S.4.2. Adjustable Components. - An adjustable

component such as a nose-iron or potentiometer shall be

held securely in adjustment. The position of a nose-iron

on a scale of more than 1000-kg (2000-lb) capacity, as
vmin #

d

N

vmin #

d

N × (scale multiple)

2.20. Scales



 See Footnote 1 to Table 3 Parameters for Accuracy 2

Classes

determined by the factory adjustment, shall be and,

accurately, clearly, and permanently defined.

(Amended 1986)

S.4.3. Multiple Load-Receiving Elements. - Except for

mechanical bench and counter scales, a scale with a

single indicating or recording element, or a combination

indicating-recording element, that is coupled to two or

more load-receiving elements with independent weighing

systems, shall be provided with means to prohibit the

activation of any load-receiving element (or elements)

not in use, and shall be provided with automatic means to

indicate clearly and definitely which load-receiving

element (or elements) is in use.

S.5. Design of Weighing Devices, Accuracy Class.

S.5.1. Designation of Accuracy Class. - Weighing

devices are divided into accuracy classes and shall be

designated as I, II, III, III L, or IIII.

[Nonretroactive as of January 1, 1986.]

S.5.2. Parameters for Accuracy Class. - The accuracy

class of a weighing device is designated by the manufacturer and shall comply with parameters shown in

Table 3.

[Nonretroactive as of January 1, 1986.]

S.5.3. Multi-Interval and Multiple Range Scales,

Division Value. - On a multi-interval scale and multiple

range scale, the value of "e" shall be equal to the value of

"d." (Added 1986) (Amended 1995) 2

S.5.4. Relationship of Load Cell Verification Interval

Value to the Scale Division. - The relationship of the

value for the load cell verification scale interval, v , to min

the scale division, d, for a specific scale installation shall

be:

(a) where N is the number of load cells in the scale for

scales without lever systems:

(b) for scales with lever systems:

[Nonretroactive as of January 1, 1994.]

[Note: When the value of the scale division, d, is different than

the verification scale division, e, for the scale, the value of e

must be used in the formulae above.]

6JKUTGSWKTGOGPVFQGUPQVCRRN[VQEQORNGVGUECNGUCPF

YGKIJKPIGNGOGPVUYJKEJUCVKUH[VJGHQNNQYKPIETKVGTKC


6JGFGXKEGJCUDGGPGXCNWCVGFHQTEQORNKCPEGYKVJ

606GORGTCVWTGWPFGTVJG0CVKQPCN6[RG

'XCNWCVKQP2TQITCO06'2


6JGFGXKEGJCUTGEGKXGFCP06'2%GTVKHKECVGQH

%QPHQTOCPEGCPF

(3) The device must be equipped with an automatic

zero-setting mechanism which cannot be made

inoperative in the normal weighing mode. (A test mode

which permits the disabling of the automatic zero-setting

mechanism is permissible, provided the scale cannot

function normally while in this mode.)

(Added 1993) (Amended 1996)

S.6. Marking Requirements. [See also G-S.1., G-S.4.,

G-S.6., G-S.7., G-UR.2.1.1., and UR.3.4.1.]

S.6.1. Nominal Capacity; Vehicle, Axle-Load, and

Livestock Scales. - For all vehicle, axle-load, and

livestock scales, the marked nominal capacity shall not

exceed the concentrated load capacity (CLC) times the

quantity of the number of sections in the scale minus 0.5.

As a formula, this is stated as:

nominal capacity < CLC x (N - 0.5)

where N = the number of sections in the scale.

(See N.1.3.4. and T.N.3.1.)

[Nonretroactive as of January 1, 1989.]

[Note: When the device is used in a combination railway

track and vehicle weighing application, the above formula

shall apply only to the vehicle scale application.]

(Added 1988) (Amended 1999)

S.6.2. Location Of Marking Information. - Scales that

are not permanently attached to an indicating element,
2.20. Scales



and for which the load-receiving element is the only

part of the weighing/load-receiving element visible

after installation, may have the marking information

required in G-S.1. of the General Code and S.6. of the

Scales Code located in an area that is accessible only

through the use of a tool; provided that the

information is easily accessible (e.g., the information

may appear on the junction box under an access

plate). The identification information for these scales

shall be located on the weighbridge (load-receiving

element) near the point where the signal leaves the

weighing element or beneath the nearest access cover.

(Added 1989)

S.6.3. Scales, Main Elements, and Components of

Scales or Weighing Systems. - Scales, main elements of

scales when not contained in a single enclosure for the

entire scale, load cells for which Certificates of

Conformance (CC) have been issued under the National

Type Evaluation Program, and other equipment

necessary to a weighing system, but having no

metrological effect on the weighing system, shall be

marked as specified in Table S.6.3.a. and explained in

the accompanying notes (Table S.6.3.b.)

(Added 1990)

S.6.4. Railway Track Scales. - A railway track scale

shall be marked with the maximum capacity of each

section of the load-receiving element of the scale. Such

marking shall be accurately and conspicuously presented

on, or adjacent to, the identification or nomenclature

plate that is attached to the indicating element of the

scale. The nominal capacity of a scale with more than

two sections shall not exceed twice its rated section

capacity. The nominal capacity of a two-section scale

shall not exceed its rated section capacity.

[Nonretroactive as of January 1, 2002]

(Amended 1988 and 2001)

N. Notes

N.1. Test Procedures.

N.1.1. Increasing-Load Test. - The increasing-load test

shall be conducted on all scales with the test loads

approximately centered on the load-receiving element of

the scale, except on a scale having a nominal capacity

greater than the total available known test load. When

the total test load is less than the nominal capacity, the

test load is used to greatest advantage by concentrating it,

within prescribed load limits, over the main load

supports of the scale.

N.1.2. Decreasing-Load Test (Automatic Indicating

Scales). - The decreasing-load test shall be conducted

with the test load approximately centered on the loadreceiving element of the scale.

N.1.2.1. Scales Marked I, II, III, or IIII. - Except

for portable wheel load weighers, decreasing-load

tests shall be conducted on scales marked I, II, III or

IIII and with n equal to or greater than 1000 with test

loads equal to the maximum test load at each tolerance

value. For example, on a Class III scale, at test loads

equal to 4000d, 2000d, and 500d; for scales with n

less than 1000, the test load shall be equal to one-half

of the maximum load applied in the increasing-load

test. (See Table 6) (Amended 1998)

N.1.2.2. All Other Scales. - On all other scales,

except for portable wheel load weighers, the decreasing-load test shall be conducted with a test load equal

to one-half of the maximum load applied in the

increasing-load test. (Amended 1998)

N.1.3. Shift Test.

N.1.3.1. Bench or Counter Scales. - A shift test

shall be conducted with a half-capacity test load

centered successively at four points equidistant

between the center and the front, left, back, and right

edges of the load-receiving element.

N.1.3.2. Dairy-Product-Test Scales. - A shift test

shall be conducted with a test load of 18 grams

successively positioned at all points on which a weight

might reasonably be placed in the course of normal

use of the scale.

N.1.3.3. Equal-Arm Scales. - A shift test shall be

conducted with a half-capacity test load positioned on

each pan as prescribed in N.1.3.1. An equal test load

shall be centered on the other pan.

N.1.3.4. Vehicle Scales, Axle-Load Scales, and

Livestock Scales With More Than Two Sections. -

At least one shift test shall be conducted with a

minimum test load of 12.5% of scale capacity and may

be performed anywhere on the load-receiving element

using the prescribed test patterns and maximum test

loads specified below. (Two-section livestock scales

shall be tested consistent with N.1.3.8.) (Amended

1991 and

2000)
5ECNGU



6CDNG

2CTCOGVGTUHQT#EEWTCE[%NCUUGU

Class

Value of the verification

scale division (d or e )1

Number of scale4

divisions (n)

Minimum Maximum

SI Units

I equal to or greater than 1 mg 50 000 --

II 1 to 50 mg, inclusive 100 100 000

equal to or greater than 100 mg 5 000 100 000

III2 0.1 to 2 g, inclusive 100 10 000

equal to or greater than 5 g 500 10 000

III L3 equal to or greater than 2 kg 2 000 10 000

IIII equal to or greater than 5 g 100 1 200

INCH-POUND Units

III 0.0002 lb to 0.005 lb, inclusive 100 10 000

0.005 oz to 0.125 oz, inclusive 100 10 000

equal to or greater than 0.01 lb 500 10 000

equal to or greater than 0.25 oz 500 10 000

III L3 equal to or greater than 5 lb 2 000 10 000

IIII greater than 0.01 lb 100 1 200

greater than 0.25 oz 100 1 200

For Class I and II devices equipped with auxiliary reading means (i.e., a rider, a vernier, or a least significant decimal 1

differentiated by size, shape, or color), the value of the verification scale division “e” is the value of the scale division

immediately preceding the auxiliary means. 

 A scale marked “For prescription weighing only” may have a scale division not less than 0.01 g. (Added 1986) 2

 The value of a scale division for crane and hopper (other than grain hopper) scales shall be not less than 0.2 kg (0.5 lb). 3

The minimum number of scale divisions shall be not less than 1 000.

1PCOWNVKRNGTCPIGQTOWNVKKPVGTXCNUECNGVJGPWODGTQHFKXKUKQPUHQTGCEJTCPIGKPFGRGPFGPVN[UJCNNPQVGZEGGFVJG 

OCZKOWOURGEKHKGFHQTVJGCEEWTCE[ENCUU6JGPWODGTQHUECNGFKXKUKQPUPHQTGCEJYGKIJKPITCPIGKUFGVGTOKPGFD[

FKXKFKPIVJGUECNGECRCEKV[HQTGCEJTCPIGD[VJGXGTKHKECVKQPUECNGFKXKUKQPGHQTGCEJTCPIG1PCUECNGU[UVGOYKVJ

OWNVKRNGNQCFTGEGKXKPIGNGOGPVUCPFOWNVKRNGKPFKECVKQPUGCEJGNGOGPVEQPUKFGTGFUJCNNPQVKPFGRGPFGPVN[GZEGGFVJG

OCZKOWOURGEKHKGFHQTVJGCEEWTCE[ENCUU+HVJGU[UVGOJCUCUWOOKPIKPFKECVQTVJGP HQTVJGUWOOGFKPFKECVKQP OCZ

UJCNNPQVGZEGGFVJGOCZKOWOURGEKHKGFHQTVJGCEEWTCE[ENCUU

=0QPTGVTQCEVKXGCUQH,CPWCT[?#OGPFGFCPF(QQVPQVG#FFGF


(QQVPQVG#OGPFGF
2.20. Scales



Table S.6.3.a.

Marking Requirements

Weighing

Equipment

To Be

Marked With

Weighing,

loadreceiving,

and

indicating

element in

same housing

or covered on

the same CC1

Indicating

element not

permanently

attached to

weighing and

load-receiving

element or

covered by a

separate CC

Weighing and

load-receiving

element not

permanently

attached to

indicating

element or

covered by a

separate CC

Load cell

with CC

(11)

Other

equipment

or device

(10)

Manufacturer’s ID (1) x x x x x

Model Designation and Prefix (1) x x x x x

Serial Number and Prefix (2) x x x x x (16)

Accuracy Class (17) x x (8) x(19) x

Nominal Capacity (3)(18)(20) x x x

Value of Scale Division, "d"(3) x x

Value of "e" (4) x x

Temperature Limits (5) x x x x

Concentrated Load Capacity

(CLC) (12)(20)

x x (9)

Special Application (13) x x x

Maximum Number of Scale

Divisions (n ) max (6)

x (8) x(19) x

Minimum Verification Scale

Division (e ) min

x(19)

"S" or "M" (7) x

Direction of Loading (15) x

Minimum Dead Load x

Maximum Capacity x

Safe Load Limit x

Load Cell Verification

Interval (v ) (21) min

x

Section Capacity (14)(20) x x

For applicable notes, see Table S.6.3.b.

(Added 1990) (Amended 1992, 1999, 2000, and 2001) (Added Footnote1 2001)

Weighing/load receiving elements and indicators which are in the same housing or which are permanently attached will 1

generally appear on the same CC. If not in the same housing, elements shall be hard wired together or sealed with a

physical seal or an electronic link. This requirement does not apply to peripheral equipment that has no input or effect

on device calibrations or configurations. 
2.20. Scales

2-15

Table S.6.3.b. Notes For Table S.6.3.a.

1. Manufacturer’s identification and model

designation and model designation prefix*.

*[Nonretroactive as of January 1, 2003.]

 (See also G-S.1.) [ Prefix lettering may be

initial capitals, all capitals or all lower

case.]

(Amended 2000)

2. Serial number [Nonretroactive as of

January 1, 1968] and prefix

[Nonretroactive as of January 1, 1986]. 

(See also G-S.1.)

3. The nominal capacity and value of the scale

division shall be shown together (e.g.,

50 000 x 5 kg, 100 000 x 10 lb, 15 x 0.005

kg , or 30 x 0.01 lb) adjacent to the weight

display when the nominal capacity and

value of the scale division are not

immediately apparent. Each scale division

value or weight unit shall be marked on

multiple range or multi-interval scales.

[Nonretroactive as of January 1, 1983]

4. Required only if different from "d."

[Nonretroactive as of January 1, 1986]

5. Required only on Class III, III L, and IIII

devices if the temperature range on the

NTEP CC is narrower then and within 

-10 EC to 40 EC (14 EF to 104 EF). 

[Nonretroactive as of January 1, 1986]

6. This value may be stated on load cells in

units of 1000; e.g., n: 10 is 10 000

divisions. [Nonretroactive as of January 1,

1988]

7. Denotes compliance for single or multiple

load cell applications. It is acceptable to

use a load cell with the “S” or Single Cell

designation in multiple load cell

applications as long as all other parameters 

meet applicable requirements. A load cell

with the “M” or Multiple Cell designation

can be used only in multiple load cell

applications.

 [Nonretroactive as of January 1, 1988]

(Amended 1999)

8. An indicating element not permanently

attached to a weighing element shall be

clearly and permanently marked with the

accuracy Class of I, II, III, III L, or IIII, as

appropriate, and the maximum number of

scale divisions, n , for which the max

indicator complies with the applicable

requirement. Indicating elements that

qualify for use in both Class III and III L

applications may be marked III/III L and

shall be marked with the maximum

number of scale divisions for which the

device complies with the applicable

requirements for each accuracy class. 

[Nonretroactive as of January 1, 1988.]

9. For vehicle, axle-load, and livestock scales

only. The CLC shall be added to the loadreceiving element of any such scale not

previously marked at the time of

modification. [Nonretroactive as of

January 1, 1989.]

10. Necessary to the weighing system but

having no metrological effect, e.g.,

auxiliary remote display, keyboard, etc.

11. The markings may be either on the load

cell or in an accompanying document;

except that, if an accompanying document

is provided, the serial number shall

appear both on the load cell and in the

document. [Nonretroactive as of January

1, 1988] The manufacturer's name or

trademark, the model designation, and

identifying symbols for the model and

serial numbers as required by paragraph

G.-S.1.shall also be marked both on the

load cell and in any accompanying

document. [Nonretroactive as of January

1, 1991]
2.20. Scales



Table S.6.3.b. Notes For Table S.6.3.a. (Continued)

12. Required on the indicating element and the

load-receiving element of vehicle, axle load,

and livestock scales. Such marking shall be

identified as "concentrated load capacity"

or by the abbreviation "CLC".*

[*Nonretroactive as of January 1, 1989]

13. A scale designed for a special application

rather than general use shall be

conspicuously marked with suitable words

visible to the operator and customer

restricting its use to that application, e.g.,

postal scale, prepack scale, weight

classifier, etc.* When a scale is installed

with an operational counting feature, the

scale shall be marked on both the operator

and customer side with the statement "The

counting feature is not legal for trade." 

[*Nonretroactive as of January 1, 1986]

(Amended 1994)

14. Required on the indicating element of

railway track scales only. When marked on

vehicle, axle-load, and livestock scales

manufactured before January 1, 1989, it may

be used as the CLC.

15. Required if the direction of loading the load

cell is not obvious. [Nonretroactive as of

January 1, 1988]

16. Serial number [Nonretroactive as of

January 1, 1968] and prefix

[Nonretroactive as of January 1, 1986].

(See also G-S.1.) Modules without

"intelligence" on a modular system (e.g.,

printer, keyboard module, cash drawer, and

secondary display in a point-of-sale system)

are not required to have serial numbers.

17. The accuracy Class of a device shall be

marked on the device with the appropriate

designation as I, II, III, III L, or IIII. 

[Nonretroactive as of January 1, 1986.]

18. The nominal capacity shall be conspicuously

marked as follows:

(a) on any scale equipped with unit weights

or weight ranges;

(b) on any scale with which counterpoise

or equal-arm weights are intended to be

used;

(c) on any automatic-indicating or

recording scale so constructed that the

capacity of the indicating or recording

element, or elements, is not immediately

apparent;

(d) on any scale with a nominal capacity

less than the sum of the reading elements;

and

(e) on the load-receiving element (weighbridge) of vehicle, axle-load, and livestock

scales.*

[*Nonretroactive as of January 1, 1989]

19. Nonretroactive as of January 1, 1988.

(Amended 1992)

20. Combination vehicle/railway track scales

must be marked with both the nominal

capacity and CLC for vehicle weighing

and the nominal capacity and section

capacity for railway weighing. All other

requirements relating to these markings

will apply. [Nonretroactive as of

January 1, 2000]

(Added 1999)

21. The value of the load cell verification

interval (v ) must be stated in mass units. min

In addition to this information, a device

may be marked with supplemental

representations of v . min

[Nonretroactive January 1, 2001]

(Added 1999)
2.20. Scales



(a) Prescribed Test Pattern. The normal prescribed

test pattern shall be an area of 1.2 m (4 ft) in length

and 3.0 m (10 ft) in width or the width of the scale

platform, whichever is less. Multiple test patterns N.1.3.6. Monorail Scales, Static Test. A shift test

may be utilized when loaded in accordance with shall be conducted with a test load equal to the largest

Paragraph (b). load that can be anticipated to be weighed in a given

(Part (a) Amended 1997 and 2001) installation, but never less than one-half scale

(b) Maximum Loading. When loading the scale

for testing, one side of the test pattern shall be

loaded to no more than half of the concentrated

load capacity or test load before loading the N.1.3.6.1. Dynamic Monorail Weighing Systems.

other side. The area covered by the test load Dynamic tests with livestock carcasses should be

may be less than 1.2 m (4 ft) x the width of the conducted to duplicate actual use conditions. No

scale; for test patterns less than 1.2 m (4 ft) in less than 20 test loads using carcasses or portions

length the maximum loading shall meet the of carcasses of the type normally weighed should

formula: [(wheel base of test cart or length of be used in the dynamic test; two additional test

test load divided by 48 in) x 0.9 x CLC]. The loads may be included in the test run for use in the

maximum test load applied to each test pattern event that one or two test loads are rendered

shall not exceed the concentrated load capacity unusable during the dynamic test. Prior to starting

of the scale. When the test pattern exceeds 1.2 the dynamic test, the test carcasses must be

m (4 ft), the maximum test load applied shall not positioned far enough ahead of the scale so that

exceed the concentrated load capacity times the their swaying motion settles to duplicate the normal

largest “r” factor in Table UR.3.2.1. for the sway of a continuously running plant chain. If the

length of the area covered by the test load. For plant conveyer chain does not space or prevent the

weighing elements installed prior to January 1, carcasses from touching one another, dynamic tests

1989, the rated section capacity may be should not be conducted until this condition has

substituted for concentrated load capacity to been corrected.

determine maximum loading. #PGZCORNGQHC

RQUUKDNGVGUVRCVVGTPKUUJQYPDGNQY

					

Section Midway Section Midway Section

 1 between 2 between 3

sections sections

1 and 2 2 and 3


2CTVD#OGPFGF

(c) Multiple Pattern Loading. To test to the nominal

capacity, multiple patterns may be simultaneously

loaded in a manner consistent with the method of

use.

(d) Other Designs. Special design scales and those

that are wider than 3.7 m (12 ft) shall be tested in a

manner consistent with the method of use but

following the principles described above.

(Amended 1988)

N.1.3.5. Railway Track Scales Weighing

Individual Cars in Single Drafts. A shift test shall

be conducted with at least two different test loads, if

available, distributed over, to the right and left of,

each pair of main levers or other weighing elements

supporting each section of the scale.

capacity. The load shall be placed successively on the

right end, the left end, and the center of the live rail. 

(Added 1985)

All carcasses shall be individually weighed

statically on either the same scale being tested

dynamically or another monorail scale with the

same or smaller divisions and in close proximity.

(The scale selected for weighing the carcasses shall

first be tested statically with test weights.)

If the scale being tested is used for weighing

freshly slaughtered animals, (often referred to as a

"hot scale"), care must be taken to get a static

weighment as quickly as possible before or

following the dynamic weighment to avoid loss due

to shrink. If multiple dynamic tests are conducted

using the same carcasses, static weights should be

obtained before and after multiple dynamic tests. If

the carcass changes weight between static tests, the

amount of weight change should be taken into

account, or the carcass should be disregarded for

tolerance purposes.

(Note: For a dynamic monorail test, the reference

scale shall comply with the principles in the 
2.20. Scales



Fundamental Considerations Paragraph 3.2.

Tolerances for Standards.)

(Added 1996) (Amended 1999)

N.1.3.7. Vehicle On-Board Weighing Systems. - tomary."

The shift test for a vehicle on-board weighing system (Added 1986)

shall be conducted in a manner consistent with its

normal use. For systems that weigh as part of the

lifting cycle, the center of gravity of the load may be

shifted in the vertical direction as well as from side to

side. In other cases, the center of gravity may be

moved to the extremes of the load-receiving element

using loads of a magnitude that reflect normal use

(i.e., the load for the shift test may exceed one-half

scale capacity), and may, in some cases, be equal to

the capacity of the scale. The shift test may be

conducted when the weighing system is out of level to

the extent that the weighing system remains

operational.

(Added 1992)

N.1.3.8. All Other Scales Except Crane Scales,

Hanging Scales, Hopper Scales, Wheel-Load

Weighers, and Portable Axle-Load Weighers. - A

shift test shall be conducted with a half-capacity test

load centered, as nearly as possible, successively at

the center of each quarter of the load-receiving

element, or with a quarter-capacity test load centered,

as nearly as possible, successively over each main

load support.

(Amended 1987)

N.1.4. Sensitivity Test. - A sensitivity test shall be

conducted on nonautomatic-indicating (weighbeam)

scales only, with the weighing device in equilibrium at

zero-load and at maximum test load. The test shall be

conducted by increasing or decreasing the test load in an

amount equal to the applicable value specified in T.2. or

T.N.6.

N.1.5. Discrimination Test. - A discrimination test shall

be conducted on all automatic indicating scales with the

weighing device in equilibrium at zero load and at

maximum test load, and under controlled conditions in

which environmental factors are reduced to the extent

that they will not affect the results obtained.

[Nonretroactive as of January 1, 1986.]

(Added 1985)

N.1.5.1. Digital Device. - On a digital device, this

test is conducted from just below the lower edge of

the zone of uncertainty for increasing load tests, or

from just above the upper edge of the zone of

uncertainty for decreasing-load tests.

 

N.1.6. RFI Susceptibility Tests, Field Evaluation. -

An RFI test shall be conducted at a given installation

when the presence of RFI has been verified and characterized if those conditions are considered "usual and cusN.1.7. Ratio Test. - A ratio test shall be conducted on

all scales employing counterpoise weights and on

nonautomatic-indicating equal-arm scales.

N.1.8. Material Tests. - A material test shall be

conducted on all customer-operated bulk weighing

systems for recycled materials using bulk material for

which the device is used. Insert into the device, in a

normal manner, several accurately pre-weighed samples

(free of foreign material) in varying amounts

approximating average drafts.

N.1.9. Zero-Load Balance Change. - A zero-load

balance change test shall be conducted on all scales after

the removal of any test load. The zero-load balance

should not change by more than the minimum tolerance

applicable. (Also see G-UR.4.2.) (Renumbered 1988)

N.2. Verification (Testing) Standards. - Field standard

weights used in verifying weighing devices shall comply

with requirements of NIST Handbook 105-1 (Class F) or

the tolerances expressed in Fundamental Considerations,

paragraph 3.2. (i.e., one-third of the smallest tolerance

applied). (Amended 1986)

N.3. Minimum Test Weights and Test Loads*. - The

minimum test weights and test loads for in-service tests

(except railway track scales) are shown in Table 4. [See

Table 4 for *.]

(Added 1984, Amended 1988)

N.3.1. Minimum Test-Weight Load and

Recommended Strain-Load Test for Railway Track

Scales. (Amended 1990)

N.3.1.1. Approval. - The test-weight load shall be

not less than 35 000 kg (80 000 lb). A strain-load test

conducted up to the used capacity of the weighing

system is recommended. (Added 1990)

N.3.1.2. Interim Approval. - A test-weight load of

not less than 13 500 kg (30 000 lb) and a strain-load

test up to at least 25 percent of scale capacity may be

used to return a scale into service following repairs.

(Added 1990)
2.20. Scales



Table 4.

Minimum Test Weights and Test Loads1

Device capacity

minimums

(in terms of device capacity) (where practicable)

Test weights (greater of) Test loads2

0 to 150 kg

(0 to 300 lb)

100%

151 to 1 500 kg

(301 to 3 000 lb)

25% or 150 kg (300 lb) 75% Test weights to dial face capacity,

1 000d, or test load to used capacity, if

greater than minimums specified

During initial verification, a scale

should be tested to capacity.

1 501 to 20 000 kg

(3 001 to 40 000 lb) 12.5% or 500 kg (1 000 lb) 50%

20 001 kg+

(40 001 lb+)

12.5% or 5 000 kg (10 000 lb) 25%3

 If the amount of test weight in Table 4 combined with the load on the scale would result in an unsafe condition, then 1

the appropriate load will be determined by the official with statutory authority.

The term "test load" means the sum of the combination of field standard test weights and any other applied load used 2

in the conduct of a test using substitution test methods. Not more than three substitutions shall be used during

substitution testing, after which the tolerances for strain load tests shall be applied to each set of test loads.

3

 The scale shall be tested from zero to at least 12.5% of scale capacity using known test weights and then to at least 25%

of scale capacity using either a substitution or strain load test that utilizes known test weights of at least 12.5% of scale

capacity. Whenever practical, a strain load test should be conducted to the used capacity of the scale. When a strain load

test is conducted, the tolerance applies only to the known test load.

(Amended 1988, 1989, 1994)

Note: The length of time the scale may be used

following an interim test is at the discretion of the

official with statutory authority.

(Added 1990)

N.3.1.3. Enforcement Action for Inaccuracy. - To

take enforcement action on a scale that is found to be

inaccurate, a minimum test load of 13 500 kg

(30 000 lb) must be used. (Added 1990)

N.4. Coupled-in-Motion Railroad Weighing Systems (a) These weighing systems shall be tested using a 3

N.4.1. Weighing Systems Used to Weigh Trains of over the scale a minimum of five times in each

Less Than 10 Cars. - These weighing systems shall be mode of operation following final calibration; or

tested using a consecutive-car test train consisting of the

number of cars weighed in the normal operation run over (b) if the official with statutory authority determines it

the weighing system a minimum of five times in each necessary, the As Used Test Procedures outlined in

mode of operation following the final calibration. N.4.3.1. shall be used.

(Added 1990; Amended 1992) (Added 1990; Amended 1992)

N.4.2. Weighing Systems Placed in Service Prior to N.4.3.1. As Used Test Procedures - A weighing

January 1, 1991, and Used to Weigh Trains of 10 or system shall be tested in a manner that represents the

More Cars. - The minimum test train shall be a

consecutive-car test train of no less than 10 cars run over

the scale a minimum of five times in each mode of

operation following final calibration.

(Added 1990; Amended 1992)

N.4.3. Weighing Systems Placed in Service on or

After January 1, 1991, and Used to Weigh Trains of

10 or More Cars. -

consecutive-car test train of no less than 10 cars run
2.20. Scales



A test weight car that is representative of one of the types of cars 3

typically weighed on the scale under test may be used wherever

reference weight cars are specified.

normal method of operation and length(s) of trains

normally weighed. The weighing systems may be

tested using either:

(1) a consecutive-car test train of a length typical of

train(s) normally weighed; or

(2) a distributed-car test train of a length typical of

train(s) normally weighed.

However, a consecutive-car test train of a shorter

length may be used provided that initial verification

test results for the shorter consecutive-car test train

agree with the test results for the distributed-car or

full-length consecutive-car test train as specified in

N.4.3.1.1.

The official with statutory authority shall be

responsible for determining the minimum test train

length to be used on subsequent tests.

(Added 1990; Amended 1992)

N.4.3.1.1. Initial Verification. - Initial

verification tests should be performed on any new

weighing system and whenever either the track

structure or the operating procedure changes. If a

consecutive-car test train of length shorter than

trains normally weighed is to be used for

subsequent verification, the shorter consecutive-car

test train results shall be compared either to a

distributed-car or to a consecutive-car test train of

length(s) typical of train(s) normally weighed.

The difference between the total train weight of the

train(s) representing the normal method of

operation and the weight of the shorter consecutivecar test train shall not exceed 0.15 percent. If the

difference in test results exceeds 0.15 percent, the

length of the shorter consecutive-car test train shall

be increased until agreement within 0.15 percent is

achieved. Any adjustments to the weighing system

based upon the use of a shorter consecutive-car test

train shall be offset to correct the bias that was

observed between the full-length train test and the

shorter consecutive-car test train.

(Added 1990; Amended 1992, 1993)

N.4.3.1.2. Subsequent Verification. - The test

train may consist of either a consecutive-car test

train with a length not less than that used in initial

verification, or a distributed-car test train representing the number of cars used in the normal

operation.

(Added 1990)

N.4.3.1.3. Distributed Car Test Trains. -

(a) The length of the train shall be typical of trains

that are normally weighed.

(b) The reference weight cars shall be split into

three groups, each group consisting of 10 cars or 10

percent of the train length, whichever is less.

(Amended 1991)

(c) The test groups shall be placed near the front,

around the middle, and near the end of the train.

(d) Following the final adjustment, the distributedcar test train shall be run over the scale at least

three times or shall produce 50 weight values,

whichever is greater.

(e) The weighing system shall be tested in each

mode of operation.

(Added 1990; Amended 1992)

N.4.3.1.4. Consecutive-Car Test Trains. -

(a) A consecutive-car test train shall consist of at

least 10 cars.

(b) If the consecutive-car test train consists of

between 10 cars and 20 cars, inclusive, it shall

be run over the scale a minimum of five times

in each mode of operation following the final

calibration.

(c) If the consecutive-car test train consists of

more than 20 cars, it shall be run over the

scale a minimum of three times in each mode

of operation.

(Added 1990; Amended 1992)

N.5. Uncoupled-in-Motion Railroad Weighing System. -

An uncoupled-in-motion scale shall be tested statically

before being tested in motion by passing railroad reference

weight cars over the scale. When an uncoupled-in-motion

railroad weighing system is tested, the car speed and the

direction of travel shall be the same as when the scale is in

normal use. The minimum in-motion test shall be three

reference weight cars passed over the scale three times.
2.20. Scales



The cars shall be selected to cover the range of weights that

are normally weighed on the system and to reflect the types

of cars normally weighed.

(Added 1993)

N.6. Nominal Capacity of Prescription Scales. - The

nominal capacity of a prescription scale shall be assumed to

be 1/2 apothecary ounce, unless otherwise marked.

(Applicable only to scales not marked with an accuracy

class.)

T. Tolerances Applicable to Devices

not Marked I, II, III, III L, or IIII

T.1. Tolerance Values.

T.1.1. General. - The tolerances applicable to devices

not marked with an accuracy class shall have the

tolerances applied as specified in Table T.1.1. 

(Amended 1990)

T.1.2. Postal and Parcel Post Scales. - The tolerances

for postal and parcel post scales are given in Table T.1.1.

and Table 5.

(Amended 1990)

T.2. Sensitivity Requirement (SR)

T.2.1. Application. - The sensitivity requirement (SR)

is applicable to all nonautomatic-indicating scales not

marked I, II, III, III L, or IIII, and is the same whether

acceptance or maintenance tolerances apply.

T.2.2. General. - Except for scales specified in

paragraphs T.2.3. through T.2.8.: 2d, 0.2 percent of the

scale capacity, or 40 lb, whichever is least.

T.2.3. Prescription Scales. 6 mg (0.1 grain).

T.2.4. Jewelers’ Scales.

T.2.4.1. With One-Half Ounce Capacity or Less. 6

mg (0.1 grain).

T.2.4.2. With More Than One-Half Ounce Capacity.

1d or 0.05 percent of the scale capacity, whichever is

less.

T.2.5. Dairy-Product-Test Scales

T.2.5.1. Used in Determining Butterfat Content. 32

mg (0.5 grain). 

T.2.5.2. Used in Determining Moisture Content. - 19

mg (0.3 grain).

T.2.6. Grain Test Scales. The sensitivity shall be as

stated in T.N.6.

(Amended 1987)

T.2.7. Vehicle, Axle-Load, Livestock, and Animal

Scales.

T.2.7.1. Equipped With Balance Indicators. 1d.

T.2.7.2. Not Equipped With Balance Indicators. 2d

or 0.2 percent of the scale capacity, whichever is less.

T.2.8. Railway Track Scales. 3d or 100 lb, whichever

is less.

T.3. Sensitivity Requirement, Equilibrium Change

Required.

The minimum change in equilibrium with test loads equal to

the values specified in T.2. shall be as follows:

(a) Scale With a Trig Loop but Without a Balance

Indicator. - The position of rest of the weighbeam

shall change from the center of the trig loop to the top

or bottom, as the case may be.

(b) Scale With a Single Balance Indicator and Having a

Nominal Capacity of Less Than 250 kg (500 lb). - The

position of rest of the indicator shall change 1.0 mm

(0.04 in) or one division on the graduated scale,

whichever is greater.

(c) Scale With a Single Balance Indicator and Having a

Nominal Capacity of 250 kg (500 lb) or Greater. - The

position of rest of the indicator shall change 6.4

mm (0.25 in) or one division on the graduated scale or

the width of the central target area, whichever is

greater. However, the indicator on a batching scale

shall change 3.2 mm (0.125 in) or one division on the

graduated scale, whichever is greater.

(d) Scale With Two Opposite-Moving Balance Indicators. - The position of rest of the two indicators

moving in opposite directions shall change 1.0 mm

(0.04 in) with respect to each other.

(e) Scale With Neither a Trig Loop nor a Balance

Indicator. - The position of rest of the weighbeam or

lever system shall change from the horizontal, or

midway between limiting stops, to either limit of

motion.

T.4. Radio Frequency Interference (RFI) and Other

Electromagnetic Interference Susceptibility. - The

difference between the weight indication with the
2.20. Scales



disturbance and the weight indication without the values apply regardless of the influence factors in effect

disturbance, shall not exceed one scale division (d) or the at the time of the conduct of the examination. (Also see

equipment shall: G-N.2.)

(a) blank the indication, or

 

(b) provide an error message, or

 

(c) the indicator shall be so completely unstable that it

could not be interpreted, or transmitted into memory

or to a recording element, as a correct measurement

value. (Added 1986)

T.5. Operating Temperature. - An indicating or recording

element shall not display or record any usable values until

the operating temperature necessary for accurate weighing

and a stable zero-balance condition has been attained.

[Nonretroactive and effective January 1, 1981.]

(Added 1986)

T.N. Tolerances Applicable to Devices

Marked I, II, III, III L, & IIII.

T.N.1. Principles.

T.N.1.1. Design. - The tolerance for a weighing device

is a performance requirement independent of the design

principle used.

T.N.1.2. Accuracy Classes. - Weighing devices are

divided into accuracy classes according to the number of

scale divisions (n) and the value of the scale division (d).

T.N.1.3. Scale Division. - The tolerance for a weighing

device is related to the value of the scale division (d) or

the value of the verification scale division (e) and is

generally expressed in terms of d or e.

T.N.2. Tolerance Application.

T.N.2.1. General. - The tolerance values are positive

(+) and negative (-) with the weighing device adjusted to

zero at no load. When tare is in use, the tolerance values

are applied from the tare zero reference; the tolerance

values apply to certified test loads only.

T.N.2.2. Type Evaluation Examinations. - For type

evaluation examinations, the tolerance values apply to

increasing and decreasing load tests within the temperature, power supply, and barometric pressure limits

specified in T.N.8.

T.N.2.3. Subsequent Verification Examinations. - For

subsequent verification examinations, the tolerance

T.N.2.4. Multi-Interval and Multiple Range

(Variable Division-Value) Scales . - For multi-interval

and multiple range scales, the tolerance values are based

on the value of the scale division of the range in use.

T.N.2.5. Ratio Tests. - For ratio tests, the tolerance

values are 0.75 of the applicable tolerances.

T.N.3. Tolerance Values.

T.N.3.1. Maintenance Tolerance Values. - The

maintenance tolerance values are as specified in Table 6.

T.N.3.2. Acceptance Tolerance Values. - The

acceptance tolerance values shall be one-half the

maintenance tolerance values.

T.N.3.3. Wheel-Load Weighers and Portable

Axle-Load Weighers of Class IIII. - The tolerance

values are two times the values specified in T.N.3.1. and

T.N.3.2. (Amended 1986)

T.N.3.4. Crane and Hopper (Other than Grain

Hopper) Scales. - The maintenance and acceptance

tolerances shall be as specified in T.N.3.1. and T.N.3.2.

for Class III L, except that the tolerance for crane and

construction materials hopper scales shall not be less

than 1d or 0.1 percent of the scale capacity, whichever is

less. (Amended 1986)

T.N.3.5. Separate Main Elements: Load

Transmitting Element, Indicating Element, Etc. - If a

main element separate from a weighing device is

submitted for type evaluation, the tolerance for the

element is 0.7 that for the complete weighing device.

This fraction includes the tolerance attributable to the

testing devices used.

T.N.3.6. Coupled-In-Motion Railroad Weighing

Systems. - The maintenance and acceptance tolerance

values for the group of weight values appropriate to the

application must satisfy the following conditions:

(Amended 1990 and 1992)
2.20. Scales



Table T.1.1. Tolerances for Unmarked Scales

Type of Device Subcategory Min. Tol. Accept. Tol. Maint. Tol. Decreasing Lo

Multiplier 

1

Vehicle, axle-load, livestock, railwaytrack (weighing statically), crane, and Class III L, T.N.3.1 (Table 6) and T.N.3.2. 1.0hopper (other than grain hopper)Grain test scales 1.0 n # 10 000 Class III, T.N.3.1. (Table 6) and T.N.3.2. n > 10 000 Class II, T.N.3.1. (Table 6) and T.N.3.2.

Railway track scales weighing in T.N.3.6. except that for T.N.3.6.2. (a), no single error shall exceedmotion four times the maintenance tolerance. 1.0

Monorail Scales, In-Motion T.N.3.8. 1.0Customer-Operated Bulk-WeighingSystems for Recycled 1.0Materials

± 5% of applied material test load.Average error on 10 or more test loads # 2.5%.

Wheel-load weighers and portable Tested individually or 0.5d or 50 lb,axle-load scales in pairs whichever is greater 2 1% of test load 2% of test load 1.53

Prescription scales 0.1 grain (6 mg) 0.1% of test load 0.1% of test load 1.5Jewelers' scales 0.05% of test load 0.05% of test load 1.5

Graduated 0.5dUngraduated smallest weight,

Sensitivity orwhichever is less

Dairy-product-test scale 1.5 Loads < 18 g 0.2 grain 0.2 grain 0.2 grain 18 g load 0.2 grain 0.3 grain 0.5 grain

Postal and parcel post scalesDesigned/used to weigh 1.5loads < 2 lb

Loads < 2 lb 1/32 oz, 0.03 oz, or 15 grain, 1 g, 1/32 oz, 15 grain, 1 g, 1/32 oz, 0.03 oz, or 0.002 lb 0.03 oz, or 0.002 lb 15 grain, 1 g, 0.002 lb

Loads $ 2 lb Table 5 Table 5 Table 5

Other postal and parcel post scales Table 5 Table 5 Table 5 1.5All other scales

n > 5000 scale capacity, 0.05% of test load 0.1% of test load 1.5 

0.5d or 0.05% of whichever is less 

n # 5000 Class III, T.N.3.1., Table 6 and T.N.3.2. 1.0 

The decreasing load test applies only to automatic indicating scales. 1

If marked and tested as a pair, the tolerance shall be applied to the sum of the indications. (Table Added 

2

6JGFGETGCUKPINQCFVGUVFQGUPQVCRRN[VQRQTVCDNGYJGGNNQCFYGKIJGTU 2.20. Scales



Table 5.

Maintenance and Acceptance Tolerances for Unmarked

Postal and Parcel Post Scales

Scale

capacity

Test

loads

Maintenance

tolerance

(±)

Acceptance

tolerance

(±)

(lb) (lb) (oz) (lb) (oz) (lb)

0 to 4, inclusive*0 to 1, inclusive 1/32 0.002 1/32 0.002

over 1 1/8 0.008 1/16 0.004

over 4* 0 to 7, inclusive 3/16 0.012 3/16 0.012

7+ to 24, inclusive 3/8 0.024 3/16 0.012

24+ to 30, inclusive 1/2 0.030 1/4 0.015

over 30 0.1% of Test Load 0.05% of Test Load

*See Table T.1.1. for scales designed and/or used to weigh loads less than 2 lb.

Table 6.

Maintenance Tolerances

(All values in this table are in scale divisions)

Tolerance in scale divisions

1 2 35

Class Test Load

I 0 - 50 000 50 001 - 200 000 200 001 +

II 0 - 5 000 5 001 - 20 000 20 001 +

III 0 - 500 501 - 2 000 2 001 - 4 000 4 001 +

IIII 0 - 50 51 - 200 201 - 400 401 +

III L 0 - 500 501 - 1 000 (Add 1d for each additional 500d or fraction thereof)

T.N.3.6.1. - For any group of weight values, the

difference in the sum of the individual in-motion car

weights of the group as compared to the sum of the

individual static weights shall not exceed

0.2 percent.

(Amended 1990)

T.N.3.6.2. - If a weighing system is used to weigh

trains of five or more cars, and if the individual car

weights are used, any single weight value within the

group must meet the following criteria:

(a) no single error may exceed three times the

static maintenance tolerance;

(b) not more than 5 percent of the errors may

exceed two times the static maintenance

tolerance; and
2.20. Scales



(c) not more than 35 percent of the errors may combination, and equipped with component parts such

exceed the static maintenance tolerance. as unit weights, weighbeam and weights, or multiple

(Amended 1990 and 1992) weighbeams that can be used in combination to

T.N.3.6.3. - For any group of weight values wherein

the sole purpose is to determine the sum of the

group, T.N.3.6.1. alone applies. (Amended 1990)

T.N.3.6.4. - For a weighing system used to weigh

trains of less than five cars, no single car weight

within the group may exceed the static maintenance

tolerance. (Amended 1990 and 1992)

T.N.3.7. Uncoupled-in-Motion Railroad Weighing

Systems. - The maintenance and acceptance tolerance

values for any single weighment within a group of

non-interactive (i.e., uncoupled) loads, the weighment

error shall not exceed the static maintenance tolerance.

(Amended 1992)

T.N.3.8. Dynamic Monorail Weighing System. -

Acceptance tolerance shall be the same as the

maintenance tolerance shown in Table 6. On a

dynamic test of 20 or more individual test loads, 10

percent of the individual test loads may be in error,

each not to exceed two times the tolerance. The error

on the total of the individual test loads shall not exceed

± 0.2 percent. (See also Note in N.1.3.6.1.) For

equipment undergoing type evaluation, a tolerance

equal to one-half the maintenance tolerance values

shown in Table 6 shall apply.

[Nonretroactive January 1, 2002]

 (Added 1986) (Amended 1999 and 2001)

T.N.3.9. Materials Test on Customer-Operated

Bulk Weighing Systems for Recycled

Materials. - The maintenance and acceptance tolerance

shall be ± 5 percent of the applied materials test load

except that the average error on 10 or more test

materials test loads shall not exceed ± 2.5 percent.

(Added 1986)

T.N.4. Agreement of Indications.

T.N.4.1. Multiple Indicating/Recording

Elements. - In the case of a scale or weighing system

equipped with more than one indicating element or

indicating element and recording element combination,

where the indicators or indicator/recorder combination T.N.6.1. Test Load.

are intended to be used independently of one another,

tolerances shall be applied independently to each

indicator or indicator/recorder combination.

(Amended 1986)

T.N.4.2. Single Indicating/Recording Element. - In

the case of a scale or weighing system with a single

indicating element or an indicating/recording element

indicate a weight, the difference in the weight value

indications of any load shall not be greater than the

absolute value of the applicable tolerance for that load,

and shall be within tolerance limits.

(Amended 1986)

T.N.4.3. Single Indicating Element/Multiple

Indications. - In the case of an analog indicating

element equipped with two or more indicating means

within the same element, the difference in the weight

indications for any load other than zero shall not be

greater than one-half the value of the scale division (d)

and be within tolerance limits.

(Amended 1986)

T.N.4.4. Shift or Section Tests. - The range of the

results obtained during the conduct of a shift test or a

section test shall not exceed the absolute value of the

maintenance tolerance applicable and each test result

shall be within applicable tolerances.

(Added 1986)

T.N.4.5. Time Dependence. - At constant test

conditions, the indication 20 seconds after the

application of a load and the indication after 1 hour

shall not differ by more than:

(a) one-half of the absolute value of the applicable

tolerance for the applied load for class III L

devices; and

(b) the absolute value of the applicable tolerance for

the applied load for all other devices.

(Amended 1989)

T.N.5. Repeatability. - The results obtained from several

weighings of the same load under reasonably static test

conditions shall agree within the absolute value of the

maintenance tolerance for that load, and shall be within

applicable tolerances. 

T.N.6. Sensitivity. - This section is applicable to all

nonautomatic-indicating scales marked I, II, III, III L, or

IIII. 

(a) The test load for sensitivity for nonautomaticindicating vehicle, axle-load, livestock, and

animal scales shall be 1d for scales equipped with

balance indicators, and 2d or 0.2 percent of the

scale capacity, whichever is less, for scales not

equipped with balance indicators.
2.20. Scales



(b) For all other nonautomatic-indicating scales, the (c) all devices manufactured after January 1, 1988, shall

test load for sensitivity shall be 1d at zero and 2d comply with the requirements of this section.

at maximum test load. (Amended 1985)

T.N.6.2. Minimum Change of Indications. - The T.N.8.1. Temperature. - Devices shall satisfy the

addition or removal of the test load for sensitivity shall tolerance requirements under the following temperature

cause a minimum permanent change as follows: conditions:

(a) for a scale with trig loop but without a balance

indicator, the position of the weighbeam shall

change from the center to the outer limit of the

trig loop;

(b) for a scale with balance indicator, the position of

the indicator shall change one division on the

graduated scale, the width of the central target

area, or the applicable value as shown below,

whichever is greater:

 

Scale of Class I or II: 1 mm (0.04 in),

Scale of Class III or IIII with a maximum

capacity of 30 kg (70 lb) or less: 2 mm (0.08 in),

Scale of Class III, III L, or IIII with a maximum

capacity of more than 30 kg (70 lb): 5 mm

0.20 in);

(c) for a scale without a trig loop or balance

indicator, the position of rest of the weighbeam or

lever system shall change from the horizontal or

midway between limiting stops to either limit of

motion.

(Amended 1987)

T.N.7. Discrimination.

T.N.7.1. Analog Automatic Indicating (i.e.,

Weighing Device With Dial, Drum, Fan, Etc.). - A

test load equivalent to 1.4d shall cause a change in the

indication of at least 1.0d. (See N.1.5.)

T.N.7.2. Digital Automatic Indicating. - A test load

equivalent to 1.4d shall cause a change in the indicated

or recorded value of at least 2.0d. This requires the

zone of uncertainty to be not greater than three-tenths

of the value of the scale division. (See N.1.5.1.)

T.N.8. Influence Factors. - The following factors are

applicable to tests conducted under controlled conditions

only, provided that:

(a) types of devices approved prior to January 1, 1986,

and manufactured prior to January 1, 1988, need not

meet the requirements of this section, and

(b) new types of devices submitted for approval after

January 1, 1986, shall comply with the requirements

of this section, and

T.N.8.1.1. If not specified in the operating

instructions for Class I or II scales, or if not marked

on the device for Class III, III L, or IIII scales, the

temperature limits shall be:

-10 EC to 40 EC (14 EF to 104 EF)

T.N.8.1.2. If temperature limits are specified for the

device, the range shall be at least that specified in

Table T.N.8.1.2.

Table T.N.8.1.2.

Class Temperature Range

I 5 EC (9 EF)

II 15 EC (27 EF)

III, III L, & IIII 30 EC (54 EF)

T.N.8.1.3. Temperature Effect on Zero-Load

Balance. - The zero-load indication shall not vary

by more than:

(a) three divisions per 5 EC (9 EF) change in

temperature for Class III L devices; or

(b) one division per 5 EC (9 EF) change in

temperature for all other devices.

(Amended 1990)

T.N.8.1.4. Operating Temperature. - Except for

Class I and II devices, an indicating or recording

element shall not display nor record any usable

values until the operating temperature necessary for

accurate weighing and a stable zero balance condition have been attained.

T.N.8.2. Barometric Pressure. - Except for Class I

scales, the zero indication shall not vary by more than

one scale division for a change in barometric pressure

of 1 kPa over the total barometric pressure range of

95 kPa to 105 kPa (28 in to 31 in of Hg).
2.20. Scales



2WTEJCUGTUCPFWUGTUQHUECNGUUWEJCUTCKNYC[ 4

VTCEMJQRRGTCPFXGJKENGUECNGUUJQWNFDGCYCTG

QHRQUUKDNGCFFKVKQPCNTGSWKTGOGPVUHQTVJGFGUKIP

CPFKPUVCNNCVKQPQHUWEJFGXKEGU


(QQVPQVG#FFGF

T.N.8.3. Electric Power Supply.

T.N.8.3.1. Power Supply, Voltage and

Frequency.

(a) Weighing devices that operate using 

alternating current must perform within the (a) For devices marked with a class designation, the

conditions defined in paragraphs T.N.3. typical class or type of device for particular

through T.N.7., inclusive, over the line voltage weighing applications is shown in Table 7a.

range of 100 V to 130 V or 200 V to 250 V

rms as appropriate, and over the frequency (b) For devices not marked with a class designation,

range of 59.5 Hz to 60.5 Hz. Table 7b applies.

(b) Battery operated instruments shall not indicate

nor record values outside the applicable

tolerance limits when battery power output is

excessive or deficient.

T.N.8.3.2. Power Interruption. - A power Division. - The value of the scale division as recorded

interruption shall not cause an indicating or shall be the same as the division value indicated.

recording element to display or record any values [Nonretroactive as of January 1, 1986]

outside the applicable tolerance limits. (Added 1995) (Amended 1999)

604CFKQ(TGSWGPE[+PVGTHGTGPEG4(+CPF UR.1.3.1. Exceptions. - The provisions of

1VJGT'NGEVTQOCIPGVKE+PVGTHGTGPEG5WUEGRVKDKNKV[

6JGFKHHGTGPEGDGVYGGPVJGYGKIJVKPFKECVKQPFWGVQ

VJGFKUVWTDCPEGCPFVJGYGKIJVKPFKECVKQPYKVJQWVVJG

FKUVWTDCPEGUJCNNPQVGZEGGFQPGUECNGFKXKUKQPFQT

VJGGSWKROGPVUJCNN


CDNCPMVJGKPFKECVKQPQT


D RTQXKFGCPGTTQTOGUUCIGQT


EVJGKPFKECVKQPUJCNNDGUQEQORNGVGN[WPUVCDNG

VJCVKVECPPQVDGKPVGTRTGVGFQTVTCPUOKVVGFKPVQ

OGOQT[QTVQCTGEQTFKPIGNGOGPVCUCEQTTGEV

OGCUWTGOGPVXCNWG

6JGVQNGTCPEGKP60KUVQDGCRRNKGF

KPFGRGPFGPVN[QHQVJGTVQNGTCPEGU(QTGZCORNGKH

KPFKECVKQPUCTGCVCNNQYCDNGDCUKEVQNGTCPEGGTTQT

NKOKVUYJGPVJGFKUVWTDCPEGQEEWTUVJGPKVKU

CEEGRVCDNGHQTVJGKPFKECVKQPVQGZEGGFVJGCRRNKECDNG

DCUKEVQNGTCPEGUFWTKPIVJGFKUVWTDCPEG='FKVQTU	

0QVG(QNNQYKPIVJG0%9/#PPWCN/GGVKPI

VJGVGZVKPVJKURCTCITCRJYCUTGXKUGFYKVJ

EQPEWTTGPEGQHVJG56%QOOKVVGGVQENCTKH[KVU

CRRNKECVKQP?


#OGPFGF

UR. User Requirements

UR.1. Selection Requirements. - Equipment shall be

suitable for the service in which it is used with respect to

elements of its design, including but not limited to, its

capacity, number of scale divisions, value of the scale

division or verification scale division, minimum capacity,

and computing capability.4

UR.1.1. General.

UR.1.2. Grain Hopper Scales. - The minimum

number of scale divisions for a Class III Hopper Scale

used for weighing grain shall be 2000.

UR.1.3. Value of the Indicated and Recorded Scale

UR.1.3.Value of the Indicated and Recorded Scale

Division shall not apply to: 

(a) Class I scales, or

(b) Dynamic monorail weighing systems when the

value of d is less than the value of e.

[Nonretroactive as of January 1, 1986.]

(Added 1999)

UR.1.4. Grain-Test Scales: Value of the Scale

Divisions. - The scale division for grain-test scales

shall not exceed 0.2 g for loads through 500 g, and

shall not exceed 1 g for loads above 500 g through

1000 g.

(Added 1992)

UR.1.5. Recording Element, Class III L Railway

Track Scales. - Class III L Railway Track Scales must

be equipped with a recording element.

[Nonretroactive as of January 1, 1996.]

(Added 1995)

UR.2. Installation Requirements.
2.20. Scales

2-29

Table 7b.

Applicable to Devices not Marked With a Class Designation

Scale Type or Design Maximum Value of d

Retail Food Scales, 50-lb capacity and less 1 ounce

Animal Scales 1 pound

Grain Hopper Scales

Capacity up to and incl. 50 000 lb

Capacity over 50 000 lb

10 pounds (not greater than 0.05 % of capacity)

20 pounds

Crane Scales not greater than 0.2 % of capacity

Vehicle and Axle-Load Scales Used in Combination

Capacity up to and including 200 000 lb

Capacity over 200 000 lb

20 pounds

50 pounds

Railway Track Scales

With weighbeam

Automatic indicating

20 pounds

100 pounds

Scales with capacities greater than 500 lb except otherwise

specified

0.1 % capacity (but not greater than 50 lb)

Wheel-Load Weighers 0.25 % capacity (but not greater than 50 lb)

Note: For scales not specified in this table, G-UR.1.1. and UR.1. apply.

(Added 1985) (Amended 1989)

Table 7a.

Typical Class or Type of Device for Weighing Operations

Class Weighing Application or Scale Type

I Precision laboratory weighing

II Laboratory weighing, precious metals and gem weighing, grain test scales

III All commercial weighing not otherwise specified, grain test scales, retail precious metals and

semi-precious gem weighing, animal scales, postal scales, scales used to determine laundry

charges, and vehicle on-board weighing systems

III L Vehicle, axle-load, livestock, railway track scales, crane, hopper (other than grain hopper) scales,

and vehicle on-board weighing systems

IIII Wheel-load weighers and portable axle-load weighers used for highway weight enforcement

Note: A scale with a higher accuracy class than that specified as "typical" may be used.

(Amended 1985, 1986, 1987, 1988, 1992, and 1995)
2.20. Scales



UR.2.1. Supports. - A scale that is portable and

that is being used on a counter, table, or the floor

shall be so positioned that it is firmly and securely

supported.

UR.2.2. Suspension of Hanging Scale. - A hanging

scale shall be freely suspended from a fixed support

when in use.

UR.2.3. Protection From Environmental Factors. -

The indicating elements, the lever system or load cells,

and the load-receiving element of a permanently

installed scale, and the indicating elements of a scale

not intended to be permanently installed, shall be

adequately protected from environmental factors such

as wind, weather, and RFI that may adversely affect the

operation or performance of the device.

UR.2.4. Foundation, Supports, and Clearance. -

The foundation and supports of any scale installed in a

fixed location shall be such as to provide strength,

rigidity, and permanence of all components, and

clearance shall be provided around all live parts to the

extent that no contacts may result when the

load-receiving element is empty, nor throughout the

weighing range of the scale. On vehicle and livestock

scales, the clearance between the load-receiving

elements and the coping at the bottom edge of the

platform shall be greater than at the top edge of the

platform.

[Nonretroactive as of January 1, 1973.]

UR.2.5. Access to Weighing Elements. - Adequate

provision shall be made for ready access to the pit of a

vehicle, livestock, animal, axle-load, or railway track

scale for the purpose of inspection and maintenance.

Any of these scales without a pit shall be installed with

adequate means for inspection and maintenance of the

weighing elements.

(Amended 1985)

UR.2.6. Approaches.

UR.2.6.1. Vehicle Scales. - On the entrance and UR.3.1. Recommended Minimum Load. - A

exit ends of a vehicle scale installed in any one recommended minimum load is specified in Table 8

location for a period of 6 months or more, there since the use of a device to weigh light loads is likely

shall be a straight approach as follows: to result in relatively large errors.

(a) the width at least the width of the platform,

(b) the length at least one-half the length of the

platform but not required to be more than

12 m (40 ft), and

(c) not less than 3 m (10 ft) of any approach

adjacent to the platform shall be constructed

of concrete or similar durable material to

ensure that this portion remains smooth and

level and in the same plane as the platform.

However, grating of sufficient strength to

withstand all loads equal to the concentrated

load capacity of the scale may be installed in

this portion. Any slope in the remaining

portion of the approach shall ensure (1) ease

of vehicle access, (2) ease for testing purposes,

and (3) drainage away from the scale.

[Nonretroactive as of 1976.]

(Amended 1977, 1983, 1993)

UR.2.6.2. Axle-Load Scales. - At each end of an

axle-load scale there shall be a straight paved

approach in the same plane as the platform. The

approaches shall be the same width as the platform

and of sufficient length to insure the level positioning of vehicles during weight determinations.

UR.2.7. Stock Racks. - A livestock or animal scale

shall be equipped with a suitable stock rack, with gates

as required, which shall be securely mounted on the

scale platform. Adequate clearances shall be

maintained around the outside of the rack. 

UR.2.8. Hoists. - On vehicle scales equipped with

means for raising the load-receiving element from the

weighing element for vehicle unloading, means shall be

provided so that it is readily apparent to the scale

operator when the load receiving element is in its

designed weighing position.

UR.2.9. Provision for Testing Dynamic Monorail

Weighing Systems. - Provisions shall be made at the

time of installation of a dynamic monorail weighing

systems for testing in accordance with N.1.3.6.1. (a rail

around or other means for returning the test carcasses

to the scale being tested). 

[Nonretroactive as of January 1, 1998]

(Added 1997) (Amended 1999)

UR.3. Use Requirements.

UR.3.1.1. Minimum Load, Grain Dockage

Determination. - When determining the quantity of

foreign material (dockage) in grain, the weight of

the sample shall be equal to or greater than 500 scale

divisions.

(Added 1985)
2.20. Scales



Prepackaging scales (and other commercial devices) 5

used for putting up packages in advance of sale are

acceptable for use in commerce if all appropriate

provisions of Handbook 44 are met. Users of such

devices must be alert to the legal requirements relating to

the declaration of quantity on a package. Such requirements are to the effect that, on the average, the contents of

the individual packages of a particular commodity

comprising a lot, shipment, or delivery must contain at

least the quantity declared on the label. The fact that a

prepackaging scale may overregister, but within

established tolerances, and is approved for commercial

service is not a legal justification for packages to contain,

on the average, less than the labeled quantity.

UR.3.2. Maximum Load. - A scale shall not be used

to weigh a load of more than the nominal capacity of

the scale.

74/CZKOWO.QCFKPIHQT8GJKENG

5ECNGU#XGJKENGUECNGUJCNNPQVDGWUGFVQYGKIJ

NQCFUGZEGGFKPIVJGOCZKOWONQCFECRCEKV[QHKVU

URCPCUURGEKHKGFKP6CDNG74

(Added 1996)

UR.3.3. Single-Draft Vehicle Weighing. - A vehicle

or a coupled vehicle combination shall be

commercially weighed on a vehicle scale only as a

single draft. That is, the total weight of such a vehicle

or combination shall not be determined by adding

together the results obtained by separately and not

simultaneously weighing each end of such vehicle or

individual elements of such coupled combination.

However:

(a) the weight of a coupled combination may be

determined by uncoupling the various elements

(tractor, semitrailer, trailer), weighing each unit

separately as a single draft, and adding together

the results, or

(b) the weight of a vehicle or coupled-vehicle

combination may be determined by adding UR.3.9. Use of Manual Gross Weight Entries. -

together the weights obtained while all individual Manual gross weight entries are permitted for use in

elements are resting simultaneously on more than the following applications only: (1) when credit is

one scale platform. given for a weighed item on point-of-sale systems

Note: This paragraph does not apply to highway-lawenforcement scales and scales used for the collection of

statistical data.

(Added 1992)

UR.3.4. Wheel-Load Weighing.

UR.3.4.1. Use in Pairs. - When wheel-load

weighers or portable axle-load weighers are to be

regularly used in pairs, both weighers of each such

pair shall be appropriately marked to identify them

as weighers intended to be used in combination.

UR.3.4.2. Level Condition. - A vehicle of which

either an axle-load determination or a gross-load

determination is being made utilizing wheel-load

weighers or portable axle-load weighers, shall be in

a reasonably level position at the time of such

determination.

UR.3.5. Special Designs. - A scale designed and

marked for a special application (such as a 

prepackaging scale) shall not be used for other than its intended purpose .5

UR.3.6. Wet Commodities. - Wet commodities not in

watertight containers shall be weighed only on a scale

having a pan or platform that will drain properly.

(Amended 1988)

UR.3.7. Minimum Load on a Vehicle Scale. - A

vehicle scale shall not be used to weigh net loads

smaller than:

(a) 10d when weighing scrap material for recycling;

(b) 50d for all other weighing.

As used in this paragraph, scrap materials for recycling

shall be limited to ferrous metals, paper (including

cardboard), textiles, plastic, and glass.

(Amended 1988 and 1992)

UR.3.8. Minimum Load for Weighing Livestock.-

A scale with scale divisions greater than 2 kg (5 lb)

shall not be used for weighing net loads smaller than

500d.

(Amended 1989)

interfaced with scales; (2) when a device or system

generates labels for standard weight packages; (3)

when postal scales or weight classifiers generate

manifests for packages to be picked up at a later time;

or (4) when livestock and vehicle scale systems

generate weight tickets to correct erroneous tickets.

(Added 1992) (Amended 2000)

UR.3.10. Dynamic Monorail Weighing Systems. -

When the value of d is different from the value of e,

the commercial transaction must be based on e. UR.4. Maintenance Requirements.

(Added 1999) 
2.20. Scales



Table 8.

Recommended Minimum Load

Class

Value of

scale division

(d or e*)

Recommended

minimum load

(d or e*)

I equal to or greater than 0.001 g 100

II 0.001 g to 0.05 g, inclusive 20

equal to or greater than 0.1 g 50

III All** 20

III L All 50

IIII All 10

*For Class I and II devices equipped with auxiliary reading means (i.e., a rider, a vernier, or

a least significant decimal differentiated by size, shape or color), the value of the verification

scale division "e" is the value of the scale division immediately preceding the auxiliary

means. For Class III and IIII devices the value of "e" is specified by the manufacturer as

marked on the device; "e" must be less than or equal to "d."

**A minimum load of 10d is recommended for a weight classifier marked in accordance

with a statement identifying its use for special applications.

(Amended 1990)

UR.4.1. Balance Condition. - The zero-load

adjustment of a scale shall be maintained so that, with

no load on the load-receiving element and with all

load-counterbalancing elements of the scale (such as

poises, drop weights, or counterbalance weights) set to

zero, the scale shall indicate or record a zero balance

condition. A scale not equipped to indicate or record

a zero-load balance shall be maintained in balance

under any no-load condition.

UR.4.2. Level Condition. - If a scale is equipped with

a level-condition indicator, the scale shall be

maintained in level.

UR.4.3. Scale Modification.- The dimensions (e.g.,

length, width, thickness, etc.) of the load receiving

element of a scale shall not be changed beyond the

manufacturer’s specifications, nor shall the capacity of

a scale be increased beyond its design capacity by

replacing or modifying the original primary indicating

or recording element with one of a higher capacity,

except when the modification has been approved by a

competent engineering authority, preferably that of the

engineering department of the manufacturer of the

scale, and by the weights and measures authority

having jurisdiction over the scale.

(Amended 1996)

UR.5. Coupled-in-Motion Railroad Weighing

Systems. - A coupled-in-motion weighing system placed

in service on or after January 1, 1991, should be tested in

the manner in which it is operated, with the locomotive

either pushing or pulling the cars at the designed speed

and in the proper direction. The cars used in the test train

should represent the range of gross weights that will be

used during the normal operation of the weighing system.

Except as provided in N.4.2. and N.4.3.(a), normal operating procedures should be simulated as nearly as practical.

Approach conditions for a train length in each direction of

the scale site are more critical for a weighing system used

for individual car weights than for a unit-train-weightsonly facility, and should be considered prior to

installation.

(Added 1990) (Amended 1992) 
2.20. Scales

2-32

Table UR.3.2.1.

Span Maximum Load

Distance in feet between the Ratio of CLC to maximum load (“r” factor) carried on any group of 2 or more consecutive axles extremes of any group of 2

or more consecutive axles

2 axles 3 axles 4 axles 5 axles 6 axles 7 axles 8 axles 9 axles

4 1.000 1

5 1.000 1

6 1.000 INSTRUCTIONS: 1

1. Determine the scale’s CLC.

2. Count the number of axles on the vehicle in a given span and determine the distance in feet between the first and last

axle in the span. 

3. Multiply the CLC by the corresponding multiplier in the table*.

4. The resulting number is the scale's maximum concentrated load for a single span based on the vehicle configuration.

*See note and formula on next page.

7 1.000 1

8 and less 1.000 1.000 1

More than 8 1.118 1.235 1

9 1.147 1.257

10 1.176 1.279

11 1.206 1.301

12 1.235 1.324 1.471 1.632

13 1.265 1.346 1.490 1.651

14 1.294 1.368 1.510 1.669

15 1.324 1.390 1.529 1.688 1.853

16 1.353 1.412 1.549 1.706 1.871

17 1.382 1.434 1.569 1.724 1.888

18 1.412 1.456 1.588 1.743 1.906

19 1.441 1.478 1.608 1.761 1.924

20 1.471 1.500 1.627 1.779 1.941

21 1.500 1.522 1.647 1.798 1.959

22 1.529 1.544 1.667 1.816 1.976

23 1.559 1.566 1.686 1.835 1.994

24 1.588 1.588 1.706 1.853 2.012 2.176

25 1.618 1.610 1.725 1.871 2.029 2.194

26 1.632 1.745 1.890 2.047 2.211

27 1.654 1.765 1.908 2.065 2.228

28 1.676 1.784 1.926 2.082 2.245 2.412

29 1.699 1.804 1.945 2.100 2.262 2.429

30 1.721 1.824 1.963 2.118 2.279 2.445

31 1.743 1.843 1.982 2.135 2.297 2.462

32 1.765 1.863 2.000 2.153 2.314 2.479 2.647

33 1.882 2.018 2.171 2.331 2.496 2.664

34 1.902 2.037 2.188 2.348 2.513 2.680

35 1.922 2.055 2.206 2.365 2.529 2.697

36 2.000 2.074 2.224 2.382 2.546 2.713 2

37 2.000 2.092 2.241 2.400 2.563 2.730 2

38 2.000 2.110 2.259 2.417 2.580 2.746 2

39 2.000 2.129 2.276 2.434 2.597 2.763

40 2.020 2.147 2.294 2.451 2.613 2.779

41 2.039 2.165 2.312 2.468 2.630 2.796

42 2.059 2.184 2.329 2.485 2.647 2.813

43 2.078 2.202 2.347 2.502 2.664 2.829

44 2.098 2.221 2.365 2.520 2.681 2.846

45 2.118 2.239 2.382 2.537 2.697 2.862

46 2.137 2.257 2.400 2.554 2.714 2.879

47 2.157 2.276 2.418 2.571 2.731 2.895

48 2.176 2.294 2.435 2.588 2.748 2.912

49 2.196 2.313 2.453 2.605 2.765 2.928

50 2.216 2.331 2.471 2.623 2.782 2.945

51 2.235 2.349 2.488 2.640 2.798 2.961

52 2.255 2.368 2.506 2.657 2.815 2.978
W ' r × 500 LN

N & 1 % 12N % 36

2.20. Scales

Table UR.3.2.1.

Span Maximum Load

Distance in feet between the Ratio of CLC to maximum load (“r” factor) carried on any group of 2 or more consecutive axles extremes of any group of 2

or more consecutive axles

2 axles 3 axles 4 axles 5 axles 6 axles 7 axles 8 axles 9 axles

2-33

53 2.275 2.386 2.524 2.674 2.832 2.994

54 2.294 2.404 2.541 2.691 2.849 3.011

55 2.314 2.423 2.559 2.708 2.866 3.028

56 2.333 2.441 2.576 2.725 2.882 3.044

57 2.353 2.460 2.594 2.742 2.899 3.061 3

58 2.478 2.612 2.760 2.916 3.077

59 2.496 2.629 2.777 2.933 3.094

60 2.515 2.647 2.794 2.950 3.110

*Note: This table was developed based upon the following formula. Values may be rounded in some cases for ease of use.

Tandem Axle Weight. 1

Exception - These values in the third column correspond to the maximum loads in which the inner bridge dimensions of 2

36, 37, and 38 feet are considered to be equivalent to 39 feet. This allows a weight of 68 000 lb on axles 2 through 5.

Corresponds to the Interstate Gross Weight Limit. 3
Index



Scales Code Index

Acceptance ............. 2-9, 2-21, 2-22, 2-24, 2-25

Acceptance tolerance ............... 2-9, 2-22, 2-25

Access to weighing elements ................. 2-28

Accuracy . . . . . . . . . . . . 2-9-11, 2-14, 2-16, 2-21, 2-22

Accuracy class .............. 2-11, 2-14, 2-16, 2-21

Adjustable components ................. 2-7, 2-10

Adjustment .......... 2-7, 2-8, 2-10, 2-11, 2-20, 2-31

Approaches .............................. 2-28

Audit trail . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 2-7

Automatic indicating scales ......... 2-12, 2-18, 2-23

Automatic zero-setting mechanism ........ 2-8, 2-11

Axle-load scales ............... 2-3, 2-9, 2-12, 2-23

Balance condition ..... 2-3, 2-7, 2-9, 2-22, 2-26, 2-31

Balance indicator ......... 2-3, 2-9, 2-10, 2-21, 2-26

Balance indicators ................ 2-21, 2-25, 2-26

Balance position . . . . . . . . . . . . . . . . . . . . . . . . . . . . 2-4

Barometric pressure ................... 2-22, 2-26

Capacity .... 2-5, 2-6, 2-8-12, 2-14, 2-16-19, 2-21-24,

2-26-28, 2-30, 2-31

Capacity indication . . . . . . . . . . . . . . . . . . . . . . . . . 2-6

Center-of-zero . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 2-3

Certificate of Conformance .................. 2-11

Class . . . . . 2-3, 2-4, 2-7-12, 2-14, 2-16, 2-18, 2-21-27,

2-31

CLC . . . . . . . . . . . . . . . . . . . . . . . . . . . 2-11, 2-14, 2-16

Clearance ......................... 2-4, 2-9, 2-28

Computing scale ....................... 2-6, 2-9

Concentrated Load Capacity . . 2-11, 2-14, 2-16, 2-17,

2-28

Consecutive-car test ............... 2-19, 2-19, 2-20

Coupled-in-motion ................ 2-19, 2-22, 2-31

Coupled-in-motion railroad weighing systems . . 2-22,

2-31

Crane and hopper (other than grain hopper) ..... 2-22

Crane scales .............................. 2-18

Customer’s indications . . . . . . . . . . . . . . . . . . . . . . . 2-6

Dairy-product-test ..................... 2-9, 2-23

Damping means ........................... 2-10

Decreasing load test . . . . . . . . . . . . . . . . . . . . . . . . 2-23

Direct sales ...................... 2-5, 2-6, 2-8-10

Discrimination test ......................... 2-18

Distributed-car test . . . . . . . . . . . . . . . . . . . . . . . . . 2-20

Electric power supply ...................... 2-27

Environmental factors ................. 2-18, 2-28

Foundation ............................... 2-28

Graduations . . . . . . . . . . . . . . . . . . . . . . . . . . 2-4-6, 2-9

Grain test scales ...................... 2-21, 2-23

Grain-test . . . . . . . . . . . . . . . . . . . . . . . . . . . . 2-9, 2-27

Hanging scale ............................. 2-28

Hanging scales ............................ 2-18

Hoists . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 2-28

Hopper scales .................... 2-18, 2-22, 2-27

I . . . 2-3, 2-4, 2-7-12, 2-16, 2-18, 2-21, 2-22, 2-24-27,

2-31

II . . . . . . . . . . . . . . . . . 2-3, 2-8-12, 2-16, 2-21-26, 2-31

III . . . . . . 2-4, 2-8, 2-9, 2-11, 2-12, 2-16, 2-21-27, 2-31

III L ................ 2-8, 2-11, 2-16, 2-21-27, 2-31

IIII . . . 2-4, 2-11, 2-12, 2-16, 2-21, 2-22, 2-24-26, 2-31

Index of an indicator . . . . . . . . . . . . . . . . . . . . . . . . 2-4

Indicating element . . 2-11, 2-12, 2-14, 2-16, 2-22, 2-25

Indicator . 2-3-5, 2-7, 2-9, 2-10, 2-21, 2-22, 2-25, 2-26,

2-31

Influence factors ..................... 2-22, 2-26

Initial verification .................... 2-19, 2-20

Initial zero-setting mechanism ............. 2-6, 2-9

Installation requirements .................... 2-27

Jewelers’ scales ...................... 2-21, 2-23

Level . . . . . . . . . 2-7, 2-9, 2-10, 2-18, 2-28, 2-30, 2-31

Level condition ...................... 2-30, 2-31

Level-indicating means . . . . . . . . . . . . . . . . . . . . . . 2-9

Livestock scales ......... 2-8, 2-11, 2-12, 2-16, 2-28

Load cell ....................... 2-11, 2-14, 2-16

Load cell verification interval ........... 2-11, 2-16

Main elements ....................... 2-12, 2-22

Maintenance . . . . . . . . . . . . . . . . . . . . . 2-21, 2-28, 2-31

Maintenance requirements ................... 2-31

Maintenance tolerance . . . . . . . . . . . . . . . . . . . . . . 2-22

Manual gross weight entries . . . . . . . . . . . . . 2-7, 2-30

Marking requirements ................. 2-11, 2-14

Material test ......................... 2-18, 2-23

Minimum load . . . . . . . . . . . . . . . . . . . 2-28, 2-30, 2-31

Minimum test weights . . . . . . . . . . . . . . . . . 2-18, 2-19

Money-value computation . . . . . . . . . . . . . . . . . . . . 2-6

Money-value graduations . . . . . . . . . . . . . . . . . . . . 2-6

Monorail scale ........................ 2-9, 2-17

Multiple Range ...................... 2-11, 2-22

Multi-interval ........................ 2-11, 2-22

Nominal capacity ..... 2-5, 2-10-12, 2-14, 2-16, 2-17,

2-21, 2-30

No-load reference value . . . . . . . . . . . . . . . . . . . . . . 2-3

Operating temperature ................. 2-22, 2-26

Parallax . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 2-4

Parcel post scales ................. 2-21, 2-23, 2-24

Poise . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 2-5

Portable axle-load weighers . . . . . . . . . . . . . . . . . . . 2-3

Postal . . . . . . . . . 2-3, 2-6, 2-16, 2-21, 2-23, 2-24, 2-30

Power interruption ......................... 2-27

Prepackaging scale ..................... 2-7, 2-30

Prescription . . . . . . . . . . . . . . . . . 2-9, 2-10, 2-21, 2-23

Provision for sealing . . . . . . . . . . . . . . . . . . . . . . . . 2-7

Railway track scales . . 2-8, 2-10, 2-12, 2-16-18, 2-21,

2-23, 2-27

Ratio test ................................ 2-18

Readability . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 2-5

Recorded representations ................. 2-3, 2-7

Recording elements ................ 2-3, 2-10, 2-25

Repeatability ............................. 2-25

RFI . . . . . . . . . . . . . . . . . . . . . . . 2-18, 2-22, 2-27, 2-28

Scale modification ......................... 2-31

Sealing ............................... 2-7, 2-8

Security seal . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 2-7

Sensitivity requirement ..................... 2-21

Sensitivity test ............................ 2-18

Shift test . . . . . . . . . . . . . . . . . . . 2-12, 2-17, 2-18, 2-25
Index



Stock racks . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 2-28

Strain-load test . . . . . . . . . . . . . . . . . . . . . . . . . . . . 2-18

Subsequent verification ................ 2-20, 2-22

Supports ............................ 2-12, 2-28

Symmetry . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 2-4

Tare . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 2-7, 2-9, 2-22

Temperature ................ 2-11, 2-14, 2-22, 2-26

Temperature effect on zero-load .............. 2-26

Test loads . . . . . . . . . . . . . 2-12, 2-17-19, 2-21-23, 2-25

Time dependence .......................... 2-25

Tolerances . . . . 2-9, 2-10, 2-18, 2-19, 2-21, 2-27, 2-30

Travel . . . . . . . . . . . . . . . . . . . . . . . 2-4, 2-5, 2-10, 2-20

Type evaluation ............. 2-11, 2-12, 2-22, 2-25

Uncoupled-in-motion . . . . . . . . . . . . . . . . . . 2-20, 2-25

Uncoupled-in-motion railroad weighing system . . 2-20

Unit weights . . . . . . . . . . . . . . . . . . . . . . 2-6, 2-16, 2-25

Use requirements .......................... 2-28

Value of scale division .................. 2-3, 2-14

Vehicle on-board weighing system ..... 2-7, 2-9, 2-18

Vehicle scale .................... 2-11, 2-28, 2-30

Vehicle scales .......... 2-8, 2-12, 2-27, 2-28, 2-30

Verification scale interval ............... 2-3, 2-11

Weighbeam . . . . . . . . . 2-4, 2-5, 2-18, 2-21, 2-25, 2-26

Weighing elements .......... 2-10, 2-11, 2-17, 2-28

Weight ranges . . . . . . . . . . . . . . . . . . . . . . . . 2-6, 2-16

Weight units . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 2-3

Wheel-load weighers ........... 2-3, 2-9, 2-23, 2-30

Zero indication ........................ 2-3, 2-26

Zero-load adjustment . . . . . . . . . . . . . . . . . . . . . . . . 2-7

Zero-load balance .................. 2-7, 2-9, 2-26
5ECNGU



6CDNG

2CTCOGVGTUHQT#EEWTCE[%NCUUGU

Class

Value of the verification

scale division (d or e )1

Number of scale4

divisions (n)

Minimum Maximum

SI Units

I equal to or greater than 1 mg 50 000 --

II 1 to 50 mg, inclusive 100 100 000

equal to or greater than 100 mg 5 000 100 000

III2 0.1 to 2 g, inclusive 100 10 000

equal to or greater than 5 g 500 10 000

III L3 equal to or greater than 2 kg 2 000 10 000

IIII equal to or greater than 5 g 100 1 200

INCH-POUND Units

III 0.0002 lb to 0.005 lb, inclusive 100 10 000

0.005 oz to 0.125 oz, inclusive 100 10 000

equal to or greater than 0.01 lb 500 10 000

equal to or greater than 0.25 oz 500 10 000

III L3 equal to or greater than 5 lb 2 000 10 000

IIII greater than 0.01 lb 100 1 200

greater than 0.25 oz 100 1 200

For Class I and II devices equipped with auxiliary reading means (i.e., a rider, a vernier, or a least significant decimal 1

differentiated by size, shape, or color), the value of the verification scale division “e” is the value of the scale division

immediately preceding the auxiliary means. 

 A scale marked “For prescription weighing only” may have a scale division not less than 0.01 g. (Added 1986) 2

 The value of a scale division for crane and hopper (other than grain hopper) scales shall be not less than 0.2 kg (0.5 lb). 3

The minimum number of scale divisions shall be not less than 1 000.

1PCOWNVKRNGTCPIGQTOWNVKKPVGTXCNUECNGVJGPWODGTQHFKXKUKQPUHQTGCEJTCPIGKPFGRGPFGPVN[UJCNNPQVGZEGGFVJG 

OCZKOWOURGEKHKGFHQTVJGCEEWTCE[ENCUU6JGPWODGTQHUECNGFKXKUKQPUPHQTGCEJYGKIJKPITCPIGKUFGVGTOKPGFD[

FKXKFKPIVJGUECNGECRCEKV[HQTGCEJTCPIGD[VJGXGTKHKECVKQPUECNGFKXKUKQPGHQTGCEJTCPIG1PCUECNGU[UVGOYKVJ

OWNVKRNGNQCFTGEGKXKPIGNGOGPVUCPFOWNVKRNGKPFKECVKQPUGCEJGNGOGPVEQPUKFGTGFUJCNNPQVKPFGRGPFGPVN[GZEGGFVJG

OCZKOWOURGEKHKGFHQTVJGCEEWTCE[ENCUU+HVJGU[UVGOJCUCUWOOKPIKPFKECVQTVJGP HQTVJGUWOOGFKPFKECVKQP OCZ

UJCNNPQVGZEGGFVJGOCZKOWOURGEKHKGFHQTVJGCEEWTCE[ENCUU

=0QPTGVTQCEVKXGCUQH,CPWCT[?#OGPFGFCPF(QQVPQVG#FFGF


(QQVPQVG#OGPFGF
2.20. Scales

2-15

Table S.6.3.b. Notes For Table S.6.3.a.

1. Manufacturer’s identification and model

designation and model designation prefix*.

*[Nonretroactive as of January 1, 2003.]

 (See also G-S.1.) [ Prefix lettering may be

initial capitals, all capitals or all lower

case.]

(Amended 2000)

2. Serial number [Nonretroactive as of

January 1, 1968] and prefix

[Nonretroactive as of January 1, 1986]. 

(See also G-S.1.)

3. The nominal capacity and value of the scale

division shall be shown together (e.g.,

50 000 x 5 kg, 100 000 x 10 lb, 15 x 0.005

kg , or 30 x 0.01 lb) adjacent to the weight

display when the nominal capacity and

value of the scale division are not

immediately apparent. Each scale division

value or weight unit shall be marked on

multiple range or multi-interval scales.

[Nonretroactive as of January 1, 1983]

4. Required only if different from "d."

[Nonretroactive as of January 1, 1986]

5. Required only on Class III, III L, and IIII

devices if the temperature range on the

NTEP CC is narrower then and within 

-10 EC to 40 EC (14 EF to 104 EF). 

[Nonretroactive as of January 1, 1986]

6. This value may be stated on load cells in

units of 1000; e.g., n: 10 is 10 000

divisions. [Nonretroactive as of January 1,

1988]

7. Denotes compliance for single or multiple

load cell applications. It is acceptable to

use a load cell with the “S” or Single Cell

designation in multiple load cell

applications as long as all other parameters 

meet applicable requirements. A load cell

with the “M” or Multiple Cell designation

can be used only in multiple load cell

applications.

 [Nonretroactive as of January 1, 1988]

(Amended 1999)

8. An indicating element not permanently

attached to a weighing element shall be

clearly and permanently marked with the

accuracy Class of I, II, III, III L, or IIII, as

appropriate, and the maximum number of

scale divisions, n , for which the max

indicator complies with the applicable

requirement. Indicating elements that

qualify for use in both Class III and III L

applications may be marked III/III L and

shall be marked with the maximum

number of scale divisions for which the

device complies with the applicable

requirements for each accuracy class. 

[Nonretroactive as of January 1, 1988.]

9. For vehicle, axle-load, and livestock scales

only. The CLC shall be added to the loadreceiving element of any such scale not

previously marked at the time of

modification. [Nonretroactive as of

January 1, 1989.]

10. Necessary to the weighing system but

having no metrological effect, e.g.,

auxiliary remote display, keyboard, etc.

11. The markings may be either on the load

cell or in an accompanying document;

except that, if an accompanying document

is provided, the serial number shall

appear both on the load cell and in the

document. [Nonretroactive as of January

1, 1988] The manufacturer's name or

trademark, the model designation, and

identifying symbols for the model and

serial numbers as required by paragraph

G.-S.1.shall also be marked both on the

load cell and in any accompanying

document. [Nonretroactive as of January

1, 1991]
2.20. Scales

2-16
2.20. Scales

2-29

Table 7b.

Applicable to Devices not Marked With a Class Designation

Scale Type or Design Maximum Value of d

Retail Food Scales, 50-lb capacity and less 1 ounce

Animal Scales 1 pound

Grain Hopper Scales

Capacity up to and incl. 50 000 lb

Capacity over 50 000 lb

10 pounds (not greater than 0.05 % of capacity)

20 pounds

Crane Scales not greater than 0.2 % of capacity

Vehicle and Axle-Load Scales Used in Combination

Capacity up to and including 200 000 lb

Capacity over 200 000 lb

20 pounds

50 pounds

Railway Track Scales

With weighbeam

Automatic indicating

20 pounds

100 pounds

Scales with capacities greater than 500 lb except otherwise

specified

0.1 % capacity (but not greater than 50 lb)

Wheel-Load Weighers 0.25 % capacity (but not greater than 50 lb)

Note: For scales not specified in this table, G-UR.1.1. and UR.1. apply.

(Added 1985) (Amended 1989)

Table 7a.

Typical Class or Type of Device for Weighing Operations

Class Weighing Application or Scale Type

I Precision laboratory weighing

II Laboratory weighing, precious metals and gem weighing, grain test scales

III All commercial weighing not otherwise specified, grain test scales, retail precious metals and

semi-precious gem weighing, animal scales, postal scales, scales used to determine laundry

charges, and vehicle on-board weighing systems

III L Vehicle, axle-load, livestock, railway track scales, crane, hopper (other than grain hopper) scales,

and vehicle on-board weighing systems

IIII Wheel-load weighers and portable axle-load weighers used for highway weight enforcement

Note: A scale with a higher accuracy class than that specified as "typical" may be used.

(Amended 1985, 1986, 1987, 1988, 1992, and 1995)
2.20. Scales

2-32

Table UR.3.2.1.

Span Maximum Load

Distance in feet between the Ratio of CLC to maximum load (“r” factor) carried on any group of 2 or more consecutive axles extremes of any group of 2

or more consecutive axles

2 axles 3 axles 4 axles 5 axles 6 axles 7 axles 8 axles 9 axles

4 1.000 1

5 1.000 1

6 1.000 INSTRUCTIONS: 1

1. Determine the scale’s CLC.

2. Count the number of axles on the vehicle in a given span and determine the distance in feet between the first and last

axle in the span. 

3. Multiply the CLC by the corresponding multiplier in the table*.

4. The resulting number is the scale's maximum concentrated load for a single span based on the vehicle configuration.

*See note and formula on next page.

7 1.000 1

8 and less 1.000 1.000 1

More than 8 1.118 1.235 1

9 1.147 1.257

10 1.176 1.279

11 1.206 1.301

12 1.235 1.324 1.471 1.632

13 1.265 1.346 1.490 1.651

14 1.294 1.368 1.510 1.669

15 1.324 1.390 1.529 1.688 1.853

16 1.353 1.412 1.549 1.706 1.871

17 1.382 1.434 1.569 1.724 1.888

18 1.412 1.456 1.588 1.743 1.906

19 1.441 1.478 1.608 1.761 1.924

20 1.471 1.500 1.627 1.779 1.941

21 1.500 1.522 1.647 1.798 1.959

22 1.529 1.544 1.667 1.816 1.976

23 1.559 1.566 1.686 1.835 1.994

24 1.588 1.588 1.706 1.853 2.012 2.176

25 1.618 1.610 1.725 1.871 2.029 2.194

26 1.632 1.745 1.890 2.047 2.211

27 1.654 1.765 1.908 2.065 2.228

28 1.676 1.784 1.926 2.082 2.245 2.412

29 1.699 1.804 1.945 2.100 2.262 2.429

30 1.721 1.824 1.963 2.118 2.279 2.445

31 1.743 1.843 1.982 2.135 2.297 2.462

32 1.765 1.863 2.000 2.153 2.314 2.479 2.647

33 1.882 2.018 2.171 2.331 2.496 2.664

34 1.902 2.037 2.188 2.348 2.513 2.680

35 1.922 2.055 2.206 2.365 2.529 2.697

36 2.000 2.074 2.224 2.382 2.546 2.713 2

37 2.000 2.092 2.241 2.400 2.563 2.730 2

38 2.000 2.110 2.259 2.417 2.580 2.746 2

39 2.000 2.129 2.276 2.434 2.597 2.763

40 2.020 2.147 2.294 2.451 2.613 2.779

41 2.039 2.165 2.312 2.468 2.630 2.796

42 2.059 2.184 2.329 2.485 2.647 2.813

43 2.078 2.202 2.347 2.502 2.664 2.829

44 2.098 2.221 2.365 2.520 2.681 2.846

45 2.118 2.239 2.382 2.537 2.697 2.862

46 2.137 2.257 2.400 2.554 2.714 2.879

47 2.157 2.276 2.418 2.571 2.731 2.895

48 2.176 2.294 2.435 2.588 2.748 2.912

49 2.196 2.313 2.453 2.605 2.765 2.928

50 2.216 2.331 2.471 2.623 2.782 2.945

51 2.235 2.349 2.488 2.640 2.798 2.961

52 2.255 2.368 2.506 2.657 2.815 2.978
W ' r × 500 LN

N & 1 % 12N % 36

2.20. Scales

Table UR.3.2.1.

Span Maximum Load

Distance in feet between the Ratio of CLC to maximum load (“r” factor) carried on any group of 2 or more consecutive axles extremes of any group of 2

or more consecutive axles

2 axles 3 axles 4 axles 5 axles 6 axles 7 axles 8 axles 9 axles

2-33

53 2.275 2.386 2.524 2.674 2.832 2.994

54 2.294 2.404 2.541 2.691 2.849 3.011

55 2.314 2.423 2.559 2.708 2.866 3.028

56 2.333 2.441 2.576 2.725 2.882 3.044

57 2.353 2.460 2.594 2.742 2.899 3.061 3

58 2.478 2.612 2.760 2.916 3.077

59 2.496 2.629 2.777 2.933 3.094

60 2.515 2.647 2.794 2.950 3.110

*Note: This table was developed based upon the following formula. Values may be rounded in some cases for ease of use.

Tandem Axle Weight. 1

Exception - These values in the third column correspond to the maximum loads in which the inner bridge dimensions of 2

36, 37, and 38 feet are considered to be equivalent to 39 feet. This allows a weight of 68 000 lb on axles 2 through 5.

Corresponds to the Interstate Gross Weight Limit. 3
2.21. Belt-Conveyor Scale Systems



Sec. 2.21. Belt-Conveyor Scale Systems

 A. Application

A.1. This code applies to belt-conveyor scale systems used

for the weighing of bulk materials.

A.2. The code does not apply to:

(a) devices used for discrete weighing while moving on

conveyors;

(b) devices that measure quantity on a time basis;

(c) check-weighers; or

(d) controllers or other auxiliary devices except as they

may affect the weighing performance of the belt-conveyor scale.

A.3. See also General Code requirements.

 S. Specifications

S.1. Design of Indicating and Recording Elements.

S.1.1. General. - A belt-conveyor scale shall be

equipped with a primary indicating element in the form

of a master weight totalizer and shall also be equipped

with a recording element, and a rate of flow indicator

and recorder (which may be analog).* An auxiliary

indicator shall not be considered part of the master

weight totalizer.

*[Nonretroactive as of January 1, 1986.]

(Amended 1986)

 

S.1.2. Units. - A belt-conveyor scale shall indicate and

record weight units in terms of pounds, tons, long tons,

metric tons, or kilograms. The value of a scale division

(d) expressed in a unit of weight shall be equal to:

(a) 1, 2, or 5, or

 

(b) a decimal multiple or submultiple of 1, 2, or 5.

S.1.3. Value of the Scale Division.

S.1.3.1. For Scales Installed After January 1, 1986.

- The value of the scale division shall not be greater

than 0.1 percent (1/1000) of the minimum totalized

load. 

[Nonretroactive as of January 1, 1986.]

S.1.3.2. For Scales Installed Before January 1,

1986. - The value of the scale division shall not be

greater than 1/1200 of the rated capacity of the

device. However, provision shall be made so that

compliance with the requirements of the zero-load test

as prescribed in N.3.1. may be readily and accurately

determined in 20 minutes of operation.

S.1.4. Recording Elements and Recorded

Representations. - The value of the scale division of

the recording element shall be the same as that of the

indicating element. The belt-conveyor scale system

shall record the initial indication and the final

indication of the master weight totalizer*, the quantity

delivered*, the unit of measurement (i.e., kilograms,

tonnes, pounds, tons, etc.), the date, and time. This

information shall be recorded for each delivery*.

[Nonretroactive as of January 1, 1986.]

[*Nonretroactive as of January 1, 1994]

(Amended 1993)

S.1.5. Rate of Flow Indicators and Recorders. - A

belt-conveyor scale shall be equipped with a rate of flow

indicator and an analog or digital recorder. Permanent

means shall be provided to produce an audio or visual

signal when the rate of flow is equal to or less than 35

percent and when the rate of flow is equal to or greater

than 98 percent of the rated capacity of the scale. The

type of alarm (audio or visual) shall be determined by

the individual installation.

[Nonretroactive as of January 1, 1986.]

(Amended 1989)
2.21. Belt-Conveyor Scale Systems



S.1.6. Advancement of Primary Indicating or

Recording Elements. - The master weight totalizer shall

advance only when the belt conveyor is in operation and

under load.

(Amended 1989)

S.1.7. Master Weight Totalizer. - The master weight

totalizer shall not be resettable without breaking a

security means.

[Nonretroactive as of January 1, 1986.]

 

S.1.8. Power Loss. - In the event of a power failure of

up to 24 hours, the accumulated measured quantity on

the master weight totalizer of an electronic digital

indicator shall be retained in memory during the power

loss.

[Nonretroactive as of January 1, 1986.]

(Amended 1989)

S.2. Design of Weighing Elements. - A belt-conveyor

scale system shall be designed to combine automatically

belt travel with belt load to provide a determination of the

weight of the material that has passed over the scale.

S.2.1. Speed Measurement. - A belt-conveyor scale

shall be equipped with a belt speed or travel sensor that

will accurately sense the belt speed or travel whether the

belt is empty or loaded.

 

S.2.2. Adjustable Components. - An adjustable

component that can affect the performance of the device

(except as prescribed in S.3.1) shall be held securely in

adjustment.

(Amended 1998)

S.2.3. Overload Protection. - The load-receiving

elements shall be equipped with means for overload

protection of not less than 150 percent of rated capacity.

The accuracy of the scale in its normal loading range,

shall not be affected by overloading.

S.3. Zero Setting.

S.3.1. Design of Zero-Setting Mechanism. - The range

of the zero-setting mechanism shall not be greater than

±2 percent of the rated capacity of the scale without

breaking the security means. Automatic and

semiautomatic zero-setting mechanisms shall be so

constructed that the resetting operation is carried out 

only after a whole number of belt revolutions and the

completion of the setting or the whole operation is

indicated. An audio or visual signal shall be given when

the automatic and semiautomatic zero-setting

mechanisms reach the limit of adjustment of the zerosetting mechanism.*

[*Nonretroactive as of January 1, 1990]

(Amended 1989)

S.3.2. Sensitivity at Zero Load (For Type

Evaluation). - When a system is operated for a time

period equal to the time required to deliver the minimum

test load and with a test load calculated to indicate two

scale divisions applied directly to the weighing element,

the totalizer shall advance not less than one or more

than three scale divisions. An alternative test of

equivalent sensitivity, as specified by the manufacturer,

shall also be acceptable.

[Nonretroactive as of January 1, 1986.]

S.4. Marking Requirements. - A belt-conveyor scale shall

be marked with the following: (Also see G-S.1.)

(a) the rated capacity in units of weight per hour

(minimum and maximum);

(b) the value of the scale division; 

(c) the belt speed in terms of feet (or meters) per minute

at which the belt will deliver the rated capacity;

(d) the load in terms of pounds per foot or kilograms per

meter (determined by materials tests);

(e) the operational temperature range if other than 

-10 EC to 40 EC (14 EF to 104 EF). 

[Nonretroactive as of January 1, 1986.]

S.5. Provision for Sealing. - A device shall be designed

using the format set forth in Table S.5. with provision(s) for

applying a security seal that must be broken, or for using

other approved means of providing security (e.g. data

change audit trail available at the time of inspection),

before any change that affects the metrological integrity of

the device can be made to any electronic mechanism.

[Nonretroactive as of January 1, 1999]

(Added 1998)
2.21. Belt-Conveyor Scale Systems



6CDNG5%CVGIQTKGUQH&GXKEGCPF/GVJQFUQH5GCNKPI

%CVGIQTKGUQH&GXKEGU /GVJQFQH5GCNKPI

%CVGIQT[0QTGOQVGEQPHKIWTCVKQPECRCDKNKV[ 5GCND[RJ[UKECNUGCNQTVYQGXGPVEQWPVGTUQPGHQT

ECNKDTCVKQPRCTCOGVGTUCPFQPGHQTEQPHKIWTCVKQP

RCTCOGVGTU

%CVGIQT[4GOQVGEQPHKIWTCVKQPECRCDKNKV[ #PGXGPVNQIIGTKUTGSWKTGFKPVJGFGXKEGKVOWUV

KPENWFGCPGXGPVEQWPVGTVQVJGRCTCOGVGT

+&VJGFCVGCPFVKOGQHVJGEJCPIGCPFVJGPGY

XCNWGQHVJGRCTCOGVGT#RTKPVGFEQR[QHVJG

KPHQTOCVKQPOWUVDGCXCKNCDNGVJTQWIJVJGFGXKEGQT

VJTQWIJCPQVJGTQPUKVGFGXKEG6JGGXGPVNQIIGT

UJCNNJCXGCECRCEKV[VQTGVCKPTGEQTFUGSWCNVQVGP

VKOGUVJGPWODGTQHUGCNCDNGRCTCOGVGTUKPVJG

FGXKEGDWVPQVOQTGVJCPTGEQTFUCTGTGSWKTGF


0QVG&QGUPQVTGSWKTGEJCPIGUVQDGUVQTGF

HQTGCEJRCTCOGVGT

=0QPTGVTQCEVKXGCUQH,CPWCT[?6CDNG#FFGF

N. Notes

N.1. General. - Belt-conveyor scales are capable of

weighing bulk material accurately. (See Tolerances.)

However, their performance can be detrimentally affected

by the conditions of the installation. (See User Requirements.)

N.1.1. Official Test. - An official test of a belt-conveyor

scale system shall be a materials test.

N.1.2. Simulated Test. - Simulated loading conditions

as recommended by the manufacturer and approved by

the official with statutory authority may be used to

properly monitor the system operational performance

between official tests, but shall not be used for official

certification.

(Amended 1991)

N.2. Conditions of Tests. - A belt-conveyor scale shall be

tested after it is installed on the conveyor system with

which it is to be used and under such environmental

conditions as may normally be expected. It shall be tested

at normal use capacity and may also be tested at any other

rate of flow that may be used at the installation. Each test

shall be conducted for:

(a) not less than 1000 scale divisions,

(b) at least three revolutions of the belt, and

(c) at least 10 minutes of operation, or for a normal

weighment.

(Amended 1986)

N.3. Test Procedures.

N.3.1. Zero Load Tests. - If a belt-conveyor scale

system has been idle for a period of 2 hours or more, the

system shall be run for not less than 30 minutes when the

temperature is above 5 EC (41 EF). When the

temperature is below 5 EC (41 EF), additional warmup

time, depending upon conditions, is required before

beginning the zero-load tests. The variation between the

beginning and ending indication of the master weight

totalizer shall not be more than ± 1 scale division when

the instrument is operated at no load for a period of time

equivalent to that required to deliver the minimum

totalized load of 1000 scale divisions.

 

The zero-load test shall be conducted over a whole

number of belt revolutions, but not less than three

revolutions or 10 minutes' operation, whichever is

greater.

 
2.21. Belt-Conveyor Scale Systems



During any portion of the zero-load test, the totalizer

shall not change more than three scale divisions from its

initial indication.

(Amended 1989)

 (f) The test shall not be conducted if the weight of the

N.3.2. Material Tests. - Use bulk material, preferably

that material for which the device is normally used.

Either pass a quantity of pre-weighed material over the

belt-conveyor scale in a manner as similar as feasible to

actual loading conditions, or weigh all material that has

passed over the belt-conveyor scale. Means for weighing

the material test load will depend on the capacity of the

belt-conveyor scale and availability of a suitable scale for

the test. To assure that the test load is accurately

weighed and determined, the following precautions shall

be observed:

 (a) The containers, whether railroad cars, trucks, or N.3.2.1. Accuracy of Material. - The quantity of

boxes, must not leak, and shall not be overloaded to material used to conduct a material test shall be

the point that material will be lost. weighed on a reference scale to an accuracy within 0.1

 (b) The actual empty or tare weight of the containers Class III and III L scales or a scale without a class

shall be determined at the time of the test. designation as described in Handbook 44, Section

Stenciled tare weight of railway cars or trucks shall 2.20., Table T.1.1.

not be used. Gross and tare weights shall be (Added 1989)(Amended 1991, 1993, 1998 and 2000)

determined on the same scale.

(c) When a pre-weighed test load is passed over the

scale, the belt loading hopper shall be examined (a) As required by the official with statutory authority,

before and after the test to assure that the hopper is simulated load tests as recommended by the

empty and that only the material of the test load has manufacturer are to be conducted between material

passed over the scale. tests to monitor the system’s operational

(d) Where practicable, a reference scale should be certification.

tested within 24 hours preceding the determination (Amended 1991)

of the weight of the test load used for a

belt-conveyor scale material test. (b) A simulated load test consisting of at least three

A reference scale which is not "as found" within possible, but not more than 12 hours after the

maintenance tolerance should have its accuracy completion of the material test, to establish the

re-verified after the belt-conveyor test with a factor to relate the results of the simulated load test

suitable known weight load if the "as found" error to the results of the material tests.

of the belt-conveyor scale material test exceeds (Added 1990)

maintenance tolerance values. 

(e) If any suitable known weight load other than a within 0.1 percent.

certified test weight load is used for re-verification (Added 1990)

of the reference scale accuracy, its weight shall be (Amended 1989 and 1990)

determined on the reference scale after the

reference scale certification and before

commencing the belt scale material test.

Note: Even if the reference scale is within

maintenance tolerance it may require adjusting

to be able to meet paragraph N.3.2.1.

test load has been affected by environmental

conditions.

(g) On initial verification, at least three individual tests

shall be conducted. On subsequent verifications, at

least two individual tests shall be conducted. The

performance of the equipment is not to be

determined by averaging the results of the

individual tests. The results of all of these tests

shall be within the tolerance limits. 

(Amended 1986, 1989, 1998 and 2000)

percent. Scales typically used for this purpose include

N.3.3. Simulated Load Tests. - 

performance, but shall not be used for official

consecutive test runs shall be conducted as soon as

(c) The results of the simulated load test shall repeat
2.21. Belt-Conveyor Scale Systems



6JGXCTKCDNGUCPFWPEGTVCKPVKGUKPENWFGFKPVJGTGNCVKXG 

VQNGTCPEGTGRTGUGPVQPN[RCTVQHVJGXCTKCDNGUVJCVCHHGEV

VJGCEEWTCE[QHVJGOCVGTKCNYGKIJGFQPDGNVEQPXG[QT

UECNGU+HVJKUVQNGTCPEGYCUDCUGFQPCPGTTQTCPCN[UKU

DGIKPPKPIYKVJOCUUUVCPFCTFUVJTQWIJCNNQHVJGVGUV

RTQEGUUGUCPFHQNNQYKPIVJGRTKPEKRNGGZRTGUUGFKP5GEVKQP

QHVJG(WPFCOGPVCN%QPUKFGTCVKQPUKP#RRGPFKZ#

VJGVQNGTCPEGYQWNFDGRGTEGPV


#FFGF

T. Tolerances

T.1. Tolerance Values . - Maintenance and acceptance 1

tolerances on materials tests, relative to the weight of the

material, shall be ± 0.25 percent of the test load.

(Amended 1993)

T.2. Tolerance Values, Repeatability Tests. - The

variation in the values obtained during the conduct of UR.1. Use Requirements. - A belt-conveyor scale system

materials tests shall not be greater than 0.25 percent shall be operated between 35 percent and 98 percent of its

(1/400). rated capacity.

T.3. Influence Factors. - The following factors are UR.1.1. Minimum Totalized Load. - Delivered

applicable to tests conducted under controlled conditions quantities of less than the minimum test load shall not be

only, provided that: considered a valid weighment.

(a) types of devices approved prior to January 1, 1986,

and manufactured prior to January 1, 1988, need not

meet the requirements of this Section; and 

(b) new types of devices submitted for approval after

January 1, 1986, shall comply with the requirements

of the Section; and 

 

(c) all devices manufactured after January 1, 1988, shall

comply with the requirements of this Section.

T.3.1. Temperature. - Devices shall satisfy the

tolerance requirements at temperatures from -10 EC to

40 EC (14 EF to 104 EF).

T.3.1.1. Effect on Zero-Load Balance. - The zeroload indication shall not change by more than 0.07

percent of the rated capacity of the scale (without the

belt) for a change in temperature of 10 EC (18 EF) at

a rate not to exceed 5 EC (9 EF) per hour.

T.3.1.2. Temperature Limits. - If a temperature range

other than -10 EC to 40 EC (14 EF to 104 EF) is

specified for the device, the range shall be at least

30 EC (54 EF).

[Nonretroactive as of January 1, 1990] (Added 1989)

T.3.2. Power Supply, Voltage and Frequency. - A

belt-conveyor scale system shall satisfy the tolerance

requirements over a range of 100 V to 130 V or 200 V to

250 V as appropriate and over a frequency range of

59.5 Hz to 60.5 Hz.

UR. User Requirements

UR.1.2. Security Means. - When a security means has

been broken, it shall be reported to the official with

statutory authority.

(Amended 1991)

UR.2. Installation Requirements.

UR.2.1. Protection from Environmental Factors. -

The indicating elements, the lever system or load cells,

and the load-receiving element of a belt-conveyor scale

shall be adequately protected from environmental factors

such as wind, moisture, dust, weather, and radio

frequency interference (RFI) and electromagnetic

interference (EMI) that may adversely affect the

operation or performance of the device.

UR.2.2. Conveyor Installation. - The design and

installation of the conveyor leading to and from the beltconveyor scale is critical with respect to scale performance. Installation shall be in accordance with the scale

manufacturer's instructions and the following:

(a) Installation - General. - A belt-conveyor scale

shall be so installed that neither its performance

nor operation will be adversely affected by any

characteristic of the foundation, supports, or any

other equipment.

(b) Live Portions of Scale. - All live portions of the

scale shall be protected by appropriate guard

devices to prevent accidental interference with

the weighing operation.

(c) Storage of Simulated Load Equipment. -

Suitable protection shall be provided for storage

of any simulated load equipment.
2.21. Belt-Conveyor Scale Systems



+PUVCNNKPIVJGDGNVUECNGKFNGTURCEGUHTQOVJGVCKNRWNNG[QTVJG 

KPHGGFUMKTVKPIYKNNDGKPVJGCTGCQHNGCUVDGNVVGPUKQPQPVJG

EQPXG[QTCPFUJQWNFRTQFWEGVJGDGUVCEEWTCE[6JGRGTHQTOCPEG

QHCDGNVEQPXG[QTUECNGOC[DGCFXGTUGN[CHHGEVGFD[CEQPECXG

EWTXGKPVJGEQPXG[QTVJCVKUNQECVGFDGVYGGPVJGNQCFKPIRQKPV

CPFVJGUECNG6JGTGHQTGYJGPGXGTRQUUKDNGCDGNVEQPXG[QT

UECNGUJQWNFPQVDGKPUVCNNGFYKVJCEQPECXGEWTXGKPVJGEQPXG[QT

DGVYGGPVJGNQCFKPIRQKPVCPFVJGUECNG


#OGPFGFCPF

(d) Take-up Device. - If the belt length is such that (i) Conveyor Orientation. - The conveyor may be

a take-up device is required, this device shall be horizontal or inclined, but, if inclined, the angle

of the counter-weighted type for either vertical shall be such that slippage of material along the

or horizontal travel. belt does not occur. 

(e) Scale Location and Training Idlers. - The (j) Conveyor Stringers. - Conveyor stringers at the

scale shall be so installed that the first weigh scale and for not less than 6 m (20 ft) before and

idler of the scale is at least 6 m (20 ft) or 5 idler beyond the scale shall be continuous or securely

spaces, whichever is greater, from loading point, joined and of sufficient size and so supported as

skirting, head or tail pulley, or convex curve in to eliminate relative deflection between the scale

the conveyor. Any training idler shall be located and adjacent idlers when under load. The

at least 18 m (60 ft) from the center line of the conveyor stringers should be so designed that

weigh span of the scale. Training idlers shall not the deflection between any two adjacent idlers

be restrained at any time in order to force belt within the weigh area does not exceed 0.6 mm

alignment.(Amended 1998) (0.025 in) under load.

(f) Concave Curve. - If there is a concave curve in (k) Identification of Scale Area. - The scale area

the conveyor, before or after the scale, the scale and 5 idlers on both ends of the scale shall be of

shall be installed so that the belt is in contact a contrasting color, or other suitable means shall

with all the idler rollers at all times for at least 6 be used to distinguish the scale from the

m (20 ft) or 5 idler spaces, whichever is greater, remainder of the conveyor installation, and the

before and after the scale. A concave curve scale shall be readily accessible. 2

shall start no closer than 12 m (40 ft) from the (Amended 1998)

scale to the tangent point of the concave curve.

(Amended 1998)

(g) Tripper and Movable Pulleys. - There shall be required for normal use. In a loaded or unloaded

no tripper or movable head pulleys in the condition, the belt shall make constant contact

conveyor. with horizontal and wing rollers of the idlers in

(h) Conveyor Length. - The conveyor shall be no

longer than 300 m (1000 ft) nor shorter than

12 m (40 ft) from head to tail pulley.

[Nonretroactive as of January 1, 1986.]

(l) Belt Composition and Maintenance. -

Conveyor belting shall be no heavier than is

the scale area. Splices shall not cause any undue

disturbance in scale operation (see N.3.).

(Amended 1998, 2000, and 2001)

(m) Uniformity of Belt Loading and Flow. - The

conveyor loading mechanism shall be designed

to provide uniform belt loading. The distance

from the loading point to the scale shall allow

for adequate settling time of the material on the

belt before it is weighed. Feeding mechanisms

shall have a positive closing or stopping action

so that material leakage does not occur. Feeders

shall provide an even flow over the scale

through the full range of scale operation.

Sufficient impact idlers shall be provided in the

conveyor under each loading point to prevent

deflection of the belt during the time material is

being loaded.
2.21. Belt-Conveyor Scale Systems



(n) Belt Alignment. - The belt shall not extend

beyond the edge of the idler roller in any area of

the conveyor.

(Amended 1998)

UR.2.3. Material Test. - A belt-conveyor scale shall be

installed so that a material test can be conveniently

conducted.

[Nonretroactive as of January 1, 1981.]

 

UR.2.4. Belt Travel (Speed or Velocity). - The belt

travel sensor shall be so positioned that it accurately

represents the travel of the belt over the scale for all flow

rates between the maximum and minimum values. The

belt travel sensor shall be so designed and installed that

there is no slip.

UR.3. Use Requirements.

UR.3.1. Loading. - The feed of material to the scale

shall be controlled to assure that, during normal

operation, the material flow is in accordance with

manufacturer’s recommendation for rated capacity.

 

UR.3.2. Maintenance. - Belt-conveyor scales and

idlers shall be maintained and serviced in accordance

with manufacturer’s instructions and the following:

(a) The scale and area surrounding the scale shall be

kept clean of debris or other foreign material that

can detrimentally affect the performance of the

system.

 (b) Simulated load tests or material tests shall be

conducted at periodic intervals between official

tests in order to provide reasonable assurance that

the device is performing correctly. The action to be

taken as a result of the tests is as follows:

(Amended 2000)

- if the error is less than 0.25 percent, no adjustment

is to be made;

- if the error is at least 0.25 percent but not more than

0.6 percent, adjustment may be made if the official

with statutory authority is notified;

(Amended 1991)

 - if the error is greater than 0.6 percent but does not

exceed 0.75 percent, adjustments shall be made

only by a competent service person and the official

with statutory authority shall be notified. After

such an adjustment, if the results of a subsequent

test require adjustment in the same direction, an

official test shall be conducted;

(Amended 1991)

- if the error is greater than 0.75 percent, an official

test is required.

(Amended 1987)

(c) Scale Alignment. - Alignment checks shall be

conducted in accordance with the manufacturer’s

recommendation when conveyor work is performed

in the scale area. A material test is required after

any realignment. 

(Amended 1986 and 2000)

(d) Simulated Load Equipment. - Simulated load

equipment shall be clean and properly maintained.

(e) Records. - Records of calibration and maintenance,

including conveyor alignment, shall be maintained

on site for at least three current years to develop a

history of scale performance. Copies of any report

as a result of a test or repair shall be mailed to the

official with statutory authority as required. The

current date and correction factor(s) for simulated

load equipment shall be recorded and maintained in

the scale cabinet.

(Amended 1991)

UR.4. Compliance. - Prior to initial verification, the scale

manufacturer or installer shall certify to the owner that the

scale meets code requirements. Prior to initial verification

and each subsequent verification, the scale owner or his

agent shall notify the official with statutory authority in

writing that the belt-conveyor scale system is in compliance

with this specification and ready for material testing.

(Amended 1991)
2.22. Automatic Bulk Weighing Systems


6KVNGCOGPFGF 



Sec. 2.22. Automatic Bulk Weighing Systems1

 A. Application

A.1. General. - This code applies to automatic bulk

weighing systems, that is, weighing systems adapted to the

automatic weighing of a commodity in successive drafts of

predetermined amounts automatically recording the no-load

and loaded weight values and accumulating the net weight

of each draft.

(Amended 1987)

A.2. Also see General Code Requirements.

 S. Specifications

S.1. Design of Indicating and Recording Elements and

Recorded Representations.

S.1.1. Zero Indication. - Provisions shall be made to

indicate and record a no-load reference value and, if the

no-load reference value is a zero value indication, to

indicate and record an out-of-balance condition on both

sides of zero.

S.1.1.1. Digital Zero Indication. - A digital zero

indication shall represent a balance condition that is

within ±1/2 the value of the scale division.

S.1.2. Value of Scale Division (d). - The value of the

scale division (d), expressed in a unit of weight, shall be

equal to: 

(a) 1, 2, or 5; or

(b) a decimal multiple or submultiple of 1, 2, or 5; or

(c) a binary submultiple of a unit of weight.

Examples: Scale divisions may be 0.01, 0.02, or 0.05;

0.1, 0.2, or 0.5; 1, 2, or 5; 10, 20, or 50; or 1/2, 1/4, 1/8,

1/16, etc.

[Nonretroactive as of January 1, 1986.]

(Amended 1987)

S.1.3. Capacity Indication and Recorded

Representation. - An indicating or recording element

shall not indicate or record any values when the gross

load is in excess of 105 percent of the capacity of the

system.

S.1.4. Weighing Sequence. - For systems used to

receive (weigh in), the no-load reference value shall be

determined and recorded only at the beginning of each

weighing cycle. For systems used to deliver (weigh out),

the no-load reference value shall be determined and

recorded only after the gross load reference value for

each weighing cycle has been indicated and recorded.

S.1.5. Recording Sequence. - Provision shall be made

so that all weight values are indicated until the

completion of the recording of the indicated value.

S.1.6. Provision for Sealing Adjustable Components

on Electronic Devices. - Provision shall be made for

applying a security seal in a manner that requires the

security seal to be broken before an adjustment can be

made to any component affecting the performance of the

device.

S.2. Design of Balance and Damping Mechanism.

S.2.1. Zero-Load Adjustment. - The weighing system

shall be equipped with manual or semiautomatic means

by which the zero-load balance or no-load reference

value indication may be adjusted. An automatic zero

setting mechanism is prohibited.

S.2.1.1. Manual. - A manual zero-load or no-load

reference value setting mechanism shall be operable

or accessible only by a tool outside of or entirely

separate from this mechanism or enclosed in a cabinet.

S.2.1.2. Semiautomatic. - A semiautomatic zero-load

or no-load reference value setting mechanism shall

meet the provisions of S.2.1.1. or shall be operable

only when:
2.22. Automatic Bulk Weighing Systems



(a) the indication is stable within ±3 scale divisions,

and

(b) cannot be operated during a weighing operation.

S.2.2. Damping Means. - A system shall be equipped

with effective means necessary to bring the indications

quickly to a readable, stable equilibrium. Effective

means shall also be provided to permit the recording of

weight values only when the indication is stable within

plus or minus three scale divisions for devices with

10 000 scale divisions, or plus or minus one division for

devices with less than 10 000 scale divisions.

S.3. Interlocks and Gate Control.

S.3.1. Gate Position. - Provision shall be made to

clearly indicate to the operator the position of the gates

leading directly to and from the weigh hopper.

S.3.2. Interlocks. - Each automatic bulk weighing

system shall have operating interlocks to provide for the

following:

(a) Product cannot be cycled and weighed if the weight

recording element is disconnected or subjected to a

power loss.

(b) The recording element cannot print a weight if

either of the gates leading directly to or from the

weigh hopper is open.

(c) A "low paper" sensor, when provided, is activated.

(d) The system will operate only in the proper

sequence in all modes of operation.

(e) When an overfill alarm is activated, the system

shall indicate and record an overfill condition.

(Amended 1993)

51XGTHKNN5GPUQT


C The weigh hopper shall be equipped with an

overfill sensor which will cause the feed gate to

close, activate an alarm, and inhibit weighing until

the overfill condition has been corrected.


2CTVC#FFGF


D +HVJGU[UVGOKUGSWKRRGFYKVJCNQYGTICTPGTQT

UWTIGDKPVJCVICTPGTUJCNNCNUQDGGSWKRRGFYKVJ

CPQXGTHKNNUGPUQTYJKEJYKNNECWUGVJGICVGQHVJG

YGKIJJQRRGTVQTGOCKPQRGPCEVKXCVGCPCNCTO

CPFKPJKDKVYGKIJKPIWPVKNVJGQXGTHKNNEQPFKVKQP

JCUDGGPEQTTGEVGF

=0QPTGVTQCEVKXGCUQH,CPWCT[?


2CTVD#OGPFGF

S.4. Design of Weighing Elements.

S.4.1. Antifriction Means. - At all points at which a

live part of the mechanism may come into contact with

another part in the course of normal usage, frictional

effects shall be reduced to a minimum by means of

suitable antifriction means, opposing surfaces and points

being properly shaped, finished, and hardened.

S.4.2. Adjustable Components. - An adjustable

component, such as a potentiometer, shall be held

securely in adjustment and, except for a component for

adjusting level or a no-load reference value, shall not be

adjustable from the outside of the device.

S.4.3. Multiple Load-Receiving Elements. - A system

with a single indicating or recording element, or a

combination indicating recording element, that is coupled

to two or more load-receiving elements with independent

weighing systems, shall be provided with means to

prohibit the activation of any load-receiving element (or

elements) not in use, and shall be provided with

automatic means to indicate clearly and definitely which

load-receiving element (or elements) is in use.

S.4.4. Venting. - All weighing systems shall be vented

so that any internal or external pressure will not affect

the accuracy or operation of the system.

S.5. Marking Requirements. (See also G-S.1.)

S.5.1. Capacity and Value of the Scale Division. - The

capacity of the weighing system and the value of the

scale division shall be clearly and conspicuously marked

on the indicating element near the weight value

indications.

S.5.2. Weighing Elements. - On a weighing element not

permanently attached to an indicating element, there shall

be clearly and permanently marked for the purposes of

identification, the name, initials, or trademark of the

manufacturer, the manufacturer's designation that

positively identifies the pattern or design, and the

nominal capacity.
2.22. Automatic Bulk Weighing Systems



S.5.3. Temperature Limits. - Unless the temperature N.1.5. Discrimination Test. - A discrimination test shall

range is -10 EC to +40 EC (14 EF to 104 EF), the be conducted on all automatic indicating scales with the

temperature range shall be marked on the device. [Non- weighing device in equilibrium at zero-load and at

retroactive as of January 1, 1986.] maximum test load, and under controlled conditions in

(Added 1985) which environmental factors are reduced to the extent

S.5.4. Accuracy Class. -

(a) All systems used to weigh grain shall be marked

Class III*.

(b) All other systems shall be marked either Class III

or III L*.

(*See Section 2.20 Scales Code for the parameters for

these accuracy classes for scales. The specific

requirements for automatic bulk weighing systems

apply to these devices when there is a conflict between

the Scales Code and the Automatic Bulk Weighing

Systems Code.)

[Nonretroactive as of January 1, 1986.]

(Added 1985) (Amended 1992)

 N. Notes

N.1. Testing Procedures.

N.1.1. Test Weights. - The increasing load test shall be

conducted using test weights equal to at least 10 percent

of the capacity of the system:

(a) on automatic grain bulk-weighing systems installed

after January 1, 1984, and

(b) on other automatic bulk-weighing systems installed

after January 1, 1986.

(Amended 1987)

N.1.2. Increasing-Load Test. - An increasing-load test

consisting of substitution and strain-load tests shall be

conducted up to the used capacity of the weighing

system.

(Amended 1987)

N.1.3. Decreasing-Load Test. - A decreasing-load test

shall be conducted on devices used to weigh out.

(Added 1986)

N.1.4. Zero Balance or No-Load Reference Value

Change Test. - A test for change of zero-balance or noload reference value shall be conducted on all scales

after the removal of any test load. The change shall not

be more than the minimum tolerance applicable.

that they will not affect the results obtained.

[Nonretroactive as of January 1, 1986.]

N.1.5.1. Digital Device. - On a digital device, this

test is conducted from just below the lower edge of

the zone of uncertainty for increasing-load tests, or

from just above the upper edge of the zone of

uncertainty for decreasing-load tests.

(Added 1987)

N.2. Verification (Testing) Standards. - Standard weights

and masses used in verifying weighing devices shall comply

with requirements of NIST Handbook 105-1 (Class F) or

the tolerances expressed in Appendix A, Fundamental

Considerations, paragraph 3.2. (i.e., one-third of the

smallest tolerance applied).

 T. Tolerances

T.1. Tolerance Application. - Tolerance values shall be

applied to all indications and recorded representations of a

weighing system.

T.1.1. To Errors of Underregistration and

Overregistration. - The tolerances hereinafter

prescribed shall be applied equally to errors of underregistration and errors of overregistration.

 

T.1.2. To Increasing-Load Tests. - Basic tolerances

shall be applied.

T.1.3. To Decreasing-Load Tests. - Basic tolerances

shall be applied to systems used to weigh out.

(Added 1986)

T.1.4. To Tests Involving Digital Indications or

Representations. - To the tolerances that would

otherwise be applied, there shall be added an amount

equal to one-half the value of the scale division. This

does not apply to digital indications or recorded

representations that have been corrected for rounding

using error weights.

(Added 1986)

 

T.2. Minimum Tolerance Values. - The minimum

tolerance value shall not be less than half the value of the

scale division.

 
2.22. Automatic Bulk Weighing Systems



T.2.1. For Systems used to Weigh Construction

Materials. - The minimum maintenance and acceptance

tolerance shall be 0.1 percent of the weighing capacity of

the system, or the value of the scale division, whichever

is less.

(Added 1986)

T.3. Basic Tolerance Values.

 T.7.1.1. If not marked on the device, the temperature

T.3.1. Acceptance Tolerance. - The basic acceptance

tolerance shall be one-half the basic maintenance

tolerance.

 

T.3.2. For Systems used to Weigh Grain. - The basic

maintenance tolerance shall be 0.1 percent of test load.

T.3.3. For all Other Systems. - The basic maintenance

tolerance shall be 0.2 percent of test load.

(Amended 1986)

T.4. Time Dependence. - At constant test conditions, the

indication 20 seconds after the application of a load and

the indication after 1 hour shall not differ by more than the

absolute value of the applicable tolerance for the applied

load.

[Nonretroactive and enforceable as of January 1, 1987.]

(Added 1986)

T.5. Repeatability. - The results obtained by several

weighings of the same load under reasonably static test

conditions shall agree within the absolute value of the

maintenance tolerance for that load, and shall be within

applicable tolerances.

(Added 1986)

T.6. Discrimination, Digital Automatic Indicating

Scales. - A test load equivalent to 1.4d shall cause a change

in the indicated or recorded value of at least 2.0d. This

requires the zone of uncertainty to be not greater than 0.3

times the value of the scale division. 

(Added 1985)

T.7. Influence Factors. - The following factors are

applicable to tests conducted under controlled conditions

only, provided that:

(a) types of devices approved prior to January 1, 1986,

and manufactured prior to January 1, 1988, need not

meet the requirements of this section; and

(b) new types of devices submitted for approval after

January 1, 1986, shall comply with the requirements

of this section; and

(c) all devices manufactured after January 1, 1988, shall

comply with the requirements of this section.

[Nonretroactive as of January 1, 1986.]

T.7.1. Temperature. - Devices shall satisfy the tolerance

requirements under the following temperature

conditions: 

limits shall be:

 

-10 EC to 40 EC (14 EF to 104 EF)

 

T.7.1.2. If temperature limits are specified for the

device, the range shall be at least 30 EC (54 EF).

 

T.7.1.3. Temperature Effect on Zero-Load Balance.

- The zero-load indicator shall not vary by more than

1 division per 5 EC (9 EF) change in temperature.

 

T.7.1.4. Operating Temperature. - An indicating or

recording element shall not display or record any

usable values until the operating temperature

necessary for accurate weighing and a stable

zero-balance condition has been attained.

[Nonretroactive as of January 1, 1986.]

T.7.2. Barometric Pressure. - The zero indication shall

not vary by more than one scale division for a change in

barometric pressure of 1 kPa over the total barometric

range of 95 kPa to 105 kPa (28 in to 31 in of mercury).

[Nonretroactive as of January 1, 1986.] 

T.7.3. Electric Power Supply.

 

T.7.3.1. Power Supply, Voltage and Frequency

 

(a) Weighing devices that operate using alternating

current must perform within the conditions

defined in paragraphs T.2. through T.7. inclusive

over the line voltage range of 100 V to 130 V or

200 V to 250 V rms as appropriate and over the

frequency range of 59.5 Hz to 60.5 Hz.

(b) Battery-operated instruments shall not indicate

nor record values outside the applicable

tolerance limits when battery power output is

excessive or deficient.

T.7.3.2. Power Interruption. - A power interruption

shall not cause an indicating or recording element to
2.22. Automatic Bulk Weighing Systems



display or record any values outside the applicable [Nonretroactive as of January 1, 1987.]

tolerance limits. (Added 1986)

[Nonretroactive as of January 1, 1986.]

(Added 1985)

UR. User Requirements

UR.1. Selection Requirements.

 

UR.1.1. For Systems used to Weigh Grain. - The

number of scale divisions of a weighing system shall not

be less than 2 000 nor greater than 10 000 divisions.

[Nonretroactive as of January 1, 1984.]

(Amended 1986 and 1992)

U.R.1.2. For Systems used to Weigh Commodities

other than Grain. - The number of scale divisions shall

not be less than 500 nor greater than 10 000.

[Nonretroactive as of January 1, 1987.]

(Added 1986)

UR.2. Installation Requirements.

UR.2.1. Protection From Environmental Factors. -

The indicating elements, the lever system or load cells,

the load-receiving element, and any permanently

installed test weights shall be adequately protected from

environmental factors such as wind, weather, and RFI

that may adversely affect the operation or performance of

the system.

UR.2.2. Foundation, Supports, and Clearance. - The

foundation and supports of any system shall be such as to

provide strength, rigidity, and permanence of all

components, and clearance shall be provided around all

live parts so that no contact can result before or during

operation of the system.

UR.3. Loading Requirements.

UR.3.1. For Systems Used to Weigh Grain. - A

system shall not be used to weigh drafts less than

40 percent of the weighing capacity of the system except

for a final partial draft. Loads shall not normally be

retained on the weighing element for a period longer than

a normal weighing cycle.

(Amended 1986)

UR.3.2. For Systems Used to Weigh Commodities

Other than Grain. - A system shall not be used to weigh

drafts less than 20 percent of the weighing capacity of

the system except for a final partial draft. Loads shall

not normally be retained on the weighing element for a

period longer than a normal weighing cycle.

UR.4. System Modification. - The weighing system shall

not be modified except when the modification has been

approved by a competent engineering authority, preferably

that of the engineering department of the manufacturer of

the scale, and the official with statutory authority having

jurisdiction over the scale.

(Amended 1991)
2.23. Weights



Sec. 2.23. Weights

 A. Application

A.1. This code applies to commercial weights; that is,

weights used in connection with commercial weighing

devices.

A.2. This code does not apply to test weights or to other

"standards" of mass.

A.3. See also General Code requirements.

 S. Specifications

S.1. Material. - The material used for weights shall be as

follows:

(a) Weights of 6 g or 100 gr and larger shall be made of

a metal, or a metal alloy, not softer than brass.

(b) Weights of less than 6 g or 100 gr may be made of

aluminum, but shall not be made of iron or of unplated

steel, except stainless steel.

S.2. Design.

S.2.1. Surface. - The surface of a weight shall be

smooth and shall not be coated with thick, soft, or brittle

material. A weight of more than 2 g or 30 gr or shall not

have sharp edges, points, or corners.

 

S.2.2. Ring. - A ring on a weight shall not be split or

removable. T.1. In Excess and in Deficiency. - The tolerances

S.3. Adjusting Material. - Adjusting material shall be

securely positioned and shall not project beyond the surface

of the weight. T.2. On Avoirdupois Weights. - The maintenance

S.4. Marking Requirements.

S.4.1. General. - A weight shall be marked to show T.3. On Metric Weights. - The maintenance tolerances

clearly its nominal value, which shall include shall be as shown in Table 2. Acceptance tolerances shall

identification of the unit; however, the nominal value of be one-half the maintenance tolerances.

a weight of 30 gr or 2 g, or less, may be designated by

dots, lines, figures, distinctive shape, or other appropriate

means.

 

S.4.2. Apothecaries’ Weights. - On apothecaries’ dram,

ounce, and pound weights, the letters "ap" shall be used

in combination with the nominal value and the

appropriate abbreviation of or symbol for the unit.

S.4.3. Troy Weights. - On troy ounce and pound

weights, the letter "t" shall be used in combination with

the nominal value and the appropriate symbol of the unit.

S.4.4. Metric Weights. - On metric weights, the

symbols "kg," "g," and "mg" shall be used in

combination with the nominal value of kilograms, grams,

and milligrams, respectively.

S.4.5. Carat Weights. - On carat weights, the letter "c"

shall be used in combination with the nominal value.

S.4.6. Counterpoise Weight. - A counterpoise weight

shall be marked to show clearly both its nominal value

and the value it represents when used on the multiplyinglever scale for which it is intended.

 N. Notes

N.1. Testing Procedures. - Commercial weights should be

tested on a precision balance using standard weights, the

errors of which, when used without correction, do not

exceed 1/3 of the smallest tolerance to be applied. (See

Appendix A; Fundamental Considerations paragraphs 3.2.

and 3.3.)

 T. Tolerances

hereinafter prescribed shall be applied equally to errors in

excess and errors in deficiency.

tolerances shall be as shown in Table 1. Acceptance

tolerances shall be one-half the maintenance tolerances.

T.4. On Carat Weights. - The maintenance tolerances

shall be as shown in Table 2. Acceptance tolerances shall

be one-half the maintenance tolerances.
2.23. Weights



T.5. On Apothecaries’ and Troy Weights. - The

maintenance tolerances shall be as shown in Table 3.

Acceptance tolerances shall be one-half the maintenance

tolerances.

Table 1.

Maintenance Tolerance for Avoirdupois Weights

Maintenance Tolerance

Nominal Equal-Arm

Value Weights

Counterpoise Weights

For scales with multiples of For scales with multiples of 1000

less than 1000 or over

oz grains mg grains mg grains mg

1/64 0.1 6

1/32 0.3 19

1/16 0.4 26

1/8 0.5 32

1/4 1.0 65

1/2 1.5 97 1.0 65

1 1.7 110 1.0 65

2 2.0 130 1.0 65

3 2.0 130 1.5 97

4 3.0 190 1.5 97 1.0 65

5 3.5 230 1.5 97 1.0 65

6 3.5 230 1.5 97

8 4.0 260 2.0 130 1.5 97

10 4.0 260 2.5 160 2.0 130

12 5.0 320 2.5 160 2.0 130

lb grains mg grains mg grains mg

1 5.0 320 3.0 190 2.5 160

2 7.0 450 6.0 390 4.0 260

3 9.0 580 9.0 580 5.0 320

4 11.0 710 11.0 710 6.0 390

5 15 970 12.0 780 6.5 420

6 17 1190

7 19 1200

8 21 1400 15.0 970 9.0 580

9 23 1500

10 25 1600 18.0 1160 10.0 650

15 28 1800

20 30 1900

25 35 2300

30 40 2600

40 45 2900

50 50 3200
2.23. Weights



Table 2.

Maintenance Tolerances for Metric Weights

Nominal Maintenance Nominal Value Maintenance

Value Tolerance (g) Tolerance

(mg) (mg) (mg)

5 or less 0.1 1 4

10 0.3 2 6

20 0.4 3 8

30 0.6 5 10

50 0.8 10 15

100 1.0 20 20

200 1.5 30 30

300 2.0 50 40

500 3.0 100 70

200 100

300 150

500 175

Nominal Maintenance Nominal Maintenance

Value Tolerance Value Tolerance

(kg) (mg) (carats) (mg)

1 250 0.25* 0.6

2 400 0.5** 1.0

3 500 1.0 1.5

5 800 2.0 2.0

10 1000 3.0 3.0

20 1500 5.0 4.0

10.0 6.0

20.0 10.0

30.0 12.0

50.0 15.0

100.0 25.0

*25 points or less

**50 points
2.23. Weights



Table 3.

Maintenance Tolerances

For Apothecaries’ and Troy Weights

Nominal Nominal

Value Value Maintenance Tolerance Maintenance Tolerance

grains grains mg oz grains mg

1 0.01 0.6 1 0.4 25.0

2 0.02 1.3 2 0.6 40.0

3 0.03 2.0 3 1.0 65.0

5 0.03 2.0 4 1.5 100.0

10 0.04 2.5 5 1.6 105.0

20 0.06 4.0

scruples grains mg oz grains mg

1 0.06 4.0 6 1.8 115.0

2 0.10 6.5 7 1.9 125.0

8 2.0 130.0

9 2.1 135.0

10 2.2 145.0

dr grains mg oz grains mg

0.5 0.07 4.5 11 2.4 155.0

1.0 0.10 6.5 12 2.5 160.0

2.0 0.20 13.0 20 2.9 190.0

3.0 0.30 20.0 30 3.7 240.0

4.0 0.40 25.0 50 5.4 350.0

5.0 0.50 30.0

6.0 0.60 40.0

dwt grains mg oz grains mg

1 0.06 4.0 100 7.7 500.0

2 0.10 6.5 200 12.3 800.0

3 0.15 10.0 300 15.4 1000.0

4 0.20 13.0 500 23.1 1500.0

5 0.30 20.0 1000 38.6 2500.0

10 0.40 25.0
 2.24. Automatic Weighing Systems



Section 2.24. Automatic Weighing Systems - Tentative Code

This tentative code has only a trial or experimental status and is not intended to be enforced by weights and

measures officials. The requirements are designed for study prior to the development and adoption of a final

Code for Automatic Weighing Systems. The tentative code is intended to be used by the National Type

Evaluation Program for type evaluation of automatic weighing systems. If upgraded to become a permanent

code, all requirements, except those for tolerances, will be nonretroactive as of the effective date of the

permanent code; tolerance requirements will apply retroactively as of the effective date of the permanent code.

(Tentative Code Added 1995) (Amended 1998)

A. Application S. Specifications

A.1. - This code applies to devices used to weigh or fill S.1. Design of Indicating and Recording Elements and

packages while the object is in motion. Some weigh- of Recorded Representations.

labelers may also include a scale that is incorporated in a

conveyor system that weighs packages in a static weighing S.1.1. Zero Indication. 

mode.

This includes:

(a) Weigh-labelers, static and dynamic

(b) Automatic checkweighers

(Amended 1997)

A.2. - This code does not apply to:

(a) Belt-conveyor Scale Systems

(b) Railway Track Scales

(c) Monorail Scales

(d) Automatic Bulk-Weighing Systems

(e) Devices that measure quantity on a time basis

(f) Controllers or other auxiliary devices except as they

may affect the weighing performance

A.3. - Also see General Code requirements.

A.4. Type Evaluation. - The National Type Evaluation

Program will accept for type evaluation only those devices

that comply with all requirements of this code.

(Added 1998)

(a) A weigh-labeler shall be equipped with an indicating

or recording element. It shall either indicate or record a

zero-balance condition and an out-of-balance condition

on both sides of zero.

(b) An automatic checkweigher may be equipped with

an indicating or recording element.

(c) A zero-balance condition may be indicated by other

than a continuous digital zero indication, provided that

effective automatic means is provided to inhibit a

weighing operation or to return to a continuous digital

indication when the device is in an out-of-balance

condition.

S.1.1.1. Digital Indicating Elements. 

(a) A digital zero indication shall represent a balance

condition that is within ± 1⁄2 scale division.

(b) A digital indicating device shall either

automatically maintain a "center of zero" condition to

± 1⁄4 scale division or less, or have an auxiliary or

supplemental "center-of-zero" indicator that defines a

zero-balance condition to ± 1⁄4 scale division or less.

(c) Verification of the accuracy of the center of zero

indication to ± 1⁄4 scale division or less during dynamic

operation is not required on automatic checkweighers.

S.1.2. Value of Division Units. - The value of a

division "d" expressed in a unit of weight shall be equal

to:

 

(a) 1, 2, or 5; or 
2.24. Automatic Weighing Systems



6CDNG5

%CVGIQTKGUQH&GXKEGCPF/GVJQFUQH5GCNKPI

%CVGIQTKGUQH&GXKEG /GVJQFQH5GCNKPI

%CVGIQT[0Q4GOQVGEQPHKIWTCVKQPECRCDKNKV[ 5GCND[RJ[UKECNUGCNQTVYQGXGPVEQWPVGTUQPGHQT

ECNKDTCVKQPRCTCOGVGTUCPFQPGHQTEQPHKIWTCVKQP

RCTCOGVGTU

%CVGIQT[4GOQVGEQPHKIWTCVKQPECRCDKNKV[DWV

CEEGUUKUEQPVTQNNGFD[RJ[UKECNJCTFYCTG

&GXKEGUJCNNENGCTN[KPFKECVGVJCVKVKUKPVJGTGOQVG

EQPHKIWTCVKQPOQFGCPFTGEQTFUWEJOGUUCIGKHECRCDNG

QHRTKPVKPIKPVJKUOQFG

6JGJCTFYCTGGPCDNKPICEEGUUHQTTGOQVG

EQOOWPKECVKQPOWUVDGCVVJGFGXKEGCPFUGCNGFWUKPIC

RJ[UKECNUGCNQTVYQGXGPVEQWPVGTUQPGHQTECNKDTCVKQP

RCTCOGVGTUCPFQPGHQTEQPHKIWTCVKQPRCTCOGVGTU

%CVGIQT[4GOQVGEQPHKIWTCVKQPECRCDKNKV[CEEGUU

OC[DGWPNKOKVGFQTEQPVTQNNGFVJTQWIJCUQHVYCTG

UYKVEJGIRCUUYQTF

#PGXGPVNQIIGTKUTGSWKTGFKPVJGFGXKEGKVOWUV

KPENWFGCPGXGPVEQWPVGTVQVJGRCTCOGVGT+&

VJGFCVGCPFVKOGQHVJGEJCPIGCPFVJGPGYXCNWGQH

VJGRCTCOGVGT#RTKPVGFEQR[QHVJGKPHQTOCVKQPOWUV

DGCXCKNCDNGVJTQWIJVJGFGXKEGQTVJTQWIJCPQVJGTQP

UKVGFGXKEG6JGGXGPVNQIIGTUJCNNJCXGCECRCEKV[VQ

TGVCKPTGEQTFUGSWCNVQVGPVKOGUVJGPWODGTQHUGCNCDNG

RCTCOGVGTUKPVJGFGXKEGDWVPQVOQTGVJCP

TGEQTFUCTGTGSWKTGF0QVG&QGUPQVTGSWKTG

EJCPIGUVQDGUVQTGFHQTGCEJRCTCOGVGT

(b) a decimal multiple or submultiple of 1, 2, or 5; inside the device. After sealing, neither the mechanism

S.1.2.1. Weight Units. - A device shall indicate

weight values using only a single unit of measure. S.1.5. Adjustable Components. - Adjustable

S.1.3. Provision for Sealing. 

(a) Automatic Weighing Systems, Except Automatic

Checkweighers. - A device shall be designed with

provision(s) as specified in Table S.1.3., "Categories of

Device and Methods of Sealing," for applying a security

seal that must be broken, or for using other approved

means of providing security (e.g., data change audit trail

available at the time of inspection), before any change

that detrimentally affects the metrological integrity of the

device can be made to any electronic mechanism.

(b) For Automatic Checkweighers. - Security seals are

not required in field applications where it would prohibit

an authorized user from having access to the calibration

functions of a device. 

S.1.4. Automatic Calibration - A device may be fitted

with an automatic or a semi-automatic calibration

mechanism. This mechanism shall be incorporated

nor the calibration process shall facilitate fraud.

components shall be held securely in adjustment and,

except for a zero-load balance mechanism, shall be

located within the housing of the element.

S.2. Design of Zero and Tare Mechanisms.

S.2.1. Zero Load Adjustment.

S.2.1.1. Automatic Zero-Setting Mechanism. -

Except for automatic checkweighers, under normal

operating conditions the maximum load that can be

"rezeroed," when either placed on or removed from

the platform all at once, shall be 1.0 scale division.

S.2.1.2. Initial Zero-Setting Mechanism. - Except

for automatic checkweighers, an initial zero-setting

mechanism shall not zero a load in excess of

20 percent of the maximum capacity of the automatic

weighing system unless tests show that the scale meets
XOKP #

F

0

 2.24. Automatic Weighing Systems



all applicable tolerances for any amount of initial load indications are not valid for determining the actual

compensated by this device within the specified range. weight of a package (e.g., only appropriate for use in

S.2.2. Tare. - On any automatic weighing system the

value of the tare division shall be equal to the value of

the division. The tare mechanism shall operate only in a

backward direction (i.e., in a direction of

underregistration) with respect to the zero-load balance

condition of the automatic weighing system. A device

designed to automatically clear any tare value shall also

be designed to prevent the automatic clearing of tare

until a complete transaction has been indicated.

Note: On a computing automatic weighing system, this

requires the input of a unit price, the display of the unit

price, and a computed positive total price at a readable

equilibrium. Other devices require a complete weighing

operation, including tare, net, and gross weight

determination.

S.3.1. Multiple Range and Multi-Interval Automatic

Weighing System. The value of "e" shall be equal to the

value of "d."

S.3.2. Load Cell Verification Interval Value. - The

relationship of the value for the load cell verification

scale interval, v , to the scale division "d" for a specific min

scale installation shall be:

where N is the number of load cells in the scale.

Note: When the value of the scale division "d" differs

from the verification scale division "e" for the scale, the

value of "e" must be used in the formula above.

S.3.3. - For automatic checkweighers, the value of "e"

shall be specified by the manufacturer and may be larger

than "d," but in no case can "e" be more than 10 times

the value of "d."

S.4. Weight Indicators, Weight Displays, Reports, and

Labels.

S.4.1. Weight Units. - An indicating or recording

element shall indicate weight values using only a single

unit of measure.

S.4.2. Additional Digits in Displays. - Auxiliary digital

displays that provide additional digits for use during

performance evaluation may be included on automatic

checkweighers. However, in cases where these

statistical process control programs by users) they shall

be clearly and distinctly differentiated from valid weight

displays by indicating them to the user. 

For example, the additional digits may be differentiated

by color, partially covered by placing crosshatch overlays

on the display, or made visible only after the operator

presses a button or turns a key to set the device in a mode

which enables the additional digits.

S.4.3. Damping. - An indicating element equipped with

other than automatic recording elements shall be

equipped with effective means to permit the recording of

weight values only when the indication is stable within

plus or minus one scale division. The values recorded

shall be within applicable tolerances.

S.4.4. Over Capacity Indication. - An indicating or

recording element shall not display nor record any values

when the scale capacity is exceeded by nine scale

divisions.

S.4.5. Label Printer. - A device that produces a printed

ticket to be used as the label for a package shall print all

values digitally and of such size, style of type, and color

as to be clear and conspicuous on the label.

S.4.5.1. Label Printing. - If an automatic

checkweigher prints a label containing weight

information that will be used in a commercial

transaction, it must conform to all of the requirements

specified for weigh-labelers so that the printed ticket

meets appropriate requirements.

S.5. Accuracy Class.

S.5.1. Marking. - Weigh-labelers and automatic

checkweighers shall be Class III devices and shall be

marked accordingly, except that a weigh-labeler marked

Class IIIS may be used in package shipping applications.

(Amended 1997)

S.6. Parameters for Accuracy Classes. - The number of

divisions for device capacity is designated by the

manufacturer and shall comply with parameters shown in

Table S.6.

S.7. Marking Requirements. [See also G-S.1., G-S.4.,

G-S.6., G-S.7., G-U.2.1.1., and UR.3.3.]

S.7.1. Location of Marking Information. - Automatic

weighing systems which are not permanently attached to

an indicating element, and for which the load-receiving
2.24. Automatic Weighing Systems



6CDNG5

2CTCOGVGTUHQT#EEWTCE[%NCUUGU

0WODGTQH

FKXKUKQPUP

%NCUU 8CNWGQHVJGXGTKHKECVKQPFKXKUKQPFQTG

/KPKOWO /CZKOWO

5+7PKVU

+++

VQIKPENWUKXG  

GSWCNVQQTITGCVGTVJCPI  

+0%*2170&7PKVU

+++ NDVQNDKPENWUKXG  

Q\VQQ\KPENWUKXG 

GSWCNVQQTITGCVGTVJCPNDU  

GSWCNVQQTITGCVGTVJCPQ\  

+++5 ITGCVGTVJCPND  

ITGCVGTVJCPQ\  

(QT%NCUU+++FGXKEGUVJGXCNWGQHGKUURGEKHKGFD[VJGOCPWHCEVWTGTCUOCTMGFQPVJGFGXKEGFUJCNNPQV

DGUOCNNGTVJCPGGUJCNNDGFKHHGTGPVKCVGFHTQOFD[UK\GUJCRGQTEQNQT

#OGPFGF

element is the only part of the weighing/load-receiving (a) Main elements and components when not contained

element visible after installation, may have the marking in a single enclosure for the entire automatic weighing

information required in G-S.1. of the General Code and system;

Table S.7.a. and S.7.b. of the Automatic Weighing

Systems Code located in an area that is accessible only (b) Load cells for which Certificates of Conformance

through the use of a tool; provided that the information (CC) have been issued under the National Type

is easily accessible (e.g., the information may appear on Evaluation Program; and

the junction box under an access plate). The

identification information for these automatic weighing (c) Other equipment necessary to a weighing system but

systems shall be located on the weighbridge (load- having no metrological effect on the weighing system.

receiving element) near the point where the signal leaves

the weighing element or beneath the nearest access

cover.

S.7.2. Marking Required on Components of

Automatic Weighing Systems. - The following

components of automatic weighing systems shall be

marked as specified in Tables S.7.a. and S.7.b.:
2.24. Automatic Weighing Systems



Table S.7.a. Marking Requirements

Weighing

Equipment

To Be

Marked With

Weighing,

loadreceiving,

and

indicating

element in

same housing

Indicating

element not

permanently

attached to

weighing and

load-receiving

element

Weighing and

load-receiving

element not

permanently

attached to

indicating

element

Load

cell

with CC

(10)

Other

equipment or

device

(9)

Manufacturer’s ID (1) x x x x x

Model Designation(1) x x x x x

Serial Number and

Prefix(2)

x x x x x(13)

Accuracy Class (14) x x (8) x x

Nominal Capacity(3)(15) x x x

Value of Division, d (3) x x

Value of "e" (4) x x

Temperature Limits (5) x x x x

Special Application (11) x x x

Maximum Number of 

Scale Divisions

n (6) max

x (8) x x

Minimum Verification

Division

(e ) min

x 

"S" or "M" (7) x

Direction of Loading(12) x

Minimum Dead Load x

Maximum Capacity

(Max)

x x

Minimum Capacity

(Min)

x

Safe Load Limit x

Load Cell Verification

Interval (v ) min

x

Maximum Belt Speed

(m/sec or m/min)

x x

Note: See Table S.7.b. for applicable parenthetical notes.

(Amended 1999)
2.24. Automatic Weighing Systems



Table S.7.b.

Notes for Table S.7.a.

1. Manufacturer’s identification and model

designation. (See G-S.1.)

2. Serial number and prefix. (See G-S.1.)

3. The nominal capacity and value of the automatic

weighing system division shall be shown together

(e.g., 50 000 x 5 kg, or 30 x 0.01 lb) adjacent to

the weight display when the nominal capacity and

value of the automatic weighing system division

are not immediately apparent. Each division value

or weight unit shall be marked on variable-division

value or division-unit automatic weighing systems.

4. Required only if different from "d."

5. Required only on automatic weighing systems if

the range is other than -10 EC to 40 EC (14 EF to

104 EF).

6. This value may be stated on load cells in units of

1000; (e.g., n: 10 is 10 000 divisions.)

7. Denotes compliance for single or multiple load

cell applications.

8. An indicating element not permanently attached to

a weighing element shall be clearly and

permanently marked with the accuracy Class III, or

IIIS and the maximum number of divisions, n . max

9. Necessary to the weighing system but having no

metrological effect, e.g., auxiliary remote display,

keyboard, etc.

10. The markings may be either on the load cell or

in an accompanying document; except that, if an

accompanying document is provided, the serial

number shall appear both on the load cell and in

the document. The manufacturer’s name or

trademark, the model designation, and

identifying symbol for the serial number shall

also be marked both on the load cell and in any

accompanying document.

11. An automatic weighing system designed for a

special application rather than general use shall

be conspicuously marked with suitable words

visible to the operator and customer restricting

its use to that application.

12. Required if the direction of loading the load cell

is not obvious. 

13. Serial number and prefix (See G-S.1) Modules

without "intelligence" on a modular system

(e.g., printer, keyboard module, cash drawer,

and secondary display in a point-of-sale system)

are not required to have serial numbers.

14. The accuracy Class of a device shall be marked

on the device with the appropriate designation.

15. The nominal capacity shall be conspicuously

marked on any automatic-indicating or

recording automatic weighing system so

constructed that the capacity of the indicating or

recording element, or elements, is not

immediately apparent.
2.24. Automatic Weighing Systems



N. Notes N.3.1.1. Increasing-Load Test. - The

N.1. Test Requirements for Automatic Weighing loads approximately centered on the load-receiving

Systems. element of the scale.

N.1.1. Test Pucks and Packages. -


C 6GUVRWEMUCPFRCEMCIGUUJCNNDG


KTGRTGUGPVCVKXGQHVJGV[RGUK\GCPFYGKIJV

TCPIGUVQDGYGKIJGFQPCFGXKEGCPF


KK UVCDNGYJKNGKPOQVKQPJGPEGVJGNGPIVJCPF maximum capacity shall be placed in the center of

YKFVJQHCRWEMQTRCEMCIGUJQWNFDGITGCVGT

VJCPKVUJGKIJV


D (QTV[RGGXCNWCVKQPVJGOCPWHCEVWTGTUJCNNUWRRN[

VJGVGUVRWEMUQTRCEMCIGUHQTGCEJTCPIGQHVGUV

NQCFU


#OGPFGF

N.1.2. Accuracy of Test Pucks or Packages. - The

error in any test puck or package shall not exceed onefourth (1/4) of the acceptance tolerance. If packages are

used to conduct field tests on automatic weighing

systems, the package weights shall be determined on a

reference scale or balance with an inaccuracy that does

not exceed one-fifth (1/5) of the smallest tolerance that

can be applied to the device under test.

N.1.3. Verification (Testing) Standards. - Field

standard weights shall comply with requirements of

NIST Handbook 105-1 (Class F) or the tolerances

expressed in Fundamental Considerations, paragraph 3.2.

(i.e., one-third of the smallest tolerance applied).

N.2. Test Requirements for Automatic Weighing

Systems.

N.2.1. Tests Loads. - A performance test shall consist

of four separate test runs conducted at different test loads

according to Table N.3.2.

N.2.2. Influence Factor Testing. - Influence factor

testing shall be conducted statically.

N.3. Test Procedures - Weigh-Labelers. - If the device is

designed for use in static weighing, it shall be tested

statically using mass standards.

Note: If the device is designed for only dynamic weighing,

it shall only be tested dynamically.

N.3.1. Laboratory - Static Tests.

increasing-load test shall be conducted with the test

N.3.1.2. Decreasing-Load Test. - The

decreasing-load test shall be conducted with the test

loads approximately centered on the load- receiving

element of the scale.

N.3.1.3. Shift Test. - To determine the effect of offcenter loading, a test load equal to one-half (1⁄2)

each of the four points equidistant between the center

and front, left, back, and right edges of the load

receiver. 

N.3.1.4. Discrimination Test. - A discrimination test

shall be conducted with the weighing device in

equilibrium at zero load and at maximum test load,

and under controlled conditions in which environmental factors are reduced to the extent that they will

not affect the results obtained. This test is conducted

from just below the lower edge of the zone of

uncertainty for increasing load tests, or from just

above the upper edge of the zone of uncertainty for

decreasing-load tests.

N.3.1.5. Zero-Load Balance Change. - A zero-load

balance change test shall be conducted on all

automatic weighing systems after the removal of any

test load. The zero-load balance should not change by

more than the minimum tolerance applicable. (Also

see G-UR.4.2.) 

N.3.1.6. Influence Factor Testing. - Influence factor

testing shall be conducted.

N.3.2. Laboratory - Dynamic Tests. - The device shall

be tested at the highest speed for each weight range using

standardized test pucks or packages. Test runs shall be

conducted using four test loads as described in Table

N.3.2. Each test load shall be run a minimum of 10

consecutive times.
 2.24. Automatic Weighing Systems



6CDNG06GUV.QCFU

#VQTPGCTOKPKOWOECRCEKV[

#VQTPGCTOCZKOWOECRCEKV[

#VVYQETKVKECNRQKPVUDGVYGGP

OKPKOWOCPFOCZKOWOECRCEKV[

6GUVOC[DGEQPFWEVGFCVQVJGTNQCFUKHVJGFGXKEG

KUKPVGPFGFHQTWUGCVQVJGTURGEKHKEECRCEKVKGU

N.3.2.1. Shift Test. - To determine the effect of

eccentric loading, for devices without a means to align

packages, a test load equal to One-third (1/3)

maximum capacity shall be passed over the load

receiver or transport belt (1) halfway between the

center and front edge, and (2) halfway between the

center and back edge.

N.3.3. Field Test Procedures

N.3.3.1. Static Tests. - If the automatic weighing

system is designed to operate statically, and used in

that manner, during normal use operation, it shall be

tested statically using mass standards. The device shall

not be tested statically if it is used only dynamically.

N.3.3.2. Dynamic Tests. - The device shall be tested

at the normal operating speed using packages. Test

runs should be conducted using at least two test loads

distributed over its normal weighing range (e.g., at the

lowest and highest ranges in which the device is

typically operated.) Each test load should be run a

minimum of 10 consecutive times.

N.4. Test Procedures - Automatic Checkweigher

N.4.1. Laboratory - Static Tests. - If the scale is

designed to operate statically during normal user

operation, it shall be tested statically using the applicable

weigh-labeler requirements.

N.4.2. Laboratory - Dynamic Tests. - The device shall

be tested at the highest speed in each weight range using

standardized test pucks or packages. Test runs shall be

conducted using four test loads. The number of

consecutive test weighments shall be as described in

Table N.4.2.

6CDNG00WODGTQH5CORNG9GKIJVURGT6GUV

HQT#WVQOCVKE%JGEMYGKIJGTU

9GKIJKPI4CPIG

OOCUUQHVGUVNQCF

0WODGTQHUCORNG

YGKIJVURGTVGUV

FKXKUKQPUOMI

FKXKUKQPUOND



MIOMI

NDOND



MIOMI

NDOND



MINDO 

N.4.3. Field Test Procedures.

N.4.3.1. Static Tests. - If the scale is designed to

operate statically during normal user operation, it shall

be tested statically according to Sections N.3.1.1.

through N.3.1.5.

N.4.3.2. Dynamic Tests. - The device shall be tested

dynamically at the highest normal operating speed

using packages at two test loads distributed over its

normal weighing range. The number of consecutive

weighments shall be one-half (1⁄2) of those specified in

Table N.4.2., but not less than 10.

T. Tolerances

T.1. Principles.

T.1.1. Design. - The tolerance for a weighing device is

a performance requirement independent of the design

principle used.

T.1.2. Scale Division. - The tolerance for a weighing

device is related to the value of the scale division (d) or

the value of the verification scale division (e) and is

generally expressed in terms of d or e. The random

tolerance for automatic checkweighers is expressed in

terms of Maximum Allowable Variance (MAV).

T.2. Tolerance Application.

T.2.1. General. - The tolerance values are positive (+)

and negative (-) with the weighing device adjusted to

zero at no load. When tare is in use, the tolerance values

are applied from the tare zero reference; the tolerance

values apply to certified test loads only.

T.2.2. Type Evaluation Examinations. - For type

evaluation examinations, the tolerance values apply to
2.24. Automatic Weighing Systems



increasing and decreasing load tests within the temperature, power supply, and barometric pressure limits

specified in T.7.

T.2.3. Multiple Range and Multi-Interval Automatic

Weighing System - For multiple range and multiinterval devices, the tolerance values are based on the

value of the scale division of the range in use.

T.3. Tolerance Values. 

T.3.1. Tolerance Values - Class III Weigh-Labeler.

(See T.3.2. for Class IIIS Weigh-Labelers)

T.3.1.1. Static Tests. - Tolerance values shall be as

specified in Table T.3., Class III - Tolerances in

Divisions.

6CDNG6%NCUU+++6QNGTCPEGUKP&KXKUKQPUF

/CKPVGPCPEG    

#EEGRVCPEG    

%NCUU 6GUV.QCFKP&KXKUKQPU

+++    


T.3.1.2. Dynamic Tests. - Acceptancre tolerance

values shall be the same as maintenance tolerance

values specified in Table T.3., Class III - Tolerances

in Divisions.

66QNGTCPEG8CNWGU%NCUU+++59GKIJNCDGNGTU

KP2CEMCIG5JKRRKPI#RRNKECVKQPU


#FFGF

65VCVKE6GUVU6QNGTCPEGXCNWGUUJCNNDGCU

URGEKHKGFKP6CDNG65VCVKE6QNGTCPEGUHQT

%NCUU+++59GKIJNCDGNGTU.

6&[PCOKE6GUVU6QNGTCPEGXCNWGUURGEKHKGFKP

6CDNG6&[PCOKE6QNGTCPEGUHQT%NCUU+++5

9GKIJNCDGNGTUUJCNNDGCRRNKGF

Table T.3.2.1. Static Tolerance for Class IIIS

Weigh-labelers

Test Load in Tolerance in Divisions

Divisions

Class IIIS Acceptance Maintenance

0 - 50 ± 0.5 ± 1

51 - 200 ± 1 ± 2

201 - 1000 ± 1.5 ± 3


#FFGF

Table T.3.2.2. Dynamic Tolerance

for Class IIIS Weigh-labelers

Test Load in Tolerance in Divisions

Divisions

Class IIIS Acceptance Maintenance

0 - 50 ± 1.5 ± 2

51 - 200 ± 2 ± 3

201 - 1000 ± 2.5 ± 4

(Added 1997)

T.3.3. Tolerance Values - Automatic Checkweighers.

T.3.3.1. Laboratory Tests for Automatic

Checkweighers. 

T.3.3.1.1. Static Tests. The acceptance tolerance

values specified in Table T.3., Class III-Tolerances in

Divisions, shall be applied.

T.3.3.1.2. Dynamic Tests.

(a) The systematic error for each test run must be

within the acceptance tolerances for the test load as

specified in Table N.3.2.

(b) The standard deviation of the results shall not

exceed one-ninth (1/9) of the Maximum Allowable

Variation (MAV) for specific package weights (3

standard deviations cannot exceed one-third (1/3) of

the MAV value) as required in the 4th Edition of

NIST Handbook 133. This value does not change

regardless of whether acceptance, or maintenance

tolerances are being applied to the device under test.

(i) For U.S. Department of Agriculture (USDA)

inspected meat and poultry products packaged at a

plant subject to inspection by the USDA Food
 2.24. Automatic Weighing Systems



Safety and Inspection Service, use Table 2-9, U.S. for Packages Labeled by Liquid or Dry

Department of Agriculture, Meat and Poultry, Volume.

Groups and Lower Limits for Individual Packages,

or

(ii) for all other packages with a labeled net

quantity in terms of weight use Table 2-5,

Maximum Allowable Variations for Packages

Labeled by Weight.

(iii) For all packages with a labeled net quantity in

terms of liquid or dry volume use Table 2-6,

Maximum Allowable Variations for Packages

Labeled by Liquid or Dry Volume.

T.3.3.2. Field Tests for Automatic

Checkweighers.

T.3.3.2.1. Static Test Tolerances. - The

tolerance values shall be as specified in Table

T.3., Class III-Tolerances in Divisions.

T.3.3.2.2. Dynamic Test Tolerances. -

(a) The systematic error requirement is not

applied in a field test.

(b) The standard deviation of the test results

shall not exceed one-ninth (1/9) of the Maximum

Allowable Variation (MAV) for specific

package weights (3 standard deviations cannot

exceed one-third (1/3) of the MAV value) as

required in the 4th Edition of NIST Handbook

133. This value does not change regardless of

whether acceptance or maintenance tolerances

are being applied to the device under test.

(i) For U.S. Department of Agriculture

(USDA) inspected meat and poultry products

packaged at a plant subject to inspection by

the USDA Food Safety and Inspection

Service, use Table 2-9, U.S. Department of

Agriculture, Meat and Poultry, Groups and

Lower Limits for Individual Packages, or

(ii) for all other packages with a labeled net

quantity in terms of weight use Table 2-5,

Maximum Allowable Variations for Packages

Labeled by Weight.

(iii) For all packages with a labeled net

quantity in terms of liquid or dry volume use

Table 2-6. Maximum Allowable Variations

T.4. Agreement of Indications. - In the case of a

weighing system equipped with more than one indicating

element or indicating element and recording element

combination, the difference in the weight value indications

of any load shall not be greater than the absolute value of

the applicable tolerance for that load, and shall be within

tolerance limits.

T.5. Repeatability. - The results obtained from several

weighings of the same load under reasonably static test

conditions shall agree within the absolute value of the

maintenance tolerance for that load, and shall be within

applicable tolerances.

T.6. Discrimination. - A test load equivalent to 1.4 d shall

cause a change in the indicated or recorded value of at least

2.0 d. This requires the zone of uncertainty to be not greater

than 0.3 d (See N.3.1.4.)

T.7. Influence Factors. - The following factors are

applicable to tests conducted under controlled conditions

only.

T.7.1. Temperature. - Devices shall satisfy the

tolerance requirements under the following temperature

conditions:

T.7.1.1. - If not specified in the operating instructions

or if not marked on the device, the temperature limits

shall be: -10 EC to 40 EC (14 EF to 104 EF).

T.7.1.2. - If temperature limits are specified for the

device, the range shall be at least 30 EC (54 EF).

T.7.1.3. Temperature Effect on Zero-Load

Balance. - The zero-load indication shall not vary by

more than one division per 5 EC (9 EF) change in

temperature.

T.7.1.4. Operating Temperature. - The indicating

or recording element shall not display nor record any

usable values until the operating temperature

necessary for accurate weighing and a stable zero

balance condition have been attained.

T.7.2. Barometric Pressure. - The zero indication shall

not vary by more than one division for a change in barometric pressure of 1 kPa over the total barometric

pressure range of 95 kPa to 105 kPa (28 in to 31 in of

Hg).
2.24. Automatic Weighing Systems



T.7.3. Electric Power Supply. - UR.1.2. Value of the Indicated and Recorded Scale

T.7.3.1. Power Supply, Voltage and Frequency. the same as the division value indicated.

(a) Alternating Current. - Weighing devices that

operate using alternating current must perform within

the conditions defined in paragraphs T.3. through

T.7., inclusive, over the line voltage range of 100 V to

130 V or 200 V to 250 V rms as appropriate, and over

the frequency range of 59.5 Hz to 60.5 Hz.

Note: This requirement applies only to metrologically

significant voltage supplies.

(Amended 2001)

(b) Battery. - Battery operated instruments shall not

indicate nor record values outside the applicable

tolerance limits when battery power output is

excessive or deficient.

T.7.3.2. Power Interruption. - A power interruption contacts may result when the load-receiving element is

shall not cause an indicating or recording element to empty, nor throughout the weighing range of the scale.

display or record any values outside the applicable

tolerance limits.

T.8. Radio Frequency Interference (RFI) and Other weighed load to the weighing zone and that carries the

Electromagnetic Interference Susceptibility. - The weighed load away from the weighing zone shall be

difference between the weight indication with the maintained per the manufacturers recommendations.

disturbance and the weight indication without the

disturbance shall not exceed one scale division (d) or the

equipment shall:

(a) blank the indication, or

(b) provide an error message, or

(c) the indication shall be so completely unstable that it

could not be interpreted, or transmitted into memory or to a

recording element, as a correct measurement value.

UR. User Requirements

UR.1. Selection Requirements. - Equipment shall be

suitable for the service in which it is used with respect to

elements of its design, including but not limited to, its

capacity, number of scale divisions, value of the scale

division or verification scale division, minimum capacity,

and computing capability.

UR.1.1. General. - Automatic Weighing Systems shall

be designated by the manufacturer for that service.

Division. - The value of the division as recorded shall be

UR.2. Installation Requirements.

UR.2.1. Protection From Environmental Factors. -

The indicating elements, the lever system or load cells,

and the load-receiving element of a permanently installed

scale, and the indicating elements of a scale not intended

to be permanently installed, shall be adequately protected

from environmental factors such as wind, weather, and

RFI that may adversely affect the operation or performance of the device.

UR.2.2. Foundation, Supports, and Clearance. - The

foundation and supports of any scale installed in a fixed

location shall be such as to provide strength, rigidity, and

permanence of all components, and clearance shall be

provided around all live parts to the extent that no

UR.2.3. Entry and Departure From Weighing Area.

- The belt or other conveyance that introduces the

UR.3. Use Requirements.

UR.3.1. Minimum Load. - The minimum load shall be

as specified by the manufacturer, but not less than 20

divisions since the use of a device to weigh light loads is

likely to result in relatively large errors.

74/KPKOWO.QCFHQT%NCUU+++59GKIJ

NCDGNGTU6JGOKPKOWONQCFUJCNNDGCUURGEKHKGF

D[VJGOCPWHCEVWTGTDWVPQVNGUUVJCPFKXKUKQPU

UKPEGVJGWUGQHCFGXKEGVQYGKIJNKIJVNQCFUKUNKMGN[

VQTGUWNVKPTGNCVKXGN[NCTIGGTTQTU


#FFGF

UR.3.2. Maximum Load. - An automatic weighing

system shall not be used to weigh a load of more than the

maximum capacity of the automatic weighing system.

UR.3.3. Special Designs. - An automatic weighing

system designed and marked for a special application

shall not be used for other than its intended purpose.

UR.3.4. Use of Manual Gross Weight Entries. -

Manual entries are permitted only when a device or

system is generating labels for standard weight packages.
U ' 

P & 

E 
ZK & Z



U ' 

P& 
j Z



K & 
j ZK





P



Z

Z ' j Z

P

 2.24. Automatic Weighing Systems



UR.4. Maintenance Requirements.

UR.4.1. Balance Condition. - If an automatic weighing

system is equipped with a zero-load display, the

zero-load adjustment of an automatic weighing system

shall be maintained so that the device indicates or

records a zero balance condition.

UR.4.2. Level Condition. - If an automatic weighing

system is equipped with a level-condition indicator, the

automatic weighing system shall be maintained in level.

UR.4.3. Automatic Weighing System Modification. -

The length or the width of the load-receiving element of

an automatic weighing system shall not be increased

beyond the manufacturer’s design dimension, nor shall

the capacity of an automatic weighing system be

increased beyond its design capacity by replacing or

modifying the original primary indicating or recording

element with one of a higher capacity, except when the

modification has been approved by competent

engineering authority, preferably that of the engineering

department of the manufacturer of the automatic

weighing system, and by the weights and measures

authority having jurisdiction over the automatic

weighing system.

D. Definitions

automatic checkweigher. - A dynamic automatic weighing

system used to subdivide items of different weights into one

or more subgroups, such as identifying packages that have

acceptable or unacceptable fill levels. These systems may

be used to fill standard packages for compliance with net

weight requirements.

automatic weighing system (AWS). - An automatic

weighing system is a weighing device that, in combination

with other hardware and/or software components,

automatically weighs discrete items. Examples include, but

are not limited to, weigh-labelers and checkweighers.

package rate. - PPM - Packages per minute.

random error(s). - The sample standard deviation of the

error (indicated values) for a number of consecutive

automatic weighings of a load, or loads, passed over the

load receptor, shall be expressed mathematically as:

or

where:

x = error of a load indication

n = the number of loads

systematic (average) error ( ). - The mean value of

the error (of indication) for a number of consecutive

automatic weighings of a load, or loads, passed over the

load receiving element (e.g., weightable), shall be expressed

mathematically as:

where:

x = error of a load indication

n = the number of loads

test puck. - Metal or plastic object used to simulate a

package. Pucks can be made in a variety of dimensions and

have different weights to represent a wide range of package

sizes. Metal versions may be covered with rubber cushions

to eliminate the possibility of damage to weighing and

handling equipment. The puck mass is adjusted to specific

accuracy so that pucks can be used to conduct performance

tests.
2.24. Automatic Weighing Systems



weigh-labeler. - An automatic weighing system that

determines the weight of a package and prints a label or

other document bearing a weight declaration for each

discrete item (usually a label also includes unit and total

price declarations). Typically, this type of weighing system

determines the weight of packages dynamically, but may

also include a scale that is incorporated in a conveyor

system that weighs packages in a static weighing mode.

Weigh-labelers are sometimes used to weigh and label

standard and random packages (also called "Prepackaging

Scales").


Section 3

Page

3.30. Liquid-Measuring Devices 3-3

3.31. Vehicle-Tank Meters 3-19

3.32. LPG and Anhydrous Ammonia Liquid-Measuring Devices 3-27

3.33. Hydrocarbon Gas Vapor-Measuring Devices 3-35

3.34. Cryogenic Liquid-Measuring Devices 3-43

3.35. Milk Meters 3-49

3.36. Water Meters 3-55

3.37. Mass Flow Meters 3-59

3.38. Carbon Dioxide Liquid-Measuring Devices - Tentative Code 3-69

Note: In this section of Handbook 44, the reference temperature for the temperature compensation of refined petroleum

products is shown as "15 EC (60 EF)." Although these values are not exact equivalents, they reflect industry usage when the

SI and inch-pound units are used in measurements.

3.30. Liquid-Measuring Devices



Sec. 3.30. Liquid-Measuring Devices

A. Application

A.1. - This code applies to:

(a) devices used for the measurement of liquids, including

liquid fuels and lubricants, and

(b) wholesale devices used for the measurement and

delivery of agri-chemical liquids such as fertilizers,

feeds, herbicides, pesticides, insecticides, fungicides,

and defoliants.

(Added 1985)

A.2. - This code does not apply to:

(a) meters mounted on vehicle tanks (see Sec. 3.31. Code

for Vehicle-Tank Meters),

(b) devices used for dispensing liquefied petroleum gases

(see Sec. 3.32. Code for Liquefied Petroleum Gas and

Anhydrous Ammonia Liquid-Measuring Devices),

(c) devices used for dispensing other liquids that do not

remain in a liquid state at atmospheric pressures and

temperatures,

(d) water meters,

(e) devices used solely for dispensing a product in

connection with operations in which the amount

dispensed does not affect customer charges, or

(f) mass flow meters (see Sec. 3.37. Code for Mass Flow

Meters.)

(Added 1994)

A.3. - In addition to the requirements of this code, liquidmeasuring devices shall meet the requirements of

Section 1.10. General Code.

S. Specifications

S.1. Indicating and Recording Elements and Recorded

Representations.

S.1.1. General. - A liquid-measuring device:

(a) shall be equipped with a primary indicating

element, and

(b) may be equipped with a primary recording element.

S.1.2. Units. - A liquid-measuring device shall indicate,

and record if the device is equipped to record, its

deliveries in liters, gallons, quarts, pints, or binarysubmultiples or decimal subdivisions of the liter or

gallon.

(Amended 1987, 1994)

S.1.2.1. Retail Motor-Fuel Devices. - Deliveries

shall be indicated and recorded, if the device is

equipped to record, in liters or gallons and decimal

subdivisions or fractional equivalents thereof.

(Added 1979)

S.1.2.2. Agri-Chemical Liquid Devices.

S.1.2.2.1. Liquid Measure. - Deliveries shall be

indicated and recorded in liters or gallons and

decimal subdivisions or fractional equivalents

thereof.

S.1.2.3. Value of Smallest Unit. - The value of the

smallest unit of indicated delivery, and recorded

delivery if the device is equipped to record, shall not

exceed the equivalent of:

(a) 0.5 L (1 pt) on retail devices;

(b) 5 L (1 gal) on wholesale devices.

This requirement does not apply to manually operated

devices equipped with stops or stroke-limiting means.

(Amended 1983 and 1986)

S.1.3. Advancement of Indicating and Recording

Elements. - It shall not be possible to advance primary

indicating and recording elements except by the

mechanical operation of the device. Clearing a device by

advancing its elements to zero is permitted, but only if:

(a) once started, the advancement movement cannot be

stopped until zero is reached, and
3.30. Liquid-Measuring Devices



(b) in the case of indicating elements only, such

elements are automatically obscured until the

elements reach the correct zero position. (a) The index of an indicator shall not be wider than

S.1.4. Graduations.

S.1.4.1. Length. - Graduations shall be varied in

length so that they may be conveniently read.

S.1.4.2. Width. - In a series of graduations, the width

of:

(a) every graduation shall be at least 0.2 mm

(0.008 in) but not greater than the minimum

clear interval between graduations, and

(b) main graduations shall be not more than 50

percent greater than the width of subordinate S.1.5.5. Parallax. - Parallax effects shall be reduced

graduations. to the practical minimum.

S.1.4.3. Clear Interval Between Graduations. -The S.1.6. Operating Requirements, Retail Devices

clear interval between graduations shall be not less (Except Slow Flow Meters).

than 1.0 mm (0.04 in). If the graduations are not

parallel, the measurement shall be made: S.1.6.1. Indication of Delivery. - The device shall

(a) along the line of movement of the tip of the condition and the quantity delivered (up to the

index of the indicator as it passes over the nominal capacity). 

graduations, or

(b) if the indicator extends over the entire length of and its associated total sales price need not be

the graduations, at the point of widest separation indicated.

of the graduations. (Amended 1982)

S.1.5. Indicators. S.1.6.2. Provisions for Power Loss.

S.1.5.1. Symmetry. - The portion of the index of an

indicator associated with the graduations shall be

symmetrical with respect to the graduations.

S.1.5.2. Length.

(a) If the indicator and the graduations are in

different planes, the index of the indicator shall

extend to each graduation with which it is to be

used. S.1.6.2.2. User Information. - The device memory

(b) If the indicator is in the same plane as the dispensed and the sales price totals during power

graduations, the distance between the index of loss.

the indicator and the ends of the graduations, [Nonretroactive as of January 1, 1983.]

measured along the line of the graduations, shall

be not more than 1.0 mm (0.04 in).

S.1.5.3. Width.

the width of the narrowest graduation. 

[Nonretroactive as of January 1, 2002.]

(Amended 2000)

(b) If the index of an indicator extends over the

entire length of a graduation, it shall be of

uniform width throughout the portion that

coincides with the graduation.

S.1.5.4. Clearance. - If the indicator and the

graduations are in different planes, the clearance

between the index of an indicator and the plane of the

graduations shall be no greater than 1.5 mm (0.06 in).

automatically show on its face the initial zero

However, the first 0.03 L (or 0.009 gal) of a delivery

S.1.6.2.1. Transaction Information. - In the event

of a power loss, the information needed to complete

any transaction in progress at the time of the

power loss (such as the quantity and unit price, or

sales price) shall be determinable for at least 15

minutes at the dispenser or at the console if the

console is accessible to the customer.

[Nonretroactive as of January 1, 1983.]

shall retain information on the quantity of fuel

S.1.6.3. Return to Zero.

(a) The primary indicating elements, and primary

recording elements if the device is equipped to

record, shall be readily returnable to a definite
3.30. Liquid-Measuring Devices



zero indication. However, a key-lock operated or

other self-operated device may be equipped with

cumulative indicating or recording elements, (a) A computing device shall compute the total sales

provided that it is also equipped with a zero- price at any single-purchase unit price (i.e.,

return indicating element. excluding fleet sales, other price contract sales,

(b) It shall not be possible to return primary trucks) for which the product being measured is

indicating elements, or primary recording offered for sale at any delivery possible within

elements beyond the correct zero position. either the measurement range of the device or

(Amended 1972) the range of the computing elements, whichever

S.1.6.4. Display of Unit Price and Product

Identity.

S.1.6.4.1. Unit Price.

(a) A computing or money-operated device shall mathematically computed price (quantity x unit

be able to display on each face the unit price price = total sales price) by an amount greater

at which the device is set to compute or to than the value in Table 1.

dispense. (Amended 1984, 1989, and 1993)

(b) Whenever a grade, brand, blend, or mixture

is offered for sale from a device at more than

one unit price, then all of the unit prices at

which that product is offered for sale shall be

displayed or shall be capable of being

displayed on the dispenser using controls

available to the customer prior to the delivery

of the product. It is not necessary that all of

the unit prices for all grades, brands, blends,

or mixtures be simultaneously displayed prior

to the delivery of the product. This subsection

shall not apply to fleet sales, other contract

sales, or truck refueling sales (e.g., sales from

dispensers used to refuel trucks).

[Effective and nonretroactive as of January 1,

1991.]

(Amended 1989 and 1997)

S.1.6.4.2. Product Identity.

(a) A device shall be able to conspicuously

display on each side the identity of the

product being dispensed.

(b) A device designed to dispense more than one

grade, brand, blend, or mixture of product

also shall be able to display on each side the

identity of the grade, brand, blend, or mixture

being dispensed.

S.1.6.5. Money-Value Computations.

and truck stop dispensers used only to refuel

is less.

[Effective and nonretroactive as of January 1,

1991].

(b) The analog sales price indicated for any

delivered quantity shall not differ from a

S.1.6.5.1. Money-Value Divisions, Analog. - The

values of the graduated intervals representing

money values on a computing type device shall be

no greater than those in Table 1.

(Amended 1991)

S.1.6.5.2. Money-Value Divisions, Digital. - A

computing type device with digital indications shall

comply with the requirements of paragraph G.S.5.5.

Money Values, Mathematical Agreement, and the

total price computation shall be based on quantities

not exceeding 0.05 L for devices indicating in

metric units and 0.01-gal. intervals for devices

indicating in inch-pound units.

(Added 1980)

S.1.6.5.3. Auxiliary Elements. - If a system is

equipped with auxiliary indications, all indicated

money value divisions of the auxiliary element shall

be identical with those of the primary element.

[Nonretroactive and enforceable as of January 1,

1985.]

S.1.6.5.4. Selection of Unit Price. - Except for

dispensers used exclusively for fleet sales, other

price contract sales, and truck refueling (e.g., truck

stop dispensers used only to refuel trucks), when a

product or grade is offered for sale at more than

one unit price through a computing device, the

selection of the unit price shall be made prior to

delivery using controls on the device or other

customer-activated controls. A system shall not

permit a change to the unit price during delivery of

product.
3.30. Liquid-Measuring Devices



6CDNG

/QPG[8CNWG&KXKUKQPUCPF

/CZKOWO#NNQYCDNG8CTKCVKQPUHQT/QPG[8CNWG

%QORWVCVKQPUQP/GEJCPKECN#PCNQI%QORWVGTU

7PKV2TKEG

/QPG[8CNWG

&KXKUKQP

/CZKOWO

#NNQYCDNG8CTKCVKQP

(TQO 6QCPFKPENWFKPI &GUKIP

6GUV

(KGNF

6GUV

 NKVGTQT

ICNNQP c vc vc

NKVGTQT

ICNNQP

NKVGTQT

ICNNQP cQTc vc vc

NKVGTQT

ICNNQP

NKVGTQT

ICNNQP cQTc vc vc

NKVGTQT

ICNNQP

NKVGTQT

ICNNQP c vc vc

[Nonretroactive as of January 1, 1991.]

(Added 1989)(Amended 1991, 1992, 1993, and

1996)

S.1.6.5.5. Display of Quantity and Total Price. - through a built-in or separate recording element for all

When a delivery is completed, the total price and transactions conducted with point-of-sale systems or

quantity for that transaction shall be displayed on devices activated by debit cards, credit cards, and/or

the face of the dispenser for at least 5 minutes or cash:

until the next transaction is initiated by using

controls on the device or other customer-activated (a) the total volume of the delivery,

controls.

[Nonretroactive as of January 1, 1994.] (b) the unit price,

(Added 1992)(Amended 1996)

S.1.6.6. Agreement Between Indications. - When a

quantity value indicated or recorded by an auxiliary

element is a derived or computed value based on data

received from a retail motor fuel dispenser, the value

may differ from the quantity value displayed on the

dispenser, provided the following conditions are met:

(a) all total money values for an individual sale that The indicator shall move at least 2.5 cm (1 in) in

are indicated or recorded by the system agree; relation to the graduations, if provided, for a delivery

and of 0.5 L (1 pt).

(b) within each element, the values indicated or

recorded meet the formula (quantity x unit price

= total sales price) to the closest cent.

[Nonretroactive as of January 1, 1988.]

(Added 1985)(Amended 1987 and 1988)

S.1.6.7. Recorded Representations. - Except for fleet

sales and other price contract sales, a printed receipt

providing the following information shall be available

(c) the total computed price, and

(d) the product identity by name, symbol, abbreviation,

or code number.

[Nonretroactive as of January 1, 1986.]

(Added 1985) (Amended 1997)

S.1.6.8. Lubricant Devices, Travel of Indicator. -
3.30. Liquid-Measuring Devices



S.1.7. Operating Requirements, Wholesale Devices

Only.

S.1.7.1. Travel of Indicator. - A wholesale device

shall be readily operable to deliver accurately any

quantity from 200 L (50 gal) to the capacity of the

device. If the most sensitive element of the indicating

system utilizes an indicator and graduations, the

relative movement of these parts corresponding to a

delivery of 4 L (1 gal) shall be not less than 5 mm

(0.20 in).

(Amended 1987)

S.1.7.2. Money Values-Mathematical Agreement. -

Any digital money-value indication and any recorded

money value on a computing-type device shall be in

mathematical agreement with its associated quantity

indication or representation to within one cent of

money value.

S.2. Measuring Elements.

S.2.1. Vapor Elimination.

(a) A liquid-measuring device shall be equipped with (Amended 1983)

a vapor or air eliminator or other automatic means

to prevent the passage of vapor and air through the

meter.

(b) Vent lines from the air or vapor eliminator shall be

made of metal tubing or other rigid material.

(Amended 1975)

S.2.1.1. Vapor Elimination on Loading Rack (b) adjustable to provide for deliveries within

Metering Systems. tolerances.

(a) A loading rack metering system shall be

equipped with a vapor or air eliminator or

other automatic means to prevent the

passage of vapor and air through the meter

unless the system is designed or

operationally controlled by a method,

approved by the weights and measures

jurisdiction having control over the device,

such that air and/or vapor cannot enter the

system.

(b) Vent lines from the air or vapor eliminator

(if present) shall be made of metal tubing

or other rigid material.

(Added 1994)

S.2.2. Provision for Sealing. - Adequate provision shall

be made for an approved means of security (e.g., data

change audit trail) or physically applying security seals in

such a manner that no adjustment may be made of:

(a) any measurement element, or

(b) any adjustable element for controlling delivery rate

when such rate tends to affect the accuracy of

deliveries.

When applicable, the adjusting mechanism shall be

readily accessible for purposes of affixing a security seal.

(c) Audit trails shall use the format set forth in Table

S.2.2. [Nonretroactive and enforceable as of

January 1, 1995.]

(Amended 1991, 1993, and 1995)

S.2.3. Directional Flow Valves. - Valves intended to

prevent reversal of flow shall be automatic in operation.

S.2.4. Stop Mechanism.

S.2.4.1. Indication. - The delivery for which the

device is set shall be conspicuously indicated.

S.2.4.2. Stroke Limiting Elements. - Stops or other

stroke limiting elements subject to direct pressure or

impact shall be: 

(a) made secure by positive, nonfrictional engagement of these elements; and

(Amended 1983) 

S.2.4.3. Setting. - If two or more stops or other

elements may be selectively brought into operation to

permit predetermined quantities of deliveries,

(a) the position for the proper setting of each such

element shall be accurately defined; and

(b) any inadvertent displacement from the proper

setting shall be obstructed.

(Amended 1983)
3.30. Liquid-Measuring Devices



6CDNG5%CVGIQTKGUQH&GXKEGCPF/GVJQFUQH5GCNKPI

%CVGIQTKGUQH&GXKEG /GVJQFQH5GCNKPI

%CVGIQT[0QTGOQVGEQPHKIWTCVKQP

ECRCDKNKV[

5GCND[RJ[UKECNUGCNQTVYQGXGPVEQWPVGTUQPGHQTECNKDTCVKQP

RCTCOGVGTUCPFQPGHQTEQPHKIWTCVKQPRCTCOGVGTU

Category 2: Remote configuration capability,

but access is controlled by physical hardware. 

Device shall clearly indicate that it is in the

remote configuration mode and record such

message if capable of printing in this mode. 

[Category 2 applies only to devices

manufactured prior to January 1, 2005. 

Devices with remote configuration capability

manufactured after that date must meet the

sealing requirements outlined in Category 3. 

Devices without remote configuration

capability manufactured after that date will be

required to meet the minimum criteria outlined

in Category 1.]

=6JGJCTFYCTGGPCDNKPICEEGUUHQTTGOQVGEQOOWPKECVKQPOWUV

DGQPUKVG6JGJCTFYCTGOWUVDGUGCNGFWUKPICRJ[UKECNUGCN

QTCPGXGPVEQWPVGTHQTECNKDTCVKQPRCTCOGVGTUCPFCPGXGPV

EQWPVGTHQTEQPHKIWTCVKQPRCTCOGVGTU6JGGXGPVEQWPVGTUOC[

DGNQECVGFGKVJGTCVVJGKPFKXKFWCNOGCUWTKPIFGXKEGQTCVVJG

U[UVGOEQPVTQNNGTJQYGXGTCPCFGSWCVGPWODGTQHEQWPVGTU

OWUVDGRTQXKFGFVQOQPKVQTVJGECNKDTCVKQPCPFEQPHKIWTCVKQP

RCTCOGVGTUQHVJGKPFKXKFWCNFGXKEGUCVCNQECVKQP+HVJG

EQWPVGTUCTGNQECVGFKPVJGU[UVGOEQPVTQNNGTTCVJGTVJCPCVVJG

KPFKXKFWCNFGXKEGOGCPUOWUVDGRTQXKFGFVQIGPGTCVGCJCTF

EQR[QHVJGKPHQTOCVKQPVJTQWIJCPQPUKVGFGXKEG?

=0QPTGVTQCEVKXGCUQH,CPWCT[?

%CVGIQT[4GOQVGEQPHKIWTCVKQPECRCDKNKV[

CEEGUUOC[DGWPNKOKVGFQTEQPVTQNNGFVJTQWIJ

CUQHVYCTGUYKVEJGIRCUUYQTF

6JGFGXKEGUJCNNENGCTN[KPFKECVGVJCVKVKUKPVJG

TGOQVGEQPHKIWTCVKQPOQFGCPFTGEQTFUWEJ

OGUUCIGKHECRCDNGQHRTKPVKPIKPVJKUOQFGQT

UJCNNPQVQRGTCVGYJKNGKPVJKUOQFG

=0QPTGVTQCEVKXGCUQH,CPWCT[?

0QPTGVTQCEVKXGCUQH,CPWCT[CNN

FGXKEGUYKVJTGOQVGEQPHKIWTCVKQPECRCDKNKV[

OWUVEQORN[YKVJVJGUGCNKPITGSWKTGOGPVUQH

%CVGIQT[

#PGXGPVNQIIGTKUTGSWKTGFKPVJGFGXKEGKVOWUVKPENWFGCP

GXGPVEQWPVGTVQVJGRCTCOGVGT+&VJGFCVGCPFVKOG

QHVJGEJCPIGCPFVJGPGYXCNWGQHVJGRCTCOGVGT#RTKPVGF

EQR[QHVJGKPHQTOCVKQPOWUVDGCXCKNCDNGVJTQWIJVJGFGXKEGQT

VJTQWIJCPQVJGTQPUKVGFGXKEG6JGGXGPVNQIIGTUJCNNJCXGC

ECRCEKV[VQTGVCKPTGEQTFUGSWCNVQVGPVKOGUVJGPWODGTQH

UGCNCDNGRCTCOGVGTUKPVJGFGXKEGDWVPQVOQTGVJCP

TGEQTFUCTGTGSWKTGF0QVG&QGUPQVTGSWKTGEJCPIGU

VQDGUVQTGFHQTGCEJRCTCOGVGT

=0QPTGVTQCEVKXGCUQH,CPWCT[?6CDNG#FFGF#OGPFGFCPF
3.30. Liquid-Measuring Devices



S.2.5. Zero-Set-Back Interlock, Retail Motor-Fuel S.2.7.3. Provision for Sealing Automatic TemDevices. - A device shall be constructed so that: perature Compensating Systems. - Provision shall

(a) after a delivery cycle has been completed by that an automatic temperature-compensating system

moving the starting lever to any position that shuts cannot be disconnected and that no adjustment may be

off the device, an automatic interlock prevents a made to the system without breaking the seal.

subsequent delivery until the indicating elements,

and recording elements if the device is equipped

and activated to record, have been returned to their

zero positions;

(b) the discharge nozzle cannot be returned to its designed

hanging position (that is, any position where the tip of

the nozzle is placed in its designed receptacle and the

lock can be inserted) until the starting lever is in its

designed shut-off position and the zero-set-back

interlock has been engaged; and

(c) in a system with more than one dispenser supplied

by a single pump, an effective automatic control

valve in each dispenser prevents product from

being delivered until the indicating elements on that

dispenser are in a correct zero position.

(Amended 1981 and 1985)

S.2.6. Temperature Determination and Wholesale

Devices. - For test purposes, means shall be provided to

determine the temperature of the liquid either:

(a) in the liquid chamber of the meter, or 

(b) immediately adjacent to the meter in the meter inlet

or discharge line.

[Nonretroactive as of January 1, 1985.]

(Added 1984)(Amended 1986)

S.2.7. Wholesale Devices Equipped with Automatic

Temperature Compensators.

S.2.7.1. Automatic Temperature Compensation. -

A device may be equipped with an automatic means

for adjusting the indication and registration of the

measured volume of product to the volume at 15 EC

(60 EF).

S.2.7.2. Provision for Deactivating. - On a device

equipped with an automatic temperature-compensating

mechanism that will indicate or record only in terms

of gallons compensated to 15 EC (60 EF), provision

shall be made for deactivating the automatic temperature-compensating mechanism so that the meter can

indicate, and record if it is equipped to record, in

terms of the uncompensated volume.

(Amended 1972)

be made for applying security seals in such a manner

S.2.7.4. Temperature Determination with Automatic Temperature Compensation. - For test

purposes, means shall be provided (e.g., thermometer

well) to determine the temperature of the liquid either:

(a) in the liquid chamber of the meter, or 

(b) immediately adjacent to the meter in the meter

inlet or discharge line.

(Amended 1987)

S.2.8. Exhaustion of Supply, Lubricant Devices Other than Meter Types. - When the level of the supply of

lubricant becomes so low as to compromise the accuracy

of measurement, the device shall:

(a) become inoperable automatically, or

(b) give a conspicuous and distinct warning. 

S.3. Discharge Lines and Valves.

5&KXGTUKQPQH/GCUWTGF.KSWKF0QOGCPU

UJCNNDGRTQXKFGFD[YJKEJCP[OGCUWTGFNKSWKFECPDG

FKXGTVGFHTQOVJGOGCUWTKPIEJCODGTQHVJGOGVGTQT

KVUFKUEJCTIGNKPG6YQQTOQTGFGNKXGT[QWVNGVUOC[

DGKPUVCNNGFQPN[KHCWVQOCVKEOGCPUCTGRTQXKFGFVQ

GPUWTGVJCV


C NKSWKFECPHNQYHTQOQPN[QPGQWVNGVCVCVKOGCPF


D VJGFKTGEVKQPQHHNQYHQTYJKEJVJGOGEJCPKUO

OC[DGUGVCVCP[VKOGKUENGCTN[CPF

EQPURKEWQWUN[KPFKECVGF

#OCPWCNN[EQPVTQNNGFQWVNGVVJCVOC[DGQRGPGFHQT

RWTIKPIQTFTCKPKPIVJGOGCUWTKPIU[UVGOQTHQT

TGEKTEWNCVKPIRTQFWEVKPUWURGPUKQPUJCNNDGRGTOKVVGF

QPN[YJGPVJGU[UVGOKUOGCUWTKPIHQQFRTQFWEVUQT

CITKEJGOKECNU'HHGEVKXGOGCPUUJCNNDGRTQXKFGFVQ

RTGXGPVRCUUCIGQHNKSWKFVJTQWIJCP[UWEJQWVNGV

FWTKPIPQTOCNQRGTCVKQPQHVJGOGCUWTKPIU[UVGOCPFVQ

KPJKDKVOGVGTKPFKECVKQPU
QTCFXCPEGOGPVQH

KPFKECVKQPUCPFTGEQTFGFTGRTGUGPVCVKQPUYJKNGVJG

QWVNGVKUKPQRGTCVKQP


#OGPFGFCPF
3.30. Liquid-Measuring Devices



S.3.2. Exceptions. - The provisions of S.3.1. Diversion

Prohibited shall not apply to:

(a) truck refueling devices when diversion of flow to S.3.7. Antidrain Means. - In a wet-hose pressureother than the receiving vehicle cannot readily be type device, means shall be incorporated to prevent

accomplished and is readily apparent. Allowable the drainage of the discharge hose.

deterrents include, but are not limited to, physical (Amended 1990)

barriers to adjacent driveways, visible valves, or

lighting systems that indicate which outlets are in

operation, and explanatory signs;

(b) other devices, when all discharge outlets designed

to operate simultaneously are 3.8 cm (1.5 in) in

diameter or larger.

(Amended 1982, 1990, and 1991)

S.3.3. Pump-Discharge Unit. - A pump-discharge unit (b) under specific installation or operating conditions;

equipped with a flexible discharge hose shall be of the or

wet-hose type.

S.3.4. Gravity-Discharge Unit. - On a gravity-discharge unit:

(a) the discharge hose or equivalent pipe shall be of the pressure, the air pressure gauge shall show by special

dry-hose type with no shutoff valve at its outlet end graduations or other means the maximum and minimum

unless the hose or pipe drains to the same level working pressures recommended by the manufacturer.

under all conditions of use;

(b) the dry hose shall be sufficiently stiff and only as

long as necessary to facilitate drainage;

(c) an automatic vacuum breaker, or equivalent

mechanism, shall be incorporated to prevent

siphoning and to ensure rapid and complete drainage; and

(d) the inlet end of the hose or outlet pipe shall be high

enough to ensure complete drainage.

S.3.5. Discharge Hose, Reinforcement. - A discharge

hose shall be reinforced so that the performance of the

device is not affected by the expansion or contraction of

the hose.

S.3.6. Discharge Valve. - A discharge valve may be

installed in the discharge line only if the device is of the

wet-hose type. Any other shutoff valve on the discharge

side of the meter shall be of the automatic or semiautomatic predetermined-stop type or shall be operable

only:

(a) by means of a tool (but not a pin) entirely separate

from the device, or

(b) by mutilation of a security seal with which the

valve is sealed open.

S.4. Marking Requirements.

S.4.1. Limitation on Use. - The limitations on its use

shall be clearly and permanently marked on any device

intended to measure accurately only:

(a) products having particular properties; or

(c) when used in conjunction with specific accessory

equipment.

S.4.2. Air Pressure. - If a device is operated by air

S.4.3. Wholesale Devices.

S.4.3.1. Discharge Rates. - A wholesale device shall

be marked to show its designed maximum and

minimum discharge rates. However, the minimum

discharge rate shall not exceed 20 percent of the maximum discharge rate. 

S.4.3.2. Temperature Compensation. - If a device

is equipped with an automatic temperature compensator, the primary indicating elements, recording

elements, and recorded representation shall be clearly

and conspicuously marked to show that the volume

delivered has been adjusted to the volume at 15 EC

(60 EF).

S.4.4. Retail Devices. - On a retail device with a designed maximum discharge rate of 100 L (25 gal) per

minute or greater, the maximum and minimum

discharge rates shall be marked on an exterior surface

of the device and shall be visible after installation. The

minimum discharge rate shall not exceed 20 percent of

the maximum discharge rate.

[Nonretroactive as of January 1, 1985.]

(Added 1984)

S.5. Totalizers for Retail Motor-Fuel Dispensers. - Retail

motor-fuel dispensers shall be equipped with a
3.30. Liquid-Measuring Devices



nonresettable totalizer for the quantity delivered through

the metering device. [Nonretroactive as of January 1,

1995.]

(Added 1993; Amended 1994)

N. Notes

N.1. Test Liquid.

N.1.1. Type of Liquid. - The liquid used for testing a

liquid-measuring device shall be the type the device is

used to measure, or another liquid with the same general

physical characteristics.

N.1.2. Labeling. - Following the completion of a

successful examination of a wholesale device, the

weights and measures official should attach a label or tag

indicating the type of liquid used during the test.

N.2. Volume Change. - Care shall be taken to minimize

changes in volume of the test liquid due to temperature

changes and evaporation losses.

N.3. Test Drafts.

N.3.1. Retail Piston-Type and Visible-Type Devices. -

Test drafts shall include the full capacity delivery and

each intermediate delivery for which the device is

designed.

N.3.2. Slow Flow Meters. - Test drafts shall be equal to

at least four times the minimum volume that can be

measured and indicated through either a visible

indication or an audible signal.

N.3.3. Lubricant Devices. - Test drafts shall be 1 L

(1 qt). Additional test drafts may include 0.5 L (1 pt), 4 L

(4 qt), and 6 L (6 qt).

N.3.4. Other Retail Devices. - On devices with a

designed maximum discharge rate of:

(a) less than 80 L (20 gal) per minute, tests shall

include drafts of one or more amounts, including a

draft of at least 19 L (5 gal).

(b) 80 L (20 gal) per minute or greater, tests shall

include drafts of one or more amounts, including a

draft of at least the amount delivered by the device

in one minute at the maximum flow rate of the

installation. (Amended 1984)

09JQNGUCNG&GXKEGU6JGFGNKXGTGFSWCPVKV[

UJQWNFDGGSWCNVQCVNGCUVVJGCOQWPVFGNKXGTGFD[

VJGFGXKEGKPQPGOKPWVGCVKVUOCZKOWOFKUEJCTIG

TCVGCPFUJCNNKPPQECUGDGNGUUVJCP.ICN


#OGPFGFCPF

N.4. Testing Procedures.

N.4.1. Normal Tests. - The "normal" test of a device

shall be made at the maximum discharge flow rate

developed under the conditions of installation. Any

additional tests conducted at flow rates down to and

including one-half of the sum of the maximum discharge

flow rate and the rated minimum discharge flow rate

shall be considered normal tests.

(Amended 1991)

N.4.1.1. Wholesale Devices Equipped with Automatic Temperature-Compensating Systems. - On

wholesale devices equipped with automatic temperature compensating systems, normal tests shall be

conducted:

(a) by comparing the compensated volume indicated

or recorded to the actual delivered volume

corrected to 15 EC (60 EF); and 

(b) with the temperature compensating system

deactivated, comparing the uncompensated

volume indicated or recorded to the actual

delivered volume.

The first test shall be performed with the automatic

temperature-compensating system operating in the "as

found" condition.

On devices that indicate or record both the compensated and uncompensated volume for each

delivery, the tests in (a) and (b) may be performed as

a single test.

(Amended 1987)

N.4.1.2. Repeatability Tests. - Tests for repeatability

should include a minimum of three consecutive test

drafts of approximately the same size and be

conducted under controlled conditions where

variations in factors, such as temperature, pressure,

and flow rate are reduced to the extent that they will

not affect the results obtained.

(Added 2001)
3.30. Liquid-Measuring Devices



N.4.2. Special Tests. - "Special" tests, to develop the

operating characteristics of a liquid-measuring device

and any special elements and accessories attached to or

associated with the device, shall be made as N.4.2.4. Wholesale Devices. - "Special" tests shall be

circumstances require. Any test except as set forth in made to develop the operating characteristics of a

N.4.1. shall be considered a special test. measuring system and any special associated or

N.4.2.1. Slow-Flow Meters. - A "special" test shall

be made at a flow rate:

(a) not larger than twice the actual minimum flow

rate, and 

(b) not smaller than the actual minimum flow rate of

the installation.

N.4.2.2. Retail Motor-Fuel Devices.

(a) Devices with a flow-rate capacity less than in the laboratory:

100 L (25 gal) per minute shall have a "special"

test performed at the slower of the following (a) compliance with paragraph S.1.6.5., Moneyrates: Value Computations, shall be determined by

(1) 19 L (5 gal) per minute, or quantity delivered;

(2) the minimum discharge rate marked on the (b) the indicated quantity shall agree with the cone

device, or gear representation with the index of the

(3) the minimum discharge rate at which the

device will deliver when equipped with an (c) the maximum allowable variation of the

automatic discharge nozzle set at its indicated sales price shall be as shown in

slowest setting. Table 1. (Amended 1984)

(b) Devices marked with a flow-rate capacity 100 L

(25 gal) or more per minute, shall have a

"special" test performed at the slowest of the

following rates:

(1) the minimum discharge rate marked on the

device, or

(2) the minimum discharge rate at which the

device will deliver when equipped with an

automatic discharge nozzle set at its slowest

setting.

(Added 1984)

N.4.2.3. Other Retail Devices. - "Special" tests of

other retail devices shall be made at the slower of the

following rates:

(a) 50 percent of the maximum discharge rate

developed under the conditions of installation, or

(b) the minimum discharge rate marked on the

device.

attached elements and accessories. "Special" tests

shall include a test at the slower of the following rates:

(a) 20 percent of the marked maximum discharge

rate; or

(b) the minimum discharge rate marked on the

device.

N.4.3. Money-Value Computation Tests.

N.4.3.1. Laboratory Tests. - When testing the device

using the cone gear as a reference for the total

indicator within the width of the graduation; and

N.4.3.2. Field Tests. - In the conduct of field tests to

determine compliance with paragraph S.1.6.5., the 

maximum allowable variation in the indicated sales

price shall be as shown in Table 1.

(Added 1982; Amended 1984)

N.5. Temperature Correction on Wholesale Devices. -

Corrections shall be made for any changes in volume

resulting from the differences in liquid temperatures

between time of passage through the meter and time of

volumetric determination in the prover. When adjustments

are necessary, appropriate petroleum measurement tables

should be used.

(Amended 1974)

T. Tolerances

T.1. Application to Underregistration and to Overregistration. - The tolerances hereinafter prescribed shall be

applied to errors of underregistration and errors of

overregistration, whether or not a device is equipped with

an automatic temperature compensator.
3.30. Liquid-Measuring Devices



Table 2.

Tolerances for Slow-Flow Meters

Indication

Normal tests Special tests

Maintenance

tolerance

Acceptance

tolerance

Maintenance and

acceptance

tolerance

percent minims percent minims percent minims

1 gill 1.0 20 0.75 15 1.25 25

0.05 gallon 1.0 30 0.75 25 1.25 40

1/2 pint 1.0 40 0.75 30 1.25 50

0.10 gallon 1.0 60 0.75 45 1.25 75

1 pint 1.0 75 0.75 60 1.25 95

0.20 gallon 1.0 120 0.75 90 1.25 155

fl drams fl drams fl drams

1 quart 1.0 2-1/2 0.75 2 1.25 3

1/2 gallon 0.75 4 0.60 3 1.0 5

1 gallon and

over

0.75 8 per

gallon

0.60 6 per

gallon

1.0 10 per

gallon

T.2. Tolerance Values.

T.2.1. Retail Devices Except Slow-Flow Meters.

T.2.1.1. Devices Indicating in Metric Units.

(a) The maintenance tolerance on normal and tests shall be 1/2 in plus 1/2 in per indicated

special tests, shall be 20 mL, plus 4 mL per gallon and never less than 1 in .

indicated liter, and never less than 40 mL. (Amended 1981 and 1986)

(b) The acceptance tolerance on normal and special

tests shall be 10 mL, plus 2 mL per indicated

liter and never less than 20 mL.

(c) The tolerance applied to a 19-liter draft shall be

that tolerance applicable to a 20-liter draft.

(Amended 1981 and 1986)

T.2.1.2. Devices Indicating in Inch-Pound Units.

(a) The maintenance tolerance on normal and

special tests shall be 1 in plus 1 in per indicated 3 3

gallon and never less than 2 in .3

(b) The acceptance tolerance on normal and special 3 3

3

T.2.1.3. Repeatability. - When multiple tests are

conducted at approximately the same flow rate and

draft size, the range of the test results for the flow rate

shall not exceed 40 percent of the absolute value of

the maintenance tolerance and the results of each test

shall be within the applicable tolerance. See also

N.4.1.2.

(Amended 2001)
3.30. Liquid-Measuring Devices



T.2.1.4. Tolerances for Devices Designed to

Primarily Deliver Less than One Gallon. -

Maintenance tolerances and acceptance tolerances

shall be as shown in Table 2. Tolerances for

Slow-Flow Meters.

(Added 1999)

T.2.2. Slow-Flow Meters. - Maintenance tolerances and

acceptance tolerances shall be as shown in Table 2.

T.2.3. Wholesale Devices.

T.2.3.1. Measurement of Agri-Chemical Liquids. -

 Acceptance tolerances and maintenance tolerances

shall be 0.3 percent and 0.5 percent, respectively.

(Amended 1995)

T.2.3.2. Measurement of Asphalt. - Maintenance

tolerances and acceptance tolerances shall be:

Acceptance Maintenance Special Test

Asphalt 0.2 % 0.3 %

below at

50 °C or

below

0.5 % UR.1.1.1. Length. - The length of the discharge hose

Asphalt 0.3 % 0.3 % 0.5 %

above

50 °C

(Amended 2001)

T.2.3.3. Measurement of Other Liquids. -

Maintenance tolerances and acceptance tolerances

shall be:

#EEGRVCPEG/CKPVGPCPEG

0QTOCNVGUV  

5RGEKCNVGUV  

T.2.3.4. Repeatability. - When multiple tests are

conducted at approximately the same flow rate and

draft size, the range of the test results for the flow rate

shall not exceed 40 percent of the absolute value of

the maintenance tolerance and the results of each test

shall be within the applicable tolerance. This

tolerance does not apply to the test of the automatic

temperature compensating system. See also N.4.1.2.

(Added 1992) (Amended 2001)

T.2.3.5. Automatic Temperature Compensating

Systems. - The difference between the meter error

(expressed as a percentage) for results determined

with and without the automatic temperature

compensating system activated shall not exceed:

(a) 0.2 percent for mechanical automatic

temperature compensating systems; and

(b) 0.1 percent for electronic automatic temperature

compensating systems.

The delivered quantities for each test shall be

approximately the same size. The results of each test

shall be within the applicable acceptance or

maintenance tolerance.

[Nonretroactive as of January 1, 1988.]

(Added 1987) (Amended 1992 and 1996)

UR. User Requirements

UR.1. Selection Requirements.

UR.1.1. Discharge Hose.

on a retail motor-fuel device:

(a) shall be measured from its housing or outlet of

the discharge line to the inlet of the discharge

nozzle;

(b) shall be measured with the hose fully extended if

it is coiled or otherwise retained or connected

inside a housing; and

(c) shall not exceed 5.5 m (18 ft) unless it can be

demonstrated that a longer hose is essential to

permit deliveries to be made to receiving

vehicles or vessels.

An unnecessarily remote location of a device shall not

be accepted as justification for an abnormally long

hose.

(Amended 1972 and 1987)

UR.1.1.2. Marinas and Airports.

UR.1.1.2.1. Length. - The length of the discharge

hose shall be as short as practicable, and shall not

exceed 15 m (50 ft) unless it can be demonstrated

that a longer hose is essential.
3.30. Liquid-Measuring Devices



UR.1.1.2.2. Protection. - Discharge hoses

exceeding 8 m (26 ft) in length shall be adequately

protected from weather and other environmental

factors when not in use.

(Made retroactive 1974 and amended 1984)

UR.2. Installation Requirements.

UR.2.1. Manufacturer’s Instructions. - A device shall

be installed in accordance with the manufacturer’s

instructions, and the installation shall be sufficiently

secure and rigid to maintain this condition.

(Added 1987)

UR.2.2. Discharge Rate. - A device shall be installed

so that the actual maximum discharge rate will not

exceed the rated maximum discharge rate. Automatic

means for flow regulation shall be incorporated in the

installation if necessary.

UR.2.3. Suction Head. - A piston-type device shall be

installed so that the total effective suction head will not

be great enough to cause vaporization of the liquid being

dispensed under the highest temperature and lowest

barometric pressure likely to occur.

UR.2.4. Diversion of Liquid Flow. - A motor-fuel

device equipped with two delivery outlets used exclusively in the fueling of trucks shall be so installed that

any diversion of flow to other than the receiving vehicle

cannot be readily accomplished and is readily apparent.

Allowable deterrents include, but are not limited to,

physical barriers to adjacent driveways, visible valves, or

lighting systems that indicate which outlets are in

operation, and explanatory signs.

(Amended 1991)

UR.2.5. Product Storage Identification.

(a) The fill connection for any petroleum product

storage tank or vessel supplying motor-fuel devices

shall be permanently, plainly, and visibly marked as

to product contained.

(b) When the fill connection device is marked by

means of a color code, the color code key shall be

conspicuously displayed at the place of business.

(Added 1975 and Amended 1976)

UR.3. Use of Device.

UR.3.1. Return of Indicating and Recording Elements to Zero. - On any dispenser used in making retail

deliveries, the primary indicating element, and recording

element if so equipped, shall be returned to zero before

each delivery.

Exceptions to this requirement are totalizers on key-lockoperated or other self-operated dispensers and the

primary recording element if the device is equipped to

record.

UR.3.2. Unit Price and Product Identity.

(a) The following information shall be conspicuously

displayed or posted on the face of a retail dispenser

used in direct sale:

(1) except for dispensers used exclusively for fleet

sales, other price contract sales, and truck

refueling (e.g., truck stop dispensers used only to

refuel trucks), all of the unit prices at which the

product is offered for sale; and

(2) in the case of a computing type or moneyoperated type, the unit price at which the dispenser is set to compute.

Provided that the dispenser complies with

S.1.6.4.1., it is not necessary that all the unit prices

for all grades, brands, blends, or mixtures be

simultaneously displayed or posted.

(b) The following information shall be conspicuously

displayed or posted on each side of a retail

dispenser used in direct sale:

(1) the identity of the product in descriptive

commercial terms, and

(2) the identity of the grade, brand, blend, or

mixture that a multi-product dispenser is set

to deliver.

(Amended 1972, 1983, 1987, 1989, 1992, and 1993)

UR.3.3. Computing Device.

(a) Any computing device used in an application where

a product or grade is offered for sale at more than

one unit price (excluding fleet sales and other price

contract sales), shall be used only for sales for

which the device computes and displays the sales

price for the selected transaction. 

(Added 1989) (Amended 1992)

(Became Retroactive 1999)

(b) A truck stop dispenser used exclusively for

refueling trucks is exempt from the requirements in

(a) if all purchases of fuel are accompanied by a
3.30. Liquid-Measuring Devices



printed receipt of the transaction containing the temperature compensator shall show that the

applicable price per gallon, the total gallons volume delivered has been adjusted to the

delivered, and the total price of the sale. volume at 15 EC (60 EF).

(Added 1993)

(c) Unless a truck stop dispenser used exclusively for wholesale device equipped with an automatic

refueling trucks complies with S.1.6.4.1. (Display temperature compensating system shall also

of Unit Price), the price posted on the dispenser indicate: (1) the API gravity, specific gravity

and the price at which the dispenser is set to or coefficient of expansion for the product;

compute shall be the highest price for any (2) product temperature; and (3) gross

transaction which may be conducted. reading.

(Added 1993) (Amended 1987)

UR.3.4. Printed Ticket. - The total price, the total UR.3.6.2. Nonautomatic.

volume of the delivery, and the price per gallon or liter

shall be shown, either printed or in clear hand script, on UR.3.6.2.1. Temperature Determination. - If the

any printed ticket issued by a device and containing any volume of the product delivered is adjusted to the

one of these values. volume at 15 EC (60 EF), the product temperature

(Amended 2001) shall be taken during the delivery in:

UR.3.5. Steps After Dispensing. - After delivery to a

customer from a retail motor-fuel device:

(a) the starting lever shall be returned to its shutoff

position and the zero-set-back interlock engaged;

and

(b) the discharge nozzle shall be returned to its

designed hanging position unless the primary UR.3.6.2.2. Invoices. - The accompanying invoice

indicating elements, and recording elements, if the shall indicate that the volume of the product has

device is equipped and activated to record, have been adjusted for temperature variations to a

been returned to a definite zero indication. volume at 15 EC (60 EF) and shall also state the

UR.3.6. Temperature Compensation, Wholesale.

UR.3.6.1. Automatic.

UR.3.6.1.1. When to be Used. - If a device is

equipped with a mechanical automatic temperature

compensator, it shall be connected, operable, and in

use at all times. An electronic or mechanical

automatic temperature compensating system may

not be removed, nor may a compensated device be

replaced with an uncompensated device, without

the written approval of the responsible weights and

measures jurisdiction.

[Note: This requirement does not specify the

method of sale for product measured through a

meter.]

(Amended 1989)

UR.3.6.1.2. Invoices.

(a) A written invoice based on a reading of a

device that is equipped with an automatic

(b) The invoice issued from an electronic

(a) the liquid chamber of the meter, or

(b) the meter inlet or discharge line adjacent to

the meter, or 

(c) the compartment of the receiving vehicle at

the time it is loaded.

product temperature used in making the adjustment.
3.30. Liquid-Measuring Devices



Liquid-Measuring Device Code Index

Acceptance ............................ 3-13, 3-14

Adjustable element . . . . . . . . . . . . . . . . . . . . . . . . . . 3-7

Advancement ............................ 3-3, 3-9

Agreement .......................... 3-5, 3-6, 3-7

Agri-chemical .......................... 3-3, 3-14

Agri-chemical liquids .................... 3-3, 3-14

Antidrain means ............................ 3-10

Automatic . . 3-7, 3-9, 3-10, 3-11, 3-12, 3-13, 3-14, 3-15,

3-16 Retail devices ......... 3-3, 3-4, 3-10, 3-11, 3-12, 3-13

Automatic temperature compensating systems ..... 3-9,

3-11, 3-14 Retail motor-fuel dispensers .................. 3-11

Automatic temperature compensation . . . . . . . . . . . . 3-9

Automatic temperature compensators . . . . . . . . . . . . 3-9

Auxiliary elements . . . . . . . . . . . . . . . . . . . . . . . . . . . 3-5

Blend . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 3-5, 3-15

Clear interval . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 3-4

Clearance . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 3-4

Computing device . . . . . . . . . . . . . . . . . . . . . . . 3-5, 3-15

Directional flow valves . . . . . . . . . . . . . . . . . . . . . . . 3-7

Discharge hose . . . . . . . . . . . . . . . . . . . . . . . . . 3-10, 3-14

Discharge lines . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 3-9

Discharge rate . . . . . . . . . . . . . . . 3-10, 3-11, 3-12, 3-15

Discharge rates ............................. 3-10

Diversion of liquid . . . . . . . . . . . . . . . . . . . . . . . . . . 3-15

Dry hose . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 3-10

Fleet sales .......................... 3-5, 3-6, 3-15

Grade . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 3-5, 3-15

Graduations . . . . . . . . . . . . . . . . . . . . 3-4, 3-6, 3-7, 3-10

Gravity-discharge unit ....................... 3-10

Index of an indicator . . . . . . . . . . . . . . . . . . . . . . . . . 3-4

Indicating element ................... 3-3, 3-5, 3-15

Indicators . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 3-4

Installation requirements ..................... 3-15

Interlock . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 3-9, 3-16

Invoices .................................. 3-16

Key-lock . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 3-5, 3-15

Lubricant devices .................... 3-6, 3-9, 3-11

Maintenance . . . . . . . . . . . . . . . . . . . . . . . . . . . 3-13, 3-14

Marking requirements ....................... 3-10

Mass . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 3-1, 3-3

Mass flow meters ......................... 3-1, 3-3

Measuring elements . . . . . . . . . . . . . . . . . . . . . . . . . . 3-7

Money values-mathematical agreement . . . . . . . . . . 3-7

Money-value computation .................... 3-12

Money-value computations . . . . . . . . . . . . . . . . . . . . 3-5

Money-value divisions . . . . . . . . . . . . . . . . . . . . . 3-5, 3-6

Motor-fuel device ................. 3-14, 3-15, 3-16

Nonautomatic .............................. 3-16

Normal tests ........................... 3-11, 3-13

Operating requirements .................... 3-4, 3-7

Parallax . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 3-4

Power loss . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 3-4

Price contract sales .................. 3-5, 3-6, 3-15

Primary indicating element ................ 3-3, 3-15

Printed ticket .............................. 3-16

Product identity ..................... 3-5, 3-6, 3-15

Provision for sealing ...................... 3-7, 3-9

Pump-discharge unit ........................ 3-10

Quantity ............... 3-4, 3-5, 3-6, 3-7, 3-11, 3-12

Recording element ................... 3-3, 3-6, 3-15

Repeatability ..................... 3-11, 3-13, 3-14

Retail motor-fuel device ................. 3-14, 3-16

Return to zero . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 3-4

Seal . . . . . . . . . . . . . . . . . . . . . . . . . . . 3-7, 3-8, 3-9, 3-10

Sealing ............................. 3-7, 3-8, 3-9

Security . . . . . . . . . . . . . . . . . . . . . . . . . . . 3-7, 3-9, 3-10

Selection of unit price . . . . . . . . . . . . . . . . . . . . . . . . 3-5

Slow flow meters ........................ 3-4, 3-11

Special tests ........................... 3-12, 3-13

Stop mechanism . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 3-7

Temperature correction ...................... 3-12

Temperature determination ................ 3-9, 3-16

Test drafts . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . 3-11

Tolerances . . . . . . . . . . . . . . . . . . . 3-7, 3-12, 3-13, 3-14

Total price ......................... 3-5, 3-6, 3-16

Totalizers ............................. 3-11, 3-15

Travel of indicator ........................ 3-6, 3-7

Truck refueling .................... 3-5, 3-10, 3-15

Unit price . . . . . . . . . . . . . . . . . . 3-4, 3-5, 3-6, 3-15, 3-16

User information . . . . . . . . . . . . . . . . . . . . . . . . . . . . 3-4

Valves . . . . . . . . . . . . . . . . . . . . . . . 3-7, 3-9, 3-10, 3-15

Vapor elimination . . . . . . . . . . . . . . . . . . . . . . . . . . . 3-7

Wet-hose ................................. 3-10

Wholesale devices . 3-3, 3-7, 3-9, 3-10, 3-11, 3-12, 3-14

Zero-set-back interlock ................... 3-9, 3-16
3.31. Vehicle-Tank Meters



Sec. 3.31. Vehicle-Tank Meters

A. Application

A.1. - This code applies to meters mounted on vehicle tanks

including those used for the measurement and delivery of

petroleum products or agri-chemical liquids such as

fertilizers, feeds, pesticides, defoliants, and bulk deliveries

of water.

(Amended 1985 and 1995)

A.2. - This code does not apply to the following devices:

(a) Devices used for dispensing liquefied petroleum gases

(for which see Sec. 3.32; Code for Liquefied

Petroleum Gas and Anhydrous Ammonia LiquidMeasuring Devices), or other liquids that do not

remain in a liquid state at atmospheric pressures and

temperatures.

(b) Devices used solely for dispensing a product in

connection with operations in which the amount dispensed does not affect customer charges.

(c) Vehicle tanks used as measures (for which see Sec.

4.40; Code for Vehicle Tanks Used as Measures).

(d) Mass flow meters (see Sec. 3.37. Code for Mass Flow

Meters)

(Added 1994)

A.3. - See also Sec. 1.10; General Code requirements.

S. Specifications

S.1. Design of Indicating and Recording Elements and

of Recorded Representations.

S.1.1. Primary Elements.

S.1.1.1. General. - A meter shall be equipped with a

primary indicating element and may also be equipped

with a primary recording element.

[Note: Except for systems used solely for the sale of

aviation fuel into aircraft and for aircraft-related

operations, vehicle-tank meters shall be equipped with

a primary recording element as required by paragraph

UR.2.2.1.] (Amended 1993)

S.1.1.2. Units. -

(a) A meter shall indicate, and record if the meter is

equipped to record, its deliveries in terms of

liters (gallons). Fractional parts of the liter

(gallon) shall be in terms of either decimal or

binary subdivisions.

(b) When it is an industry practice to purchase and

sell milk by weight based upon 1.03 kg/L

(8.6 lb/gal), the primary indicating element may

indicate in kilograms (pounds) and decimal

kilograms (pounds). The weight value division

shall be a decimal multiple or submultiple of 1,

2, or 5. (See S.5.5.)

S.1.1.3. Value of Smallest Unit. - The value of the

smallest unit of indicated delivery, and recorded

delivery if the meter is equipped to record, shall not

exceed the equivalent of:

(a) 0.5 L (0.1 gal) or 0.5 kg (1 lb) on milk-metering

systems 

(b) 0.5 L (0.1 gal) on meters with a rated maximum

flow rate of 500 L/min (100 gal/min) or less

used for retail deliveries of liquid fuel, or

(c) 5 L (1 gal) on other meters.

(Amended 1989 and 1994)

S.1.1.4. Advancement of Indicating and Recording

Elements. - Primary indicating and recording

elements shall be susceptible to advancement only by

the mechanical operation of the meter. However, a

meter may be cleared by advancing its elements to

zero, but only if:

(a) the advancing movement, once started, cannot be

stopped until zero is reached, or

(b) in the case of indicating elements only, such

elements are automatically obscured until the

elements reach the correct zero position.

S.1.1.5. Return to Zero. - Primary indicating

elements shall be readily returnable to a definite zero

indication. Means shall be provided to prevent the
3.31. Vehicle-Tank Meters



return of primary indicating elements, and of primary When the index of an indicator extends along the

recording elements if these are returnable to zero, entire length of a graduation, that portion of the index

beyond their correct zero position. of the indicator that may be brought into coincidence

S.1.2. Graduations.

S.1.2.1. Length. - Graduations shall be so varied in

length that they may be conveniently read. S.1.3.4. Clearance. - The clearance between the

S.1.2.2. Width. - In any series of graduations, the

width of a graduation shall in no case be greater than

the width of the minimum clear interval between S.1.3.5. Parallax. - Parallax effects shall be reduced

graduations, and the width of main graduations shall to the practicable minimum.

be not more than 50 percent greater than the width of

subordinate graduations. Graduations shall in no case

be less than 0.2 mm (0.008 in) wide.

S.1.2.3. Clear Interval Between Graduations. - these parts corresponding to the smallest indicated

The clear interval shall be not less than 2.5 mm value shall not be less than 5 mm (0.20 in).

(0.10 in). If the graduations are not parallel, the

measurement shall be made:

(a) along the line of relative movement between the

graduations at the end of the indicator, or

(b) if the indicator is continuous, at the point of

widest separation of the graduations.

(Amended 1986)

S.1.3. Indicators.

S.1.3.1. Symmetry. - The index of an indicator shall

be symmetrical with respect to the graduations at least

throughout that portion of its length associated with

the graduations.

S.1.3.2. Length. - The index of an indicator shall

reach to the finest graduations with which it is used,

unless the indicator and the graduations are in the

same plane, in which case the distance between the

end of the indicator and the ends of the graduations,

measured along the line of the graduations, shall be

not more than 1.0 mm (0.04 in).

S.1.3.3. Width. - The width of the index of an and shall be accurately positioned. The value of each

indicator in relation to the series of graduations with graduated interval shall be 1 cent. On electronic

which it is used shall be not greater than: devices with digital indications, the total price may be

(a) the width of the narrowest graduation*, and the value of the smallest division indicated is equal to

[*Nonretroactive as of January 1, 2002] or less than 0.2 L (0.1 gal) or 0.2 kg (1 lb).

(Amended 2001) (Amended 1979, 1989)

(b) the width of the minimum clear interval between

graduations. - Any digital money-value indication and any recorded

with the graduation shall be of the same width

throughout the length of the index that coincides with

the graduation.

index of an indicator and the graduations shall in no

case be more than 1.5 mm (0.06 in).

S.1.3.6. Travel of Indicator. - If the most sensitive

element of the primary indicating element utilizes an

indicator and graduations, the relative movement of

S.1.4. Computing-Type Device.

S.1.4.1. Display of Unit Price. - In a device of the

computing type, means shall be provided for

displaying on the outside of the device, in a manner

clear to the operator and an observer, the unit price at

which the device is set to compute.

(Amended 1983)

S.1.4.2. Printed Ticket. - If a computing-type device

issues a printed ticket which displays the total

computed price, the ticket shall also have printed

clearly thereon the total quantity of the delivery, the

appropriate fraction of the quantity, and the price per

unit of quantity.

(Amended 1989)

S.1.4.3. Money-Value Computations. - Moneyvalue computations shall be of the full-computing type

in which the money value at a single unit price, or at

each of a series of unit prices, shall be computed for

every delivery within either the range of measurement

of the device or the range of the computing elements,

whichever is less. Value graduations shall be supplied

computed on the basis of the quantity indicated when

S.1.4.4. Money Values, Mathematical Agreement.
3.31. Vehicle-Tank Meters



money value on a computing-type device shall be in (b) the direction of flow for which the mechanism may

mathematical agreement with its associated quantity be set at any time is definitely and conspicuously

indication or representation to within 1 cent of money indicated.

value.

S.2. Design of Measuring Elements. unit, the discharge hose shall be of the wet-hose type

S.2.1. Vapor Elimination. - A metering system shall be discharge unit may be equipped also with a dry hose

equipped with an effective vapor or air eliminator or without a shutoff valve at its outlet end, but only if:

other automatic means to prevent the passage of vapor

and air through the meter. Vent lines from the air or (a) the dry hose is as short as practicable, and

vapor eliminator shall be made of metal tubing or some

other suitable rigid material. (b) there is incorporated in the discharge piping,

(Amended 1993) immediately adjacent to the meter, effective means

S.2.2. Provision for Sealing. - Except on devices for

metering milk, adequate provision shall be made for

applying security seals in such a manner that no adjustment may be made of:

(a) any measurement element, and charge unit, the discharge hose or equivalent pipe shall

(b) any adjustable element for controlling delivery rate end. The dry hose shall be of such stiffness and only of

when such rate tends to affect the accuracy of such length as to facilitate its drainage. The inlet end of

deliveries. the hose or of an equivalent outlet pipe shall be of such

S.2.2.1. Milk-Metering Systems. - Adequate

provision shall be made for applying security seals to

the adjustment mechanism and the register.

The adjusting mechanism shall be readily accessible for S.3.4. Discharge Hose. - A discharge hose shall be

purposes of affixing a security seal. adequately reinforced.

S.2.3. Directional Flow Valves. - Valves intended to S.3.5. Discharge Valve. - A discharge valve may be

prevent reversal of flow shall be automatic in operation. installed in the discharge line only if the device is of the

However, on equipment used exclusively for fueling wet-hose type, in which case such valve shall be at the

aircraft, such valves may be manual in operation. discharge end of the line. Any other shutoff valve on the

S.3. Design of Discharge Lines and Discharge Line

Valves.

(Not applicable to milk-metering systems.)

S.3.1. Diversion of Measured Liquid. - Except on

equipment used exclusively for fueling aircraft, no means

shall be provided by which any measured liquid can be

diverted from the measuring chamber of the meter or the

discharge line therefrom. However, two or more

delivery outlets may be installed if means is provided to S.3.6. Antidrain Valve. - In a wet-hose, pressure-type

insure that: device, an effective antidrain valve shall be incorporated

(a) liquid can flow from only one such outlet at one The antidrain valve shall function so as to prevent the

time, and drainage of the discharge hose. However, a device used

S.3.2. Pump-Discharge Unit. - On a pump-discharge

with a shutoff valve at its outlet end. However, a pumpto insure that liquid can flow through only one of

the discharge hoses at any one time and that the

meter and the wet hose remain full of liquid at all

times.

S.3.3. Gravity-Discharge Unit. - On a gravity-disbe of the dry-hose type with no shutoff valve at its outlet

height as to provide for proper drainage of the hose or

pipe. There shall be incorporated an automatic vacuum

breaker or equivalent means to prevent siphoning and to

insure the rapid and complete drainage.

discharge side of the meter shall be of the automatic or

semiautomatic predetermined-stop type or shall be

operable only:

(a) by means of a tool (but not a pin) entirely separate

from the device, or

(b) by mutilation of a security seal with which the

valve is sealed open.

in the discharge valve or immediately adjacent thereto.

exclusively for fueling and defueling aircraft may be of

the pressure type without an antidrain valve.
3.31. Vehicle-Tank Meters



S.4. Design of Intake Lines (for Milk-Metering Systems).

S.4.1. Diversion of Liquid to be Measured. - No

means shall be provided by which any liquid can be

diverted from the supply tank to the receiving tank

without being measured by the device.

S.4.2. Intake Hose. - The intake hose shall be:

(a) of the dry-hose type;

(b) adequately reinforced;

(c) not more than 6 m (20 ft) in length, unless it can be

demonstrated that a longer hose is essential to

permit pickups from a supply tank; and

(d) connected to the pump at horizontal or above, to

permit complete drainage of the hose.

S.5. Marking Requirements

S.5.1. Limitation of Use. - If a meter is intended to

measure accurately only liquids having particular

properties, or to measure accurately only under specific

installation or operating conditions, or to measure

accurately only when used in conjunction with specific

accessory equipment, these limitations shall be clearly

and permanently stated on the meter.

S.5.2. Discharge Rates. - A meter shall be marked to

show its designed maximum and minimum discharge

rates. However, the minimum discharge rate shall not

exceed 20 percent of the maximum discharge rate.

S.5.3. Measuring Components Milk-Metering System. - All components that affect the measurement of

milk that are disassembled for cleaning purposes shall be

clearly and permanently identified with a common serial

number.

S.5.4. Flood Volume, Milk-Metering System. - When

applicable, the volume of product necessary to flood the

system when dry shall be clearly, conspicuously, and

permanently marked on the air eliminator.

S.5.5. Conversion Factor. - When the conversion factor

of 1.03 kg/L (8.6 lb/gal) is used to convert the volume of

milk to weight, the conversion factor shall be clearly

marked on the primary indicating element and recorded

on the delivery ticket.

(Added 1989)

N. Notes

N.1. Test Liquid.

(a) A measuring system shall be tested with the liquid to

be commercially measured or with a liquid of the

same general physical characteristics. Following a

satisfactory examination, the weights and measures

official should attach a seal or tag indicating the

product used during the test.

(Amended 1975)

(b) A milk measuring system shall be tested with the type

of milk to be measured when the accuracy of the

system is affected by the characteristics of milk (e.g.,

positive displacement meters).

(Part (b) Added 1989)

N.2. Evaporation and Volume Change. - Care shall be

exercised to reduce to a minimum, evaporation losses and

volume changes resulting from changes in temperature of

the test liquid.

N.3. Test Drafts. - Test drafts should be equal to at least

the amount delivered by the device in 1 minute at its

maximum discharge rate, and shall in no case be less than

180 L (50 gal) or 225 kg (500 lb).

(Amended 1989)

N.4. Testing Procedures

N.4.1. Normal Tests. - The "normal" test of a measuring system shall be made at the maximum discharge

rate that may be anticipated under the conditions of the

installation. Any additional tests conducted at flow rates

down to and including one-half of the sum of the

maximum discharge flow rate and the rated minimum

discharge flow rate shall be considered normal tests.

(Amended 1992)

N.4.1.1. Milk Measuring System. - The "normal"

test shall include a determination of the effectiveness

of the air elimination system.

N.4.1.2. Repeatability Tests. - Tests for repeatability

should include a minimum of three consecutive test

drafts of approximately the same size and be

conducted under controlled conditions where

variations in factors such as, temperature pressure and

flow rate are reduced to the extent that they will not

affect the results obtained.

(Added 2001)
3.31. Vehicle-Tank Meters



N.4.2. Special Tests (Except Milk-Measuring Systems). - "Special" tests shall be made to develop the

operating characteristics of a measuring system and any

special elements and accessories attached to or

associated with the device. Any test except as set forth

in N.4.1. shall be considered a special test. Special test

of a measuring system shall be made as follows:

(a) at a minimum discharge rate of 20 percent of the

marked maximum discharge rate or at the minimum

discharge rate marked on the device whichever is

less;

(b) to develop operating characteristics of the measuring system during a split-compartment delivery.

(Amended 1978)

N.4.3. Antidrain Valve Test. - The effectiveness of the

antidrain valve shall be tested after the pump pressure in

the measuring system has been released and a valve

between the supply tank and the discharge valve is

closed.

N.4.4. System Capacity. - The test of a milk-measuring

system shall include the verification of the volume of

product necessary to flood the system as marked on the

air eliminator.

T. Tolerances

T.1. Application. are returnable to zero, shall be returned to zero

T.1.1. To Underregistration and to Over- pump has been activated and the product to be measured

registration. - The tolerances hereinafter prescribed has been supplied to the measuring system.

shall be applied to errors of underregistration and errors (Amended 1981)

of overregistration.

T.2. Tolerance Values. - Maintenance and acceptance mounted metering systems shall be equipped with a

tolerances shall be as shown in Tables 1, 2, 3, and 4. ticket printer which shall be used for all sales where

(Amended 1995) product is delivered through the meter. A copy of the

T.3. Tolerance Values on Meters Used for the Measurement of Agri-Chemical Liquids. - The maintenance

tolerance on normal and special tests shall be 1 percent of

the indicated quantity. The acceptance tolerance on a normal test shall be 0.5 percent of the indicated quantity and on

special tests, 1 percent of the indicated quantity.

(Added 1985, Amended 1989)

T.4. Repeatability. - When multiple tests are conducted at

approximately the same flow rate and draft size, the range

of the test results for the flow rate shall not exceed 40

percent absolute value of the maintenance tolerance and the

results of each test shall be within the applicable tolerance.

See also N.4.1.2.

(Added 1992) (Amended 2001)

UR. User Requirements

UR.1. Installation Requirements.

UR.1.1. Discharge Rate. - A meter shall be so installed

that the actual maximum discharge rate will not exceed

the rated maximum discharge rate. If necessary, means

for flow regulation shall be incorporated in the

installation, in which case this shall be fully effective and

automatic in operation.

UR.1.2. Unit Price. - There shall be displayed on the

face of a device of the computing type the unit price at

which the device is set to compute.

UR.1.3. Intake Hose. - The intake hose in a milkmetering system shall be installed to permit complete

drainage and ensure that all available product is measured following each pickup.

UR.2. Use Requirements.

UR.2.1. Return of Indicating and Recording Elements to Zero. - The primary indicating elements

(visual), and the primary recording elements, when these

immediately before each delivery is begun and after the

UR.2.2. Ticket Printer; Customer Ticket. - Vehicleticket issued by the device shall be left with the customer

at the time of delivery or as otherwise specified by the

customer.

(Added 1993) (Amended 1994)
3.31. Vehicle-Tank Meters



UR.2.2.1. Exceptions for the Sale of Aviation

Fuel. - The provisions of UR.2.2. Ticket Printer;

Customer Ticket shall not apply to vehicle-mounted

metering systems used solely for the sale of aviation

fuel into aircraft and for aircraft-related operations.

(Added 1999)

UR.2.3. Ticket in Printing Device. - A ticket shall not

be inserted into a device equipped with a ticket printer

until immediately before a delivery is begun, and in no

case shall a ticket be in the device when the vehicle is in

motion while on a public street, highway, or thoroughfare.

UR.2.4. Credit for Flood Volume. - The volume of

product necessary to flood the system as marked on the

air eliminator shall be individually recorded on the

pickup ticket of each seller affected.

Table 1.

Tolerances for Vehicle-Tank Meters Except for Vehicle-Mounted Milk Meters,

Agri-Chemical Meters, and Water Meters

Normal tests Special tests

Indication Acceptance tolerance Maintenance Maintenance and acceptance

tolerance tolerance

(Gallons) (Cubic inches) (Cubic inches) (Cubic inches)

50 50 25 50

Over 50 Add 1/2 cubic Add 1/4 cubic Add 1 cubic

inch per inch per inch per

indicated indicated indicated

gallon over 50 gallon over 50 gallon over 50
3.31. Vehicle-Tank Meters



Table 2. Tolerances for

Vehicle-Mounted Milk Meters

Indication Maintenance Acceptance

gallons gallons gallons

100 0.5 0.3

200 0.7 0.4

300 0.9 0.5

400 1.1 0.6

500 1.3 0.7

Over 500 Add 0.002 gallon per indicated gallon

over 500

Add 0.001

gallon per

indicated gallon over 500

(Added 1989)
3.31. Vehicle-Tank Meters



Table 3. Tolerances for Vehicle-Mounted Water Meters

Normal Tests

Meter size Rate of flow

(inches) (gal/min)

Maximum Rate

Meter indication Tolerance on over- and

under-registration gal ft3

5/8 15 50 5

3/4 25 50 5

1 40 100 10

1 1/2 80 300 40 1.5%

2 120 500 40

3 250 500 50

4 350 1 000 100

6 700 1 000 100

(Added 1995)

Table 4. Tolerances for Vehicle-Mounted Water Meters

Special Tests

Meter size Tolerance

(inches)

Intermediate rate Minimum rate

Rate of Rate of

flow flow

(gal/ (gal/

min) min)

Meter Meter

indication Tolerance on over- indication

and underregistration gal ft gal ft 3 3 Under- Overregistration registration

5/8 2 10 1 1/4 5 1

3/4 3 10 1 1/2 5 1

1 4 10 1 3/4 5 1

1 1/2 8 50 5 1.5% 1 1/2 10 1 5.0% 1.5%

2 15 50 5 2 10 1

3 20 50 5 4 10 1

4 40 100 10 7 50 5

6 60 100 10 12 50 5

(Added 1995)
3.32. LPG and Anhydrous Ammonia Liquid-Measuring Devices

6KVNGCOGPFGF 



Sec. 3.32. Liquefied Petroleum Gas and Anhydrous Ammonia

Liquid-Measuring Devices1

A. Application.

A.1. - This code applies to devices used for the

measurement of liquefied petroleum gas and anhydrous

ammonia in the liquid state, whether such devices are

installed in a permanent location or mounted on a vehicle.

A.2. - Insofar as they are clearly appropriate, the

requirements and provisions of the code may be applied to

devices used for the measurement of other liquids that do

not remain in a liquid state at atmospheric pressures and

temperatures.

A.3. - See also Sec. 1.10; General Code requirements.

A.4. - This code does not apply to mass flow meters (see

Sec. 3.37. Code for Mass Flow Meters)

(Added 1994)

S. Specifications

S.1. Design of Indicating and Recording Elements and

of Recorded Representations.

S.1.1. Primary Elements

S.1.1.1. General. - A device shall be equipped with

a primary indicating element and may also be

equipped with a primary recording element.

[Note: Vehicle-mounted metering systems shall be

equipped with a primary recording element as required

by paragraph UR.2.6.]

S.1.1.2. Units. - A device shall indicate, and record if

the device is equipped to record, its deliveries in terms

of gallons, quarts, pints, or binary-submultiple or

decimal subdivisions of the gallon.

(Amended 1987)

S.1.1.3. Value of Smallest Unit. - The value of the

smallest unit of indicated delivery, and recorded

delivery if the device is equipped to record, shall not

exceed the equivalent of:

(a) 0.5 L (1 pt) on retail devices, or

(b) 5 L (1 gal) on wholesale devices.

(Amended 1987)

S.1.1.4. Advancement of Indicating and Recording

Elements. - Primary indicating and recording

elements shall be susceptible to advancement only by

the mechanical operation of the device. However, a

device may be cleared by advancing its elements to

zero, but only if:

(a) the advancing movement, once started, cannot be

stopped until zero is reached, or

(b) in the case of indicating elements only, such

elements are automatically obscured until the

elements reach the correct zero position.

S.1.1.5. Money Values--Mathematical

Agreement. - Any digital money-value indication and

any recorded money value on a computing-type device

shall be in mathematical agreement with its associated

quantity indication or representation to within one

cent of money value; except that a stationary retail

computing-type device must compute and indicate to

the nearest 1 cent of money value (see Sec. 1.10., GS.5.5.).

(Amended 1984, 1988)

S.1.1.6. Printed Ticket. - Any printed ticket issued

by a device of the computing type on which there is

printed the total computed price, shall have printed

clearly thereon the total volume of the delivery in

terms of liters or gallons, and the appropriate decimal

fraction of the liter or gallon, and the corresponding

price per liter or gallon.

(Added 1979; Amended 1987)

S.1.2. Graduations.

S.1.2.1. Length. - Graduations shall be so varied in

length that they may be conveniently read.

S.1.2.2. Width. - In any series of graduations, the

width of a graduation shall in no case be greater then

the width of the minimum clear interval between

graduations, and the width of main graduations shall

be not more than 50 percent greater than the width of
3.32. LPG and Anhydrous Ammonia Liquid-Measuring Devices



subordinate graduations. Graduations shall in no case

be less than 0.2 mm (0.008 in) width.

S.1.2.3. Clear Interval Between Graduations. - The shall be constructed to show automatically its initial

clear interval shall be not less than 1.0 mm (0.04 in). zero condition and the amounts delivered up to the

If the graduations are not parallel, the measurement nominal capacity of the device.

shall be made:

(a) along the line of relative movement between the

graduations at the end of the indicator, or

(b) if the indicator is continuous, at the point of

widest separation of the graduations.

S.1.3. Indicators. zero indication if the device is equipped to

S.1.3.1. Symmetry. - The index of an indicator shall

be symmetrical with respect to the graduations, at least (c) Means shall be provided to prevent the return of

throughout that portion of its length associated with primary indicating elements, and of primary

the graduations. recording elements if these are returnable to

S.1.3.2. Length. - The index of an indicator shall

reach to the finest graduations with which it is used,

unless the indicator and the graduations are in the S.1.5. For Stationary Retail Devices Only.

same plane, in which case the distance between the

end of the indicator and the ends of the graduations, S.1.5.1. Display of Unit Price and Product

measured along the line of graduations, shall be not Identity. - In a device of the computing type, means

more than 1.0 mm (0.04 in). shall be provided for displaying on each face of the

S.1.3.3. Width. - The width of the index of an

indicator in relation to the series of graduations with

which it is used shall be not greater than:

(a) the width of the narrowest graduation* , and

[*Nonretroactive as of January 1, 2002]

(Amended 2001)

(b) the width of the minimum clear interval between

graduations. S.1.5.2. Money-Value Computations. - A

When the index of an indicator extends along the any single-purchase unit price (excluding fleet sales

entire length of a graduation, that portion of the index and other price contract sales) for which the product is

of the indicator that may be brought into coincidence offered for sale at any delivery possible within either

with the graduation shall be of the same width the measurement range of the device or the range of

throughout the length of the index that coincides with the computing elements, whichever is less. The

the graduation. analog money value indication shall not differ from

S.1.3.4. Clearance. - The clearance between the

index of an indicator and the graduations shall in no

case be more than 1.5 mm (0.06 in).

S.1.3.5. Parallax. - Parallax effects shall be reduced

to the practicable minimum.

S.1.4. For Retail Devices Only.

S.1.4.1. Indication of Delivery. - A retail device

S.1.4.2. Return to Zero. - 

(a) Primary indicating elements shall be readily

returnable to a definite zero indication.

(b) Primary recording elements on a stationary retail

device shall be readily returnable to a definite

record.

zero, beyond their correct zero position.

(Amended 1990)

device the unit price at which the device is set to

compute or to deliver as the case may be, and there

shall be conspicuously displayed on each side of the

device the identity of the product that is being dispensed. If a device is so designed as to dispense more

than one grade, brand, blend, or mixture of product,

the identity of the grade, brand, blend, or mixture

being dispensed shall also be displayed on each face

of the device.

computing device shall compute the total sales price at

the mathematically computed money value (quantity x

unit price = sales price), for any delivered quantity, by

an amount greater than the values shown in Table 1.

(Amended 1995)

S.1.5.2.1. Money-Value Divisions, Analog. - The
3.32. LPG and Anhydrous Ammonia Liquid-Measuring Devices



value of the graduated intervals representing money (b) any adjustable element for controlling delivery rate,

values on a computing-type device with analog when such rate tends to affect the accuracy of

indications shall be as follows: deliveries.

(a) Not more than 1 cent at all unit prices up to The adjusting mechanism shall be readily accessible for

and including $0.25 per liter or $1.00 purposes of affixing a security seal.

per gallon.

(b) Not more than 2 cents at unit prices greater

than $0.25 per liter or $1.00 per gallon up to

and including $0.75 per liter or $3.00

per gallon.

(c) Not more than 5 cents at all unit prices greater

than $0.75 per liter or $3.00 per gallon.

(Amended 1984)

S.1.5.2.2. Money-Value Divisions, Digital. - A

computing-type device with digital indications shall

comply with the requirements of paragraph G.-

S.5.5., Money Values, Mathematical Agreement,

and the total price computation shall be based on

quantities not exceeding 0.01-gallon intervals for

devices indicating in inch-pound units and 0.05

liter for devices indicating in metric units.

S.1.5.2.3. Money-Value Divisions, Auxiliary

Indications. In a system equipped with auxiliary

indications, all indicated money-value divisions

shall be identical.

[Nonretroactive as of January 1, 1985.]

S.1.6. For Wholesale Devices Only.

S.1.6.1. Travel of Indicator. - A wholesale device

shall be readily operable to deliver accurately any

quantity from 180 L (50 gal) to the capacity of the

device. If the most sensitive element of the indicating

system uses an indicator and graduations, the relative

movement of these parts corresponding to a delivery

of 5 L (1 gal) shall be not less than 5 mm (0.20 in).

(Amended 1987)

S.2. Design of Measuring Elements.

S.2.1. Vapor Elimination. - A device shall be equipped

with an effective vapor eliminator or other effective

means to prevent the passage of vapor through the meter.

S.2.2. Provision for Sealing. - Adequate provision shall

be made for applying security seals in such a manner that

no adjustment may be made of:

(a) any measurement element, and

S.2.3. Directional Flow Valves. - A measuring system

shall be equipped with a valve or other effective means,

automatic in operation and installed in or adjacent to the

measuring element, to prevent reversal of flow of the

product being measured.

(Amended 1982).

S.2.4. Maintenance of Liquid State. - A device shall

be so designed and installed that the product being

measured will remain in a liquid state during the passage

through the meter.

S.2.5. Thermometer Well. - For test purposes, means

shall be provided to determine the temperature of the

liquid either:

(a) in the liquid chamber of the meter, or

(b) in the meter inlet or discharge line and immediately

adjacent to the meter.

(Amended 1987)

S.2.6. Automatic Temperature Compensation. - A

device may be equipped with an adjustable automatic

means for adjusting the indication and registration of the

measured volume of product to the volume at 15 EC

(60 EF).

S.2.6.1. Provision for Deactivating. - On a device

equipped with an automatic temperature compensating

mechanism that will indicate or record only in terms

of liters compensated to 15 EC or gallons to 60 EF,

provision shall be made to facilitate the deactivation

of the automatic temperature compensating

mechanism so that the meter may indicate, and record

if it is equipped to record, in terms of the

uncompensated volume. 

(Amended 1972)

S.2.6.2. Provision for Sealing. - Provision shall be

made for applying security seals in such a manner that

an automatic temperature-compensating system cannot

be disconnected and that no adjustment may be made

to the system.
3.32. LPG and Anhydrous Ammonia Liquid-Measuring Devices



Table 1.

Money-Value Divisions and

Maximum Allowable Variations for Money-Value

Computations on Mechanical Analog Computers

Unit Price Money

Value

Division

Maximum

Allowable Variation

From To and including Design

Test

Field

Test

0 $0.25/liter or

$1.00/gallon 1¢ ± 1¢ ± 1¢

$0.25/liter or

$1.00/gallon

$0.75/liter or

$3.00/gallon 1¢ or 2¢ ± 1¢ ± 2¢

$0.75/liter or

$3.00/gallon

$2.50/liter or

$10.00/gallon 1¢ or 2¢ ± 1¢ ± 2¢

$0.75/liter or

$3.00/gallon

$2.50/liter or

$10.00/gallon 5¢ ± 2 1/2¢ ± 5¢

S.3. Design of Discharge Lines and Discharge Line

Valves.

S.3.1. Diversion of Measured Liquid. - No means shall

be provided by which any measured liquid can be S.4.1. Limitation of Use. - If a device is intended to

diverted from the measuring chamber of the meter or the measure accurately only products having particular

discharge line therefrom. However, two or more properties, or to measure accurately only under specific

delivery outlets may be permanently installed if means installation or operating conditions, or to measure

are provided to insure that: accurately only when used in conjunction with specific

(a) liquid can flow from only one such outlet at one and permanently stated on the device.

time, and

(b) the direction of flow for which the mechanism may

be set at any time is definitely and conspicuously

indicated.

In addition, a manually controlled outlet that may be

opened for the purpose of emptying a portion of the

system to allow for repair and maintenance operations

shall be permitted. Effective means shall be provided to

prevent the passage of liquid through any such outlet

during normal operation of the device and to indicate

clearly and unmistakably when the valve controls are so

set as to permit passage of liquid through such outlet.

(Amended 1975).

S.3.2. Delivery Hose. - The delivery hose of a retail

device shall be of the wet-hose type with a shutoff valve

at its outlet end.

S.4. Marking Requirements.

accessory equipment, these limitations shall be clearly

S.4.2. Discharge Rates. - A device shall be marked to

show its designed maximum and minimum discharge

rates. The marked minimum discharge rate shall not

exceed:

(a) 20 L (5 gal) per minute for stationary retail devices,

or

(b) 20 percent of the marked maximum discharge rate

for other retail devices and for wholesale devices.

(Amended 1987)

S.4.3. Temperature Compensation. - If a device is

equipped with an automatic temperature compensator,

the primary indicating elements, recording elements, and

recorded representation shall be clearly and
3.32. LPG and Anhydrous Ammonia Liquid-Measuring Devices



conspicuously marked to show that the volume delivered conducted under controlled conditions where

has been adjusted to the volume at 15 EC (60 EF). variations in factors such as, temperature pressure and

N. Notes

N.1. Test Liquid. - A device shall be tested with the liquid

to be commercially measured or with a liquid of the same

general physical characteristics.

N.2. Vaporization and Volume Change. - Care shall be

exercised to reduce to a minimum, vaporization and volume

changes.

N.3. Test Drafts. - Test drafts should be equal to at least

the amount delivered by the device in 1 minute at its normal

discharge rate.

(Amended 1982)

N.4. Testing Procedures.

N.4.1. Normal Tests. - The "normal" test of a device

shall be made at the maximum discharge flow rate

developed under the conditions of the installation. Any

additional tests conducted at flow rates down to and

including one-half the sum of the maximum discharge

flow rate and the rated minimum discharge flow rate

shall be considered normal tests.

(Amended 1998)

N.4.1.1. Automatic Temperature Compensation. -

On devices equipped with automatic temperature

compensating systems, normal tests shall be

conducted as follows:

(a) by comparing the compensated volume indicated

or recorded to the actual delivered volume

corrected to 15 EC (60 EF); and,

(b) with the temperature compensating system

deactivated, comparing the uncompensated

volume indicated or recorded to the actual

delivered volume.

The first test shall be performed with the automatic

temperature-compensating system operating in the "as

found" condition. On devices that indicate or record

both the compensated and uncompensated volume for

each delivery, the tests in (a) and (b) may be

performed as a single test.

(Amended 1987)

N.4.1.2. Repeatability Tests. - Tests for repeatability

should include a minimum of three consecutive test

drafts of approximately the same size and be

flow rate are reduced to the extent that they will not

affect the results obtained.

(Added 2001)

N.4.2. Special Tests. - "Special" tests, to develop the

operating characteristics of a device and any special

elements and accessories attached to or associated with

the device, shall be made as circumstances require. Any

test except as set forth in N.4.1. shall be considered a

special test.

N.4.2.1. For Motor-Fuel Devices. - A motor-fuel

device shall be so tested at a minimum discharge rate

of:

(a) 20 L (5 gal) per minute, or

(b) the minimum discharge rate marked on the

device, whichever is less.

N.4.2.2. For Other Retail Devices. - A retail device

other than a motor-fuel device shall be tested at a

minimum discharge rate of:

(a) the minimum discharge rate that can be

developed under the conditions of installation, or

(b) the minimum discharge rate marked on the

device, whichever is greater.

(Amended 1973)

N.4.2.3. For Wholesale Devices. - A wholesale

device shall be so tested at a minimum discharge rate

of:

(a) 40 L (10 gal) per minute for a device with a

rated maximum discharge less than 180 L

(50 gal) per minute.

(b) 20 percent of the marked maximum discharge

rate for a device with a rated maximum

discharge of 180 L (50 gal) per minute or more,

or

(c) the minimum discharge rate marked on the

device, whichever is least.

(Amended 1987)

N.4.3. Money-Value Computation Tests.

N.4.3.1. Laboratory Design Evaluation Tests. - In

the conduct of laboratory design evaluation tests,
3.32. LPG and Anhydrous Ammonia Liquid-Measuring Devices



compliance with paragraph S.1.5.2. shall be

determined by using the cone gear as a reference for

the total quantity delivered. The indicated delivered

quantity shall agree with the cone gear representation

with the index of the indicator within the width of the

graduation. The maximum allowable variation of the

indicated sales price shall be as shown in Table 1.

N.4.3.2. Field Tests. - In the conduct of field tests to

determine compliance with paragraph S.1.5.2. the (b) 0.5 percent for electronic automatic temperature

maximum allowable variation in the indicated sales compensating systems.

price shall be as shown in Table 1.

(Added 1984) The delivered quantities for each test shall be approximately

N.5. Temperature Correction. - Corrections shall be

made for any changes in volume resulting from the

differences in liquid temperatures between time of passage

through the meter and time of volumetric determination in

the test measure.

T. Tolerances

T.1. Application.

T.1.1. To Underregistration and to

Overregistration. - The tolerances hereinafter

prescribed shall be applied to errors of underregistration

and errors of overregistration, whether or not a device is

equipped with an automatic temperature compensator. UR.1.2. Length of Discharge Hose. - The length of the

T.2. Tolerance Values. - The maintenance and acceptance

tolerances for normal and special tests shall be:

#EEGRVCPEG

6QNGTCPEG

/CKPVGPCPEG

6QNGTCPEG

0QTOCN6GUVU  

5RGEKCN6GUVU  

(Amended 1988 and 1992)

T.3. Repeatability. -When multiple tests are conducted at

approximately the same flow rate and draft size, the range

of the test results for the flow rate shall not exceed 40

percent of the absolute value of the maintenance tolerance

and the results of each test shall be within applicable

tolerance. This tolerance does not apply to the test of the

automatic temperature compensating system. See also

N.4.1.2.

(Added 1992) (Amended 1997 and 2001) 

T.4. Automatic Temperature Compensating Systems. -

The difference between the meter error (expressed as a

percentage) for results determined with and without the

automatic temperature compensating system activated shall

not exceed:

(a) 1.0 percent for mechanical automatic temperature

compensating systems; and

the same size. The results of each test shall be within the

applicable acceptance or maintenance tolerance.

(Added 1991)(Amended 1992, 1996, and 1997)

 UR. User Requirements

UR.1. Installation Requirements.

UR.1.1. Discharge Rate. - A device shall be so

installed that the actual maximum discharge rate will not

exceed the rated maximum discharge rate. If necessary,

means for flow regulation shall be incorporated in the

installation, in which case this shall be fully effective and

automatic in operation.

discharge hose on a stationary motor-fuel device shall

not exceed 5.5 m (18 ft), measured from the outside of

the housing of the device to the inlet end of the discharge

nozzle, unless it can be demonstrated that a longer hose

is essential to permit deliveries to be made to receiving

vehicles or vessels. Unnecessarily remote location of a

device shall not be accepted as justification for an

abnormally long hose.

(Amended 1991)

UR.2. Use Requirements.

UR.2.1. Return of Indication and Recording

Elements to Zero. - The primary indicating elements

(visual), and the primary recording elements when these

are returnable to zero, shall be returned to zero before

each delivery.

UR.2.2. Condition of Fill of Discharge Hose. - The

discharge hose shall be completely filled with liquid

before the "zero" condition is established prior to the

start of a commercial delivery, whether this condition is

established by resetting the primary indicating elements

to zero indication or by recording the indications of the

primary indicating elements. (Also see UR.2.1.)
3.32. LPG and Anhydrous Ammonia Liquid-Measuring Devices



UR.2.3. Vapor-Return Line. - During any metered

delivery of liquefied petroleum gas from a supplier’s tank

to a receiving container, there shall be no vapor-return

line from the receiving container to the supplier’s tank:

(a) in the case of any receiving container to which UR.2.6. Ticket Printer; Customer Ticket. - Vehiclenormal deliveries can be made without the use of mounted metering systems shall be equipped with a

such vapor-return line, or ticket printer. The ticket printer shall be used for all

(b) in the case of any new receiving container when the left with the customer at the time of delivery.

ambient temperature is below 90 EF. (Added 1992)

UR.2.4. Temperature Compensation.

UR.2.4.1. Use of Automatic Temperature

Compensators. - If a device is equipped with an

automatic temperature compensator, this shall be

connected, operable, and in use at all times. Such

automatic temperature compensator may not be

removed, nor may a compensated device be replaced

with an uncompensated device, without the written

approval of the weights and measures authority having

jurisdiction over the device.

UR.2.4.2. Temperature Compensated Sale. - All

sales of liquefied petroleum gas in a liquid state, when

the quantity is determined by an approved measuring

system equipped with a temperature compensating

mechanism, or by weight and converted to liters or

gallons, or by a calibrated container, shall be in terms

of liters at 15 EC or the United States gallon of 231 in3

at 60 EF.

(Added 1984)

UR.2.4.3. Invoices. - Any invoice based on a reading

of a device that is equipped with an automatic

temperature compensator or based on a weight

converted to gallons, or based on the volume of a calibrated container, shall have shown thereon that the

volume delivered has been adjusted to the volume at

15 EC (60 EF).

(Amended 1984)

UR.2.4.4. Automated Temperature-Compensating

Systems. - Means for determining the temperature of

measured liquid in an automatic temperaturecompensating system shall be so designed and located

that, in any "usual and customary" use of the

system,the resulting indications and/or recorded

representations are within applicable tolerances.

(Added 1987)

UR.2.5. Ticket in Printing Device. - A ticket shall not

be inserted into a device equipped with a ticket printer

until immediately before a delivery is begun, and in no

case shall a ticket be in the device when the vehicle is in

motion while on a public street, highway, or

thoroughfare.

sales; a copy of the ticket issued by the device shall be
3.33. Hydrocarbon Gas Vapor-Measuring Devices

6KVNGEJCPIGF 

3-35

Sec. 3.33. Hydrocarbon Gas Vapor-Measuring Devices1

A. Application

A.1. - This code applies to devices used for the measurement of hydrocarbon gas in the vapor state, such as

propane, propylene, butanes, butylenes, ethane, methane,

natural gas and any other hydrocarbon gas/air mix.

(Amended 1984, 1986, 1988, 1991)

A.2. - This code does not apply to:

(a) Liquid-measuring devices used for dispensing

liquefied petroleum gases in liquid form (for which

see Sec. 3.32; Code for Liquefied Petroleum Gas and

Anhydrous Ammonia Liquid-Measuring Devices).

(b) Natural, liquefied petroleum, and manufactured-gasvapor meters when these are operated in a public

utility system.

(c) Mass flow meters (see Sec. 3.37. Code for Mass Flow

Meters)

(Added 1994)

A.3. - See also Sec. 1.10; General Code requirements.

S. Specifications

S.1. Design of Indicating and Recording Elements and

of Recorded Representations.

S.1.1. Primary Elements.

S.1.1.1. General. - A device shall be equipped with

a primary indicating element and may also be

equipped with a primary recording element.

S.1.1.2. Units. - A volume-measuring device shall

indicate, and record if equipped to record, its

deliveries in terms of cubic meters or cubic feet, or

multiple or decimal subdivisions of cubic meters or

cubic feet.

(Amended 1972, 1991)

S.1.1.3. Value of Smallest Unit. - VolumeMeasuring Devices: The value of the smallest unit of

indicated delivery, and recorded delivery if the device

is equipped to record, shall not exceed:

(a) 1 m (1000 dm ) (100 ft ) when the maximum 3 33

rated gas capacity is less than 100 m /h3

(10 000 ft /h); 3

(b) 10 m (1000 ft ) when the maximum rated gas 3 3

capacity is 280 m /h (10 000 ft /h) up to but not 3 3

including 1700 m /h (60 000 ft /h); 3 3

(c) 100 m (10 000 ft ) when the maximum rated gas 3 3

capacity is 1700 m /h (60 000 ft /h) or more. 3 3

(Amended 1972, 1988, 1991)

S.1.1.4. Advancement of Indicating and Recording

Elements. - Primary indicating and recording

elements shall advance digitally or continuously and

be susceptible to advancement only by the mechanical

operation of the device.

S.1.1.5. Proving Indicator. - Devices rated less than

280 m /h (10 000 ft /h) gas capacity shall be equipped 3 3

with a proving indicator measuring 0.025, 0.05, 0.1,

0.2, or 0.25 m per revolution, (1, 2, 5, or 10 ft per 3 3

revolution) for testing the meter. Devices with larger

capacities shall be equipped as follows:

(a) Devices rated 280 m (10 000 ft ) up to but not 3 3

including 1700 m /h (60 000 ft /h) gas capacity 3 3

shall be equipped with a proving indicator

measuring not greater than 1 m (100 ft ) per 3 3

revolution.

(b) Devices rated 1700 m /h (60 000 ft /h) gas 3 3

capacity or more shall be equipped with a

proving indicator measuring not more than 10 m3

(1000 ft ) per revolution. 3

The test circle of the proving indicator shall be

divided into 10 equal parts. Additional subdivisions

of one or more of such equal parts may be made.
82/ ' ##2 % )2

##2 % 0)2

3.33. Hydrocarbon Gas Vapor-Measuring Devices

3-36

(Amended 1973 and 1988)

S.1.2. Graduations. case be more than 1.5 mm (0.06 in).

S.1.2.1. Length. - Graduations shall be so varied in

length that they may be conveniently read.

S.1.2.2. Width. - In any series of graduations, the

width of a graduation shall in no case be greater than

the width of the minimum clear interval between

graduations, and in no case should it exceed 1.0 mm

(0.04 in) for indicating elements and 0.5 mm (0.02 in)

for proving circles.

S.1.2.3. Clear Interval Between Graduations. - The (a) 2740 Pa ± 685 Pa (11 in of water column

clear interval shall be not less than 1.0 mm (0.04 in). (0.40 psig) ± 2.75 in of water column (0.10 psig))

If the graduations are not parallel, the measurement for liquefied petroleum gas vapor; or

shall be made:

(a) along the line of relative movement between the ± 1.75 in of water column (0.06 psig)) for natural

graduations at the end of the indicator, or and manufactured gas.

(b) if the indicator is continuous, at the point of When vapor is measured at a pressure other than what is

widest separation of the graduations. specified above for the specific product, a volume

S.1.3. Indicators.

S.1.3.1. Symmetry. - The index of an indicator shall

be symmetrical with respect to the graduations, at least

throughout that portion of its length associated with

the graduations.

S.1.3.2. Length. - The index of an indicator shall

reach to the finest graduations with which it is used.

S.1.3.3. Width. - The width of the index of an

indicator in relation to the series of graduations with

which it is used shall be not greater than:

(a) the width of the narrowest graduation* , and

[*Nonretroactive as of January 1, 2002]

(Amended 2001)

(b) the width of the minimum clear interval between

graduations.

When the index of an indicator extends along the

entire length of a graduation, that portion of the index

of the indicator that may be brought into coincidence

with the graduation shall be of the same width

throughout the length of the index that coincides with

the graduation.

S.1.3.4. Clearance. - The clearance between the

index of an indicator and the graduations shall in no

S.1.3.5. Parallax. - Parallax effects shall be reduced

to the practicable minimum.

S.2. Design of Measuring Elements.

S.2.1. Pressure Regulation. - Except when measured as

a retail motor fuel, the vapor should be measured at a

normal gauge pressure (psig) of:

(Amended 1991)

(b) 1744 Pa ± 436 Pa (7 in of water column (0.25 psig)

multiplier shall be applied within the meter or to the

billing invoice based on the following equation:

Where

VPM = Volume pressure multiplier

AAP = Assumed atmospheric pressure in psia

GP = Gauge pressure in pascal or psig

NGP = Normal gauge pressure in pascal or psig

The assumed atmospheric pressure is to be taken from

Tables 2 and 2M.

When liquefied petroleum gas vapor is measured at a

pressure of 6900 Pa (1 psig) or more, the delivery

pressure shall be maintained within ±1725 Pa

(±0.25 psig).

Pressure variations due to regulator lock off shall not

increase the operating pressure by more than 25 percent.

(Amended 1980, 1984, 1991)

S.2.2. Provision for Sealing. - Adequate provision shall

be made for applying security seals in such a manner that

no adjustment may be made of any measurement

element.

S.2.3. Maintenance of Vapor State. - A device shall be

so designed and installed that the product being
3.33. Hydrocarbon Gas Vapor-Measuring Devices

3-37

Table 1.

Capacity of Low-Flow Test Rate Orifices

With Respect to Device Capacity

Metric Units Inch-Pound Units

Rated Capacity Low-Flow Test Rate Rated Capacity Low-Flow Test Rate

Up to and including

7 m /h3

0.007 m /h3 Up to and including 250

ft /h3

0.25 ft /h3

Over 7 m /h up to and 3

including 14 m /h3

0.014 m /h3 Over 250 ft /h up to and 3

including 500 ft /h3

0.50 ft /h3

Over 14 m /h3 0.1% of capacity rate Over 500 ft /h3 0.1% of capacity rate

measured will remain in a vapor state during passage

through the meter.

S.2.4. Automatic Temperature Compensation. - A and capacity rate of the device for the particular products

device may be equipped with an adjustable automatic that it was designed to meter as recommended by the

means for adjusting the indication and registration of the manufacturer.

measured volume of vapor product to the volume at

15 EC (60 EF).

S.3. Design of Discharge Lines

S.3.1. Diversion of Measured Vapor. - No means shall

be provided by which any measured vapor can be

diverted from the measuring chamber of the meter or the

discharge line therefrom.

S.4. Marking Requirements.

S.4.1. Limitations of Use. - If a device is intended to

measure accurately only products having particular

properties, or to measure accurately only under specific

installation or operating conditions, or to measure

accurately only when used in conjunction with specific

accessory equipment, these limitations shall be clearly

and permanently stated on the device.

S.4.2. Discharge Rates. - A volume-measuring device

shall be marked to show its rated gas capacity in cubic

meters per hour or cubic feet per hour.

S.4.3. Temperature Compensation. - If a device is

equipped with an automatic temperature compensator,

this shall be indicated on the badge or immediately

adjacent to the badge of the device and on the register.

S.4.4. Badge. - A badge affixed in a prominent position

on the front of the device shall show the manufacturer’s

name, serial number and model number of the device,

N. Notes

N.1. Test Medium. - The device shall be tested with air or

the product to be measured.

(Amended 1991)

N.2. Temperature and Volume Change. - Care should be

exercised to reduce to a minimum any volume changes.

The temperature of the air, bell-prover oil, and the meters

under test should be within 1 EC (2 EF) of one another. The

devices should remain in the proving room for at least 16

hours before starting any proving operations to allow the

device temperature to approximate the temperature of the

proving device.

N.3. Test Drafts. - Except for low-flame tests, test drafts

shall be at least equal to one complete revolution of the

largest capacity proving indicator, and shall in no case be

less than 0.05 m or 2 ft . All flow rates shall be controlled 3 3

by suitable outlet orifices.

(Amended 1973 and 1991)

N.4. Test Procedures. - If a device is equipped with an

automatic temperature compensator, the proving device

reading shall be corrected to 15 EC (60 EF), using an

approved table.

(Amended 1972)
3.33. Hydrocarbon Gas Vapor-Measuring Devices

3-38

N.4.1. Normal Tests. - The normal test of a device shall

be made at a rate not to exceed the capacity rate given on

the badge of the meter.

(Amended 1988)

N.4.1.1. Automatic Temperature Compensation. -

If a device is equipped with an automatic temperature

compensator, the quantity of the test draft indication

of the standard shall be corrected to 15 EC (60 EF).

N.4.2. Special Tests. - "Special" tests, to develop the

operating characteristics of a device, and any special

elements and accessories attached to or associated with

the device, shall be made as circumstances require. Any

test except as set forth in N.4.1. is a special test.

N.4.2.1. Slow Test. - The device shall be tested at a

rate not less than 20 percent of the marked capacity

rate, or (at the check rate) not less than the minimum

flow rate if marked on the device, whichever is less.

(Amended 1988)

N.4.2.2. Low-Flame Test. - The device shall be

tested at an extremely low-flow rate as given in Table

1. The test shall consist of passing air at a pressure of

375 Pa (1.5 in water column) through the meter for

not less than 60 minutes. The meter shall continue to

advance at the conclusion of the test period.

(Amended 1990, 1991)

N.4.2.3. Pressure Regulation Test. - On devices

operating at a pressure of 6900 Pa (1 psig) or more, a

pressure regulation test shall be made at both the

minimum and maximum use load to determine the

proper operation of the regulator and the proper sizing

of the piping and dispensing equipment. These tests

may include a test of 24 hours during which the

pressure is recorded.

(Added 1984)

N.5. Temperature Correction. - Corrections shall be

made for any changes in volume resulting from the

difference in air temperatures between time of passage

through the device and time of volumetric determination in

the proving device.

N.6. Frequency of Test. - A hydrocarbon gas vapormeasuring device shall be tested before installation and

allowed to remain in service for 10 years from the time last

tested without being retested, unless a test is requested by:

(a) the purchaser of the product being metered,

(b) the seller of the product being metered, or

(c) the weights and measures official.

T. Tolerances

T.1. Tolerance Values on Normal Tests and on Special

Tests Other Than Low-Flame Tests. - Maintenance and

acceptance tolerances for hydrocarbon gas vapor-measuring

devices shall be 3 percent (1.03 proof) of the test draft on

underregistration and 1.5 percent (0.985 proof) of the test

draft on overregistration.

(Amended 1981)

 UR. User Requirements

UR.1. Installation Requirements.

UR.1.1. Capacity Rate. - A device shall be so installed

that the actual maximum flow rate will not exceed the

capacity rate except for short durations. If necessary,

means for flow regulation shall be incorporated in the

installation, in which case this shall be fully effective and

automatic in operation.

UR.1.2. Leakage. - The metering system shall be

installed and maintained as a pressure-tight and leak-free

system.

UR.2. Use Requirements.

UR.2.1. Automatic Temperature Compensation. - A

compensated device may not be replaced with an

uncompensated device without the written approval of

the weights and measures authority having jurisdiction

over the device.

UR.2.2. Invoices. - A customer purchasing hydrocarbon

gas measured by a vapor meter for other than motor fuel

shall receive from the seller an invoice for each billing

period. The invoice shall clearly and separately show the

following:

(a) The opening and closing meter readings and the

dates of those readings.

(b) The altitude correction factor.

(c) The total cubic meters (cubic feet) billed, corrected

for elevation.

(d) The charge per cubic meter (cubic foot) after

correction for elevation.
#%( ' )2 QH ICU % ##2

DCUG RTGUUWTG

3.33. Hydrocarbon Gas Vapor-Measuring Devices

3-39

(e) All periodic charges independent of the measured weights and measures officials, a thermometer well

gas, such as meter charges, meter reading fees, (appropriately protected against freezing) shall be

service charges or a minimum charge for a installed immediately up-stream of the meter.

minimum number of cubic meters (cubic feet). (Added 1990)

(f) The total charge for the billing period.

If the vapor meter is equipped with an automatic

temperature compensator, or any other means are used to

compensate for temperature, the invoice shall show that

the volume has been adjusted to the volume at 15 EC

(60 EF).

(Amended 1988, 1991)

UR.2.3. Correction for Elevation. - The metered

volume of gas shall be corrected for changes in the

atmospheric pressure with respect to elevation to the

standard pressure of 101.56 kPa (14.73 psia). The

appropriate altitude correction factor from Table 2M or

2 shall be used. (The table is modified from NIST Handbook 117.) (Amended 1988)

Elevation correction factors (ACF) were obtained by

using the following equation:

where

GP = gauge pressure

AAP = assumed atmospheric pressure

base pressure = 101.560 kPa = 14.73 psia

2740 Pa = 11 in of water column = 0.397 psig

1744 Pa = 7 in of water column = 0.253 psig

(Added 1988)

UR.2.4. Valves and Test Tee. - All gas meter

installations shall be provided with a shut-off valve

located adjacent to and on the inlet side of the meter. In

the case of a single meter installation utilizing a

liquefied petroleum gas tank, the tank service valve may

be used in lieu of the shut-off valve. All gas meter

installations shall be provided with a test tee located

adjacent to and on the outlet side of the meter.

[Nonretroactive as of January 1, 1990.]

(Added 1989)

UR.2.5. Use of Auxiliary Heated Vaporizer Systems.-

Automatic temperature compensation shall be used on

hydrocarbon gas vapor meters equipped with an auxiliary

heated vaporizer system unless there is sufficient length

of underground piping to provide gas at a uniform

temperature to the meter inlet. When required by
3.33. Hydrocarbon Gas Vapor-Measuring Devices

3-40

Table 2M. Corrections for Altitude, Metric Units

Elevation

(meters)

Altitude

Correction

Factor

Assumed

Atmospheric

Pressure

Assumed Atmospheric

Pressure Plus

Gauge Pressure

2.74 kPa

Gauge

Pressure

1.74 kPa

Gauge

Pressure

(kPa)

2.74 kPa

Gauge

Pressure

1.74 kPa

Gauge

Pressure

above

above

above

above

above

above

above

above

above

above

above

above

above

above

above

above

above

above

above

above

above

-50 to

120 to

300 to

470 to

650 to

830 to

1020 to

1210 to

1400 to

1590 to

1790 to

2000 to

2210 to

2420 to

2640 to

2860 to

3080 to

3320 to

3560 to

3800 to

4050 to

4310 to

120

300

470

650

830

1020

1210

1400

1590

1790

2000

2210

2420

2640

2860

3080

3320

3560

3800

4050

4310

4580

1.02

1.00

0.98

0.96

0.94

0.92

0.90

0.88

0.86

0.84

0.82

0.80

0.78

0.76

0.74

0.72

0.70

0.68

0.66

0.64

0.62

0.60

1.01

0.99

0.97

0.95

0.93

0.91

0.89

0.87

0.85

0.83

0.81

0.79

0.77

0.75

0.73

0.71

0.69

0.67

0.65

0.63

0.61

0.59

100.85

98.82

96.79

94.76

92.73

90.70

88.66

86.63

84.60

82.57

80.54

78.51

76.48

74.45

72.41

70.38

68.35

66.32

64.29

62.26

60.23

58.20

103.59

101.56

99.53

97.50

95.47

93.44

91.40

89.37

87.34

85.31

83.28

81.25

79.22

77.19

75.15

73.12

71.09

69.06

67.03

65.00

62.97

60.94

102.58

100.54

98.51

96.48

94.45

92.42

90.39

88.36

86.33

84.29

82.26

80.23

78.20

76.17

74.15

72.12

70.08

68.05

66.01

63.98

61.95

59.92
3.33. Hydrocarbon Gas Vapor-Measuring Devices

3-41

Table 2. Corrections for Altitude, Inch-Pound Units

Elevation

(feet)

Altitude

Correction

Factor

Assumed

Atmospheric

Pressure

Assumed Atmospheric

Pressure Plus Gauge

Pressure

11 inch

WC

7 inch

WC

(psia) 11 inch WC

(psia)

7 inch WC

(psia)

above

above

above

above

above

above

above

above

above

above

above

above

above

above

above

above

above

above

above

above

above

-150 to

400 to

950 to

1550 to

2100 to

2700 to

3300 to

3950 to

4550 to

5200 to

5850 to

6500 to

7200 to

7900 to

8600 to

9350 to

10 100 to

10 850 to

11 650 to

12 450 to

13 250 to

14 100 to

400

950

1550

2100

2700

3300

3950

4550

5200

5850

6500

7200

7900

8600

9350

10 100

10 850

11 650

12 450

13 250

14 100

14 950

1.02

1.00

0.98

0.96

0.94

0.92

0.90

0.88

0.86

0.84

0.82

0.80

0.78

0.76

0.74

0.72

0.70

0.68

0.66

0.64

0.62

0.60

1.01

0.99

0.97

0.95

0.93

0.91

0.89

0.87

0.85

0.83

0.81

0.79

0.77

0.75

0.73

0.71

0.69

0.67

0.65

0.63

0.61

0.59

14.64

14.35

14.05

13.76

13.46

13.17

12.87

12.58

12.28

11.99

11.69

11.40

11.10

10.81

10.51

10.22

9.92

9.63

9.33

9.04

8.75

8.45

15.04

14.74

14.45

14.15

13.86

13.56

13.27

12.97

12.68

12.38

12.09

11.79

11.50

11.20

10.91

10.61

10.32

10.03

9.73

9.44

9.14

8.85

14.89

14.60

14.30

14.01

13.71

13.42

13.12

12.83

12.53

12.24

11.94

11.65

11.35

11.06

10.76

10.47

10.17

9.88

9.58

9.29

9.00

8.70
3.34. Cryogenic Liquid-Measuring Devices

3-43

Sec. 3.34. Cryogenic Liquid-Measuring

Devices

A. Application

A.1. - This code applies to devices used for the measurement of cryogenic liquids such as, but not limited to

oxygen, nitrogen, hydrogen, and argon.

(Amended 1986 and 1995)

A.2. - This code does not apply to the following:

(a) Devices used for dispensing liquefied petroleum gases

(for which see Sec. 3.32; Code for Liquefied

Petroleum Gas and Anhydrous Ammonia LiquidMeasuring Devices).

(b) Devices used solely for dispensing a product in

connection with operations in which the amount dispensed does not affect customer charges.

(c) Devices used solely for dispensing liquefied natural

gas.

(d) mass flow meters (see Sec. 3.37. Code for Mass Flow

Meters)

(Added 1994)

A.3. - See also Sec. 1.10; General Code requirements.

S. Specifications

S.1. Design of Indicating and Recording Elements and

of Recorded Representations.

S.1.1. Primary Elements.

S.1.1.1. General. - A device shall be equipped with

a primary indicating element and may also be

equipped with a primary recording element.

S.1.1.2. Units. - A device shall indicate and record, if

equipped to record, its deliveries in terms of: liters or

gallons of liquid at the normal boiling point of the

specific cryogenic product; cubic meters (cubic feet)

of gas at a normal temperature of 21 EC (70 EF) and

an absolute pressure of 101.325 kPa (14.696 psia); or

decimal subdivisions or multiples of the measured

units cited above.

S.1.1.3. Value of Smallest Unit. - The value of the

smallest unit of indicated delivery, and recorded

delivery, if the device is equipped to record, shall not

exceed the equivalent of:

(a) for small delivery devices

(1) 1 L

(2) 0.1 gal

(3) 0.1 m of gas 3

(4) 10 ft of gas 3

(b) for large delivery devices

(1) 10 L

(2) 1 gal

(3) 1 m of gas 3

(4) 100 ft of gas 3

S.1.1.4. Advancement of Indicating and Recording

Elements. - Primary indicating and recording

elements shall be susceptible to advancement only by

the normal operation of the device. However, a

device may be cleared by advancing its elements to

zero, but only if:

(a) the advancing movement, once started, cannot be

stopped until zero is reached, or

(b) in the case of indicating elements only, such

elements are automatically obscured until the

elements reach the correct zero position.

S.1.1.5. Return to Zero. - Primary indicating and

recording elements shall be readily returnable to a

definite zero indication. Means shall be provided to

prevent the return of primary indicating elements and

of primary recording elements beyond their correct

zero position.

S.1.2. Graduations.

S.1.2.1. Length. - Graduations shall be so varied in

length that they may be conveniently read.

S.1.2.2. Width. - In any series of graduations, the

width of a graduation shall in no case be greater than

the width of the minimum clear interval between

graduations, and the width of main graduations shall
3.34. Cryogenic Liquid-Measuring Devices

3-44

be not more than 50 percent greater than the width of

subordinate graduations. Graduations shall in no case

be less than 0.2 mm (0.008 in) in width.

S.1.2.3. Clear Interval Between Graduations. - The

clear interval shall be no less than 1.0 mm (0.04 in).

If the graduations are not parallel, the measurement

shall be made:

(a) along the line of relative movement between the

graduations at the end of the indicator, or

(b) if the indicator is continuous, at the point of

widest separation of the graduations.

(See also S.1.3.6.)

S.1.3. Indicators.

S.1.3.1. Symmetry. - The index of an indicator shall

be symmetrical with respect to the graduations, at least

throughout that portion of its length associated with

the graduations.

S.1.3.2. Length. - The index of an indicator shall

reach to the finest graduations with which it is used,

unless the indicator and the graduations are in the

same plane, in which case the distance between the

end of the indicator and the ends of the graduations,

measured along the line of the graduations, shall be

not more than 1.0 mm (0.04 in).

S.1.3.3. Width. - The width of the index of an - Any digital money-value indication and any

indicator in relation to the series of graduations with recorded money value on a computing type device

which it is used shall be not greater than: shall be in mathematical agreement with its associated

(a) the width of the narrowest graduation* , and cent of money value.

[*Nonretroactive as of January 1, 2002]

(Amended 2001)

(b) the width of the minimum clear interval between

graduations.

When the index of an indicator extends along the

entire length of a graduation, that portion of the index

of the indicator that may be brought into coincidence

with the graduation shall be of the same width

throughout the length of the index that coincides with

the graduation.

S.1.3.4. Clearance. - The clearance between the

index of an indicator and the graduations shall in no

case be more than 1.5 mm (0.06 in).

S.1.3.5. Parallax. - Parallax effect shall be reduced to

the practicable minimum.

S.1.3.6. Travel of Indicator. - If the most sensitive

element of the primary indicating element uses an

indicator and graduations, the relative movement of

these parts corresponding to the smallest indicated

value shall be not less than 0.5 mm (0.20 in).

S.1.4. Computing-Type Device.

S.1.4.1. Printed Ticket. - Any printed ticket issued

by a device of the computing type on which there is

printed the total computed price shall have printed

clearly thereon also the total quantity of the delivery

and the price per unit.

S.1.4.2. Money-Value Computations. - Moneyvalue computations shall be of the full-computing type

in which the money value at a single unit price, or at

each of a series of unit prices, shall be computed for

every delivery within either the range of measurement

of the device or the range of the computing elements,

whichever is less. Value graduations shall be supplied

and shall be accurately positioned. The total price

shall be computed on the basis of the quantity

indicated when the value of the smallest division

indicated is equal to or less than the values specified

in S.1.1.3.

S.1.4.3. Money Values, Mathematical Agreement.

quantity indication or representation to within one

S.2. Design of Measuring Elements.

S.2.1. Vapor Elimination. - A measuring system shall

be equipped with an effective vapor eliminator or other

effective means to prevent the measurement of vapor that

will cause errors in excess of the applicable tolerances.

(See Section T. Tolerances)

S.2.2. Directional Flow Valves. - A valve or valves or

other effective means, automatic in operation, to prevent

the reversal of flow shall be installed in or adjacent to the

measuring device.

(Amended 1978)

S.2.3. Maintenance of Liquid State. - A device shall

be so designed that the product being measured will
3.34. Cryogenic Liquid-Measuring Devices

3-45

remain in a liquid state during passage through the

device.

S.2.4. Automatic Temperature or Density

Compensation. - A device shall be equipped with

automatic means for adjusting the indication and/or

recorded representation of the measured quantity of the

product, to indicate and/or record in terms of: liters or

gallons of liquid at the normal boiling point of the

specific cryogenic product; or the equivalent cubic

meters (cubic feet) of gas at a normal temperature of 21

EC (70 EF) and an absolute pressure of 101.325 kPa

(14.696 lb/in absolute). When a compensator system 2

malfunctions, the indicating and recording elements may

indicate and record in uncompensated volume if the

mode of operation is clearly indicated, e.g., by a marked

annunciator, recorded statement, or other obvious

means.*

[*Nonretroactive as of January 1, 1992.]

(Amended 1991)

S.2.5. Provision for Sealing. - Adequate provision shall

be made for applying security seals in such a manner that

no adjustment or interchange may be made of:

(a) any measurement element,

(b) any adjustable element for controlling delivery rate

when such rate tends to affect the accuracy of

deliveries, and

(c) any automatic temperature or density compensating

system.

Any adjusting mechanism shall be readily accessible for

purposes of affixing a security seal.

S.3. Design of Discharge Lines and Discharge Line

Valves.

S.3.1. Diversion of Measured Liquid. - No means shall

be provided by which any measured liquid can be

diverted from the measuring chamber of the device or the

discharge line therefrom, except that a manually

controlled outlet that may be opened for purging or

draining the measuring system shall be permitted.

Effective means shall be provided to prevent the passage

of liquid through any such outlet during normal

operation of the device and to indicate clearly and

unmistakably when the valve controls are so set as to

permit passage of liquid through such outlet.

S.3.2. Discharge Hose. - The discharge hose of a

measuring system shall be of the completely draining

dry-hose type.

S.4. Level Condition, On-Board Weighing Systems. -

Provision shall be made for automatically inhibiting the

delivery of a cryogenic liquid when the vehicle is out of

level beyond the limit required for the performance to be

within applicable tolerance.

(Added 1986)

S.5. Marking Requirements.

S.5.1. Limitation of Use. - If a measuring system is

intended to measure accurately only liquids having

particular properties, or to measure accurately only under

specific installation or operating conditions, or to

measure accurately only when used in conjunction with

specific accessory equipment, these limitations shall be

clearly and permanently marked on the device.

S.5.2. Discharge Rates. - A meter shall be marked to

show its designed maximum and minimum discharge

rates.

S.5.3. Temperature or Density Compensation. -

Devices equipped with an automatic temperature or

density compensator, shall be clearly and conspicuously

marked on the primary indicating elements, recording

elements, and recorded representations to show that the

quantity delivered has been adjusted to the conditions

specified in S.2.4.

N. Notes

N.1. Test Liquid. - A meter shall be tested with the liquid

to be commercially measured except that, in a type

evaluation examination, nitrogen may be used.

N.2. Vaporization and Volume Change. - Care shall be

exercised to reduce to a minimum vaporization and volume

changes. When testing by weight, the weigh tank and

transfer systems shall be pre-cooled to liquid temperature

prior to the start of the test to avoid the venting of vapor

from the vessel being weighed.

N.3. Test Drafts.

N.3.1. Gravimetric Test. - Weight test drafts shall be

equal to at least the amount delivered by the device in

two minutes at its maximum discharge rate, and shall in

no case be less than 907 kg (2000 lb).
3.34. Cryogenic Liquid-Measuring Devices

3-46

N.3.2. Transfer Standard Test. - When comparing a

meter with a calibrated transfer standard, the test draft

shall be equal to at least the amount delivered by the

device in two minutes at its maximum discharge rate, and

shall in no case be less than 180 L (50 gal) or equivalent

thereof. When testing uncompensated volumetric meters N.6. Temperature Correction. - Corrections shall be

in a continuous recycle mode, appropriate corrections made for any changes in volume resulting from the

shall be applied if product conditions are abnormally differences in liquid temperature between time of passage

affected by this test mode. through the meter and time of volumetric determination of

(Amended 1976) test draft.

N.4. Density. - Temperature and pressure of the metered N.7. Automatic Temperature or Density

test liquid shall be measured during the test for the Compensation. - When a device is equipped with an

determination of density or volume correction factors when automatic temperature or density compensator, the compenapplicable. For Liquid Density and Volume Correction sator shall be tested by comparing the quantity indicated or

Factors (with respect to temperature and pressure) the recorded by the device (with the compensator connected

following publication shall apply: "Thermophysical and operating) with the actual delivered quantity corrected

Properties of Fluids. 1. Argon, Ethylene, Parahydrogen, to the normal boiling point of the cryogenic product being

Nitrogen, Nitrogen Trifluoride, and Oxygen," published in measured or to the normal temperature and pressure as

the Journal of Physical and Chemical Reference Data, applicable.

Volume 11, 1982, Supplement No. 1, and published by the

American Chemical Society and the American Institute of

Physics for the National Institute of Standards and

Technology.

(Amended 1986)

N.5. Testing Procedures.

N.5.1. Normal Tests. - The "normal" tests of a device

shall be made over a range of discharge rates that may be

anticipated under the conditions of installation.

N.5.1.1. Repeatability Tests. - Tests for repeatability

should include a minimum of three consecutive test

drafts of approximately the same size and be

conducted under controlled conditions where

variations in factors, such as, temperature pressure

and flow rate are reduced to the extent that they will

not affect the results obtained.

(Added 2001)

N.5.2. Special Tests. - Any test except as set forth in

N.5.1. shall be considered a "special" test. Tests shall be

conducted, if possible, to evaluate any special elements

or accessories attached to or associated with the device.

A device shall be tested at a minimum discharge rate of:

(a) 50 percent of the maximum discharge rate

developed under the conditions of installation, or

the minimum discharge rate marked on the device,

whichever is less, or

(b) the lowest discharge rate practicable under

conditions of installation.

Special tests may be conducted to develop any

characteristics of the device that are not normally

anticipated under the conditions of installation as circumstances require.

 T. Tolerances

T.1. Application.

T.1.1. To Underregistration and to Overregistration.

- The tolerances hereinafter prescribed shall be applied to

errors of underregistration and errors of overregistration.

T.2. Tolerance Values.

T.2.1. On Normal Tests. - The maintenance tolerance

on "normal" tests shall be two and one-half percent

(2.5%) of the indicated quantity. The acceptance

tolerance shall be one and one-half percent (1.5%) of the

indicated quantity.

T.2.2. On Special Tests. - The maintenance and

acceptance tolerance on "special" tests shall be two and

one-half percent (2.5%) of the indicated quantity.

T.3. On Tests Using Transfer Standards. - To the basic

tolerance values that would otherwise be applied, there shall

be added an amount equal to two times the standard

deviation of the applicable transfer standard when

compared to a basic reference standard.

(Added 1976)

T.4. Repeatability. - When multiple tests are conducted at

approximately the same flow rate and draft size, the range

of the test results for the flow rate shall not exceed 40

percent of the absolute value of the maintenance tolerance

and the results of each test shall be within the applicable

tolerance. See also N.5.1.1.
3.34. Cryogenic Liquid-Measuring Devices

3-47

See Appendix D, Definitions. 1

(Added 2001)

UR. User Requirements

UR.1. Installation Requirements.

UR.1.1. Discharge Rate. - A device shall be so

installed that the actual maximum discharge rate will not

exceed the rated maximum discharge rate. If necessary,

means for flow regulation shall be incorporated in the

installation.

UR.1.2. Length of Discharge Hose. - The discharge

hose shall be of such a length and design as to keep

vaporization of the liquid to a minimum.

UR.1.3. Maintenance of Liquid State. - A device shall

be so installed and operated that the product being

measured shall remain in the liquid state during passage

through the meter.

UR.2. Use Requirements.

UR.2.1. Return of Indicating and Recording

Elements to Zero. - The primary indicating elements

(visual) and the primary recording elements shall be

returned to zero immediately before each delivery.

UR.2.2. Condition of Discharge System. - The

discharge system, up to the measuring element, shall be

precooled to liquid temperatures before a "zero"

condition is established prior to the start of a commercial

delivery.

UR.2.3. Vapor Return Line. - A vapor return line shall

not be used during a metered delivery.

(Amended 1976)

UR.2.4. Drainage of Discharge Line. - On a dry-hose

system, upon completion of a delivery, the vendor shall

leave the discharge line connected to the receiving

container with the valve adjacent to the meter in the

closed position and the valve at the discharge line outlet

in the open position for a period of at least

(a) 1 minute for small delivery devices, and

(b) 3 minutes for large delivery devices,

to allow vaporization of some product in the discharge

line to force the remainder of the product in the line to

flow into the receiving container.

(Amended 1976)

UR.2.5. Conversion Factors. - Established conversion

values (see references in N.4.) shall be used whenever

metered liquids are to be billed in terms of:

(a) kilograms or pounds based on a meter indication of

liters, gallons, cubic meters of gas, or cubic feet of

gas; or,

(b) cubic meters or cubic feet of gas based on a meter

indication of liters or gallons, kilograms, or pounds;

or,

(c) liters or gallons based on a meter indication of

kilograms or pounds, cubic meters of gas or cubic

feet of gas.

All sales of cryogenic liquids shall be based on either

kilograms or pounds, liters or gallons at NBP , cubic 1

meters of gas or cubic feet of gas at NTP .1

(Amended 1986)

UR.2.6. Temperature or Density Compensation.

UR.2.6.1. Use of Automatic Temperature or

Density Compensators. - If a device is equipped with

an automatic temperature or density compensator, this

shall be connected, operable, and in use at all times.

Such automatic temperature or density compensator

may not be removed, nor may a compensated device

be replaced with an uncompensated device, without

the written approval of the weights and measures

authority having jurisdiction over the device.

UR.2.6.2. Tickets or Invoices. - Any written invoice

or printed ticket based on a reading of a device that is

equipped with an automatic temperature or density

compensator shall have shown thereon that the

quantity delivered has been adjusted to the quantity at

the NBP of the specific cryogenic product or the

equivalent volume of gas at NTP.

UR.2.6.3. Printed Ticket. - Any printed ticket issued

by a device of the computing type on which there is

printed the total computed price, the total quantity of

the delivery, or the price per unit, shall also show the

other two values (either printed or in clear hand

script).

UR.2.6.4. Ticket in Printing Device. - A ticket shall

not be inserted into a device equipped with a ticket

printer until immediately before a delivery is begun,

and in no case shall a ticket be in the device when the
3.34. Cryogenic Liquid-Measuring Devices

3-48

vehicle is in motion while on a public street, highway,

or thoroughfare.

UR.2.7. Pressure of Tanks with Volumetric Metering

Systems without Temperature Compensation. - When

the saturation pressure of the product in the vendor’s tank

exceeds 240 kPa (35 psia), a correction shall be applied

to the written invoice or printed ticket using the

appropriate tables as listed in N.4.; or the saturation

pressure shall be reduced to 207 kPa (30 psia) (if this can

be safely accomplished) prior to making a delivery.

(Added 1976)
3.35. Milk Meters



Sec. 3.35. Milk Meters

A. Application

A.1. - This code applies to devices used for the

measurement of milk; generally applicable to, but not

limited to, meters used in dairies, milk processing plants,

and cheese factories, to measure incoming bulk milk.

A.2. - See also Sec. 1.10; General Code requirements.

A.3. - This code does not apply to mass flow meters (see

Sec. 3.37. Code for Mass Flow Meters).

(Added 1994)

S. Specifications

S.1. Design of Indicating and Recording Elements and

of Recorded Representations.

S.1.1. Primary Elements

S.1.1.1. General. - A meter shall be equipped with a

primary indicating element and may also be equipped

with a primary recording element.

S.1.1.2. Units. -

(a) A meter shall indicate, and record if the meter is

equipped to record, its deliveries in terms of

liters or gallons. Fractional parts of the liter

shall be in terms of decimal subdivisions.

Fractional parts of the gallon shall be in terms of

either decimal or binary subdivisions.

(b) When it is an industry practice to purchase and

sell milk by weight based upon 1.03 kg/L

(8.6 lb/gal), the primary indicating element may

indicate in kilograms or pounds. The weight

value division shall be a decimal multiple or

submultiple of 1, 2, or 5. Fractional parts of

the kilogram or pound shall be in decimal

subdivisions. (See S.4.5.)

S.1.1.3. Value of Smallest Unit. - The value of the

smallest unit of indicated quantity and recorded

quantity, if the meter is equipped to record, shall not

exceed the equivalent of:

(a) 0.5 L or 5 kg (1 pt or 1 lb) when measuring

quantities less than or equal to 4000 L or 4000

kg (1000 gal or 8600 lb), or

(b) 5 L or 5 kg (1 gal or 10 lb) when measuring

quantities in excess of 4000 L or 4000 kg

(1000 gal or 8600 lb).

(Amended 1989)

S.1.1.4. Advancement of Indicating and Recording

Elements. - Primary indicating and recording

elements shall be susceptible to advancement only by

the mechanical operation of the meter. However, a

meter may be cleared by advancing its elements to

zero, but only if:

(a) the advancing movement, once started, cannot be

stopped until zero is reached, or

(b) in the case of indicating elements only, such

elements are automatically obscured until the

elements reach the correct zero position.

S.1.1.5. Return to Zero. - Primary indicating

elements and primary recording elements, if the device

is equipped to record, shall be readily returnable to a

definite zero indication. Means shall be provided to

prevent the return of the primary indicating elements

and the primary recording elements, if the device is so

equipped, beyond their correct zero position.

S.1.1.6. Indication of Measurement. A meter shall

be constructed to show automatically its initial zero

condition and the volume measured up to the nominal

capacity of the device.

S.1.2. Graduations.

S.1.2.1. Length. - Graduations shall be so varied in

length that they may be conveniently read.

S.1.2.2. Width. - In any series of graduations, the

width of a graduation shall in no case be greater then

the width of the minimum clear interval between

graduations, and the width of main graduations shall

be not more than 50 percent greater than the width of

subordinate graduations. Graduations shall in no case

be less than 0.2 mm (0.008 in) in width.
3.35. Milk Meters



S.1.2.3. Clear Interval Between Graduations. - The

clear interval shall be not less than 1.0 mm (0.04 in).

If the graduations are not parallel, the measurement

shall be made: S.1.4. Computing-Type Devices.

(a) along the line of relative movement between the S.1.4.1. Display of Unit Price. - In a device of the

graduations at the end of the indicator, or computing type, means shall be provided for

(b) if the indicator is continuous, at the point of proximity to the display of the total computed price,

widest separation of the graduations. the price per unit at which the device is set to

S.1.3. Indicators.

S.1.3.1. Symmetry. - The index of an indicator shall issues a printed ticket which displays the total

be symmetrical with respect to the graduations, at least computed price, the ticket also shall have printed

throughout that portion of its length associated with clearly thereon the total quantity of the delivery, the

the graduations. appropriate fraction of the quantity, and the price per

S.1.3.2. Length. - The index of an indicator shall

reach to the finest graduations with which it is used,

unless the indicator and the graduations are in the S.1.4.3. Money-Value Computations. - Moneysame plane, in which case the distance between the value computations shall be of the full-computing type

end of the indicator and the ends of the graduations, in which the money value at a single unit price, or at

measured along the line of graduations, shall be not each of a series of unit prices, shall be computed for

more than 1.0 mm (0.04 in). every delivery within either the range of measurement

S.1.3.3. Width. - The width of the index of an

indicator in relation to the series of graduations with

which it is used shall be not greater than:

(a) the width of the narrowest graduation* , and S.1.4.4. Money Values, Mathematical Agreement.

[*Nonretroactive as of January 1, 2002] Any digital money-value indication and any recorded

(Amended 2001) money value on a computing-type device shall be in

(b) the width of the minimum clear interval between indicating or representation to within 1 cent of money

graduations. value.

When the index of an indicator extends along the

entire length of a graduation, that portion of the index

of the indicator that may be brought into coincidence

with the graduation shall be of the same width

throughout the length of the index that coincides with

the graduation.

S.1.3.4. Clearance. - The clearance between the tubing or some other suitably rigid material.

index of an indicator and the graduations shall in no

case be more than 1.5 mm (0.06 in).

S.1.3.5. Parallax. - Parallax effects shall be reduced such a manner that the vapor eliminator cannot easily be

to the practicable minimum. emptied between uses.

S.1.3.6. Travel of Indicator. - If the most sensitive S.2.3. Provision for Sealing. - Adequate provision shall

element of the primary indicating element utilizes an be made for applying security seals to the adjustment

indicator and graduations, the relative movement of mechanism and the register.

these parts corresponding to the smallest indicated

value shall be not less than 5 mm (0.20 in).

displaying on the outside of the device, and in close

compute.

S.1.4.2. Printed Ticket. - If a computing-type device

unit of quantity.

(Amended 1989)

of the device or the range of the computing elements,

whichever is less. Value graduations shall be supplied

and shall be accurately positioned. The value of each

graduated interval shall be 1 cent.

mathematical agreement with its associated quantity

S.2. Design of Measuring Elements.

S.2.1. Vapor Elimination. - A metering system shall be

equipped with an effective vapor eliminator or other

effective means automatic in operation to prevent the

passage of vapor and air through the meter. Vent lines

from the air (or vapor) eliminator shall be made of metal

S.2.2. Maintaining Flooded Condition. - The vent on

the vapor eliminator shall be positioned or installed in
3.35. Milk Meters



S.2.4. Directional Flow Valves. - Valves intended to

prevent reversal of flow shall be automatic in operation.

S.3. Design of Intake Lines.

S.3.1. Diversion of Liquid to be Measured. - No

means shall be provided by which any liquid can be

diverted from the supply tank to the receiving tank

without being measured by the device. A manually

controlled outlet that may be opened for purging or

draining the measuring system shall be permitted.

Effective means shall be provided to prevent passage of

liquid through any such outlet during normal operation

of the measuring system.

(Amended 1994)

S.3.2. Intake Hose. - The intake hose shall be:

(a) of the dry-hose type,

(b) adequately reinforced,

(c) not more than 6 m (20 ft) in length unless it can be

demonstrated that a longer hose is essential to

permit transfer from a supply tank; and

(d) connected to the pump at horizontal or above to

permit complete drainage of the hose.

(Amended 1991)

S.4. Marking Requirements.

S.4.1. Limitation of Use. - If a meter is intended to

measure accurately only liquids having particular

properties, or to measure accurately only under specific

installation or operating conditions, or to measure

accurately only when used in conjunction with specific

accessory equipment, these limitations shall be clearly

and permanently stated on the meter.

S.4.2. Discharge Rates. - A meter shall be marked to

show its designed maximum and minimum discharge

rates. However, the minimum discharge rate shall not

exceed 20 percent of the maximum discharge rate.

S.4.3. Measuring Components. - All components that

affect the measurement of milk that are disassembled for

cleaning purposes shall be clearly and permanently

identified with a common serial number.

S.4.4. Flood Volume. - When applicable, the volume of

product (to the nearest minimum division of the meter)

necessary to flood the system when dry shall be clearly,

conspicuously, and permanently marked on the air

eliminator.

S.4.5. Conversion Factor. - When the conversion factor

of 1.03 kg/L (8.6 lb/gal) is used to convert the volume of

milk to weight, the conversion factor shall be clearly

marked on the primary indicating element and recorded

on the delivery ticket.

N. Notes

N.1. Test Liquid.

(a) A meter shall be tested with the liquid to be

commercially measured or with a liquid of the same

general physical characteristics. Following a satisfactory examination, the weights and measures official

should attach a seal or tag indicating the product used

during the test.

(Amended 1989)

(b) A milk measuring system shall be tested with the

type of milk to be measured when the accuracy

of the system is affected by the characteristics of

milk (e.g., positive displacement meters).

(Added 1989)

N.2. Evaporation and Volume Change. - Care shall be

exercised to reduce to a minimum, evaporation losses and

volume changes resulting from changes in temperature of

the test liquid.

N.2.1. Temperature Correction. - Corrections shall be

made for any changes in volume resulting from the

differences in liquid temperatures between time of

passage through the meter and time of volumetric determination in the test measure. When adjustments are

necessary, appropriate tables should be used.

N.3. Test Drafts. - Test drafts should be equal to at least

the amount delivered by the device in one minute at its

maximum discharge rate, and shall in no case be less than

400 L or 400 kg (100 gal or 1000 lb).

(Amended 1989)

N.4. Testing Procedures.

N.4.1. Normal Tests. - The "normal" test of a meter

shall be made at the maximum discharge rate that may be

anticipated under the conditions of the installation. The

"normal" test shall include a determination of the

effectiveness of the air elimination system.
3.35. Milk Meters



N.4.2. Special Tests. - "Special" tests to develop the

operating characteristics of a meter and any special

elements and accessories attached to or associated with

the meter, shall be made as circumstances require. Any

test except as set forth in N.4.1. shall be considered a

special test.

N.4.3. System Capacity. - The test of a milk-metering

system shall include the verification of the volume of

product necessary to flood the system as marked on the

air eliminator.

 T. Tolerances

T.1. Application.

T.1.1. To Underregistration and to Overregistration.

- The tolerances hereinafter prescribed shall be applied to

errors of underregistration and errors of overregistration.

T.2. Tolerance Values. - Maintenance and acceptance

tolerances shall be as shown in Table 1.

(Amended 1989)

UR. User Requirements

UR.1. Installation Requirements.

UR.1.1. Plumb and Level Condition. - A device

installed in a fixed location shall be installed plumb and

level, and the installation shall be sufficiently strong and

rigid to maintain this condition.

Table 1. Tolerances for Milk Meters

Indication Maintenance Acceptance

gallons gallons gallons

100 0.5 0.3

200 0.7 0.4

300 0.9 0.5

400 1.1 0.6

500 1.3 0.7

Over 500 Add 0.002 gallon per indicated

gallon over 500

Add 0.001 gallon per indicated gallon

over 500

(Added 1989)

UR.1.2. Discharge Rate. - A meter shall be so installed

that the actual maximum discharge rate will not exceed

the rated maximum discharge rate. If necessary, means

for flow regulation shall be incorporated in the

installation, in which case this shall be fully effective and

automatic in operation.

UR.1.3. Unit Price. - There shall be displayed on the

face of a device of the computing type the unit price at

which the device is set to compute.

UR.1.4. Intake Hose. - The intake hose shall be so

installed as to permit complete drainage and that all

available product is measured following each transfer.

UR.2. Use Requirements.

UR.2.1. Return of Indicating and Recording

Elements to Zero. - The primary indicating elements

(visual), and the primary recording elements when these

are returnable to zero, shall be returned to zero before

each transfer.

UR.2.2. Printed Ticket. - Any printed ticket issued by

a device of the computing type on which there is printed

the total computed price, the total quantity, or the price

per unit of quantity, shall also show the other two values

(either printed or in clear script).

(Amended 1989)

UR.2.3. Ticket in Printing Device. - A ticket shall not

be inserted into a device equipped with a ticket printer

until immediately before a transfer is begun. If the meter

is mounted on a vehicle, in no case shall a ticket be in the

device when the vehicle is in motion while on a public

street, highway, or thoroughfare.
3.35. Milk Meters



UR.2.4. Credit for Flood Volume. - The volume of

product necessary to flood the system as marked on the

air eliminator shall be individually recorded on the ticket

of each transfer affected.
3.36. Water Meters



Sec. 3.36. Water Meters

A. Application

A.1. - This code applies to devices used for the

measurement of water; generally applicable to, but not

limited to, utilities type meters installed in homes or

business establishments and meters installed in batching

systems.

A.2. - This code does not apply to:

(a) water meters mounted on vehicle tanks (for which see

Sec. 3.31; Code for Vehicle-Tank Meters).

(b) mass flow meters (see Sec. 3.37. Code for Mass Flow

Meters)

(Added 1994)

A.3. - See also Sec. 1.10; General Code requirements.

S. Specifications

S.1. Design of Indicating and Recording Elements and

of Recorded Representations.

S.1.1. Primary Elements.

S.1.1.1. General. - A water meter shall be equipped

with a primary indicating element and may also be

equipped with a primary recording element.

S.1.1.2. Units. - A water meter shall indicate and

record, if the device is equipped to record, its

deliveries in terms of liters, gallons or cubic feet or

binary or decimal subdivisions thereof except batch

plant meters, which shall indicate deliveries in terms

of liters, gallons or decimal subdivisions of the liter or

gallon only.

S.1.1.3. Value of Smallest Unit. - The value of the

smallest unit of indicated delivery and recorded

delivery, if the device is equipped to record, shall not

exceed the equivalent of:

(a) 50 L (10 gal) on utility type meters,

(b) 0.2 L (1/10 gal) on batching meters delivering

less than 375 L/min (100 gal/min), or

(c) 5 L (1 gal) on batching meters delivering

375 L/min (100 gal/min) or more.

S.1.1.4. Advancement of Indicating and Recording

Elements. - Primary indicating and recording

elements shall be susceptible to advancement only by

the mechanical operation of the device.

S.1.1.5. Return to Zero. - If the meter is so designed

that the primary indicating elements are readily returnable to a definite zero indication, means shall be

provided to prevent the return of these elements

beyond their correct zero position.

S.1.2. Graduations.

S.1.2.1. Length. - Graduations shall be so varied in

length that they may be conveniently read.

S.1.2.2. Width. - In any series of graduations, the

width of a graduation shall in no case be greater than

the width of the minimum clear interval between

graduations, and the width of main graduations shall

be not more than 50 percent greater than the width of

subordinate graduations. Graduations shall in no case

be less than 0.2 mm (0.008 in) in width.

S.1.2.3. Clear Interval Between Graduations. - The

clear interval shall not be less than 1.0 mm (0.04 in).

If the graduations are not parallel, the measurement

shall be made:

(a) along the line of relative movement between the

graduations at the end of the indicator, or

(b) if the indicator is continuous, at the point of

widest separation of the graduations.

S.1.3. Indicators.

S.1.3.1. Symmetry. - The index of an indicator shall

be symmetrical with respect to the graduations, at least

throughout that portion of its length associated with

the graduations.

S.1.3.2. Length. - The index of an indicator shall

reach to the finest graduations with which it is used,
3.36. Water Meters



the width of the minimum clear interval between

graduations, and the width of main graduations shall

be not more than 50 percent greater than the width of

subordinate graduations. Graduations shall in no case

be less than 0.2 mm (0.008 in) in width.

S.1.2.3. Clear Interval Between Graduations. - The

clear interval shall not be less than 1.0 mm (0.04 in).

If the graduations are not parallel, the measurement

shall be made:

(a) along the line of relative movement between the

graduations at the end of the indicator, or

(b) if the indicator is continuous, at the point of

widest separation of the graduations.

S.1.3. Indicators.

S.1.3.1. Symmetry. - The index of an indicator shall

be symmetrical with respect to the graduations, at least

throughout that portion of its length associated with

the graduations.

S.1.3.2. Length. - The index of an indicator shall

reach to the finest graduations with which it is used,

unless the indicator and the graduations are in the

same plane, in which case the distance between the

end of the indicator and the ends of the graduations,

measured along the line of the graduations, shall be

not more than 1.0 mm (0.04 in).

S.1.3.3. Width. - The width of the index of an

indicator in relation to the series of graduations with

which it is used shall not be greater than:

(a) the width of the narrowest graduation* , and

[*Nonretroactive as of January 1, 2002]

(Amended 2001)

(b) the width of the minimum clear interval between

graduations.

When the index of an indicator extends along the

entire length of a graduation, that portion of the index

of the indicator that may be brought into coincidence

with the graduation shall be of the same width

throughout the length of the index that coincides with

the graduation.

S.1.3.4. Clearance. - The clearance between the

index of an indicator and the graduations shall in no

case be more than 1.5 mm (0.06 in).

S.1.3.5. Parallax. - Parallax effects shall be reduced

to the practicable minimum.

S.2. Design of Measuring Elements.

S.2.1. Provision for Sealing. - Adequate provision shall

be made for applying security seals in such a manner that

no adjustment may be made of:

(a) any measurement elements, and

(b) any adjustable element for controlling delivery rate

when such rate tends to affect the accuracy of

deliveries.

The adjusting mechanism shall be readily accessible for

purposes of affixing a security seal.

S.2.2. Batching Meters Only.

S.2.2.1. Air Elimination. - Batching meters shall be

equipped with an effective air eliminator.

S.2.2.2. Directional Flow Valves. - Valves intended

to prevent reversal of flow shall be automatic in

operation.

N. Notes

N.1. Test Liquid. - A meter shall be tested with water.

N.2. Evaporation and Volume Change. - Care shall be

exercised to reduce to a minimum, evaporation losses and

volume changes resulting from changes to temperature of

the test liquid.

N.3. Test Drafts. - Test drafts should be equal to at least

the amount delivered by the device in 2 minutes and in no

case less than the amount delivered by the device in

1 minute at the actual maximum flow rate developed by the

installation. The test drafts shown in Table 1, next page,

shall be followed as closely as possible.
3.36. Water Meters



N.4. Testing Procedures.

N.4.1. Normal Tests. The normal test of a meter shall

be made at the maximum discharge rate developed by the

installation. Meters with maximum gallon per minute

ratings higher than Table 1 values may be tested up to the

meter rating.

(Amended 1990)

N.4.2. Special Tests. - Special tests to develop the

operating characteristics of meters may be made

according to the rates and quantities shown in Table 2.

N.4.3. Batching Meter Tests. - Tests on batching

meters should be conducted at the maximum and

intermediate rates only.

T. Tolerances

T.1. Tolerance Values. - Maintenance and acceptance

tolerances shall be as shown in Table 1 and Table 2.

UR. User Requirements

UR.1. Batching Meters Only.

UR.1.1. Strainer. - A filter or strainer shall be provided

if it is determined that the water contains excessive

amounts of foreign material.

UR.1.2. Siphon Breaker. - An automatic siphon

breaker or other effective means shall be installed in the

discharge piping at the highest point of outlet, in no case

below the top of the meter, to prevent siphoning of the

meter and permit rapid drainage of the pipe or hose.

UR.1.3. Provision for Testing. - Acceptable provisions

for testing shall be incorporated into all meter systems.

Such provisions shall include a two-way valve, or

manifold valving, and a pipe or hose installed in the

discharge line accessible to the proper positioning of the

test measure.
3.36. Water Meters



Table 1. Tolerances for Water Meters

Normal Tests

Meter size Rate of flow

(inches) (gal/min)

Maximum Rate

Meter Tolerance on overindication and underregistration gal ft3

5/8 15 50 5

3/4 25 50 5

1 40 100 10

1 1/2 80 300 40 1.5%

2 120 500 40

3 250 500 50

4 350 1 000 100

6 700 1 000 100

Table 2. Tolerances for Water Meters

Special Tests

Meter size indication indication

(inches)

Intermediate Rate Minimum Rate

Rate of Tolerance on Rate of

flow over- and un- flow

(gal/ derregis- (gal/

min) tration min)

Meter Meter Tolerance

gal ft gal ft registra- regis- 3 3

Under- Overtion tration

5/8 2 10 1 1/4 5 1

3/4 3 10 1 1/2 5 1

1 4 10 1 3/4 5 1

1 1/2 8 50 5 1.5% 1 1/2 10 1 5.0% 1.5%

2 15 50 5 2 10 1

3 20 50 5 4 10 1

4 40 100 10 7 50 5

6 60 100 10 12 50 5
3.37. Mass Flow Meters

3-59

Sec. 3.37. Mass Flow Meters 

A. Application

A.1. Liquids. - This code applies to devices that are

designed to dynamically measure the mass, or the mass and

density of liquids. It also specifies the relevant

examinations and tests that are to be conducted.

(Amended 1997)

A.2. Vapor (Gases). - This code applies to devices that

are designed to dynamically measure the mass of

hydrocarbon gas in the vapor state. Examples of these

products are propane, propylene, butanes, butylenes, ethane,

methane, natural gas and any other hydrocarbon gas/air mix.

S. Specifications

S.1. Indicating and Recording Elements.

S.1.1. Indicating Elements. - A measuring assembly

shall include an indicating element. Indications shall be

clear, definite, accurate, and easily read under normal

conditions of operation of the instrument.

S.1.2. Compressed Natural Gas Dispensers. - Except

for fleet sales and other price contract sales, a

compressed natural gas dispenser used to refuel vehicles

shall be of the computing type and shall indicate the

quantity, the unit price, and the total price of each

delivery. The dispenser shall display the mass measured

for each transaction either continuously on an external or

internal display accessible during the inspection and test

of the dispenser, or display the quantity in mass units by

using controls on the device.

(Added 1994)

S.1.3. Units. - 

S.1.3.1. Units of Measurement. - Deliveries shall be

indicated and recorded in grams, kilograms, metric

tons, pounds, tons, and/or liters, gallons, quarts, pints

and decimal subdivisions thereof. The indication of a

delivery shall be on the basis of apparent mass versus

a density of 8.0 g/cm3. The volume indication shall

be based on the mass measurement and an automatic

means to determine and correct for changes in product

density.

(Amended 1993 and 1997)

S.1.3.1.1. Compressed Natural Gas Used as an

Engine Fuel. - When compressed natural gas is

dispensed as an engine fuel, the delivered quantity

shall be indicated in "gasoline liter equivalent

(GLE) units" or "gasoline gallon equivalent (GGE)

units" (see definitions).

(Added 1994)

S.1.3.2. Numerical Value of Quantity-Value

Divisions. - The value of a scale interval shall be

equal to: 

- 1, 2, or 5, or

- a decimal multiple or submultiple of 1, 2, or 5.

S.1.3.3. Maximum Value of Quantity-Value

Divisions.

(a) The maximum value of the quantity-value

division for liquids shall be not greater than 0.2

percent of the minimum measured quantity.

(b) For dispensers of compressed natural gas used to

refuel vehicles, the value of the division for the

gasoline liter equivalent shall not exceed

0.01 GLE; the division for gasoline gallon

equivalent (GGE) shall not exceed 0.001 GGE.

The maximum value of the mass division shall

not exceed 0.001 kg or 0.001 lb.

(Amended 1994)

S.1.3.4. Values Defined. - Indicated values shall be

adequately defined by a sufficient number of figures,

words, symbols, or combinations thereof. A display

of "zero" shall be a zero digit for all displayed digits

to the right of the decimal mark and at least one to the

left.

S.2. Operating Requirements.

S.2.1. Return to Zero. - Except for measuring

assemblies in a pipeline:

(a) One indicator and the primary recording elements,

if the device is equipped to record, shall be

provided with a means for readily returning the

indication to zero either automatically or manually.
3.37. Mass Flow Meters

3-60

(b) It shall not be possible to return primary indicating

elements, or primary recording elements, beyond

the correct zero position.

(Amended 1993)

S.2.2. Indicator Reset Mechanism. - The reset

mechanism for the indicating element shall not be

operable during a delivery. Once the zeroing operation

has begun, it shall not be possible to indicate a value

other than the latest measurement, or "zeros" when the

zeroing operation has been completed. 

S.2.3. Nonresettable Indicator. - An instrument may

also be equipped with a nonresettable indicator if the

indicated values cannot be construed to be the indicated

values of the resettable indicator for a delivered quantity.

S.2.4. Provisions for Power Loss.

S.2.4.1. Transaction Information. - In the event of

a power loss, the information needed to complete any

transaction in progress at the time of the power loss

(such as the quantity and unit price, or sales price)

shall be determinable for at least 15 minutes at the

dispenser or at the console if the console is accessible

to the customer.

(Added 1993)

S.2.4.2. User Information. - The device memory

shall retain information on the quantity of fuel

dispensed and the sales price totals during power loss.

(Added 1993)

S.2.5. Display of Unit Price and Product Identity.

S.2.5.1. Unit Price. - A computing or moneyoperated device shall be able to display on each face

the unit price at which the device is set to compute or

to dispense.

(Added 1993)

S.2.5.2. Product Identity. - A device shall be able to

conspicuously display on each side the identity of the

product being dispensed.

(Added 1993)

55GNGEVKQPQH7PKV2TKEG'ZEGRVHQT

FKURGPUGTUWUGFGZENWUKXGN[HQTHNGGVUCNGUQVJGT

RTKEGEQPVTCEVUCNGUCPFVTWEMTGHWGNKPIGIVTWEM

UVQRFKURGPUGTUWUGFQPN[VQTGHWGNVTWEMUYJGPC

RTQFWEVQTITCFGKUQHHGTGFHQTUCNGCVOQTGVJCPQPG

WPKVRTKEGVJTQWIJCEQORWVKPIFGXKEGVJGUGNGEVKQP

QHVJGWPKVRTKEGUJCNNDGOCFGRTKQTVQFGNKXGT[WUKPI

EQPVTQNUQPVJGFGXKEGQTQVJGTEWUVQOGTCEVKXCVGF

EQPVTQNU#U[UVGOUJCNNPQVRGTOKVCEJCPIGVQVJG

WPKVRTKEGFWTKPIFGNKXGT[QHCRTQFWEV

=0QPTGVTQCEVKXGCUQH,CPWCT[?


#FFGF

5#ITGGOGPV$GVYGGP+PFKECVKQPU9JGPC

SWCPVKV[XCNWGKPFKECVGFQTTGEQTFGFD[CPCWZKNKCT[

GNGOGPVKUCFGTKXGFQTEQORWVGFXCNWGDCUGFQP

FCVCTGEGKXGFHTQOCTGVCKNOQVQTHWGNFKURGPUGTVJG

XCNWGOC[FKHHGTHTQOVJGSWCPVKV[XCNWGFKURNC[GFQP

VJGFKURGPUGTRTQXKFGFVJGHQNNQYKPIEQPFKVKQPUCTG

OGV


CCNNVQVCNOQPG[XCNWGUHQTCPKPFKXKFWCNUCNG

VJCVCTGKPFKECVGFQTTGEQTFGFD[VJGU[UVGO

CITGGCPF


DYKVJKPGCEJGNGOGPVVJGXCNWGUKPFKECVGFQT

TGEQTFGFOGGVVJGHQTOWNCSWCPVKV[ZWPKVRTKEG

VQVCNUCNGURTKEGVQVJGENQUGUVEGPV

[Nonretroactive as of January 1, 1998]


#FFGF

S.2.6. Money-Value Computations. - A computing

device shall compute the total sales price at any singlepurchase unit price (i.e., excluding fleet sales, other price

contract sales, and truck stop dispensers used only to

refuel trucks) for which the product being measured is

offered for sale at any delivery possible within either the

measurement range of the device or the range of the

computing elements, whichever is less.

(Added 1993)

S.2.6.1. Auxiliary Elements. - If a system is

equipped with auxiliary indications, all indicated

money value and quantity divisions of the auxiliary

element shall be identical with those of the primary

element.

(Added 1993)

S.2.6.2. Display of Quantity and Total Price. -

When a delivery is completed, the total price and

quantity for that transaction shall be displayed on the

face of the dispenser for at least 5 minutes or until the

next transaction is initiated by using controls on the

device or other user-activated controls.

(Added 1993)

S.2.7. Recorded Representations, Point of Sale

Systems. - The sales information recorded by cash

registers when interfaced with a retail motor-fuel

dispenser shall contain the following information for

products delivered by the dispenser:

(a) the total volume of the delivery,

(b) the unit price,
3.37. Mass Flow Meters

3-61

(c) the total computed price, and approved by the weights and measures jurisdiction

(d) the product identity by name, symbol, abbreviation, having statutory authority over the device, such that

or code number. neither air nor vapor can enter the system.

[Nonretroactive as of January 1, 1986.]

(Added 1993) (b) Vent lines from the air or vapor eliminator (if

5+PFKECVKQPQH&GNKXGT[6JGFGXKEGUJCNN

CWVQOCVKECNN[UJQYQPKVUHCEGVJGKPKVKCN\GTQEQPFKVKQP

CPFVJGSWCPVKV[FGNKXGTGFWRVQVJGPQOKPCNECRCEKV[

*QYGXGTVJGHKTUV.ICNQHCFGNKXGT[CPF

KVUCUUQEKCVGFVQVCNUCNGURTKEGPGGFPQVDGKPFKECVGF

=0QPTGVTQCEVKXGCUQH,CPWCT[?


#FFGF

S.3. Measuring Elements and Measuring Systems.

S.3.1. Maximum and Minimum Flow-Rates.

(a) The ratio of the maximum to minimum flow-rates

specified by the manufacturer for devices

measuring liquified gases shall be 5:1 or greater.

(b) The ratio of the maximum to minimum flow-rates

specified by the manufacturer for devices

measuring other than liquified gases shall be 10:1

or greater.

S.3.2. Adjustment Means. - An assembly shall be

provided with means to change the ratio between the

indicated quantity and the quantity of liquid measured by

the assembly. A bypass on the measuring assembly shall

not be used for these means.

S.3.2.1. Discontinuous Adjusting Means. When the

adjusting means changes the ratio between the

indicated quantity and the quantity of measured liquid

in a discontinuous manner, the consecutive values of

the ratio shall not differ by more than 0.1 percent.

S.3.3. Vapor Elimination. - A liquid-measuring

instrument or measuring system shall be equipped with

an effective vapor or air eliminator or other effective

means, automatic in operation, to prevent the

measurement of vapor and air. Vent lines from the air or

vapor eliminator shall be made of metal tubing or some

other suitable rigid material.

(Amended 1999)

S.3.3.1. Vapor Elimination on Loading Rack

Liquid Metering Systems. -

(a) A loading rack liquid metering system shall be

equipped with a vapor or air eliminator or other

automatic means to prevent the passage of vapor

and air through the meter unless the system is

designed or operationally controlled by a method,

present) shall be made of metal tubing or other

rigid material.

(Added 1995)

S.3.4. Maintenance of Liquid State. - A liquidmeasuring device shall be installed so that the measured

product remains in a liquid state during passage through

the instrument.

S.3.5. Provision for Sealing. - Adequate provision shall

be made for an approved means of security (e.g., data

change audit trail) or physically applying security seals in

such a manner that no adjustment may be made of:

(a) any measurement element;

(b) any adjustable element for controlling delivery rate

when such rate tends to affect the accuracy of

deliveries; or

(c) the zero adjustment mechanism.

When applicable, the adjusting mechanism shall be

readily accessible for purposes of affixing a security seal.

Audit trails shall use the format set forth in Table S.3.5.

(Amended 1992 and 1995)

S.3.6. Automatic Density Correction.

(a) An automatic means to determine and correct for

changes in product density shall be incorporated in

any mass flow metering system that is affected by

changes in the density of the product being

measured.

(b) Volume-measuring devices with automatic

temperature compensation used to measure natural

gas as a motor vehicle engine fuel shall be

equipped with an automatic means to determine and

correct for changes in product density due to

changes in the temperature, pressure, and

composition of the product.

(Amended 1994 and 1997)

S.3.7. Pressurizing the Discharge Hose. - The

discharge hose for compressed natural gas shall

automatically pressurize prior to the device beginning to

register the delivery.

(Added 1993)
3.37. Mass Flow Meters

3-62

Table S.3.5. Categories of Device and Methods of Sealing

Categories of Device Method of Sealing

Category 1: No remote configuration

capability.

Seal by physical seal or two event counters: one for

calibration parameters and one for configuration parameters.

Category 2: Remote configuration

capability, but access is controlled by

physical hardware. 

Device shall clearly indicate that it is in the

remote configuration mode and record such

message if capable of printing in this mode

or shall not operate while in this mode.

[Category 2 applies to only devices

manufactured prior to January 1, 2005. 

Devices with remote configuration capability

manufactured or in use after that date must

meet the sealing requirements outlined in

Category 3. Devices without remote

configuration capability manufactured after

that date will be required to meet the

minimum criteria outlined in Category 1].

=6JGJCTFYCTGGPCDNKPICEEGUUHQTTGOQVGEQOOWPKECVKQP

OWUVDGQPUKVG6JGJCTFYCTGOWUVDGUGCNGFWUKPIC

RJ[UKECNUGCNQTGXGPVEQWPVGTHQTECNKDTCVKQPRCTCOGVGTU

CPFCPGXGPVEQWPVGTHQTEQPHKIWTCVKQPRCTCOGVGTU6JG

GXGPVEQWPVGTUOC[DGNQECVGFGKVJGTCVVJGKPFKXKFWCN

OGCUWTKPIFGXKEGQTCVVJGU[UVGOEQPVTQNNGTJQYGXGTCP

CFGSWCVGPWODGTQHEQWPVGTUOWUVDGRTQXKFGFVQOQPKVQT

VJGECNKDTCVKQPCPFEQPHKIWTCVKQPRCTCOGVGTUQHVJG

KPFKXKFWCNFGXKEGUCVCNQECVKQP+HVJGEQWPVGTUCTGNQECVGF

KPVJGU[UVGOEQPVTQNNGTTCVJGTVJCPCVVJGKPFKXKFWCNFGXKEG

OGCPUOWUVDGRTQXKFGFVQIGPGTCVGCJCTFEQR[QHVJG

KPHQTOCVKQPVJTQWIJCPQPUKVGFGXKEG?

=0QPTGVTQCEVKXGCUQH,CPWCT[?

%CVGIQT[4GOQVGEQPHKIWTCVKQP

ECRCDKNKV[CEEGUUOC[DGWPNKOKVGFQT

EQPVTQNNGFVJTQWIJCUQHVYCTGUYKVEJGI

RCUUYQTF

6JGFGXKEGUJCNNENGCTN[KPFKECVGVJCVKVKUKP

VJGTGOQVGEQPHKIWTCVKQPOQFGCPFTGEQTF

UWEJOGUUCIGKHECRCDNGQHRTKPVKPIKPVJKU

OQFGQTUJCNNPQVQRGTCVGYJKNGKPVJKU

OQFG

=0QPTGVTQCEVKXGCUQH,CPWCT[?

Nonretroactive as of January 1, 2005, all

devices with remote configuration capability

must comply with the sealing requirements of

Category 3.

#PGXGPVNQIIGTKUTGSWKTGFKPVJGFGXKEGKVOWUVKPENWFGCP

GXGPVEQWPVGTVQVJGRCTCOGVGT+&VJGFCVGCPF

VKOGQHVJGEJCPIGCPFVJGPGYXCNWGQHVJGRCTCOGVGT#

RTKPVGFEQR[QHVJGKPHQTOCVKQPOWUVDGCXCKNCDNGVJTQWIJ

VJGFGXKEGQTVJTQWIJCPQVJGTQPUKVGFGXKEG6JGGXGPV

NQIIGTUJCNNJCXGCECRCEKV[VQTGVCKPTGEQTFUGSWCNVQVGP

VKOGUVJGPWODGTQHUGCNCDNGRCTCOGVGTUKPVJGFGXKEGDWV

PQVOQTGVJCPTGEQTFUCTGTGSWKTGF0QVG&QGUPQV

TGSWKTGEJCPIGUVQDGUVQTGFHQTGCEJRCTCOGVGT

0QPTGVTQCEVKXGCPFGPHQTEGCDNGCUQH,CPWCT[?


6CDNG#FFGF
#OGPFGFCPF

S.3.8. Zero-Set-Back Interlock, Retail Motor-Fuel

Devices. - A device shall be constructed so that:

(a) after a delivery cycle has been completed by

moving the starting lever to any position that shuts

off the device, an automatic interlock prevents a

subsequent delivery until the indicating elements,

and recording elements if the device is equipped

and activated to record, have been returned to their

zero positions;

(b) the discharge nozzle cannot be returned to its

designed hanging position (that is, any position

where the tip of the nozzle is placed in its designed

receptacle and the lock can be inserted) until the
3.37. Mass Flow Meters

3-63

starting lever is in its designed shut-off position and

the zero-set-back interlock has been engaged; and

(c) in a system with more than one dispenser supplied

by a single pump, an effective automatic control

valve in each dispenser prevents product from

being delivered until the indicating elements on that

dispenser are in a correct zero position.

(Added 1993)

S.4. Discharge Lines and Valves.

S.4.1. Diversion of Measured Product. - No means

shall be provided by which any measured product can be

diverted from the measuring instrument. However, two

or more delivery outlets may be permanently installed

and operated simultaneously, provided that any diversion

of flow to other than the intended receiving receptacle

cannot be readily accomplished or is readily apparent.

Such means include physical barriers, visible valves or

indications that make it clear which outlets are in

operation, and explanatory signs if deemed necessary.

A manually controlled outlet that may be opened for

purging or draining the measuring system shall be

permitted. Effective means shall be provided to prevent

the passage of liquid through any such outlet during

normal operation of the measuring system.

S.4.2. Pump-Discharge Unit. - A pump-discharge unit

for liquids equipped with a flexible discharge hose shall

be of the wet-hose type.

(Added 1993)

S.4.3. Directional Flow Valves. - If a reversal of flow (h) applicable range of temperature if other than -10 EC to

could result in errors that exceed the tolerance for the +50 EC;

minimum measured quantity, a valve or valves or other

effective means, automatic in operation (and equipped (i) minimum measured quantity; and

with a pressure limiting device, if necessary) to prevent

the reversal of flow shall be properly installed in the (j) product limitations, if applicable.

system. (See N.1.) [*Nonretroactive as of January 1, 1995.]

S.4.4. Discharge Valves. - A discharge valve may be S.5.1. Marking of Gasoline Volume Equivalent

installed on a discharge line only if the system is a Conversion Factor. - A device dispensing

wet-hose type. Any other shutoff valve on the discharge compressed natural gas shall have either the statement

side of the instrument shall be of the automatic or "1 Gasoline Liter Equivalent (GLE) is Equal to

semiautomatic predetermined-stop type or shall be 0.678 kg of Natural Gas" or "1 Gasoline Gallon

operable only: Equivalent (GGE) is Equal to 5.660 lb of Natural

- by means of a tool (but not a pin) entirely separate from face of the dispenser according to the method of sale

the device, or used.

- by means of a security seal with which the valve is

sealed open.

S.4.5. Antidrain Means. - In a wet-hose type device,

effective means shall be provided to prevent the drainage

of the hose between transactions.

S.4.6. Other Valves. - Check valves and closing

mechanisms that are not used to define the measured

quantity shall have relief valves (if necessary) to

dissipate any abnormally high pressure that may arise in

the measuring assembly.

S.5. Markings. - A measuring system shall be legibly and

indelibly marked with the following information:

(a) pattern approval mark (i.e., type approval number);

(b) name and address of the manufacturer or his

trademark and, if required by the weights and

measures authority, the manufacturer’s identification

mark in addition to the trademark;

(c) model designation or product name selected by the

manufacturer;

(d) nonrepetitive serial number;

(e) the accuracy class of the meter as specified by the

manufacturer consistent with Table T.2.;*

(Added 1994)

(f) maximum and minimum flow rates in pounds per unit

of time;

(g) maximum working pressure;

Gas" permanently and conspicuously marked on the

(Added 1994)

S.6. Printer. - When an assembly is equipped with means

for printing the measured quantity, the following conditions

apply:
3.37. Mass Flow Meters

3-64

(a) the scale interval shall be the same as that of the test shall include a test for accuracy using five starts and

indicator; stops during a delivery to simulate the operation of the

(b) the value of the printed quantity shall be the same part of the normal inspection and test of the meter.

value as the indicated quantity;

(c) a quantity for a delivery (other than an initial

reference value) cannot be recorded until the

measurement and delivery has been completed;

(d) the printer is returned to zero when the resettable

indicator is returned to zero; and 

(e) the printed values shall meet the requirements

applicable to the indicated values.

S.6.1. Printed Receipt. - Any delivered, printed

quantity shall include an identification number, the time

and date, and the name of the seller. This information

may be printed by the device or pre-printed on the ticket.

S.7. Totalizers for Retail Motor-Fuel Devices.- Retail

motor-fuel dispensers shall be equipped with a

nonresettable totalizer for the quantity delivered through

the metering device.

=0QPTGVTQCEVKXGCUQH,CPWCT[?


#FFGF

N. Notes

N.1. Minimum Measured Quantity. - The minimum

measured quantity shall be specified by the manufacturer.

N.2. Test Medium.

N.2.1. Liquid-Measuring Devices. - The device shall

be tested with the liquid that the device is intended to

measure or another liquid with the same general physical

characteristics.

N.2.2. Vapor-Measuring Devices. - The device shall be

tested with air or the product to be measured.

N.3. Test Drafts. - The minimum test shall be one test

draft at the maximum flow rate of the installation and one

test draft at the minimum flow rate. More tests may be

performed at these or other flow rates. (See T.3.)

N.4. Minimum Measured Quantity. - The device shall be

tested for a delivery equal to the declared minimum

measured quantity when the device is likely to be used to

make deliveries on the order of the minimum measured

quantity.

N.5. Motor Fuel Dispenser. - When a device is intended

for use as a liquid motor-fuel dispenser, the type evaluation

automatic shut-off nozzle. This test may be conducted as

N.6 Testing Procedures.

N.6.1. Normal Tests. - The normal test of a meter shall

be made at the maximum discharge rate developed by the

installation. Any additional tests conducted at flow rates

down to and including the rated minimum discharge flow

rate shall be considered normal tests.

(Added 1999)

N.6.1.1. Repeatability Tests. - Tests for repeatability

should include a minimum of three consecutive test

drafts of approximately the same size and be

conducted under controlled conditions where

variations in factors, such as, temperature pressure

and flow rate are reduced to the extent that they will

not affect the results obtained.

(Added 2001)

N.6.2. Special Tests. - Special tests to develop the

operating characteristics of a meter and any special

elements and accessories attached to or associated with

the device, shall be made as circumstances require. Any

test except as set forth in N.6.1. shall be considered a

special test. Special test of a measuring system shall be

made to develop operating characteristics of the

measuring systems during a split compartment delivery.

(See Table T.2.)

(Added 1999)
3.37. Mass Flow Meters

3-65

Table T.2. Accuracy Classes for Mass Flow Meter Applications

Accuracy Class Application or Commodity Being Measured Acceptance

Tolerance

Maintenance

Tolerance

Special

Tolerance

0.3

Loading rack meters, vehicle-tank meters,

home heating oil, heated products (except

asphalt above 50 EC), asphalt 50 EC or

below, milk and other food products, large

capacity motor-fuel dispensers (maximum

discharge flow rates greater than 100 L or

25 gallon per minute), all other liquid

applications not shown in the table where

the minimum delivery is at least 700 kg

(1500 lb)

0.2% 0.3% 0.5%

0.3A Asphalt above 50 EC 0.3% 0.3% 0.5%

0.5

Small capacity (retail) motor-fuel

dispensers, agri-chemical liquids, all other

liquid applications not shown in the table

0.3% 0.5% 0.5%

1.0 Anhydrous ammonia, LP Gas (including

vehicle tank meters) 0.6% 1.0% 1.0%

2.0 Compressed natural gas as a motor fuel 1.5% 2.0% 2.0%

2.5 Cryogenic liquid meters, liquefied

compressed gases other than LP Gas

1.5% 2.5% 2.5%

(Added 1994) (Amended 1999 and 2001)

T. Tolerances

T.1. Tolerances, General.

(a) The tolerances apply equally to errors of

underregistration and errors of overregistration.

(b) The tolerances apply to all products at all temperatures

measured at any flow rate within the rated measuring

range of the meter.

(Amended 1999)

T.2. Tolerances. - The tolerances for mass flow meters for

specific liquids, gases, and applications are listed in Table

T.2.

(Amended 1994 and 1999)

T.3. Repeatability. - When multiple tests are conducted at UR.1.1. Discharge Hose-Length. - The length of the

approximately the same flow rate and draft size, the range discharge hose on a retail motor-fuel device shall not

of the test results for the flow rate shall not exceed 40 exceed 4.6 m (15 ft) unless it can be demonstrated that a

percent of the absolute value of the maintenance tolerance longer hose is essential to permit deliveries to be made

and the results of each test shall be within the applicable to receiving vehicles or vessels.

tolerance. See also N.6.1.1. [Nonretroactive as of January 1, 1998]

(Amended 1992, 1994, and 2001)

T.4. Type Evaluation Examinations for LiquidMeasuring Devices. - For type evaluation examinations,

the tolerance values shall apply under the following

conditions:

(a) with any one liquid within the range of liquids,

(b) at any one liquid temperature and pressure within the

operating range of the meter, and

(c) at all flow rates within the range of flow rates.

(Added 1993) (Amended 1994)

UR. User Requirements

UR.1. Selection Requirements.


#FFGF

UR.1.2. Minimum Measured Quantity.
3.37. Mass Flow Meters

3-66

(a) The minimum measured quantity shall be specified (b) in the case of a computing type or money-operated

by the manufacturer. type, the unit price at which the dispenser is set to

(b) The minimum measured quantity appropriate for a (Added 1993)

transaction may be specified by the weights and

measures authority. A device may have a minimum

measured quantity smaller than that specified by the

weights and measures authority; however, the

device must perform within the performance

requirements for the declared minimum measured

quantity.

UR.2. Installation Requirements. normal deliveries can be made without the use of

UR.2.1. /CPWHCEVWTGT	U+PUVTWEVKQPU#FGXKEG

UJCNNDGKPUVCNNGFKPCEEQTFCPEGYKVJVJGOCPWHCEVWTGT U

KPUVTWEVKQPUCPFVJGKPUVCNNCVKQPUJCNNDGUWHHKEKGPVN[

UGEWTGCPFTKIKFVQOCKPVCKPVJKUEQPFKVKQP


#FFGF

UR.2.2. &KUEJCTIG4CVG#FGXKEGUJCNNDGKPUVCNNGF

UQVJCVVJGCEVWCNOCZKOWOFKUEJCTIGTCVGYKNNPQV

GZEGGFVJGTCVGFOCZKOWOFKUEJCTIGTCVG#WVQOCVKE

OGCPUQHHNQYTGIWNCVKQPUJCNNDGKPEQTRQTCVGFKPVJG

KPUVCNNCVKQPKHPGEGUUCT[


#FFGF

UR.2.3. Low-Flow Cut-Off Valve. - If a metering

system is equipped with a programmable or adjustable

"low-flow cut-off" feature:

(a) the low-flow cut-off value shall not be set at flow

rates lower than the minimum operating flow rate

specified by the manufacturer on the meter; and

(b) the system shall be equipped with flow control

valves which prevent the flow of product and stop

the indicator from registering product flow

whenever the product flow rate is less than the lowflow cut-off value.

(Added 1992)

UR.3. Use of Device.

UR.3.1. Unit Price and Product Identity for Retail

Dispensers. - The following information shall be

conspicuously displayed or posted on the face of a retail

dispenser used in direct sale:

(a) Except for dispensers used exclusively for fleet

sales, other price contract sales, and truck refueling

(e.g., truck stop dispensers used only to refuel

trucks), all of the unit prices at which the product is

offered for sale; and

compute.

UR.3.2. Vapor-Return Line. - During any metered

delivery of liquefied petroleum gas and other liquids

from a supplier’s tank to a receiving container, there shall

be no vapor-return line from the receiving container to

the supplier’s tank:

(a) in the case of any receiving container to which

such vapor-return line, or

(b) in the case of any new receiving container when the

ambient temperature is below 90 EF.

(Added 1993)

UR.3.3. Ticket Printer; Customer Ticket. - Vehiclemounted metering systems shall be equipped with a

ticket printer which shall be used for all sales where

product is delivered through the meter. A copy of the

ticket issued by the device shall be left with the customer

at the time of delivery or as otherwise specified by the

customer. [Nonretroactive as of January 1, 1995. To

become retroactive as of January 1, 1999.]

(Added 1994)

UR.3.4. Printed Ticket. - The total price, the total

quantity of the delivery, and the price per unit shall be

printed on any ticket issued by a device of the computing

type and containing any one of these values.

(Added 1993)

UR.3.5. Ticket in Printing Device. - A ticket shall not

be inserted into a device equipped with a ticket printer

until immediately before a delivery is begun, and in no

case shall a ticket be in the device when the vehicle is in

motion while on a public street, highway, or

thoroughfare.

(Added 1993)

UR.3.6. Steps After Dispensing. - After delivery to a

customer from a retail motor-fuel device:

(a) the starting lever shall be returned to its shutoff

position and the zero-set-back interlock engaged;

and
3.37. Mass Flow Meters

3-67

(b) the discharge nozzle shall be returned to its

designed hanging position unless the primary

indicating elements, and recording elements, if the

device is equipped and activated to record, have

been returned to a definite zero indication.

(Added 1993)

UR.3.7. Return of Indicating and Recording

Elements to Zero. - The primary indicating elements

(visual), and the primary recording elements when these

are returnable to zero, shall be returned to zero

immediately before each delivery. 'ZEGRVKQPUVQVJKU

TGSWKTGOGPVCTGVQVCNK\GTUQPMG[NQEMQRGTCVGFQTQVJGT

UGNHQRGTCVGFFKURGPUGTUCPFVJGRTKOCT[TGEQTFKPI

GNGOGPVKHVJGFGXKEGKUGSWKRRGFVQTGEQTF

(Added 1995) (Amended 1997)

UR.3.8. Return of Product to Storage, Retail

Compressed Natural Gas Dispensers.- Provisions at

the site shall be made for returning product to storage or

disposing of the product in a safe and timely manner

during or following testing operations. Such provisions

may include return lines, or cylinders adequate in size

and number to permit this procedure.

(Added 1998)
3.38. Carbon Dioxide Liquid-Measuring Devices

3-69

Section 3.38. Carbon Dioxide 

Liquid-Measuring Devices - Tentative Code

This tentative code has only a trial or experimental status and is not intended to be enforced by weights and measures

officials. The requirements are designed for study prior to the development and adoption of a final code for Carbon

Dioxide Liquid-Measuring Devices. The tentative code is intended to be used by the National Type Evaluation

Program for type evaluation of carbon dioxide liquid-measuring devices. If upgraded to become a permanent code,

all requirements, except those for tolerances, will be nonretroactive as of the effective date of the permanent code;

tolerance requirements will apply retroactively as of the effective date of the permanent code.

(Tentative Code Added 1996) (Amended 1998)

A. Application

A.1. This code applies to carbon dioxide liquid-measuring

devices used for the measurement of liquid carbon dioxide.

A.2. This code does not apply to devices used solely for

dispensing a product in connection with operations in which

the amount dispensed does not affect customer charges.

A.3. See also, Sec. 1.10; General Code requirements.

#6[RG'XCNWCVKQP6JG0CVKQPCN6[RG'XCNWCVKQP

2TQITCOYKNNCEEGRVHQTV[RGGXCNWCVKQPQPN[VJQUGFGXKEGU

VJCVEQORN[YKVJCNNTGSWKTGOGPVUQHVJKUEQFG


#FFGF

S. Specifications

S.1. Design of Indicating and Recording Elements and

of Recorded Representations. 

S.1.1. Primary Elements. -

S.1.1.1. General. - A device shall be equipped

with a primary indicating element and may also

be equipped with a primary recording element.

S.1.1.2. Units. - A device shall indicate and

record, if equipped to record, its deliveries in

terms of pounds or kilograms or decimal

subdivisions or multiples thereof.

S.1.1.3. Value of Smallest Unit. - The value of

the smallest unit of indicated delivery, and

recorded delivery, if the device is equipped to

record, shall not exceed the equivalent of:

(a) for small delivery devices:

(1) 1 kilogram, or

(2) 1 pound 

(b) for large delivery devices:

(1) 10 kilograms , or

(2) 10 pounds

S.1.1.4. Advancement of Indicating and

Recording Elements. - Primary indicating and

recording elements shall be susceptible to

advancement only by the normal operation of the

device. However, a device may be cleared by

advancing its elements to zero, but only if:

(a) the advancing movement, once started,

cannot be stopped until zero is reached, or

(b) in the case of indicating elements only,

such elements are automatically obscured

until the elements reach the correct zero

position.

S.1.1.5. Return to Zero. - Primary indicating

and recording elements shall be readily

returnable to a definite zero indication. Means

shall be provided to prevent the return of primary

indicating elements and of primary recording

elements beyond their correct zero position.

S.1.2. Graduations. -

S.1.2.1. Length. - Graduations shall be so

varied in length that they may be conveniently

read.

S.1.2.2. Width. - In any series of graduations,

the width of a graduation shall in no case be

greater than the width of the minimum clear

interval between graduations. The width of main

graduations shall be not more than 50 percent

greater than the width of subordinate

graduations. Graduations shall in no case be less

than 0.2 mm (0.008 in) in width.
3.38. Carbon Dioxide Liquid-Measuring Devices

3-70

S.1.2.3. Clear Interval Between Graduations. S.1.3.6. Travel of Indicator. - If the most

- The clear interval shall be not less than 1.0 mm

(0.04 in). If the graduations are not parallel, the

measurement shall be made:

(a) along the line of relative movement

between the graduations at the end of the

indicator, or

(b) if the indicator is continuous, at the point of

widest separation of the graduations.

(See also S.1.3.6.)

S.1.3. Indicators. -

S.1.3.1. Symmetry. - The index of an indicator

shall be of the same shape as the graduations at

least throughout that portion of its length

associated with the graduations.

S.1.3.2. Length. - The index of an indicator

shall reach to the finest graduations with which

it is used, unless the indicator and the

graduations are in the same plane, in which case

the distance between the end of the indicator and

the ends of the graduations, measured along the

line of the graduations, shall be not more than

1.0 mm (0.04 in).

S.1.3.3. Width. - The width of the index of the

indicator in relation to the series of graduations

with which it is used shall be not greater than:

(a) the width of the narrowest graduation*,

and

[*Nonretroactive as of January 1, 2002]

(Amended 2001)

(b) the width of the minimum clear interval

between graduations.

When the index of an indicator extends along the

entire length of a graduation, that portion of the

index of the indicator that may be brought into

coincidence with the graduation shall be of the

same width throughout the length of the index

that coincides with the graduation.

S.1.3.4. Clearance. - The clearance between

the index of an indicator and the graduations

shall in no case be more than 1.5 mm (0.06 in).

S.1.3.5. Parallax. - Parallax effects shall be

reduced to the practicable minimum.

sensitive element of the primary indicating

element utilizes an indicator and graduations, the

relative movement of these parts corresponding

to the smallest indicated value shall be no less

than 5 mm (0.20 in).

S.1.4. Computing-Type Devices. -

S.1.4.1. Printed Ticket. - Any printed ticket

issued by a device of the computing type on

which there is printed the total computed price

shall have printed clearly thereon also the total

quantity of the delivery and the price per unit.

S.1.4.2. Money-Value Computations. -

Money-value computations shall be of the fullcomputing type in which the money value at a

single unit price, or at each of a series of unit

prices, shall be computed for every delivery

within either the range of measurement of the

device or the range of the computing elements,

whichever is less.

The total price shall be computed on the basis of

the quantity indicated when the value of the

smallest division indicated is equal to or less

than the value specified in S.1.1.3.

S.1.4.3. Money-Values, Mathematical

Agreement. - Any digital money-value

indication and any recorded money value on a

computing-type device shall be in mathematical

agreement with its associated quantity indication

or representation to within one cent of money

value.

S.2. Design of Measuring Elements.

S.2.1. Vapor Elimination. - A measuring system

shall be equipped with an effective vapor eliminator

or other effective means to prevent the measurement

of vapor that will cause errors in excess of the

applicable tolerances.

S.2.2. Reverse Flow Measurement. - Effective

means, automatic in operation, shall be installed to

prevent reverse flow measurement. 

S.2.3. Maintenance of Liquid State. - A device

shall be so designed that the product being measured

will remain in a liquid state during passage through

the device.
3.38. Carbon Dioxide Liquid-Measuring Devices

3-71

S.2.4. Automatic Temperature or Density S.4.2. Discharge Rates. - A meter shall be marked to

Compensation. - A volumetric device shall be show its designed maximum and minimum discharge

equipped with automatic means for adjusting the rates. The marked minimum discharge rate shall not

indication and recorded representation of the exceed 20 percent of the maximum discharge rate.

measured quantity of the product to indicate or record

the quantity of the product measured in terms of

pounds.

S.2.5. Provision for Sealing. - Adequate provision level beyond the limit required for the performance to be

shall be made for applying security seals in such a within the applicable tolerances.

manner that no adjustment or interchange may be

made of:

(a) any measurement element,

(b) any adjustable element for controlling delivery

rate when such rate tends to affect the accuracy

of deliveries, and

(c) any automatic temperature or density

compensating system.

Any adjusting mechanism shall be readily accessible

for purposes of affixing a security seal.

S.3. Design of Discharge Lines and Discharge Line

Valves.

S.3.1. Diversion of Measured Liquid. - No means

shall be provided by which any measured liquid can be

diverted from the measuring chamber of the device or

the discharge line therefrom, except that a manually

controlled outlet that may be opened for purging or

draining the measuring system shall be permitted.

Effective means shall be provided to prevent the

passage of liquid through any such outlet during

normal operation of the device and to indicate clearly

and unmistakably when the valve controls are so set as

to permit passage of liquid through such outlet.

S.3.2. Discharge Hose. - The discharge hose of a

measuring system shall be of a wet hose type with a

shutoff valve at its outlet end.

S.4. Marking Requirements.

S.4.1. Limitation of Use . - If a measuring system is

intended to measure accurately only liquids having

particular properties, or to measure accurately only

under specific installation or operating conditions, or

to measure accurately only when used in conjunction

with specific accessory equipment, these limitations

shall be clearly and permanently marked on the device.

S.5. Level Condition, On-Board Weighing Systems. -

Provision shall be made for automatically inhibiting the

delivery of liquid carbon dioxide when the vehicle is out of

N. Notes

N.1. Test Liquid. - The test liquid shall be carbon dioxide

in a compressed liquid state.

N.2. Vaporization and Volume Change. - Care shall be

exercised to reduce vaporization and volume changes to a

minimum. When testing by weight, the weigh tank and

transfer systems shall be pre-cooled to liquid temperature

prior to the start of the test to avoid the venting of vapor

from the vessel being weighed.

N.3. Test Drafts.

N.3.1. Gravimetric Test. - Weight test drafts shall

be equal to at least the amount delivered by the device

in two minutes at its maximum discharge rate.

N.3.2. Transfer Standard Test. - When comparing

a meter with a calibrated transfer standard, the test

draft shall be equal to at least the amount delivered by

the device in two minutes at its maximum discharge

rate.

N.3.3. Volumetric Prover Test Drafts. - Test drafts

shall be equal to at least the amount delivered in one

minute at normal discharge rate.
3.38. Carbon Dioxide Liquid-Measuring Devices

3-72

N.4. Testing Procedures. T.2.1. On Normal Tests. - The maintenance

N.4.1. Normal Tests. - The "normal" test of a device percent (2.5%) of the indicated quantity. The

shall be made at the maximum discharge flow rate acceptance tolerances shall be one and one-half

developed under the conditions of installation. Any percent (1.5%) of the indicated quantity.

additional tests conducted at flow rates down to and

including one-half of the sum of the maximum

discharge flow rate and the rated minimum discharge

flow rate shall be considered normal tests.

N.4.2. Special Tests. - Any test except as set forth in

N.4.1. shall be considered a special test. Tests shall

be conducted, if possible, to evaluate any special

elements or accessories attached to or associated with

the device. A device shall be tested at a minimum

discharge rate of:

(a) not less than the minimum rated capacity or 20

percent of the maximum rated discharge rate of

the device, whichever is less, or

(b) the lowest discharge rate practicable under the

conditions of installation.

"Special" tests may be conducted to develop any

characteristics of the device anticipated under the

conditions of installation as circumstances require.

N.4.3. Density. - Temperature and pressure of the

metered test liquid shall be measured during the test

for the determination of density or volume correction

when applicable. The appropriate correction values

shall apply as specified in Table 1.

N.4.4. Automatic Temperature or Density

Compensation. - If a device is equipped with an

automatic temperature or density compensator, the

compensator shall be tested by comparing the quantity

indicated or recorded by the device (with the

compensator connected and operating) with the actual

delivered quantity. The appropriate correction values

shall apply as specified in Table 1.

T. Tolerances

T.1. Application.

T.1.1. To Underregistration and to

Overregistration. - The tolerances hereinafter

prescribed shall be applied to errors of

underregistration and errors of overregistration.

T.2. Tolerance Values.

tolerance on "normal" tests shall be two and one-half

T.2.2. On Special Tests. - The maintenance and

acceptance tolerance on "special" tests shall be two

and one-half percent (2.5%) of the indicated quantity.

T.3. On Tests Using Transfer Standards. - To the basic

tolerance values that would otherwise be applied, there shall

be added an amount equal to two times the standard

deviation of the applicable transfer standard when

compared to a basic reference standard.

UR. User Requirements

UR.1. Installation Requirements.

UR.1.1. Discharge Rate. - A device shall be so

installed that the actual maximum discharge rate will

not exceed the rated maximum discharge rate. If

necessary, means for flow regulation shall be

incorporated in the installation.

UR.1.2. Length of Discharge Hose. - The discharge

hose shall be of such a length and design as to keep

vaporization of the liquid to a minimum.

UR.1.3. Maintenance of Liquid State. - A device

shall be so installed and operated that the product

being measured shall remain in the liquid state during

passage through the meter.

UR.2. Use Requirements.

UR.2.1. Return of Indicating and Recording

Elements to Zero. - The primary indicating elements

(visual) and the primary recording elements shall be

returned to zero immediately before each delivery.

UR.2.2. Condition of Discharge System. - The

discharge hose, up to the valve at the end of the

discharge hose, shall be completely filled and

precooled to liquid temperatures before a "zero"

condition is established and prior to the start of a

commercial delivery. Means shall be provided to fill

the discharge hose with liquid prior to the start of a

delivery.

UR.2.3. Vapor Equalization Line. - A vapor

equalization line shall not be used during a metered

delivery unless the quantity of vapor displaced from

the buyer’s tank to the seller’s tank is deducted from
3.38. Carbon Dioxide Liquid-Measuring Devices

3-73

the metered quantity. The appropriate correction

values shall apply as specified in Table 2.

UR.2.4. Temperature or Density Compensation.

UR.2.4.1. Use of Automatic Temperature or

Density Compensators. - Devices equipped

with an automatic temperature or density

compensator shall have the compensator

connected, operable, and in use at all times.

Such automatic temperature or density

compensator may not be removed.

UR.2.4.2. Tickets or Invoices. - Any written

invoice or printed ticket based on a reading of a

device that is equipped with an automatic

temperature or density compensator shall have

shown thereon that the quantity delivered has

been temperature or density compensated.

UR.2.5. Ticket in Printing Device. - A ticket shall

not be inserted into a device equipped with a ticket

printer until immediately before a delivery is begun,

and in no case shall a ticket be in the device when the

vehicle is in motion while on a public street, highway,

or thoroughfare.

UR.2.6. Sale by Weight. - All quantity

determinations shall be made by means of an approved

and sealed weighing or measuring device. All sales

shall be stated in kilograms or pounds.

D. Definitions

The terms defined here have a special and technical

meaning when used in the Code for Carbon Dioxide

Liquid-Measuring Devices.

automatic temperature or density compensation.

The use of integrated or ancillary equipment to obtain,

from the output of a volumetric meter, an equivalent

mass indication.

carbon dioxide liquid-measuring device. A system

including a mechanism or machine of (a) the meter or

(b) a weighing type of device mounted on a vehicle

designed to measure and deliver liquid carbon dioxide.

Means may be provided to indicate automatically, for

one of a series of unit prices, the total money value of

the quantity measured.

large-delivery devices. Devices used primarily for

single deliveries greater than 1000 pounds or 500

kilograms.

small-delivery device. Any device other than a largedelivery device.

transfer standard. A measurement system designed

for use in proving and testing carbon dioxide liquidmeasuring devices.

vapor equalization credit. The quantity deducted

from the metered quantity of liquid carbon dioxide

when a vapor equalizing line is used to facilitate the

transfer of liquid during a metered delivery.

vapor equalization line. A hose or pipe connected

from the vapor space of the seller’s tank to the vapor

space of the buyer’s tank that is used to equalize the

pressure during a delivery.

wet-hose type. A type of device in which it is

intended that the discharge hose be completely filled

prior to each commercial delivery.
3.38. Carbon Dioxide Liquid-Measuring Devices

3-74

Table 1.

Temp Pressure Liquid Density Vapor Density Vapor

E F Displacement

PSIA PSIG lb/gal (lb-oz)/gal lb/cu ft lb/gal %

- 30.00 177.89 163.19 9.127 9 - 2.0 1.989 0.266 2.9

- 29.75 178.75 164.05 9.122 9 - 2.0 1.999 0.267 2.9

- 29.50 179.62 164.92 9.117 9 - 1.9 2.008 0.268 2.9

- 29.25 180.49 165.79 9.113 9 - 1.8 2.018 0.270 3.0

- 29.00 181.36 166.67 9.108 9 - 1.7 2.028 0.271 3.0

- 28.75 182.24 167.54 9.103 9 - 1.7 2.038 0.272 3.0

- 28.50 183.12 168.42 9.098 9 - 1.6 2.048 0.274 3.0

- 28.25 184.00 169.31 9.094 9 - 1.5 2.058 0.275 3.0

- 28.00 184.89 170.19 9.089 9 - 1.4 2.067 0.276 3.0

- 27.75 185.78 171.08 9.084 9 - 1.3 2.077 0.278 3.1

- 27.50 186.67 171.98 9.080 9 - 1.3 2.087 0.279 3.1

- 27.25 187.57 172.87 9.075 9 - 1.2 2.098 0.280 3.1

- 27.00 188.47 173.77 9.070 9 - 1.1 2.108 0.282 3.1

- 26.75 189.37 174.67 9.065 9 - 1.0 2.118 0.283 3.1

- 26.50 190.28 175.58 9.061 9 - 1.0 2.128 0.284 3.1

- 26.25 191.18 176.49 9.056 9 - 0.9 2.138 0.286 3.2

- 26.00 192.10 177.40 9.051 9 - 0.8 2.148 0.287 3.2

- 25.75 193.01 178.32 9.046 9 - 0.7 2.159 0.289 3.2

- 25.50 193.93 179.23 9.041 9 - 0.7 2.169 0.290 3.2

- 25.25 194.85 180.16 9.037 9 - 0.6 2.179 0.291 3.2

- 25.00 195.78 181.08 9.032 9 - 0.5 2.190 0.293 3.2

- 24.75 196.70 182.01 9.027 9 - 0.4 2.200 0.294 3.3

- 24.50 197.64 182.94 9.022 9 - 0.4 2.211 0.296 3.3

- 24.25 198.57 183.87 9.017 9 - 0.3 2.221 0.297 3.3

- 24.00 199.51 184.81 9.013 9 - 0.2 2.232 0.298 3.3

- 23.75 200.45 185.75 9.008 9 - 0.1 2.243 0.300 3.3

- 23.50 201.39 186.70 9.003 9 - 0.0 2.253 0.301 3.3
3.38. Carbon Dioxide Liquid-Measuring Devices

Table 1.

Temp Pressure Liquid Density Vapor Density Vapor

E F Displacement

PSIA PSIG lb/gal (lb-oz)/gal lb/cu ft lb/gal %

3-75

- 23.25 202.34 187.64 8.998 9 - 0.0 2.264 0.303 3.4

- 23.00 203.29 188.60 8.993 8 - 15.9 2.275 0.304 3.4

- 22.75 204.25 189.55 8.989 8 - 15.8 2.286 0.306 3.4

- 22.50 205.20 190.51 8.984 8 - 15.7 2.296 0.307 3.4

- 22.25 206.16 191.47 8.979 8 - 15.7 2.307 0.308 3.4

- 22.00 207.13 192.43 8.974 8 - 15.6 2.318 0.310 3.5

- 21.75 208.09 193.40 8.969 8 - 15.5 2.329 0.311 3.5

- 21.50 209.06 194.37 8.964 8 - 15.4 2.340 0.313 3.5

- 21.25 210.04 195.34 8.959 8 - 15.4 2.351 0.314 3.5

- 21.00 211.02 196.32 8.955 8 - 15.3 2.362 0.316 3.5

- 20.75 212.00 197.30 8.950 8 - 15.2 2.374 0.317 3.5

- 20.50 212.98 198.28 8.945 8 - 15.1 2.385 0.319 3.6

- 20.25 213.97 199.27 8.940 8 - 15.0 2.396 0.320 3.6

- 20.00 214.96 200.26 8.935 8 - 15.0 2.407 0.322 3.6

- 19.75 215.95 201.26 8.930 8 - 14.9 2.419 0.323 3.6

- 19.50 216.95 202.25 8.925 8 - 14.8 2.430 0.325 3.6

- 19.25 217.95 203.25 8.920 8 - 14.7 2.441 0.326 3.7

- 19.00 218.95 204.26 8.915 8 - 14.6 2.453 0.328 3.7

- 18.75 219.96 205.27 8.911 8 - 14.6 2.464 0.329 3.7

- 18.50 220.97 206.28 8.906 8 - 14.5 2.476 0.331 3.7

- 18.25 221.99 207.29 8.901 8 - 14.4 2.488 0.333 3.7

- 18.00 223.01 208.31 8.896 8 - 14.3 2.499 0.334 3.8

- 17.75 224.03 209.33 8.891 8 - 14.3 2.511 0.336 3.8

- 17.50 225.05 210.36 8.886 8 - 14.2 2.523 0.337 3.8

- 17.25 226.08 211.38 8.881 8 - 14.1 2.534 0.339 3.8

- 17.00 227.11 212.42 8.876 8 - 14.0 2.546 0.340 3.8

- 16.75 228.15 213.45 8.871 8 - 13.9 2.558 0.342 3.9

- 16.50 229.18 214.49 8.866 8 - 13.9 2.570 0.344 3.9
3.38. Carbon Dioxide Liquid-Measuring Devices

Table 1.

Temp Pressure Liquid Density Vapor Density Vapor

E F Displacement

PSIA PSIG lb/gal (lb-oz)/gal lb/cu ft lb/gal %

3-76

- 16.25 230.23 215.53 8.861 8 - 13.8 2.582 0.345 3.9

- 16.00 231.27 216.58 8.856 8 - 13.7 2.594 0.347 3.9

- 15.75 232.32 217.62 8.851 8 - 13.6 2.606 0.348 3.9

- 15.50 233.37 218.68 8.846 8 - 13.5 2.618 0.350 4.0

- 15.25 234.43 219.73 8.841 8 - 13.5 2.630 0.352 4.0

- 15.00 235.49 220.79 8.836 8 - 13.4 2.643 0.353 4.0

- 14.75 236.55 221.86 8.831 8 - 13.3 2.655 0.355 4.0

- 14.50 237.62 222.92 8.826 8 - 13.2 2.667 0.357 4.0

- 14.25 238.69 223.99 8.821 8 - 13.1 2.680 0.358 4.1

- 14.00 239.76 225.07 8.816 8 - 13.1 2.692 0.360 4.1

- 13.75 240.84 226.14 8.811 8 - 13.0 2.704 0.362 4.1

- 13.50 241.92 227.22 8.806 8 - 12.9 2.717 0.363 4.1

- 13.25 243.00 228.31 8.801 8 - 12.8 2.729 0.365 4.1

- 13.00 244.09 229.39 8.796 8 - 12.7 2.742 0.367 4.2

- 12.75 245.18 230.49 8.791 8 - 12.7 2.755 0.368 4.2

- 12.50 246.28 231.58 8.786 8 - 12.6 2.767 0.370 4.2

- 12.25 247.37 232.68 8.781 8 - 12.5 2.780 0.372 4.2

- 12.00 248.48 233.78 8.776 8 - 12.4 2.793 0.373 4.3

- 11.75 249.58 234.89 8.771 8 - 12.3 2.806 0.375 4.3

- 11.50 250.69 236.00 8.765 8 - 12.2 2.819 0.377 4.3

- 11.25 251.80 237.11 8.760 8 - 12.2 2.832 0.379 4.3

- 11.00 252.92 238.22 8.755 8 - 12.1 2.845 0.380 4.3

- 10.75 254.04 239.34 8.750 8 - 12.0 2.858 0.382 4.4

- 10.50 255.16 240.47 8.745 8 - 11.9 2.871 0.384 4.4

- 10.25 256.29 241.60 8.740 8 - 11.8 2.884 0.386 4.4

- 10.00 257.42 242.73 8.735 8 - 11.8 2.897 0.387 4.4

- 9.75 258.56 243.86 8.730 8 - 11.7 2.911 0.389 4.5

- 9.50 259.70 245.00 8.725 8 - 11.6 2.924 0.391 4.5
3.38. Carbon Dioxide Liquid-Measuring Devices

Table 1.

Temp Pressure Liquid Density Vapor Density Vapor

E F Displacement

PSIA PSIG lb/gal (lb-oz)/gal lb/cu ft lb/gal %

3-77

- 9.25 260.84 246.14 8.719 8 - 11.5 2.937 0.393 4.5

- 9.00 261.98 247.29 8.714 8 - 11.4 2.951 0.394 4.5

- 8.75 263.13 248.44 8.709 8 - 11.3 2.964 0.396 4.5

- 8.50 264.29 249.59 8.704 8 - 11.3 2.978 0.398 4.6

 - 8.25 265.44 250.75 8.699 8 - 11.2 2.991 0.400 4.6

 - 8.00 266.60 251.91 8.694 8 - 11.1 3.005 0.402 4.6

 - 7.75 267.77 253.07 8.688 8 - 11.0 3.019 0.404 4.6

 - 7.50 268.93 254.24 8.683 8 - 10.9 3.032 0.405 4.7

 - 7.25 270.11 255.41 8.678 8 - 10.8 3.046 0.407 4.7

 - 7.00 271.28 256.59 8.673 8 - 10.8 3.060 0.409 4.7

 - 6.75 272.46 257.76 8.668 8 - 10.7 3.074 0.411 4.7

 - 6.50 273.64 258.95 8.662 8 - 10.6 3.088 0.413 4.8

 - 6.25 274.83 260.13 8.657 8 - 10.5 3.102 0.415 4.8

 - 6.00 276.02 261.32 8.652 8 - 10.4 3.116 0.417 4.8

 - 5.75 277.21 262.52 8.647 8 - 10.3 3.130 0.418 4.8

 - 5.50 278.41 263.72 8.641 8 - 10.3 3.144 0.420 4.9

 - 5.25 279.61 264.92 8.636 8 - 10.2 3.159 0.422 4.9

 - 5.00 280.82 266.12 8.631 8 - 10.1 3.173 0.424 4.9

 - 4.75 282.03 267.33 8.626 8 - 10.0 3.187 0.426 4.9

 - 4.50 283.24 268.55 8.620 8 - 9.9 3.202 0.428 5.0

 - 4.25 284.46 269.76 8.615 8- 9.8 3.216 0.430 5.0

 - 4.00 285.68 270.98 8.610 8 - 9.8 3.231 0.432 5.0

 - 3.75 286.90 272.21 8.604 8 - 9.7 3.245 0.434 5.0

 - 3.50 288.13 273.44 8.599 8 - 9.6 3.260 0.436 5.1

 - 3.25 289.37 274.67 8.594 8 - 9.5 3.275 0.438 5.1

 - 3.00 290.60 275.91 8.589 8 - 9.4 3.289 0.440 5.1

 - 2.75 291.84 277.15 8.583 8 - 9.3 3.304 0.442 5.1

 - 2.50 293.09 278.39 8.578 8 - 9.2 3.319 0.444 5.2
3.38. Carbon Dioxide Liquid-Measuring Devices

Table 1.

Temp Pressure Liquid Density Vapor Density Vapor

E F Displacement

PSIA PSIG lb/gal (lb-oz)/gal lb/cu ft lb/gal %

3-78

 - 2.25 294.33 279.64 8.573 8 - 9.2 3.334 0.446 5.2

 - 2.00 295.58 280.89 8.567 8 - 9.1 3.349 0.448 5.2

 - 1.75 296.84 282.14 8.562 8 - 9.0 3.364 0.450 5.3

 - 1.50 298.10 283.40 8.556 8 - 8.9 3.379 0.452 5.3

 - 1.25 299.36 284.67 8.551 8 - 8.8 3.395 0.454 5.3

 - 1.00 300.63 285.93 8.546 8 - 8.7 3.410 0.456 5.3

 - 0.75 301.90 287.21 8.540 8 - 8.6 3.425 0.458 5.4

 - 0.50 303.18 288.48 8.535 8 - 8.6 3.440 0.460 5.4

 - 0.25 304.46 289.76 8.530 8 - 8.5 3.456 0.462 5.4

0.00 305.74 291.74 8.524 8 - 8.4 3.471 0.464 5.4

0.25 307.03 292.33 8.519 8 - 8.3 3.487 0.466 5.5

0.50 308.32 293.62 8.513 8 - 8.2 3.503 0.468 5.5

0.75 309.61 294.92 8.508 8 - 8.1 3.518 0.470 5.5

1.00 310.91 296.21 8.502 8 - 8.0 3.534 0.472 5.6

1.25 312.21 297.52 8.497 8 - 8.0 3.550 0.475 5.6

1.50 313.52 298.82 8.491 8 - 7.9 3.566 0.477 5.6

1.75 314.83 300.13 8.486 8 - 7.8 3.582 0.479 5.6

2.00 316.15 301.45 8.480 8-7.7 3.598 0.481 5.7

2.25 317.46 302.77 8.475 8 - 7.6 3.614 0.483 5.7

2.50 318.79 304.09 8.469 8 - 7.5 3.630 0.485 5.7

2.75 320.11 305.42 8.464 8 - 7.4 3.646 0.487 5.8

3.00 321.45 306.75 8.458 8 - 7.3 3.662 0.490 5.8

3.25 322.78 308.08 8.453 8 - 7.2 3.679 0.492 5.8

3.50 324.12 309.42 8.447 8 - 7.2 3.695 0.494 5.8

3.75 325.46 310.77 8.442 8 - 7.1 3.712 0.496 5.9

4.00 326.81 312.11 8.436 8 - 7.0 3.728 0.498 5.9

4.25 328.16 313.46 8.431 8 - 6.9 3.745 0.501 5.9

4.50 329.52 314.82 8.425 8 - 6.8 3.761 0.503 6.0
3.38. Carbon Dioxide Liquid-Measuring Devices

Table 1.

Temp Pressure Liquid Density Vapor Density Vapor

E F Displacement

PSIA PSIG lb/gal (lb-oz)/gal lb/cu ft lb/gal %

3-79

4.75 330.88 316.18 8.420 8 - 6.7 3.778 0.505 6.0

5.00 332.24 317.54 8.414 8 - 6.6 3.795 0.507 6.0

5.25 333.61 318.91 8.408 8 - 6.5 3.812 0.510 6.1

5.50 334.98 320.28 8.403 8 - 6.4 3.829 0.512 6.1

5.75 336.35 321.66 8.397 8 - 6.4 3.846 0.514 6.1

6.00 337.73 323.04 8.392 8 - 6.3 3.863 0.516 6.2

6.25 339.12 324.42 8.386 8 - 6.2 3.880 0.519 6.2

6.50 340.51 325.81 8.380 8 - 6.1 3.897 0.521 6.2

6.75 341.90 327.20 8.375 8 - 6.0 3.915 0.523 6.3

7.00 343.30 328.60 8.369 8 - 5.9 3.932 0.526 6.3

7.25 344.70 330.00 8.363 8 - 5.8 3.949 0.528 6.3

7.50 346.10 331.41 8.358 8 - 5.7 3.967 0.530 6.3

7.75 347.51 332.82 8.352 8 - 5.6 3.984 0.533 6.4

8.00 348.92 334.23 8.346 8 - 5.5 4.002 0.535 6.4

8.25 350.34 335.65 8.341 8 - 5.4 4.020 0.537 6.4

8.50 351.76 337.07 8.335 8 - 5.4 4.038 0.540 6.5

8.75 353.19 338.49 8.335 8-5.4 4.038 0.540 6.5

9.00 354.62 339.92 8.323 8 - 5.2 4.073 0.545 6.5

9.25 356.06 341.36 8.318 8 - 5.1 4.091 0.547 6.6

9.50 357.49 342.80 8.312 8 - 5.0 4.110 0.549 6.6

9.75 358.94 344.24 8.306 8 - 4.9 4.128 0.552 6.6

10.00 360.38 345.69 8.300 8 - 4.8 4.146 0.554 6.7

10.25 361.84 347.14 8.295 8 - 4.7 4.164 0.557 6.7

10.50 363.29 348.60 8.289 8 - 4.6 4.183 0.559 6.7

10.75 364.75 350.06 8.283 8 - 4.5 4.201 0.562 6.8

11.00 366.22 351.52 8.277 8 - 4.4 4.220 0.564 6.8

11.25 367.68 352.99 8.271 8 - 4.3 4.238 0.567 6.8
3.38. Carbon Dioxide Liquid-Measuring Devices

Table 1.

Temp Pressure Liquid Density Vapor Density Vapor

E F Displacement

PSIA PSIG lb/gal (lb-oz)/gal lb/cu ft lb/gal %

3-80

11.50 369.16 354.46 8.266 8 - 4.2 4.257 0.569 6.9

11.75 370.64 355.94 8.260 8 - 4.2 4.276 0.572 6.9

12.00 372.12 357.42 8.254 8 - 4.1 4.295 0.574 7.0

12.25 373.60 358.91 8.248 8 - 4.0 4.314 0.577 7.0

12.50 375.09 360.40 8.242 8 - 3.9 4.333 0.579 7.0

12.75 376.59 361.89 8.236 8 - 3.8 4.352 0.582 7.1

13.00 378.09 363.39 8.230 8 - 3.7 4.371 0.584 7.1

13.25 379.59 364.89 8.224 8 - 3.6 4.390 0.587 7.1

13.50 381.10 366.40 8.219 8 - 3.5 4.410 0.589 7.2

13.75 382.61 367.91 8.213 8 - 3.4 4.429 0.592 7.2

14.00 384.13 369.43 8.207 8 - 3.3 4.449 0.595 7.2

14.25 385.65 370.95 8.201 8 - 3.2 4.468 0.597 7.3

14.50 387.17 372.48 8.195 8 - 3.1 4.488 0.600 7.3

14.75 388.70 374.01 8.189 8 - 3.0 4.508 0.603 7.4

15.00 390.24 375.54 8.183 8 - 2.9 4.527 0.605 7.4

15.25 391.78 377.08 8.177 8 - 2.8 4.547 0.608 7.4

15.50 393.32 378.62 8.171 8 - 2.7 4.567 0.611 7.5

15.75 394.87 380.17 8.165 8 - 2.6 4.587 0.613 7.5

16.00 396.42 381.72 8.159 8 - 2.5 4.608 0.616 7.5

16.25 397.98 383.28 8.153 8 - 2.4 4.628 0.619 7.6

16.50 399.54 384.84 8.147 8 - 2.3 4.648 0.621 7.6

16.75 401.10 386.41 8.141 8 - 2.2 4.669 0.624 7.7

17.00 402.67 387.98 8.134 8 - 2.2 4.689 0.627 7.7

17.25 404.25 389.55 8.128 8 - 2.1 4.710 0.630 7.7

17.50 405.82 391.13 8.122 8 - 2.0 4.731 0.632 7.8

17.75 407.41 392.71 8.116 8 - 1.9 4.751 0.635 7.8

18.00 409.00 394.30 8.110 8 - 1.8 4.772 0.638 7.9
3.38. Carbon Dioxide Liquid-Measuring Devices

Table 1.

Temp Pressure Liquid Density Vapor Density Vapor

E F Displacement

PSIA PSIG lb/gal (lb-oz)/gal lb/cu ft lb/gal %

3-81

18.25 410.59 395.89 8.104 8 - 1.7 4.793 0.641 7.9

18.50 412.19 397.49 8.098 8 - 1.6 4.814 0.644 7.9

18.75 413.79 399.09 8.092 8 - 1.5 4.835 0.646 8.0

19.00 415.39 400.70 8.085 8 - 1.4 4.857 0.649 8.0

19.25 417.00 402.31 8.079 8 - 1.3 4.878 0.652 8.1

19.50 418.62 403.92 8.073 8 - 1.2 4.900 0.655 8.1

19.75 420.24 405.54 8.067 8 - 1.1 4.921 0.658 8.2

20.00 421.86 407.17 8.061 8 - 1.0 4.943 0.661 8.2


SECTION 4

Page

4.40. Vehicle Tanks Used as Measures 4-3

4.41. Liquid Measures 4-7

4.42. Farm Milk Tanks 4-9

4.43. Measure-Containers 4-15

4.44. Graduates 4-17

4.45. Dry Measures 4-21

4.46. Berry Baskets and Boxes 4-23

8GJKENG6CPMU7UGFCU/GCUWTGU



Sec. 4.40. Vehicle Tanks Used as Measures

A. Application

A.1. - This code applies to vehicle tanks when these are

used as commercial measures.

A.2. - This code does not apply to the following devices:

(a) Devices used solely for dispensing a product in

connection with operations in which the amount

dispensed does not affect customer charges.

(b) Meters mounted on vehicle tanks (for which see Sec.

3.31; Code for Vehicle-Tank Meters).

A.3. - Also see Sec. 1.10; General Code requirements.

S. Specifications

S.1. Design of Compartments.

S.1.1. Compartment Distortion. - The shell and

bulkheads of a vehicle tank shall be so constructed that

under any condition of liquid lading they will not become

distorted sufficiently to cause a change in the capacity of

any compartment (as determined by volumetric test)

equal to more than 0.25 L per 750 L (0.5 pt per 200 gal),

or fraction thereof, of the nominal compartment capacity,

or to more than 0.5 L (1 pt), whichever is greater. (This

specification prescribes a limit on permissible distortion

only, and is not to be construed as setting up a secondary

tolerance on compartment capacities to be added to the

values given in tolerance paragraph T.2.)

S.1.2. Venting. - During filling operations, effective

venting of a compartment shall be provided to permit air

to escape from all spaces designed to be filled with liquid

and to permit the influx of air to the compartment during

the discharge of liquid therefrom. Venting shall prevent

any formation of air pockets.

S.1.3. Completeness of Delivery. - A tank shall be so

constructed that, when it is standing on a level surface,

complete delivery can be made from any compartment

through its delivery faucet or valve whether other

compartments are full or empty, and whether or not the

delivery is through a manifold.

S.1.4. Fill or Inspection Opening. - The fill or

inspection opening of a compartment shall be of such

size and location that it can readily be determined by

visual inspection that the compartment has been properly

filled or completely emptied and shall be so positioned

with respect to the ends of the compartment that the

indicator may be positioned as required. In no case shall

the opening, if circular, have a diameter of less than

20 cm (7-5/8 in), or, if other than circular, have an

effective area of less than 290 cm (45 in ). 2 2

S.1.5. Dome Flange and Baffle Plates. - Any dome

flange extending into a compartment shall be provided

with sufficient perforations or openings flush with the

compartment shell to prevent any trapping of air. All

baffle plates in a compartment shall be so cut away at top

and bottom, and elsewhere as necessary, as to facilitate

loading and unloading.

S.1.6. Compartment and Piping Capacities and

Emergency Valve. - If a compartment is equipped with

an emergency (or safety) valve, this shall be positioned

at the lowest point of outlet from the compartment, and

the compartment capacity or capacities shall be construed

as excluding the capacity of the piping leading

therefrom. However, the capacity of the piping leading

from such a compartment shall be separately determined

and reported, and may be separately marked as specified

in S.4.

S.1.6.1. On Vehicle Tanks Equipped for Bottom

Loading. - On equipment designed for bottom

loading, the compartment capacity shall include the

piping of a compartment to the valve located on the

upstream side of the manifold and immediately

adjacent thereto or, if not manifolded, to the outlet

valve, provided that on or immediately adjacent to the

marking as specified in S.4. the following words or a

statement of similar meaning shall be affixed:

"Warning: Emergency valves must be opened before

checking measurement."
8GJKENG6CPMU7UGFCU/GCUWTGU



S.1.7. Expansion Space. - When a compartment is S.2.6. Adjustable Indicators. - Adequate provision

filled to the level of the highest indicator in the shall be made for conveniently affixing a security seal or

compartment, there shall remain an expansion space of at seals:

least 0.75 percent of the nominal compartment capacity

as defined by that indicator. (a) to any indicator intended to remain adjustable, so

S.2. Design of Compartment Indicators.

S.2.1. General. - An indicator shall be so designed that

it will distinctly and unmistakably define a capacity point

of its compartment when liquid is in contact with the

lowest portion of the indicator.

S.2.2. Number of Indicators. - In no case shall a compartment be provided with more than five indicators.

(Amended 1972)

S.2.3. Identification of Multiple Indicators. - If a

compartment is provided with multiple indicators, each

such indicator shall be conspicuously marked with an

identifying letter or number.

S.2.4. Location. - An indicator shall be located:

(a) midway between the sides of its compartment;

(b) as nearly as practicable midway between the ends

of its compartment, and in no case offset by more

than 10 percent of the compartment space or 15 cm

(6 in), whichever is less;

(c) so that it does not extend into, nor more than 15 cm

(6 in) from that section of the compartment defined

by a vertical projection of the fill opening;

(Amended 1974)

(d) at a depth, measuring from the top of the dome

opening, not lower than 46 cm (18 in) for fill

openings of less than 38 cm (15 in) in diameter, or,

if other than circular, an effective area of less than

1130 cm (175 in ), and not lower than 61 cm 2 2

(24 in) for larger fill openings; and

(e) to provide a clearance of not less than 5 cm (2 in)

between indicators.

 (Amended 1972)

S.2.5. Permanence. - Any indicator that is not intended

to remain adjustable and all brackets or supports shall be

securely welded in position.

that no adjustment of the indicator can be made

without mutilating or destroying the seal, and

(b) to any removable part to which an indicator may be

attached, so that the part cannot be removed

without mutilating or destroying the seal.

S.2.7. Sensitiveness. - The position of any indicator in

its compartment shall be such that at the level of the indicator a change of 1.0 mm (0.04 in) in the height of the

liquid surface will represent a volume change of not

more than the value of the tolerance for the nominal

compartment capacity as defined by that indicator.

S.3. Design of Compartment Discharge Manifold. -

When two or more compartments discharge through a

common manifold or other single outlet, effective means

shall be provided to ensure:

(a) that liquid can flow through the delivery line leading

from only one compartment at one time and that flow

of liquid from one compartment to any other is

automatically prevented, or

(b) that all compartments will discharge simultaneously.

If the discharge valves from two or more compartments are

automatically controlled so that they can only be operated

together, thus effectively connecting these compartments to

one another, such compartments shall, for purposes of this

paragraph, be construed to be one compartment.

S.4. Marking of Compartments.

S.4.1. Compartment Identification. - Each

compartment of a multiple-compartment tank shall be

conspicuously identified by a letter or number marked on

the dome or immediately below the fill opening. Such

letters or numbers shall be in regular sequence from front

to rear, and the delivery faucets or valves shall be marked

to correspond with their respective compartments.

S.4.2. Compartment Capacity, Single Indicator. - A

compartment provided with a single indicator shall be

clearly, permanently, and conspicuously marked with a

statement of its capacity as defined by its indicator.
8GJKENG6CPMU7UGFCU/GCUWTGU



S.4.3. Compartment Capacity, Multiple Indicators. -

A compartment provided with two or more indicators

shall be clearly, permanently, and conspicuously marked

with a statement identifying:

(a) each indicator by a letter or number and, immediately adjacent thereto,

(b) the capacity of the compartment as defined by the

particular indicator.

N. Notes

N.1. Test Liquid. - Water or light fuel oil shall be used as

the test liquid for a vehicle-tank compartment.

N.2. Evaporation and Volume Change. - Care shall be

exercised to reduce to a minimum, evaporation losses and

volume changes resulting from changes in temperature of

the test liquid.

N.3. To Deliver. - A vehicle-tank compartment shall be

gauged "to deliver." If the compartment is gauged by

measuring the test liquid into the tank, the inside tank walls

shall first be thoroughly wetted.

N.4. Gauging of Compartments. - When a compartment

is gauged to determine the proper position for an indicator

or to determine what a capacity marking should be, whether

on a new vehicle tank or following repairs or modifications

that might affect compartment capacities, tolerances are not

applicable, and the indicator shall be set and the

compartment capacity shall be marked as accurately as

practicable.

N.5. Adjustment and Remarking. - When a compartment

is found upon test to have an error in excess of the

applicable tolerance, the capacity of the compartment shall

be adjusted to agree with its marked capacity, or its marked

capacity shall be changed to agree with its capacity as

determined by the test.

 T. Tolerances

T.1. Application.

T.1.1. To Excess and to Deficiency. - The tolerances

hereinafter prescribed shall be applied to errors in excess

and in deficiency.

T.2. Tolerance Values. - Maintenance and acceptance

tolerances shall be as shown in Table 1.

 UR. User Requirements

UR.1. Conditions of Use.

UR.1.1. Filling. - A vehicle shall stand upon a level

surface during the filling of a compartment.

UR.1.2. Delivering. - During a delivery, a vehicle shall

be so positioned as to assure complete emptying of a

compartment. Each compartment shall be used for an

individual delivery only; that is, an individual delivery

shall consist of the entire contents of a compartment or

compartments.

(Amended 1976)
8GJKENG6CPMU7UGFCU/GCUWTGU



6CDNG

/CKPVGPCPEGCPF#EEGRVCPEG6QNGTCPEGU

QP8GJKENG6CPM%QORCTVOGPVU

0QOKPCNECRCEKV[QHEQORCTVOGPV /CKPVGPCPEGCPFCEEGRVCPEGVQNGTCPEGU

ICNNQPU 'ZRTGUUGFKPSWCTVU 'ZRTGUUGFKPICNNQPU

QTNGUU  

VQKPENWUKXG  

VQKPENWUKXG  

VQKPENWUKXG  

VQKPENWUKXG  

QXGT #FFSWCTVRGTICNNQPU

QTHTCEVKQPVJGTGQH

#FFICNNQPRGTICNNQPU

QTHTCEVKQPVJGTGQH
4.41. Liquid Measures



Sec. 4.41. Liquid Measures

A. Application

A.1. - This code applies to liquid measures; that is, to rigid

measures of capacity designed for general and repeated use

in the measurement of liquids.

A.2. - The code does not apply to test measures or other

volumetric standards.

A.3. - See also Sec. 1.10; General Code requirements.

S. Specifications

S.1. Units. - 

a. The capacity of a liquid measure shall be 0.1 L, 0.2 L,

0.5 L, 1 L, 2 L, 5 L, or a multiple of 5 L, and the measure

shall not be subdivided. 

b. The capacity of a liquid measure shall be 1 gill,

1/2 liquid pint, 1 liquid pint, 1 liquid quart, 1/2 gallon, 1-

1/4 gallon, 1-1/2 gallon, or a multiple of 1 gallon, and the

measure shall not be subdivided. However, 3-pint and 5-

pint brick molds and 2-1/2 gallon (10-quart) cans shall be

permitted when used exclusively for ice cream.

S.2. Material. - Measures shall be made of metal, glass,

earthenware, enameled ware, composition, or similar and

suitable material. If made of metal, the thickness of the

metal shall not be less than the appropriate value given in

Table 1.

S.3. Capacity Point. - The capacity of a measure shall be

determined to a definite edge, or to the lowest portion of a

plate, bar, or wire, at or near the top of the measure, and

shall not include the capacity of any lip or rim that may be

provided.

S.4. Reinforcing Rings. - Reinforcing rings, if used, shall

be attached to the outside of the measure and shall show no

divisions or lines on the inside surface of the measure.

S.5. Discharge. - A measure equipped with a discharge

faucet or valve shall be susceptible to complete discharge

through the faucet or valve when the measure is standing on

a level surface.

S.6. Marking Requirements. - A measure shall be marked

on its side with a statement of its capacity. If the capacity

is stated in terms of the pint or quart, the word "Liquid" or

the abbreviation "Liq" shall be included.

 T. Tolerances

T.1. - Maintenance tolerances in excess and in deficiency

shall be as shown in Table 2. Acceptance tolerances shall

be one-half the maintenance tolerances.
4.41. Liquid Measures



Table 1.

Minimum Thickness of Metal for Liquid Measures

Nominal capacity

Minimum thickness

For iron or steel, plated or unplated (inch)For copper or aluminum (inch)

1 pint or less 0.010 0.020

1 quart, 1/2 gallon, 1 gallon 0.014 0.028

Over 1 gallon 0.016 0.032

Table 2.

Maintenance Tolerances, in Excess and in Deficiency, for Liquid Measures

Nominal capacity

Tolerance

In excess In deficiency

fluid drams cubic inches fluid drams cubic inches

1/2 pint or less 2.0 0.4 1.0 0.2

1 pint 3.0 0.7 1.5 0.3

1 quart 4.0 0.9 2.0 0.5

1/2 gallon 6.0 1.4 3.0 0.7

fluid

ounces

cubic

inches

fluid

drams

cubic

inches

1 and 1-1/4 gallons 1.0 1.8 4.0 0.9

1-1/2 gallons 1.5 2.7 6.0 1.4

fluid

ounces

cubic

inches

fluid

ounces

cubic

inches

2 gallons 2.0 3.5 1.0 1.8

3 and 4 gallons 4.0 7.0 2.0 3.6

5 gallons 6.0 11.0 3.0 5.4

10 gallons 10.0 18.0 5.0 9.0
(CTO/KNM6CPMU



5GE(CTO/KNM6CPMU

##RRNKECVKQP

#6JKUEQFGCRRNKGUVQHCTOOKNMVCPMUQPVJG

RTGOKUGUQHRTQFWEGTUYJGPVJGUGCTGWUGFQTCTGVQDG

WUGFHQTVJGEQOOGTEKCNOGCUWTGOGPVQHOKNM

#6JKUEQFGFQGUPQVCRRN[VQVCPMUOQWPVGFQP

JKIJYC[XGJKENGU

#5GGCNUQ5GE)GPGTCN%QFGTGSWKTGOGPVU

55RGEKHKECVKQPU

5%QORQPGPVU#HCTOOKNMVCPMYJGVJGT

UVCVKQPCT[QTRQTVCDNGUJCNNDGEQPUKFGTGFUWKVCDNGHQT

EQOOGTEKCNWUGQPN[YJGPKVEQORTKUGU


C CXGUUGNYJGVJGTQTPQVKVKUGSWKRRGFYKVJOGCPU

HQTEQQNKPIKVUEQPVGPVU


D OGCPUHQTTGCFKPIVJGNGXGNQHNKSWKFKPVJGVCPM

UWEJCUCTGOQXCDNGICWIGTQFQTUWTHCEGICWIGCPF


E CEJCTVHQTEQPXGTVKPINGXGNQHNKSWKFTGCFKPIUVQ

XQNWOG

'CEJEQORCTVOGPVQHCUWDFKXKFGFVCPMUJCNNHQTVJG

RWTRQUGUQHVJKUEQFGDGEQPUVTWGFVQDGCHCTOOKNM

VCPM

5&GUKIPQH6CPM

5.GXGN#HCTOOKNMVCPMUJCNNDGFGUKIPGFVQ

DGKPPQTOCNQRGTCVKPIRQUKVKQPYJGPKVKUKPNGXGN6JG

VCPMUJCNNDGUQEQPUVTWEVGFVJCVKVYKNNOCKPVCKPKVU

EQPFKVKQPQHNGXGNWPFGTCNNPQTOCNEQPFKVKQPUQHNCFKPI

5.GXGN+PFKECVKPI/GCPU#VCPMUJCNNDG

RGTOCPGPVN[GSWKRRGFYKVJUGPUKVKXGOGCPUD[YJKEJ

VJGNGXGNQHVJGVCPMECPDGFGVGTOKPGF

51PC5VCVKQPCT[6CPM#UVCVKQPCT[VCPM

UJCNNDGRTQXKFGFYKVJUWEJNGXGNKPFKECVKPIOGCPUCU

CVYQYC[QTEKTEWNCTNGXGNCRNWODDQDVYQYC[

NGXGNKPINWIUQTVJGNKMGQTVJGVQRGFIGQTGFIGUQH

VJGVCPMUJCNNDGUQEQPUVTWEVGFVJTQWIJQWVCUVQ

RTQXKFGCPCEEWTCVGTGHGTGPEGHQTNGXGN

FGVGTOKPCVKQPURTQXKFGFVJCVYJGPNGXGNKPINWIUQT

VJGVQRGFIGQTGFIGUQHVJGVCPMCTGWUGFCUVJG

TGHGTGPEGHQTNGXGNFGVGTOKPCVKQPUVJGTGUJCNNDG

UWRRNKGFYKVJVJGVCPMCUGPUKVKXGURKTKVNGXGNQHCR

RTQRTKCVGFKOGPUKQPUCPFVJGRQUKVKQPUYJGTGUWEJ

NGXGNKUKPVGPFGFVQDGWUGFUJCNNDGRGTOCPGPVN[

OCTMGFQPVJGTGHGTGPEGUWTHCEGQHVJGVCPMCPF

RTQXKFGFHWTVJGTVJCVYJGPNGXGNKPINWIUCTGWUGF

VJG[UJCNNDGUQFGUKIPGFEQPUVTWEVGFCPFKPUVCNNGF

CVVJGHCEVQT[VJCVCP[CNVGTCVKQPQHVJGQTKIKPCN

RQUKVKQPQTEQPFKVKQPUWEJCUD[JCOOGTKPIQT

HKNKPIYQWNFDGFKHHKEWNVCPFYQWNFDGEQOGQDXKQWU

#UVCVKQPCT[VCPMYKVJCPQOKPCNECRCEKV[QH.

QTICNQTITGCVGTUJCNNDGRTQXKFGFYKVJCVNGCUV

VYQUKOKNCTNGXGNKPFKECVKPIOGCPUCPFVJGUGUJCNNDG

NQECVGFKPQRRQUKVGCPFFKUVCPVRQUKVKQPUHTQOGCEJ

QVJGTVQHCEKNKVCVGCPCEEWTCVGNGXGNFGVGTOKPCVKQPKP

DQVJFKTGEVKQPUQHVJGVCPM	UJQTK\QPVCNRNCPG


#OGPFGF

51PC2QTVCDNG6CPM#RQTVCDNGVCPM

UJCNNDGRTQXKFGFYKVJGKVJGTCVYQYC[QTCEKTEWNCT

NGXGN

52QTVCDNG6CPM#RQTVCDNGVCPMUJCNNDGQH

VJGEGPVGTTGCFKPIV[RGVJCVKUKVUJCNNDGUQFGUKIPGF

VJCVVJGICWIGTQFQTUWTHCEGICWIGYJGPRTQRGTN[

RQUKVKQPGFHQTWUGYKNNDGCRRTQZKOCVGN[KPVJGXGTVKECN

CZKUQHVJGVCPMEGPVTCNN[RQUKVKQPGFYKVJTGURGEVVQVJG

VCPMYCNNU

5%CRCEKV[#HCTOOKNMVCPMUJCNNDGENGCTN[

CPFRGTOCPGPVN[OCTMGFQPCUWTHCEGXKUKDNGCHVGT

KPUVCNNCVKQPYKVJKVUECRCEKV[CUFGVGTOKPGFD[VJG

OCPWHCEVWTGT6JGECRCEKV[UJCNNPQVGZEGGFCPCOQWPV

VJCVECPDGCIKVCVGFYKVJQWVQXGTHNQYKPICPFVJCVECP

DGOGCUWTGFCEEWTCVGN[YKVJVJGNKSWKFCVTGUV

=0QPTGVTQCEVKXGCUQH?

5&GUKIPQH+PFKECVKPI/GCPU

5)GPGTCN#VCPMUJCNNKPENWFGKPFKECVKPI

OGCPUCPFUJCNNDGECNKDTCVGFQXGTVJGGPVKTGTCPIGQH

VJG(CTO/KNM6CPMU



XQNWOGQHVJGVCPMHTQORGTEGPVQHECRCEKV[QTO OQWPVGFKPCXGTVKECNRQUKVKQPCPFGSWKRRGFYKVJC 


ICNYJKEJGXGTKUNGUUVQKVUOCZKOWOECRCEKV[ UNKFKPIOGEJCPKUOVQCUUKUVKPFGVGTOKPKPIVJGNKSWKF

=0QPTGVTQCEVKXGCUQH,CPWCT[? NGXGN


#FFGF

5)CWIG4QF$TCEMGVQT5WRRQTVU+HCVCPMKU DQTQUKNKECVGINCUUQTCRRTQXGFTKIKFRNCUVKEQTTKIKFN[

FGUKIPGFHQTWUGYKVJCICWIGTQFCUWDUVCPVKCNCPF UWRRQTVGFHNGZKDNGVWDKPIYKVJCWPKHQTOKPVGTPCN

TKIKFICWIGTQFDTCEMGVQTQVJGTUWKVCDNGUWRRQTVKPI FKCOGVGTPQVNGUUVJCPEO
KP+VUJCNNDG

GNGOGPVUHQTRQUKVKQPKPIVJGICWIGTQFUJCNNDG FGUKIPGFCPFEQPUVTWEVGFUQVJCVCNNRTQFWEVKPVJG

RTQXKFGF#ICWIGTQFCPFKVUDTCEMGVUQTQVJGT ICWIGECPDGFKUECTFGFKPUWEJCOCPPGTVJCVPQ

UWRRQTVKPIGNGOGPVUUJCNNDGUQEQPUVTWEVGFVJCV RTQFWEVKPVJGICWIGVWDGYKNNGPVGTVJGFKUEJCTIG

YJGPGXGTVJGTQFKURNCEGFKPGPICIGOGPVYKVJVJG NKPGQTVCPM

DTCEMGVQTUWRRQTVUCPFTGNGCUGFVJGTQFYKNN 
#OGPFGF

CWVQOCVKECNN[UGCVKVUGNHCVCHKZGFJGKIJVCPFKPC

XGTVKECNRQUKVKQP9JGPCICWIGTQFKURTQRGTN[UGCVGF 55ECNG2NCVG6JGUECNGRNCVGUJCNNDG

QPKVUDTCEMGVUQTUWRRQTVUVJGTGUJCNNDGCENGCTCPEGQH OQWPVGFCFLCEGPVVQCPFRCTCNNGNYKVJVJGICWIGVWDG

CVNGCUVEOKPDGVYGGPVJGITCFWCVGFHCEGQHVJG CPFDGPQOQTGVJCPOOKPHTQOVJGVWDG

TQFCPFCP[VCPMYCNNQTQVJGTUWTHCEGVJCVKVHCEGU

5)CWIG4QF9JGPRTQRGTN[UGCVGFKP UJCNNDGENGCTCPFGCUKN[TGCFCDNGCPFUJCNNEQORN[

RQUKVKQPCTQFUJCNNPQVVQWEJVJGDQVVQOQHVJGVCPM YKVJVJGTGSWKTGOGPVUQHRCTCITCRJUKPENWFGFWPFGT

WPNGUUVJKUKUTGSWKTGFD[VJGFGUKIPQHVJGUWRRQTVKPI 5)TCFWCVKQPU

GNGOGPVU6JGTQFUJCNNDGITCFWCVGFVJTQWIJQWVCP

KPVGTXCNEQTTGURQPFKPIVQVJGXQNWOGTCPIGYKVJKP 58GPVKPI#PGZVGTPCNICWIGVWDGUJCNNDG

YJKEJTGCFKPIUQHNKSWKFNGXGNCTGVQDGOCFG CFGSWCVGN[XGPVGFCVVJGVQRQRGPVQVJGCVOQURJGTG

55WTHCEG)CWIG$TCEMGVQT5WRRQTVU+HC CHHGEVVJGQRGTCVKQPQHVJKUXGPV

VCPMKUFGUKIPGFHQTWUGYKVJCUWTHCEGICWIGC 
#FFGF

UWDUVCPVKCNCPFTKIKFUWTHCEGICWIGDTCEMGVQTQVJGT

UWKVCDNGUWRRQTVKPIGNGOGPVUHQTRQUKVKQPKPIVJGUWTHCEG 5)TCFWCVKQPU

ICWIGUJCNNDGRTQXKFGF#UWTHCEGICWIGCPFKVU

DTCEMGVUQTQVJGTUWRRQTVKPIGNGOGPVUUJCNNDGUQ 55RCEKPICPF9KFVJQH)TCFWCVKQPU1P

EQPUVTWEVGFVJCVYJGPGXGTVJGICWIGCUUGODN[KU CICWIGTQFQTUWTHCEGICWIGVJGURCEKPIQHVJG

RNCEGFKPGPICIGOGPVYKVJVJGDTCEMGVQTUWRRQTVUVJG ITCFWCVKQPUEGPVGTVQEGPVGTUJCNNDGPQVOQTGVJCP

KPFKECVQTKHPQVRGTOCPGPVN[OQWPVGFQPVJGVCPMYKNN OOKPQTKPCPFUJCNNPQVDGNGUU

CWVQOCVKECNN[UGCVKVUGNHKPEQTTGEVQRGTCVKPIRQUKVKQP VJCPOO
KPQTKP6JG

CPFVJGITCFWCVGFGNGOGPVYKNNDGXGTVKECNN[RQUKVKQPGF ITCFWCVKQPUUJCNNPQVDGNGUUVJCPOO

CPFYKNNDGUGEWTGN[JGNFCVCP[JGKIJVVQYJKEJKVOC[ 
KPKPYKFVJCPFVJGENGCTKPVGTXCNDGVYGGP

DGOCPWCNN[UGV CFLCEGPVGFIGUQHUWEEGUUKXGITCFWCVKQPUUJCNNDGPQV

55WTHCEG)CWIG9JGPRTQRGTN[GPICIGFYKVJ

KVUDTCEMGVCPFUGVVQKVUNQYGUVRQUKVKQPCUWTHCEG 58CNWGUQH)TCFWCVKQPU1PCICWIGTQF

ICWIGUJCNNPQVVQWEJVJGDQVVQOQHVJGVCPM6JG QTUWTHCEGICWIGVJGITCFWCVKQPUOC[DGFGUKIPCVGF

ICWIGUJCNNDGITCFWCVGFVJTQWIJQWVCPKPVGTXCN KPKPEJGUQTKPEGPVKOGVGTUCPFHTCEVKQPUVJGTGQHQT

EQTTGURQPFKPIVQVJGXQNWOGTCPIGYKVJKPYJKEJ OC[DGKFGPVKHKGFKPCPWOGTKECNUGTKGUYKVJQWV

TGCFKPIUQHNKSWKFNGXGNCTGVQDGOCFG TGHGTGPEGVQKPEJGUQTEGPVKOGVGTUQTHTCEVKQPU

5'ZVGTPCN)CWIG#UUGODNKGU#FFGF RTQXKFGFHQTGCEJUWEJTQFQTICWIGCPFGCEJVCPM

5&GUKIPCPF+PUVCNNCVKQP6JGICWIG QHVJGITCFWCVKQPQPVJGTQFQTICWIG+HCTQFQT

CUUGODN[UJCNNDGFGUKIPGFVQOGGVUCPKVCT[TG ICWIGKUCUUQEKCVGFYKVJDWVQPGVCPMKPNKGWQH

SWKTGOGPVUCPFUJCNNDGTGCFKN[CEEGUUKDNGHQT NKPGCTQTPWOGTKECNUGTKGUITCFWCVKQPUCPFXQNWOG

ENGCPKPIRWTRQUGU6JGICWIGCUUGODN[UJCNNDG

5)CWIG6WDG6JGICWIGVWDGUJCNNDG

55ECNG)TCFWCVKQPU6JGITCFWCVKQPNKPGU

#P[CVVCEJOGPVVQVJGICWIGVWDGUJCNNPQVCFXGTUGN[

NGUUVJCPOOKPQTKP

VJGTGQH+PGKVJGTECUGCXQNWOGEJCTVUJCNNDG

YKVJYJKEJKVKUCUUQEKCVGFUJQYKPIXCNWGUKPVGTOU
(CTO/KNM6CPMU



EJCTVXCNWGUKPVGTOUQHXQNWOGQHNKSWKFKPVJGVCPM CPFVJGHCEGQHVJGEJCTVUJCNNDGUQRTQVGEVGFVJCVKVU

OC[DGUJQYPFKTGEVN[QPVJGTQFQTICWIG NGVVGTKPICPFHKIWTGUYKNNPQVVGPFGCUKN[VQDGEQOG

58CNWGQH)TCFWCVGF+PVGTXCN6JGXCNWG =0QPTGVTQCEVKXGCUQH,CPWCT[?

QHCITCFWCVGFKPVGTXCNQPCICWIGTQFQTUWTHCEG 
#OGPFGF

ICWIGGZENWUKXGQPVJGKPVGTXCNHTQOVJGDQVVQOQH

VJGVCPMVQVJGNQYGUVITCFWCVKQPUJCNNPQVGZEGGF 5(QTC6CPMQH.KVGTUQT)CNNQPU


C .HQTCVCPMQHCPQOKPCNECRCEKV[QH. ECRCEKV[QH.QTICNQTNGUUUJCNNUJQY

QTNGUUICNHQTCVCPMQHCPQOKPCNECRCEKV[ XCNWGUCVNGCUVVQVJGPGCTGUV.QTICN

QHICNQTNGUU

5(QTC6CPMQH.KVGTUVQ.KVGTUQT


D .HQTCVCPMQHCPQOKPCNECRCEKV[QH. VQ)CNNQPU6JGXQNWOGEJCTVHQTCVCPMQH

VQ.KPENWUKXGICNHQTCVCPMQHC PQOKPCNECRCEKV[QH.VQ.QTICNVQ

PQOKPCNECRCEKV[QHICNVQICN ICNKPENWUKXGUJCNNUJQYXCNWGUCVNGCUVVQVJG

KPENWUKXG PGCTGUV.QTICN


E .HQTCVCPMQHCPQOKPCNECRCEKV[QH. 5(QTC6CPMQH)TGCVGTVJCP.KVGTUQT

VQ.KPENWUKXGICNHQTCVCPMQHC )CNNQPU6JGXQNWOGEJCTVHQTCVCPMQHPQOKPCN

PQOKPCNECRCEKV[QHICNVQICN ECRCEKV[QHITGCVGTVJCP.QTICNUJCNNUJQY

KPENWUKXG XCNWGUCVNGCUVVQVJGPGCTGUVICNNQPQT.


F .HQTCVCPMQHCPQOKPCNECRCEKV[QH.

VQ.KPENWUKXGICNHQTCVCPMQHC 5)CWIKPI

PQOKPCNECRCEKV[QHICNVQICNKP

ENWUKXG5.GXGN#HCTOOKNMVCPMUJCNNDGNGXGNCU


G .RNWU.HQTGCEJCFFKVKQPCN.QT QTKIKPCNICWIKPIQRGTCVKQP

HTCEVKQPVJGTGQHHQTVCPMUQHPQOKPCNECRCEKV[

CDQXG.QTICNRNWUICNHQTGCEJ 56Q&GNKXGT#HCTOOKNMVCPMUJCNNDG

CFFKVKQPCNICNQTHTCEVKQPVJGTGQHHQT QTKIKPCNN[ICWIGFVQFGNKXGT+HVJGVCPMKUICWIGFD[

VCPMUYKVJPQOKPCNECRCEKV[CDQXGICN OGCUWTKPIVJGVGUVNKSWKFKPVQVJGVCPMVJGKPUKFGVCPM


#OGPFGF YCNNUUJCNNHKTUVDGVJQTQWIJN[YGVVGFCPFVJGVCPMUJCNN

5&GUKIPQH+PFKECVKPI/GCPUQP6CPMUYKVJC HNQYJCUEGCUGF

%CRCEKV[)TGCVGTVJCP.KVGTUQTICNNQP

#P[HCTOOKNMVCPMYKVJCECRCEKV[ITGCVGTVJCP 52TGRCTCVKQPQH8QNWOG%JCTV9JGPCVCPM

.QTICNUJCNNDGGSWKRRGFYKVJCPGZVGTPCN KUICWIGFHQTVJGRWTRQUGUQHRTGRCTKPICXQNWOGEJCTV

ICWIGCUUGODN[ VQNGTCPEGUCTGPQVCRRNKECDNGCPFVJGEJCTVUJCNNDG

=0QPTGVTQCEVKXGCPFCRRNKECDNGQPN[VQVCPMU RTGRCTGFCUCEEWTCVGN[CURTCEVKECDNG

OCPWHCEVWTGFCHVGT,CPWCT[?


#FFGF 5+FGPVKHKECVKQP#VCPMCPFCP[ICWIGTQFUWTHCEG

5&GUKIPQH8QNWOG%JCTV VJGTGYKVJUJCNNDGOWVWCNN[KFGPVKHKGFCUD[CEQOOQP

5)GPGTCN#XQNWOGEJCTVUJCNNUJQYXQNWOG

XCNWGUQPN[QXGTVJGGPVKTGTCPIGQHVJGXQNWOGQHVJG

VCPMHTQORGTEGPVQHECRCEKV[QTO 
ICN 

YJKEJGXGTKUNGUUVQKVUOCZKOWOECRCEKV[#NNNGVVGTU

CPFHKIWTGUQPVJGEJCTVUJCNNDGFKUVKPEVCPFGCUKN[

TGCFCDNG6JGEJCTVUJCNNDGUWDUVCPVKCNN[EQPUVTWEVGF

QDNKVGTCVGFQTKNNGIKDNG

QT.GUU6JGXQNWOGEJCTVHQTCVCPMQHPQOKPCN


#OGPFGF

UJQYPD[VJGNGXGNKPFKECVKPIOGCPUFWTKPIVJG

VJGPDGFTCKPGFHQTUGEQPFUCHVGTVJGOCKPFTCKPCIG

ICWIGURKTKVNGXGNCPFXQNWOGEJCTVKPVGPFGFVQDGWUGF

UGTKCNPWODGTKPCRTQOKPGPVCPFRGTOCPGPVOCPPGT

00QVGU

06GUV.KSWKF9CVGTUJCNNDGWUGFCUVJGVGUVNKSWKF

HQTCHCTOOKNMVCPM
(CTO/KNM6CPMU



0'XCRQTCVKQPCPF8QNWOG%JCPIG%CTGUJCNNDG 
#FFGF

GZGTEKUGFVQTGFWEGVQCOKPKOWOGXCRQTCVKQPNQUUGUCPF

XQNWOGEJCPIGUTGUWNVKPIHTQOEJCPIGUKPVGORGTCVWTGQH

VJGVGUVNKSWKF

06Q&GNKXGT#HCTOOKNMVCPMUJCNNDGVGUVGFVQ

FGNKXGT+HVJGVCPMKUICWIGFD[OGCUWTKPIVJGVGUV

NKSWKFFGNKXGTGFKPVQVJGVCPMVJGKPUKFGVCPMYCNNUUJCNN

HKTUVDGVJQTQWIJN[YGVVGFCPFVJGVCPMVJGPUJCNNDG

FTCKPGFHQTUGEQPFUCHVGTVJGOCKPFTCKPCIGHNQYJCU

EGCUGF

0.GXGN#HCTOOKNMVCPMUJCNNDGNGXGNCUUJQYP

D[VJGNGXGNKPFKECVKPIOGCPUFWTKPIICWIKPICPFVGUVKPI

06GUV/GVJQFU#EEGRVCPEGVGUVUQHOKNMVCPMUOC[

DGQHGKVJGTVJGRTQXGTOGVJQFQTVJGOCUVGTOGVGTOGVJQF

RTQXKFGFVJCVVJGOCUVGTOGVGTKPIU[UVGOKUECRCDNGQH

QRGTCVKPIYKVJKPRGTEGPVQHVJGCRRNKECDNGVQNGTCPEG

HQWPFKP65WDUGSWGPVVGUVUOC[DGQHGKVJGTVJG

RTQXGTOGVJQFQTVJGOCUVGTOGVGTOGVJQFRTQXKFGFVJCV

VJGOCUVGTOGVGTKPIU[UVGOKUECRCDNGQHQRGTCVKPIYKVJKP

RGTEGPVQHVJGCRRNKECDNGVQNGTCPEGHQWPFKP6

08GTKHKECVKQPQH/CUVGT/GVGTKPI5[UVGOU#

OCUVGTOGVGTKPIU[UVGOWUGFVQICWIGCOKNMVCPMUJCNN

DGXGTKHKGFDGHQTGCPFCHVGTVJGICWIKPIRTQEGUU#

OCUVGTOGVGTKPIU[UVGOWUGFVQECNKDTCVGCOKNMVCPM

UJCNNDGXGTKHKGFDGHQTGUVCTVKPIVJGECNKDTCVKQPCPF

TGXGTKHKGFCVNGCUVGXGT[SWCTVGTQHVJGVCPMECRCEKV[QT

GXGT[.ICNYJKEJGXGTKUITGCVGT


#FFGF

06GORGTCVWTG%JCPIGUKP9CVGT5WRRN[

9JGPWUKPICOCUVGTOGVGTKPIU[UVGOVQICWIGQT

ECNKDTCVGCOKNMVCPMVJGQHHKEKCNUJCNNOQPKVQTVJG

VGORGTCVWTGQHVJGYCVGTDGHQTGCPFCHVGTEJCPIKPI

UQWTEGUQHUWRRN[+HVJGYCVGTVGORGTCVWTGQHVJGPGY

UQWTEGEJCPIGUD[OQTGVJCPu%u(HTQOVJG

RTGXKQWUUWRRN[VJGQHHKEKCNUJCNNTGXGTKH[VJGCEEWTCE[

QHVJGOCUVGTOGVGTKPIU[UVGOCUUQQPCURQUUKDNGCHVGT

VJGU[UVGOTGCEJGUVGORGTCVWTGGSWKNKDTKWOYKVJVJG

PGYUWRRN[UQWTEG


#FFGF

04GCFKPIVJG/GPKUEWU9JGPCTGCFKPIQTUGVVKPI

KUVQDGQDVCKPGFHTQOCOGPKUEWUHQTOGFD[OKNMQTQVJGT

QRCSWGNKSWKFVJGKPFGZQTTGCFKPINKPGKUVJGRQUKVKQPQH

VJGJKIJGUVRQKPVQHVJGEGPVGTQHVJGOGPKUEWU9JGP

ECNKDTCVKPICFGXKEGYKVJYCVGTCPFVJGFGXKEGKUVQDG

WUGFYKVJCPQRCSWGNKSWKFVJGTGCFKPIUJQWNFDGQDVCKPGF

CEEQTFKPIN[VJCVKUVJGRQUKVKQPQHVJGJKIJGUVRQKPVQHVJG

EGPVGTQHVJGOGPKUEWU

66QNGTCPEGU

6#RRNKECVKQP6JGVQNGTCPEGUJGTGKPCHVGTRTGUETKDGF

UJCNNDGCRRNKGFGSWCNN[VQGTTQTUKPGZEGUUCPFGTTQTUKP

FGHKEKGPE[

6/KPKOWO6QNGTCPEG8CNWGU1PCRCTVKEWNCT

VCPMVJGOCKPVGPCPEGCPFCEEGRVCPEGVQNGTCPEGCRRNKGF

UJCNNDGPQVUOCNNGTVJCPVJGXQNWOGEQTTGURQPFKPIVQVJG

ITCFWCVGFKPVGTXCNCVVJGRQKPVQHVGUVFTCHVQPVJG

KPFKECVKPIOGCPUQT.ICNYJKEJGXGTKUITGCVGT


#OGPFGF

6$CUKE6QNGTCPEG8CNWGU6JGDCUKEOCKPVGPCPEG

CPFCEEGRVCPEGVQNGTCPEGUJCNNDGRGTEGPVQHVJG

XQNWOGQHVGUVNKSWKFKPVJGVCPMCVGCEJVGUVFTCHV


#OGPFGF

6$CUKE6QNGTCPEG8CNWGU/CUVGT/GVGT/GVJQF

6JGDCUKEOCKPVGPCPEGCPFCEEGRVCPEGVQNGTCPEGHQTVCPMU

VGUVGFD[VJGOCUVGTOGVGTOGVJQFUJCNNDGRGTEGPVQH

VJGXQNWOGQHVGUVNKSWKFKPVJGVCPMCVGCEJVGUVFTCHV


#FFGF

747UGT4GSWKTGOGPVU

74+PUVCNNCVKQP#UVCVKQPCT[VCPMUJCNNDGTKIKFN[

KPUVCNNGFKPNGXGNYKVJQWVVJGWUGQHTGOQXCDNGDNQEMUQT

UJKOUWPFGTVJGNGIU+HUWEJVCPMKUPQVOQWPVGF

RGTOCPGPVN[KPRQUKVKQPVJGEQTTGEVRQUKVKQPQPVJGHNQQT

HQTGCEJNGIUJCNNDGENGCTN[CPFRGTOCPGPVN[FGHKPGF

74.GXGN%QPFKVKQP

745VCVKQPCT[6CPM#UVCVKQPCT[HCTOOKNM

VCPMUJCNNDGOCKPVCKPGFKPNGXGN

74.GXGNKPI.WIU+HNGXGNKPINWIUCTG

RTQXKFGFQPCUVCVKQPCT[VCPMUWEJNWIUUJCNNPQVDG

JCOOGTGFQTHKNGFVQGUVCDNKUJQTEJCPIGCNGXGN

EQPFKVKQPQHVJGVCPM

742QTVCDNG6CPM1PCRQTVCDNGVCPM

OGCUWTGOGPVTGCFKPIUUJCNNDGOCFGQPN[YJGPVJG

VCPMKUCRRTQZKOCVGN[NGXGNVJCVKUYJGPKVKUPQVQWV

QHNGXGND[OQTGVJCPRGTEGPVQTCRRTQZKOCVGN[

FGITGGUKPCP[FKTGEVKQP

749GKIJV%JCTV#PCWZKNKCT[YGKIJVEJCTVOC[

DGRTQXKFGFQPYJKEJUJCNNDGRTQOKPGPVN[FKURNC[GFVJG
(CTO/KNM6CPMU



YGKIJVRGTWPKVXQNWOGXCNWGWUGFVQFGTKXGVJGYGKIJV

XCNWGUHTQOVJGQHHKEKCNXQNWOGEJCTV

747UG#HCTOOKNMVCPMUJCNNPQVDGWUGFVQ

OGCUWTGSWCPVKVKGUITGCVGTVJCPCPCOQWPVVJCVECPDG

CIKVCVGFYKVJQWVQXGTHNQYKPI
4.43. Measure-Containers



Sec. 4.43. Measure-Containers

A. Application

A.1. - This code applies to measure-containers, including

lids or closures if such are necessary to provide total

enclosure of the measured commodity, as follows:

(a) Retail measure-containers intended to be used only

once to determine at the time of retail sale, and from

bulk supply, the quantity of commodity on the basis of

liquid measure. The retail measure-container serves

as the container for the delivery of the commodity.

(b) Prepackaged measure-containers intended to be used

only once to determine in advance of sale the quantity

of a commodity (such as ice cream, ice milk, or

sherbet) on the basis of liquid measure. The

prepackaged measure-container serves as the

container for the delivery of the commodity, in either

a wholesale or a retail marketing unit.

A.2. - This code does not apply to rigid containers used for

milk, cream, or other fluid dairy products, which are

covered by packaging requirements.

A.3. - See also Sec. 1.10; General Code requirements.

S. Specifications

S.1. Units. - The capacity of a measure-container shall be

a multiple of or a binary submultiple of a quart or a liter,

and the measure shall not be subdivided. However, for

prepackaged measure-containers, any capacity less than

1/4 L or 1/2 liquid pint shall be permitted.

(Amended 1979)

S.2. Capacity Point. - The capacity of a measure-container

shall be sharply defined by:

(a) the top edge,

(b) a line near the top edge, or

(c) the horizontal cross-sectional plane established by the

bottom surface of the removable lid or cap when

seated in the container.

S.3. Shape. - A measure-container shall be designed as

some suitable geometrical shape, and its capacity shall be

determined without distortion from its normal assembled

shape.

S.4. Marking.

S.4.1. Capacity Point. - If the capacity point of a measure-container is defined by a line, the container shall be

marked conspicuously on its side with a suitable

statement clearly identifying this line as the capacity

point.

S.4.2. Capacity Statement. - A measure-container shall

be clearly and conspicuously marked with a statement of

its capacity in terms of one of the units prescribed in S.1.

N. Notes

N.1. Test Liquid. - Water shall be used as the test liquid

for a measure-container.

N.2. Preparation of Container for Test.

N.2.1. General. - Before an actual test is begun, a

measure-container shall, if necessary, be so restrained

that it will maintain its normal assembled shape and that

its sides will not bulge when it is filled with water.

N.2.2. Restraining Form for Test.

N.2.2.1. For Rectangular Containers of One Liter,

One Quarter Less. - Bulging of the sides of a rectangular measure-container of 1-liter (1-quart) capacity or

less may be controlled by holding against each side of

the container, with a cord, rubber bands, or tape, a

metal plate or a piece of heavy cardboard slightly

smaller than the side of the container.

(Amended 1979)
4.43. Measure-Containers



6CDNG

#EEGRVCPEG6QNGTCPEGU

KP'ZEGUUCPFKP&GHKEKGPE[HQT/GCUWTG%QPVCKPGTU

0QOKPCN%CRCEKV[

6QNGTCPEG

+PGZEGUU +PFGHKEKGPE[

OKNNKNKVGTU OKNNKNKVGTU

NKVGTQTNGUU  

NKVGT  

NKVGT  

1XGTNKVGT #FFRGTNKVGT

OKNNKNKVGTU

#FFRGTNKVGT

OKNNKNKVGTU

HNWKFFTCOU EWDKEKPEJGU HNWKFFTCOU EWDKEKPEJGU

RKPVQTNGUU    

RKPV    

SWCTV    

SWCTVU    

SWCTVU    

SWCTVU    

1XGTSWCTVU #FFRGTSWCTV

HNWKFFTCOU

#FFRGTSWCTV

EWDKEKPEJ

#FFRGTSWCTV

HNWKFFTCOU

#FFRGTSWCTV

EWDKEKPEJ

N.2.2.2. For Rectangular Prepackaged Measure--

container of Two Quarts or Two Liters or Greater.

- A rectangular prepackaged measure-container of

2-liter (2-quart) capacity or greater shall be supported

during a test by a rigid restraining form. This form

shall restrain not less than the entire area of the central

two-thirds of each side of the container, measured

from bottom to top. The inside width dimension of

any side panel of the restraining form shall be 1.6 mm

(1/16 in) greater than the corresponding outside

dimension of the container. (The outside width

dimension of any side panel of the container shall be

established by adding to the inner side center-of-score

to center-of-score dimension two thicknesses of the

board used, and the sum thus obtained shall be

rounded off to the nearest 0.4 mm (1/64 in).

(Amended 1979)

T. Tolerances

T.1. Tolerances on an Individual Measure. - The

acceptance tolerances in excess and in deficiency on an

individual measure shall be as shown in Table 1.

T.2. Tolerance on Average Capacity. - The average

capacity on a random sample of ten measures selected from

a lot of 25 or more shall be equal to or greater than the

nominal capacity.

(Amended 1979)

 UR. User Requirements

UR.1. Limitation of Use. - The use of a measure-container

with a rectangular cross section of a capacity of 2 L (2 qt)

or greater shall be limited to the packaging, in advance of

sale, of ice cream, sherbet, or other similar frozen desserts.

(Amended 1979)
)TCFWCVGU



6CDNG

&GUKIP&GVCKNUHQT)TCFWCVGU

0QOKPCN

ECRCEKV[ 6QDGITCFWCVGFDGVYGGP 8CNWGQH

ITCFWCVGFKPVGTXCNU

0WODGTCVGCEJ

ITCFWCVKQPFKXKUKDNG

D[

OKNNKNKVGTU OKNNKNKVGTU OKNNKNKVGTU OKNNKNKVGTU















CPF

CPF

CPF

CPF

CPF

CPF

CPF











































OKPKOU OKPKOU OKPKOU OKPKOU





CPF

CPF









C

D

HNWKFFTCOU HNWKFFTCOU HNWKFFTCOU HNWKFFTCOU





CPF

CPF













HNWKFQWPEGU HNWKFQWPEGU HNWKFQWPEGU HNWKFQWPEGU











CPF

CPF

CPF

CPF

CPF































#PFKPCFFKVKQPCVVJGHKTUVOKPKOITCFWCVKQP C

#PFKPCFFKVKQPCVVJGHKTUVOKPKOITCFWCVKQP D

5GE)TCFWCVGU

##RRNKECVKQP 55RGEKHKECVKQPU

#6JKUEQFGCRRNKGUVQUWDFKXKFGFINCUUOGCUWTGUQH 57PKVU0QOKPCNECRCEKVKGUITCFWCVKQPTCPIGU

ECRCEKV[GKVJGTE[NKPFTKECNQTEQPKECNKPUJCRG XCNWGUQHITCFWCVGFKPVGTXCNUCPFPWODGTGFITCFWCVKQPU

#5GGCNUQ5GE)GPGTCN%QFGTGSWKTGOGPVU RQTVKQPUQHFQWDNGUECNGITCFWCVGUUJCNNDGCUUJQYPKP

CRRNKECDNGVQUKPINGUECNGITCFWCVGUCPFVQVJGCRRTQRTKCVG

6CDNG
)TCFWCVGU



5+PKVKCN+PVGTXCN#ITCFWCVGUJCNNJCXGCPKPKVKCN GZVGPFCVNGCUVQPGJCNHVJGFKUVCPEGCTQWPFVJG

KPVGTXCNVJCVKUPQVUWDFKXKFGFGSWCNVQPQVNGUUVJCPQPG ITCFWCVG

HKHVJCPFPQVOQTGVJCPQPGHQWTVJQHVJGECRCEKV[QHVJG

ITCFWCVG 51PC&QWDNG5ECNGQTC&WRNGZ)TCFWCVG

55JCRG#ITCFWCVGQHCECRCEKV[QHOQTGVJCP ENGCTURCEGDGVYGGPVJGGPFUQHVJGOCKPITCFWCVKQPU

O.HNWKFFTCOUOC[DGQHGKVJGTVJGE[NKPFTKECNQT QPVJGVYQUECNGUCPFVJKUURCEGUJCNNDGCRRTQZKOCVGN[

EKTEWNCTEQPKECNV[RG#ITCFWCVGQHCECRCEKV[QHO. FGITGGUHTQOVJGNKRQHVJGITCFWCVGCPFUJCNN


HNWKFFTCOUQTNGUUUJCNNDGQHVJGUKPINGUECNG EQPHQTOVQVJGTGSWKTGOGPVUQH6CDNG

E[NKPFTKECNV[RG

5/CVGTKCN#ITCFWCVGUJCNNDGOCFGQHIQQF

SWCNKV[VJQTQWIJN[CPPGCNGFENGCTVTCPURCTGPVINCUUHTGG

HTQODWDDNGUCPFUVTGCMUVJCVOKIJVCHHGEVVJGCEEWTCE[QH

OGCUWTGOGPV6JGINCUUUJCNNDGWPKHQTOKPVJKEMPGUUCPF

UJCNNPQVDGGZEGUUKXGN[VJKEM

5&KOGPUKQPCN2TQRQTVKQPU

51PC%KTEWNCT%QPKECN)TCFWCVG6JGKPUKFG

OGCUWTGOGPVHTQOVJGDQVVQOQHCEKTEWNCTEQPKECN

ITCFWCVGVQVJGECRCEKV[ITCFWCVKQPUJCNNDGPQVNGUU

VJCPVYQVKOGUVJGKPUKFGFKCOGVGTCVVJGECRCEKV[

ITCFWCVKQP6JGKPUKFGOGCUWTGOGPVHTQOVJGDQVVQOQH

VJGITCFWCVGVQVJGRQKPVTGRTGUGPVKPIQPGHQWTVJQHVJG

ECRCEKV[UJCNNDGPQVNGUUVJCPVJGKPUKFGFKCOGVGTCVVJCV

RQKPV

51PC%[NKPFTKECN)TCFWCVG6JGKPUKFG

OGCUWTGOGPVHTQOVJGDQVVQOQHCE[NKPFTKECNITCFWCVG

VQVJGECRCEKV[ITCFWCVKQPUJCNNDGPQVNGUUVJCPHKXG

VKOGUVJGKPUKFGFKCOGVGTCVVJGECRCEKV[ITCFWCVKQP

5$CUG6JGDCUGQHVJGITCFWCVGUJCNNDG

RGTRGPFKEWNCTVQVJGXGTVKECNCZKUQHVJGITCFWCVG6JG

FKCOGVGTQHVJGDCUGUJCNNDGQHUWEJUK\GVJCVVJGGORV[

ITCFWCVGYKNNTGOCKPUVCPFKPIQPCPKPENKPGFUWTHCEGQH

RGTEGPVQTCRRTQZKOCVGN[FGITGGUHTQOVJGJQTK\QP

VCN

5&GUKIPQH)TCFWCVKQPU

5)GPGTCN)TCFWCVKQPUUJCNNDGRGTRGPFKEWNCT

VQVJGXGTVKECNCZKUQHVJGITCFWCVGCPFRCTCNNGNVQGCEJ

QVJGT)TCFWCVKQPUUJCNNDGEQPVKPWQWUQHWPKHQTO

VJKEMPGUUPQVITGCVGTVJCPOOKPENGCTN[

XKUKDNGRGTOCPGPVCPFKPFGNKDNGWPFGTPQTOCN

EQPFKVKQPUQHWUG


#OGPFGF

51PC5KPING5ECNG)TCFWCVG1PCUKPING

UECNGITCFWCVGVJGOCKPITCFWCVKQPUUJCNNEQORNGVGN[

GPEKTENGVJGITCFWCVGCPFUWDQTFKPCVGITCFWCVKQPUUJCNN

1PCFQWDNGUECNGQTFWRNGZITCFWCVGVJGTGUJCNNDGC

6CDNG

%NGCT5RCEG$GVYGGP'PFUQH/CKP

)TCFWCVKQPUQP&QWDNG5ECNG)TCFWCVGU

+PUKFGFKCOGVGTQH

ITCFWCVGCVVJG

ITCFWCVKQPU


KPEJGU

%NGCTURCEG

DGVYGGPGPFUQH

OCKPITCFWCVKQPU


KPEJ

.GUUVJCP VQ

VQKPENWUKXG VQ

1XGT VQ

5$CUKUQH)TCFWCVKQP#ITCFWCVGUJCNNDG

ITCFWCVGFVQFGNKXGTYJGPVJGVGORGTCVWTGQHVJG

ITCFWCVGKUE%
E(CPFUJCNNDGOCTMGF

CEEQTFKPIN[KPCRGTOCPGPVCPFEQPURKEWQWUOCPPGT

5/CTMKPI4GSWKTGOGPVU'CEJOCKPITCFWCVKQP

UJCNNDGOCTMGFVQUJQYKVUXCNWG+PVGTOGFKCVG

ITCFWCVKQPUUJCNNPQVDGOCTMGF8CNWGHKIWTGUUJCNNDG

WPKHQTON[RQUKVKQPGFGKVJGTFKTGEVN[WRQPQTKOOGFKCVGN[

CDQXGVJGITCFWCVKQPUVQYJKEJVJG[TGHGT(KIWTGURNCEGF

WRQPITCFWCVKQPUUJCNNDGUGVKPHTQOVJGGPFUQHVJG

ITCFWCVKQPUCUWHHKEKGPVFKUVCPEGVQCNNQYVJGGPFUQHVJG

ITCFWCVKQPUVQDGWUGFKPOCMKPICUGVVKPI

00QVGU

06GUV.KSWKF9CVGTUJCNNDGWUGFCUVJGVGUVNKSWKF

HQTITCFWCVGU

06GORGTCVWTG%QPVTQN&WTKPIVJGVGUVQHC

ITCFWCVGCRRTQRTKCVGRTGECWVKQPUUJCNNDGGZGTEKUGFVQ

TGFWEGCP[FGVTKOGPVCNVGORGTCVWTGGHHGEVUVQVJGRTCE

VKECDNGOKPKOWO
)TCFWCVGU



6CDNG

/CKPVGPCPEGCPF#EEGRVCPEG6QNGTCPEGUKP'ZEGUUCPF

KP&GHKEKGPE[HQT)TCFWCVGU

+PUKFGFKCOGVGTQHITCFWCVG

6QNGTCPEG

+PUKFGFKCOGVGTQH

ITCFWCVG

6QNGTCPEG

(TQO VQDWVPQV

KPENWFKPI (TQO VQDWVPQV

KPENWFKPI

OKNNKOGVGTU OKNNKNKVGTU KPEJGU OKPKOU







































































































































































































66QNGTCPEGU

6/CKPVGPCPEGCPFCEEGRVCPEGVQNGTCPEGUKPGZEGUU

CPFKPFGHKEKGPE[UJCNNDGCUUJQYPKP6CDNGHQT

ITCFWCVGUVJCVCTGITCFWCVGFVQEQPVCKPQTVQFGNKXGT


6JGVQNGTCPEGVQDGCRRNKGFCVCP[ITCFWCVKQPKU

FGVGTOKPGFD[VJGKPUKFGFKCOGVGTQHVJGITCFWCVGCVVJG

ITCFWCVKQPKPSWGUVKQP


6CDNG#OGPFGF
&T[/GCUWTGU



5GE&T[/GCUWTGU

##RRNKECVKQP 56QR4GKPHQTEGOGPV6JGVQRGFIGQHCOGCUWTG

#6JKUEQFGCRRNKGUVQTKIKFOGCUWTGUQHECRCEKV[ DCUMGVQHCECRCEKV[QHSWCTVQTOQTGVJKUTGKPHQT

FGUKIPGFHQTIGPGTCNCPFTGRGCVGFWUGKPVJGOGCUWTGOGPV EGOGPVUJCNNDGKPVJGHQTOQHCHKTON[CVVCEJGFOGVCN

QHUQNKFUKPENWFKPIECRCEKVKGUQHDWUJGNQTOQTG DCPF

#6JKUEQFGFQGUPQVCRRN[VQUVCPFCTFEQPVCKPGTU

WUGFHQTVJGOGCUWTGOGPVQHHTWKVUCPFXGIGVCDNGUCPFCU

UJKRRKPIEQPVCKPGTUVJGTGQH

#6JKUEQFGFQGUPQVCRRN[VQDGTT[DCUMGVUCPF

DQZGUHQTYJKEJUGG5GE%QFGHQT$GTT[$CUMGVU

CPF$QZGU


#FFGF

#5GGCNUQ5GE)GPGTCN%QFGTGSWKTGOGPVU

55RGEKHKECVKQPU

57PKVU6JGECRCEKV[QHCOGCUWTGUJCNNDGDWUJGN

COWNVKRNGQHVJGDWUJGNQTCDKPCT[UWDOWNVKRNGQHVJG

DWUJGNCPFVJGOGCUWTGUJCNNPQVDGUWDFKXKFGFQTFQWDNG

GPFGF

5/CVGTKCN#FT[OGCUWTGUJCNNDGOCFGQHCP[

UWKVCDNGOCVGTKCNVJCVYKNNTGVCKPKVUUJCRGFWTKPIPQTOCN

WUCIG

55JCRG#OGCUWTGQVJGTVJCPCDCUMGVQHC

ECRCEKV[QHDWUJGNQTNGUUUJCNNDGE[NKPFTKECNQT

EQPKECNKPUJCRG6JGVQRFKCOGVGTUJCNNKPPQECUGDGNGUU

VJCPVJGCRRTQRTKCVGOKPKOWOFKCOGVGTUJQYPKP6CDNG


PGZVRCIG6JGDQVVQOQHCOGCUWTGQVJGTVJCPC

DCUMGVUJCNNDGRGTRGPFKEWNCTVQVJGXGTVKECNCZKUQHVJG

OGCUWTGCPFUJCNNDGHNCVGZEGRVVJCVCOGVCNDQVVQOOC[

DGUNKIJVN[EQTTWICVGF6JGDQVVQOQHCOGCUWTGUJCNNPQV

DGCFLWUVCDNGQTOQXCDNG

5%QPKECN&T[/GCUWTG+HEQPKECNVJGVQR

FKCOGVGTUJCNNGZEGGFVJGDQVVQOFKCOGVGTD[PQVOQTG

VJCPRGTEGPVQHVJGDQVVQOFKCOGVGT

5%CRCEKV[2QKPV6JGECRCEKV[QHCOGCUWTGUJCNN

DGFGVGTOKPGFD[VJGVQRGFIGQHVJGOGCUWTG

UJCNNDGTGKPHQTEGF1PCYQQFGPOGCUWTGQVJGTVJCPC

6CDNG

/KPKOWO6QR&KCOGVGTUHQT

&T[/GCUWTGUQVJGTVJCP$CUMGVU

0QOKPCNECRCEKV[ /KPKOWOVQRFKCOGVGT

KPEJGU

RKPV 

SWCTV 

SWCTVU 

RGEM 

RGEM 

DWUJGN 

5/CTMKPI4GSWKTGOGPVU#OGCUWTGUJCNNDG

EQPURKEWQWUN[OCTMGFQPKVUUKFGYKVJCUVCVGOGPVQHKVU

ECRCEKV[+HVJGECRCEKV[KUUVCVGFKPVGTOUQHVJGRKPVQT

SWCTVVJGYQTF&T[UJCNNDGKPENWFGF6JGECRCEKV[

UVCVGOGPVUJCNNDGKPNGVVGTUQHVJGHQNNQYKPIFKOGPUKQPU


C #VNGCUVKPEJJKIJCPFKPEJYKFGQPC

OGCUWTGQHCP[ECRCEKV[DGVYGGPRKPVCPF

RGEM


D #VNGCUVKPEJJKIJCPFKPEJYKFGQPCOGCUWTG

QHCECRCEKV[QHDWUJGNQTOQTG


E 1PCOGCUWTGQHCECRCEKV[QHRKPVQTNGUUVJG

UVCVGOGPVUJCNNDGCURTQOKPGPVCURTCEVKECDNG

EQPUKFGTKPIVJGUK\GCPFFGUKIPQHUWEJOGCUWTG
&T[/GCUWTGU



6CDNG

/CKPVGPCPEG6QNGTCPEGUKP'ZEGUUCPFKP&GHKEKGPE[

HQT&T[/GCUWTGU

0QOKPCNECRCEKV[

6QNGTCPEG

+PGZEGUU +PFGHKEKGPE[

EWDKEKPEJGU EWDKEKPEJGU

RKPVQTNGUU  

RKPV  

RKPV  

RKPV  

RKPV  

RKPV  

SWCTV  

SWCTVU  

RGEM  

RGEM  

DWUJGN  

DWUJGN  

00QVGU 66QNGTCPEGU

06GUVKPI/GFKWO 6/CKPVGPCPEGVQNGTCPEGUKPGZEGUUCPFKPFGHKEKGPE[

09CVGTVKIJV&T[/GCUWTGU9CVGTUJCNNDG DGQPGJCNHVJGOCKPVGPCPEGVQNGTCPEGU

WUGFCUVJGVGUVKPIOGFKWOHQTYCVGTVKIJVFT[OGCUWTGU

00QPYCVGTVKIJV&T[/GCUWTGU#FT[

OGCUWTGUJCNNDGVGUVGFGKVJGTXQNWOGVTKECNN[WUKPITCRG

UGGFCUCVGUVKPIOGFKWOQTIGQOGVTKECNN[VJTQWIJ

KPUKFGOGCUWTGOGPVCPFECNEWNCVKQP


#OGPFGF

UJCNNDGCUUJQYPKP6CDNG#EEGRVCPEGVQNGTCPEGUUJCNN
$GTT[$CUMGVUCPF$QZGU



6CDNG

/CKPVGPCPEGCPF#EEGRVCPEG6QNGTCPEGU

KP'ZEGUUCPFKP&GHKEKGPE[

0QOKPCN

%CRCEKV[

6QNGTCPEG

+PGZEGUU

EWDKEKPEJGU

+PFGHKEKGPE[

EWDKEKPEJGU

RKPV  

RKPV  

SWCTV  

5GE$GTT[$CUMGVUCPF$QZGU

##RRNKECVKQP 00QVGU

#6JKUEQFGCRRNKGUVQDCUMGVUCPFDQZGUHQTDGTTKGU 0/GVJQFQH6GUV#DGTT[DCUMGVQTDQZOC[DG

CPFUOCNNHTWKVUKPECRCEKVKGUQHFT[SWCTVCPFNGUU VGUVGFGKVJGTXQNWOGVTKECNN[WUKPITCRGUGGFCUVJGVGUVKPI

#5GGCNUQ5GE)GPGTCN%QFGTGSWKTGOGPVU UKQPOGCUWTGOGPVCPFECNEWNCVKQP

55RGEKHKECVKQPU 66QNGTCPEGU

57PKVU6JGECRCEKV[QHCDGTT[DCUMGVQTDQZUJCNN 66QNGTCPEGUQP+PFKXKFWCN/GCUWTGU/CKPVGPCPEG

DGFT[RKPVFT[RKPVQTFT[SWCTV CPFCEEGRVCPEGVQNGTCPEGUKPGZEGUUCPFFGHKEKGPE[QPCP

5/CVGTKCNU#DGTT[DCUMGVQTDQZUJCNNDGOCFGQH

CP[UWKVCDNGOCVGTKCNUVJCVYKNNTGVCKPKVUUJCRGFWTKPI 66QNGTCPEGUQP#XGTCIG%CRCEKV[6JGCXGTCIG

PQTOCNHKNNKPIUVQTCIGCPFJCPFNKPI ECRCEKV[QPCTCPFQOUCORNGQHOGCUWTGUUGNGEVGF

5%CRCEKV[2QKPV6JGECRCEKV[QHCDGTT[DCUMGVQT VJGPQOKPCNECRCEKV[

DQZUJCNNDGFGVGTOKPGFD[KVUVQRGFIGU 
#OGPFGF

OGFKWOQTIGQOGVTKECNN[VJTQWIJCEEWTCVGKPUKFGFKOGP

KPFKXKFWCNOGCUWTGUJCNNDGCUUJQYPKP6CDNG

HTQOCNQVQHQTOQTGUJCNNDGGSWCNVQQTITGCVGTVJCP


 Section 5

Page

5.50. Fabric-Measuring Devices 5-3

5.51. Wire- and Cordage-Measuring Devices 5-7

5.52. Linear Measures 5-9

5.53. Odometers 5-11

5.54. Taximeters 5-13

5.55. Timing Devices 5-19

5.56.(a) Grain Moisture Meters 5-22

5.56.(b) Grain Moisture Meters 5-31

5.57. Near-Infrared Grain Analyzers - Tentative Code 5-37

5.58. Multiple Dimension Measuring Devices - Tentative Code 5-43

5.50. Fabric-Measuring Devices



Sec. 5.50. Fabric-Measuring Devices

A. Application

A.1. - This code applies only to mechanisms and machines

designed to indicate automatically (with or without

value-computing capabilities) the length of fabric passed

through the measuring elements.

A.2. - Insofar as they are clearly appropriate, the

requirements and provisions of this code apply also to

devices designed for the commercial measurement of other

material similar to fabrics, in sheet, roll, or bolt form.

A.3. - See also Sec. 1.10; General Code requirements.

S. Specifications

S.1. Units. - A fabric-measuring device shall indicate

lengths in terms of 10 cm (1/8 yd), 25 cm (1/4 yd), 50 cm

(1/2 yd), and meters (yd). In addition, lengths may be

indicated in terms of any or all of the following

subdivisions: 30 cm (1/3 yd), 6 cm (1/16 yd), meters and

centimeters (feet and inches). Digital indicators may

indicate values in decimal fractions. 

(Amended 1977)

S.2. Design of Indicating Elements.

S.2.1. Graduations.

S.2.1.1. Length. - Graduations shall be so varied in

length that they may be conveniently read.

S.2.1.2. Width. - In any series of graduations, the

width of a graduation shall in no case be greater than

the width of the minimum clear interval between

graduations, and the width of main graduations shall

be not more than 50 percent greater than the width of

subordinate graduations. Graduations shall in no case

be less than 0.2 mm (0.008 in) in width.

S.2.1.3. Clear Interval Between Graduations. - The

clear interval between graduations shall be at least 6

mm for cm graduations (1/4 in) for 1/8-yard graduations), and 3 mm for 20-cm graduations (1/8 inch for

1-inch graduations).

S.2.2. Indicator.

S.2.2.1. Symmetry. - The index of an indicator shall

be symmetrical with respect to the graduations, at least

throughout that portion of its length associated with

the graduations.

S.2.2.2. Length. - The index of an indicator shall

reach to the finest graduations with which it is used,

unless the indicator and the graduations are in the

same plane, in which case the distance between the

end of the indicator and the ends of the graduations,

measured along the line of the graduations, shall be

not more than 1.0 mm (0.04 in).

S.2.2.3. Width. - The index of an indicator shall not

be wider than the narrowest graduations with which it

is used, and shall in no case exceed 0.4 mm (0.015 in).

S.2.2.4. Clearance. - The clearance between the

index of an indicator and the graduations shall in no

case be more than 1.5 mm (0.06 in).

S.2.2.5. Parallax. - Parallax effects shall be reduced

to the practicable minimum.

S.2.3. Money-Value Computations.

S.2.3.1. Full-Computing Type. - In this type, the

money value at each of a series of unit prices shall be

computed automatically for every length within the

range of measurement of the fabric-measuring device.

Value graduations shall be provided and shall be

accurately positioned. The value of each graduated

interval shall be 1 cent at all prices per yard of 30

cents and less, and shall not exceed 2 cents at higher

prices per yard. Five-cent intervals may be

represented in the 2-cent range by special graduations,

but these shall not be positioned in the clear intervals

between graduations of the regular series.
5.50. Fabric-Measuring Devices



S.2.3.2. Limited-Computing Type. - In this type, the

money value at each of a series of unit prices shall be

computed automatically only for lengths corresponding to a definite series of length graduations. There

shall be no value graduations. At no position that the

chart can assume shall two value figures at the same

price per yard be completely and clearly exposed to

view at one time. Money values shown shall be

mathematically accurate, except that a fraction of less

than 1/2 cent shall be dropped and the next higher cent

shall be shown in the case of a fraction of 1/2 cent or

more. One of the following requirements shall be

met:

(a) There shall be a money-value computation for

each length graduation within the range of

measurement of the device.

(b) No money-value computation shall be exposed

to view except at such times as the device shows

a length indication for which a corresponding

series of value indications is computed.

(c) Each column or row of money-value computations shall be marked to show the length to

which the computations correspond, the device

shall be marked to show the character and

limitations of the computations, and there shall

be computations corresponding to at least 10 cm

(1/8 yd) throughout the range of measurement of

the device.

S.2.4. Return to Zero. - Primary indicating elements

shall be readily returnable to a definite zero indication.

Means shall be provided to prevent the return of the

indicating elements beyond their correct zero positions.

S.3. Marking Requirements. - If a device will not

accurately measure all fabrics, it shall be marked to indicate

clearly its limitations.

S.4. Design Accuracy. - Indications of length and money

value shall be accurate whether the values of the indications

are being increased or decreased.

 N. Notes

N.1. Testing Medium. - A fabric-measuring device shall

be tested with a suitable testing tape approximately 7.62 cm

(3 in) wide and with a graduated length of at least 11 m

(12 yd), made from such material and having such surface

finish as to provide dimensional stability and reduce

slippage to the practicable minimum.

 T. Tolerances

T.1. Tolerance Values. - Maintenance and acceptance

tolerances shall be as shown in Table 1.

UR. User Requirements

UR.1. Installation Requirements.

UR.1.1. Installation. - A fabric-measuring device shall

be securely supported and firmly fixed in position.

UR.2. Use Requirements.

UR.2.1. Limitation of Use. - A fabric-measuring device

shall be used to measure only those fabrics that it was

designed to measure, and in no case shall it be used to

measure a fabric that a marking on the device indicates

should not be measured.

UR.2.2. Return of Indicating Elements to Zero. - The

primary indicating elements shall be returned to zero

before each measurement.
5.50. Fabric-Measuring Devices



Table 1.

Maintenance and Acceptance Tolerances

for Fabric-Measuring Devices

Indication of

device

(yards)

Maintenance tolerance Acceptance tolerance

On underregistration

(inches)

On overregistration

(inches)

On underregistration

(inches)

On overregistration

(inches)

2 or less 3/8 1/4 1/4 1/8

3 3/8 5/16 1/4 5/32

4 1/2 5/16 1/4 5/32

5 5/8 3/8 5/16 3/16

6 3/4 3/8 3/8 3/16

7 and 8 1 1/2 1/2 1/4

9 1-1/4 5/8 5/8 5/16

10 and 11 1-1/2 3/4 3/4 3/8

12 and 13 1-3/4 7/8 7/8 7/16

14 and 15 2 1 1 1/2

Over 15 Add 1/8 inch per

indicated yard

Add 1/16 inch per

indicated yard

Add 1/16 inch

per indicated yard

Add 1/32 inch per

indicated yard
5.51. Wire- and Cordage-Measuring Devices



Sec. 5.51. Wire- and Cordage-Measuring Devices

A. Application

A.1. - This code applies to mechanisms and machines

designed to indicate automatically the length of cordage,

rope, wire, cable, or similar flexible material passed through

the measuring elements.

A.2. - See also Sec. 1.10; General Code requirements.

 S. Specifications

S.1. Units. - A wire- or cordage-measuring device shall

indicate lengths in terms of feet, yards, or meters, or combinations of units of the same measurement system, and shall

have minimum increments with values that do not exceed

the equivalent of 0.1 meter or 0.1 yard.

(Amended 1989)

S.2. Design of Indicating Elements.

S.2.1. Graduations.

S.2.1.1. Length. - Graduations shall be so varied in

length that they may be conveniently read.

S.2.1.2. Width. - In any series of graduations, the

width of a graduation shall in no case be greater than

the width of the minimum clear interval between

graduations, and the width of main graduations shall

be not more than 50 percent greater than the width of

subordinate graduations. Graduations shall in no case

be less than 0.2 mm (0.008 in), nor more than 1.0 mm

(0.04 in), in width.

S.2.1.3. Clear Interval Between Graduations. - The

clear interval between graduations shall be at least as

wide as the widest graduation, and in no case less than

0.8 mm (0.03 in).

S.2.2. Indicator.

S.2.2.1. Symmetry. - The index of an indicator shall

be symmetrical with respect to the graduations, at least

throughout that portion of its length associated with

the graduations.

S.2.2.2. Length. - The index of an indicator shall

reach to the finest graduations with which it is used,

unless the indicator and the graduations are in the

same plane, in which case the distance between the

end of the indicator and the ends of the graduations,

measured along the line of the graduations, shall be

not more than 1.0 mm (0.04 in).

S.2.2.3. Width. - The index of an indicator shall not

be wider than the narrowest graduations with which it

is used, and shall in no case exceed 0.4 mm (0.015 in).

S.2.2.4. Clearance. - The clearance between the

index of an indicator and the graduations shall in no

case be more than 1.5 mm (0.06 in).

S.2.2.5. Parallax. - Parallax effects shall be reduced

to the practicable minimum.

S.2.3. Zero Indication. - Primary indicating elements

shall be readily returnable to a definite zero indication.

S.3. Design of Measuring Elements.

S.3.1. Sensitiveness. - If the most sensitive element of

the indicating system utilizes an indicator and

graduations, the relative movement of these parts

corresponding to a measurement of 30 cm (1 ft) shall be

not less than 6 mm (1/4 in).

S.3.2. Slippage. - The measuring elements of a wire- or

cordage-measuring device shall be so designed and

constructed as to reduce to the practicable minimum any

slippage of material being measured and any lost motion

in the measuring mechanism.

S.3.3. Accessibility. - A wire- or cordage-measuring

device shall be so constructed that the measuring

elements are readily visible and accessible, without

disassembly of any supporting frame or section of the

main body, for purposes of cleaning or removing any

foreign matter carried into the mechanism by the material

being measured.

S.4. Marking Requirements.

S.4.1. Limitation of Use. - If a device will measure

accurately only certain configurations, diameters, types,

or varieties of materials, or with certain accessory

equipment, all limitations shall be clearly and

permanently stated on the device.

S.4.2. Operating Instructions. - Any necessary

operating instructions shall be clearly stated on the

device.
5.51. Wire- and Cordage-Measuring Devices



Table 1.

Maintenance and Acceptance Tolerances for

Wire- and Cordage-Measuring Devices

Indication of device

(feet)

Acceptance and maintenance tolerances

On underregistration

(inches)

On overregistration

20

Over 20 to 30

Over 30 to 40

Over 40 to 50

Over 50

6

8

10

12

Add 2 inches per indicated 10 feet

3

4

5

6

Add 1 inch per indicated 10 feet

S.4.3. Indications. - Indicating elements shall be

identified by suitable words or legends so that the values

of the indications will be unmistakable.

S.5. Design Accuracy. - Indications of length shall be

accurate whether the values of the indications are being

increased or decreased.

 N. Notes

N.1. Testing Medium. - Wherever feasible, a wire- or

cordage-measuring device shall be tested with a steel tape

not less than 10 mm (3/8 in) in width and at least 15 m

(50 ft) in length. When a device cannot be tested in this

manner because of the design of the device, it shall be

tested with a dimensionally stable material appropriately

marked and compared at frequent periodic intervals with a UR.2.2. Return to Zero. - The primary indicating

steel tape in order to assure that any marked interval is not elements of a wire- or cordage-measuring device shall be

in error by more than 1/3 of the tolerance of the device at returned to zero before each measurement.

that particular interval.

(Amended 1981)

N.2. Minimum Test. - Tests shall be conducted at a a manner as to cause slippage or inaccurate measminimum initial increment of 5 m (20 ft) and appropriate urement.

increments up to at least 15 m (50 ft).

T. Tolerances

T.1. Tolerance Values. - Maintenance and acceptance

tolerances shall be as shown in Table 1.

UR. User Requirements

UR.1. Installation Requirements.

UR.1.1. Installation. - A wire- or cordage-measuring

device shall be securely supported and firmly fixed in

position.

UR.2. Use Requirements.

UR.2.1. Limitation of Use. - A wire- or

cordage-measuring device shall be used to measure only

those materials that it was designed to measure, and in no

case shall it be used to measure a material that a marking

on the device indicates should not be measured.

UR.2.3. Operation of Device. - A wire- or

cordage-measuring device shall not be operated in such

UR.2.4. Cleanliness. - The measuring elements of a

wire- or cordage-measuring device shall be kept clean to

prevent buildup of dirt and foreign material that would

adversely affect the measuring capability of the device.
5.52. Linear Measures



Sec. 5.52. Linear Measures

A. Application

A.1. - This code applies to any linear measure or measure of

length, whether flexible or inflexible, permanently installed

or portable.

A.2. - See also Sec. 1.10; General Code requirements.

 S. Specifications

S.1.M. Units. - A linear measure may be in total length, and

the total length may be subdivided in any or all of the

following:

(a) centimeters and tenths of the centimeter;

(b) meters; and

(c) multiples of meters

A 1-m measure may be graduated, in addition, to show 0.1-

m and multiples of 0.1-m subdivisions.

S.1. Units. - A linear measure may be in total length, and

the total length may be subdivided in any or all of the

following:

(a) inches and binary submultiples of the inch;

(b) feet;

(c) yards and multiples of yards.

A 1-yd measure may be graduated, in addition, to show

1/3-yd and 2/3-yd subdivisions. A flexible tape may be

graduated in tenths or hundredths of a foot, or both tenths

and hundredths of a foot. (Any other subdivisions are

allowable only on measures of special purposes and when

required for such purposes.)

S.2. Material.

S.2.1. Flexible Tape. - A flexible tape shall be made of

metal.

S.2.2. End Measure. - If an end measure is made of

material softer than brass, the ends of the measure shall

be protected by brass (or other metal at least equally

hard) securely attached.

S.3. Finish. - Measures shall be smoothly finished.

S.4. Design.

S.4.1. Rigid Measure. - A rigid measure shall be

straight.

S.4.2. Folding Measure. - A folding measure shall open

to a definite stop, and when so opened shall be straight.

S.5. Graduations.

S.5.1. General. - Graduations shall be perpendicular to

the edge of the measure.

S.5.2. Width. - The width of the graduations on any

measure shall not exceed one-half the width of the

smallest graduated interval on the measure, and in no

case shall be wider than 0.75 mm (0.03 in).

(Amended 1982)

 T. Tolerances

T.1. For Measures Except Metal Tapes. - Maintenance

tolerances in excess and in deficiency for measures except

metal tapes shall be as shown in Table 1. Acceptance

tolerances shall be one-half the maintenance tolerances.

Table 1.

Maintenance Tolerances, in Excess

and in Deficiency, for Linear Measures Except

Metal Tapes

Nominal interval

from zero Tolerance

feet inch

1/2 or less

1

2

3

4

5

6

1/64

1/32

1/16

3/32

1/8

5/32

3/16
5.52. Linear Measures



T.2. For Metal Tapes. - Maintenance and acceptance

tolerances in excess and in deficiency for metal tapes shall

be as shown in Table 2. Tapes of 10 m (25 ft) or over shall

be tested at a tension resulting from a load of 5 kg (10 lb).

Tapes less than 10 m (25 ft) shall be tested at a tension

resulting from a load of 2.5 kg (5 lb). However, flexible

metal tapes of 10 m (25 ft) or less that are not normally

used under tension shall be tested with no tension applied.

All tapes shall be supported throughout on a horizontal flat

surface whenever tested. 

(Amended 1972)

Table 2.

Maintenance and Acceptance Tolerances,

in Excess and in Deficiency, for Metal Tapes

Nominal interval

from zero

Tolerance

feet inch

6 or less

7 to 30, inclusive

31 to 55, inclusive

56 to 80, inclusive

81 to 100, inclusive

1/32

1/16

1/8

3/16

1/4
1FQOGVGTU



5GE1FQOGVGTU

##RRNKECVKQP

#6JKUEQFGCRRNKGUVQQFQOGVGTUVJCVCTGWUGFQTCTG

VQDGWUGFVQFGVGTOKPGVJGEJCTIGUHQTTGPVQTJKTGQH

RCUUGPIGTXGJKENGUCPFVTWEMUCPFDWUGU9JGPQHHKEKCN

GZCOKPCVKQPUCTGWPFGTVCMGPQPQFQOGVGTUVJCVHQTOVJG

DCUKUHQTVJGRC[OGPVQHHGGUQTVCZGUVQQTVJG

RTGRCTCVKQPQHTGRQTVUHQTIQXGTPOGPVCNCIGPEKGUCPFKP

UKOKNCTECUGUVJGTGSWKTGOGPVUQHVJKUEQFGUJCNNDG

CRRNKGFKPUQHCTCUVJG[CTGCRRNKECDNGCPFCRRTQRTKCVGVQ

VJGEQPFKVKQPUQHUWEJURGEKCNWUGU


#OGPFGF

#6JKUEQFGFQGUPQVCRRN[VQVCZKOGVGTUHQTYJKEJ

UGG5GE%QFGHQT6CZKOGVGTU


#OGPFGF

#5GGCNUQ5GE)GPGTCN%QFGTGSWKTGOGPVU

55RGEKHKECVKQPU

5&GUKIPQH+PFKECVKPI'NGOGPVU

5)GPGTCN6JGRTKOCT[KPFKECVKPIGNGOGPVQH

CPQFQOGVGTOC[DG


C VJGFKUVCPEGVTCXGNGFRQTVKQPQHVJGURGGFQOGVGT

CUUGODN[QHCOQVQTXGJKENG


D CURGEKCNECDNGFTKXGPFKUVCPEGKPFKECVKPIFGXKEG

QT


E CJWDQFQOGVGTCVVCEJGFVQVJGJWDQHCYJGGNQP

COQVQTXGJKENG


#OGPFGF

57PKVU#PQFQOGVGTUJCNNKPFKECVGKPVGTOUQH

OKNGUQTMKNQOGVGTU


#OGPFGF

5/KPKOWO+PFKECVGF8CNWG6JGXCNWGQHVJG

KPVGTXCNQHKPFKECVGFFKUVCPEGUJCNNDG


C HQTQFQOGVGTUKPFKECVKPIKPMKNQOGVGTU

MKNQOGVGTQT


D HQTQFQOGVGTUKPFKECVKPIKPOKNGUOKNG


#OGPFGF

5#FXCPEGOGPVQH+PFKECVKPI'NGOGPVU6JG

OQUVUGPUKVKXGKPFKECVKPIGNGOGPVUQHCPQFQOGVGTOC[

CFXCPEGEQPVKPWQWUN[QTKPVGTOKVVGPVN[CNNQVJGT

GNGOGPVUUJCNNCFXCPEGKPVGTOKVVGPVN['ZEGRVYJGPVJG

KPFKECVKQPUCTGDGKPITGVWTPGFVQ\GTQVJGKPFKECVKQPUQH

CPKPUVCNNGFQFQOGVGTUJCNNDGUWUEGRVKDNGVQ

CFXCPEGOGPVQPN[D[VJGTQVCVKQPQHVJGXGJKENGYJGGN

QTYJGGNU#OGPFGF

54GCFCDKNKV[&KUVCPEGHKIWTGUCPFVJGKT

DCEMITQWPFUJCNNDGQHUJCTRN[EQPVTCUVKPIEQNQTU

(KIWTGUKPFKECVKPIVGPVJWPKVUUJCNNDGFKHHGTGPVKCVGF

HTQOQVJGTHKIWTGUYKVJFKHHGTGPVEQNQTUQTYKVJC

FGEKOCNRQKPVQTD[QVJGTGSWCNN[GHHGEVKXGOGCPU

'ZEGRVFWTKPIVJGRGTKQFQHCFXCPEGQHCP[FGECFGVQ

VJGPGZVJKIJGTKPFKECVKQPQPN[QPGHKIWTGKPGCEJ

FGECFGUJCNNDGGZRQUGFVQXKGY#P[RTQVGEVKXG

EQXGTKPIKPVGPFGFVQDGVTCPURCTGPVUJCNNDGKPUWEJ

EQPFKVKQPVJCVKVECPDGOCFGVTCPURCTGPVD[QTFKPCT[

ENGCPKPIQHKVUGZRQUGFUWTHCEG


#OGPFGF

5&KIKVCN+PFKECVKQPUCPF4GRTGUGPVCVKQP

&KIKVCNKPFKECVKPIQFQOGVGTU
FKUEQPVKPWQWU

TGIKUVTCVKQPUJCNNTQWPFQHHKPFKECVKQPUVQVJGPGCTGUV

OKPKOWOFKXKUKQPQTVTWPECVGKPFKECVKQPUVQVJGNQYGT

OKPKOWOFKXKUKQP


#FFGF

00QVGU

06GUVKPI2TQEGFWTGU

06GUV/GVJQFU6QFGVGTOKPGEQORNKCPEGYKVJ

FKUVCPEGVQNGTCPEGUCFKUVCPEGVGUVQHCPQFQOGVGTUJCNN

DGEQPFWEVGFWUKPIQPGQTOQTGQHVJGHQNNQYKPIVGUV

OGVJQFU


C 4QCF6GUV#TQCFVGUVEQPUKUVUQHFTKXKPIVJG

XGJKENGQXGTCRTGEKUGN[OGCUWTGFTQCFEQWTUG


D (KHVJ9JGGN6GUV#HKHVJYJGGNVGUVEQPUKUVUQH

FTKXKPIVJGXGJKENGQXGTCP[TGCUQPCDNGTQCF

EQWTUGCPFFGVGTOKPKPIVJGFKUVCPEGCEVWCNN[
1FQOGVGTU



VTCXGNGFVJTQWIJVJGWUGQHCOGEJCPKUOMPQYPCU 08GJKENG.QCFKPI

CHKHVJYJGGNVJCVKUCVVCEJGFVQVJGXGJKENGCPF

VJCVKPFGRGPFGPVN[OGCUWTGUCPFKPFKECVGUVJG 
C 2CUUGPIGT.QCF&WTKPIVJGFKUVCPEGVGUVQH

FKUVCPEG CPQFQOGVGTVJGXGJKENGOC[ECTT[VYQ


E 5KOWNCVGF4QCF6GUV#UKOWNCVGFTQCFVGUV

EQPUKUVUQHFGVGTOKPKPIVJGFKUVCPEGVTCXGNGFD[ 
D 6TWEM%CTIQ.QCF6TWEMQFQOGVGTUUJCNN

WUGQHCTQNNGTFGXKEGQTD[EQORWVCVKQPHTQO DGVGUVGFD[QPGQHVJGHQNNQYKPIOGVJQFU

TQNNKPIEKTEWOHGTGPEGCPFYJGGNVWTPFCVC


#OGPFGF 
 VJGVTWEMKUNQCFGFYKVJQPGJCNHQHVJG

06GUV4WPU0QVNGUUVJCPVYQVGUVTWPUUJCNN

DGEQPFWEVGF#EEGNGTCVKQPCPFFGEGNGTCVKQPUJCNNDG 
 WPNQCFGFKHWPNQCFGFVGUVVQNGTCPEGUCTG

ECTGHWNN[EQPVTQNNGFVQCXQKFURKPPKPIQTUMKFFKPIVJG CRRNKGF

YJGGNU
#OGPFGF


#OGPFGF

0(QT&GXKEGU+PFKECVKPIKP/KNGU6JG

VGUVTWPUUJCNNDGOKNGUKPNGPIVJUJCNNUVCTVHTQO

CPFHKPKUJCVCFGCFUVQRYKVJCOKPKOWOQH

RGTEGPVQHVJGTWPDGVYGGPOKNGURGTJQWTCPF

OKNGURGTJQWT


#FFGF

0(QT&GXKEGU+PFKECVKPIKP-KNQOGVGTU

6JGVGUVTWPUUJCNNDGMKNQOGVGTUKPNGPIVJUJCNN

UVCTVHTQOCPFHKPKUJCVCFGCFUVQRYKVJC

OKPKOWOQHRGTEGPVQHVJGTWPDGVYGGPMKNQ

OGVGTURGTJQWTCPFMKNQOGVGTURGTJQWT#FFGF



06GUV%QPFKVKQPU

06KTG5VCDKNK\CVKQP4QCFVGUVUQT

HKHVJYJGGNVGUVUUJCNNDGRTGEGFGFD[CTWPQHCVNGCUV

MKNQOGVGTUQTOKNGUHQTVJGRWTRQUGQH

UVCDKNK\KPIVKTGRTGUUWTGU5KOWNCVGFTQCFVGUVUQPC

TQNNGTFGXKEGUJCNNDGOCFGCVUVCDNGVKTGRTGUUWTGU


#OGPFGF

06KTG2TGUUWTG#VVJGEQORNGVKQPQHVJG

VGUVTWPQTTWPUVJGVKTGUQHVJGXGJKENGWPFGTVGUV

UJCNNDGEJGEMGFVQFGVGTOKPGVJCVVJGVKTGRTGUUWTGKU

VJCVQRGTCVKPIVKTGRTGUUWTGRQUVGFKPVJGXGJKENG+H

PQVVJGVKTGRTGUUWTGUJQWNFDGCFLWUVGFVQVJGRQUVGF

VKTGRTGUUWTGCPFHWTVJGTVGUVUOC[DGEQPFWEVGFVQ

FGVGTOKPGVJGQRGTCVKPIEJCTCEVGTKUVKEUQHVJG

QFQOGVGT


#OGPFGF

RGTUQPU

OCZKOWOECTIQNQCFQT

66QNGTCPEGU

66Q7PFGTTGIKUVTCVKQPCPFVQ1XGTTGIKUVTCVKQP

6JGVQNGTCPEGUJGTGKPCHVGTRTGUETKDGFUJCNNDGCRRNKGFVQ

GTTQTUQHWPFGTTGIKUVTCVKQPCPFGTTQTUQHQXGTTGIKUVTCVKQP

66QNGTCPEG8CNWGU'ZEGRVHQTWPNQCFGFVTWEMU

OCKPVGPCPEGCPFCEEGRVCPEGVQNGTCPEGUQPQFQOGVGTUUJCNN

DGRGTEGPVQHVJGKPVGTXCNWPFGTVGUV


#OGPFGF

66QNGTCPEGUHQT7PNQCFGF6TWEMU

/CKPVGPCPEGCPFCEEGRVCPEGVQNGTCPEGUQPVTWEM

QFQOGVGTUUJCNNDGRGTEGPVHQTWPFGTTGIKUVTCVKQPCPF

RGTEGPVHQTQXGTTGIKUVTCVKQPQHVJGKPVGTXCNWPFGTVGUV


#FFGF

747UGT4GSWKTGOGPVU

74+PHNCVKQPQH8GJKENG6KTGU6JGQRGTCVKQPCNVKTG

RTGUUWTGQHRCUUGPIGTXGJKENGCPFVTWEMVKTGUUJCNNDG

RQUVGFKPVJGXGJKENGCPFVKTGUUJCNNDGOCKPVCKPGFCVVJG

RQUVGFRTGUUWTG


#OGPFGF
5.54. Taximeters



 Sec. 5.54. Taximeters

A. Application

A.1. This code applies to taximeters; that is, to devices that

automatically calculate at a predetermined rate or rates and

indicate the charge for hire of a vehicle.

A.2. This code does not apply to odometers on vehicles that are

rented on a distance basis (for which see Sec. 5.53 Code for

Odometers). (Amended 1977)

A.3. See also Sec. 1.10; General Code requirements.

 S. Specifications

S.1. Design of Indicating and Recording Elements.

S.1.1. General. - A taximeter shall be equipped with a

primary indicating element and may be equipped with a

recording element.

(Amended 1988)

S.1.2. Advancement of Indicating Elements. - Except when

a taximeter is being cleared, the primary indicating and

recording elements shall be susceptible of advancement only

by the movement of the vehicle or by the time mechanism.

(Amended 1988)

S.1.3. Visibility of Indications. - The indications of fare,

including extras, and the mode of operation, such as "time" or

"hired," shall be constantly displayed whenever the meter is in

operation. All indications of passenger interest shall be easily

read from a distance of 1.2 m (4 ft) under any condition of

normal operation.

(Amended 1986, 1977, and 1988)

S.1.3.1. Minimum Height of Figures, Words, and Symbols. - The minimum height of the figures used to indicate

the fare shall be 10 mm and for extras, 8 mm. The minimum height of the figures, words, or symbols used for other

indications, including those used to identify or define, shall

be 3.5 mm.

(Added 1986)

S.1.3.2. Lighting of Indications. Integral lighting shall be

provided to illuminate the fare, extras, the rate or rate code,

and the taximeter status (i.e., vacant, hired, and time off).

[Nonretroactive as of January 1, 1989.]

(Added 1988)(Amended 1990)

S.1.4. Actuation of Fare-Indicating Mechanism. -

When a taximeter designed to calculate fares upon the

basis of a combination of distance traveled and time

elapsed is operative with respect to fare indication, the

fare-indicating mechanism shall be actuated by the

distance mechanism whenever the vehicle is in motion at

such a speed that the rate of distance revenue equals or

exceeds the time rate, and may be actuated by the time

mechanism whenever the vehicle speed is less than this

and when the vehicle is not in motion. Means shall be

provided for the vehicle operator to render the time

mechanism either operative or inoperative with respect to

the fare-indicating mechanism.

(Amended 1977)

S.1.5. Operating Condition.

S.1.5.1. General. - When a taximeter is cleared, the

indication "Not Registering," "Vacant," or an equivalent

expression shall be shown. Whenever a taximeter is set

to register charges, it shall indicate "Registering,"

"Hired," or an equivalent expression and the rate at

which it is set shall be automatically indicated (Rate 1

or Rate A, for example).

(Amended 1988)

S.1.5.2. Time not Recording. - When a taximeter is set

for fare registration with the time mechanism

inoperative, it shall indicate "Time Not Recording" or

an equivalent expression.

(Amended 1988)

S.1.6. Fare Identification. - Fare indications shall be

identified by the word "Fare" or by an equivalent

expression. Values shall be defined by suitable words or

monetary signs.

S.1.7. Extras. - Extras shall be indicated as a separate

item and shall not be included in the fare indication. They

shall be identified by the word "Extras" or by an equivalent

expression. Values shall be defined by suitable words or

monetary signs. Means may be provided to totalize the

fare and extras if the totalized amount returns to separate

indications of fare and extras within 5 seconds or less.

(Amended 1988)

S.1.7.1. Nonuse of Extras. - If and when taximeter

extras are prohibited by legal authority or are
5.54. Taximeters



discontinued by a vehicle operator, the extras mechanisms displayed for ten seconds, or less. If the information

shall be rendered inoperable or the extras indications shall consists of a list of information, the list may be displayed

be effectively obscured by permanent means. one item after another, provided that each item is

displayed for ten seconds, or less.

S.1.8. Protection of Indications. - Indications of fare and

extras shall be displayed through and entirely protected by

glass or other suitable transparent material securely attached to

the housing of the taximeter. S.2. Basis of Fare Calculations. - A taximeter shall

S.1.9. Recorded Representation. - A printed receipt issued

from a taximeter, whether through an integral or separate

recording element, shall include the following:

(a) date

(b) unique vehicle identification number, such as the

medallion number, taxi number, vehicle identification

number (VIN) or permit number*

(c) start and end time of trip*

(d) distance traveled, maximum increment of 0.1 kilometer

(0.1 mile)*

(e) fare in $

(f) for multi-rate taximeters, each rate at which fare was

computed and the associated fare at that rate*

(g) additional charges where permitted such as extras,

surcharge, telephone use, tip and tax shall be identified

and itemized*

(h) total fare in $ (total charge)*

[Nonretroactive as of January 1, 1989.] *[Nonretroactive as

of January 1, 2000 ]

(Added 1988)(Amended 1999)

S.1.9.1. Multiple Recorded Representations.

S.1.9.1.1. Duplicate Receipts. - A recording element may

produce a duplicate receipt for the previous transaction

provided the information printed is identical to the original

with the exception of time issued. The duplicate receipt

shall include the words "duplicate" or "copy." The feature

to print a duplicate receipt shall be deactivated at the time

the meter is hired for the next fare.

[Nonretroactive as of January 1, 2000]

(Added 1999)

S.1.10. Non-fare Information. - The fare and extras displays

may be used to display auxiliary information provided the

meter is in the vacant condition and such information is only

[Nonretroactive as of January 1, 2002]

(Added 2000)

calculate fares only upon the basis of:

(a) distance traveled,

(b) time elapsed, or

(c) a combination of distance traveled and time elapsed.

(Amended 1977)

S.2.1. Initial Time and Distance Intervals. - The time

and distance intervals of a taximeter shall be directly

proportional as expressed in the following formula:

Seconds of Initial Time Interval =

Seconds per Non-Initial Time Interval

Distance of Initial Mileage Interval 

Distance per Non-Initial Mileage Interval

(Added 1990)

S.3. Design of Operating Control.

S.3.1. Positions of Control. The several positions of the

operating controls shall be clearly defined and shall be so

constructed that accidental or inadvertent changing of the

operating condition of the taximeter is improbable.

Movement of the operating controls to an operating

position immediately following movement to the cleared

position shall be delayed enough to permit the taximeter to

come to a complete rest in the cleared position.

(Amended 1988)

S.3.2. Flag. - If the control for the operating condition is

a lever-arm and flag, the flag shall be at its highest position

when the taximeter is cleared, and in this position the

whole of the flag shall be above the level of the taximeter

housing.

S.3.3. Control for Extras Mechanism. - The knob,

handle, or other means provided to actuate the extras

mechanism shall be inoperable whenever the taximeter is

cleared.

S.4. Interference. - The design of a taximeter shall be such

that there will be no interference between the time and the
5.54. Taximeters



distance portions of the mechanism device at any speed of (a) The meter shall either filter out the inconsistent distance

operation. input signals or cease to increment fare based on

(Amended 1977 and 1988) distance until the distance input signal returns to

S.5. Provision for Security Seals. - Adequate provision shall be

made to provide security for a taximeter. Security may be

provided either by:

(a) Affixing security seals to the taximeter and to all other

components required for service operation of a complete

installation on a vehicle, so that no adjustments, alterations, or

replacements affecting accuracy or indications of the device or

the assembly can be made without mutilating the seal or seals; or

(b) Using a combination of security seals described in paragraph

(a) and, in the case of a component that may be removed from a

vehicle (e.g., slide mounting the taximeter), providing a physical

or electronic link between components affecting accuracy or

indications of the device to ensure that its performance is not

affected and operation is permitted only with those components

having the same unique properties. 

The sealing means shall be such that it is not necessary to

disassemble or remove any part of the device or of the vehicle to

apply or inspect the seals. 

(Amended 1988 and 2000)

S.6. Power Interruption, Electronic Taximeters.

(a) After a power interruption of 3 seconds or less, the fare and

extras indications shall return to the previously displayed

indications and may be susceptible to advancement without

the taximeter being cleared.

(b) After a power interruption exceeding 3 seconds, the fare

and extras indications shall return to the previously

displayed indications and shall not be susceptible to

advancement until the taximeter is cleared.

After restoration of power following an interruption

exceeding 3 seconds, the previously displayed fare shall be

displayed for a maximum of 1 minute at which time the fare

shall automatically clear and the taximeter shall return to

the vacant condition.*

*[Nonretroactive as of January 1, 2002]

(Added 1988) (Amended 1989, 1990, and 2000)

S.7. Anti-fraud Provisions, Electronic Taximeters. - An

electronic taximeter may have provisions to detect and eliminate

distance input that is inconsistent with output of the vehicle’s

distance sensor. When a taximeter equipped with this feature

detects input inconsistent with the distance sensor:

normal. If the meter ceases to increment fare based on

distance, the taximeter may continue to increment fare

based on elapsed time;

(b) The taximeter shall provide a visible or audible signal

that inconsistent input signals are being detected; and

(c) The taximeter shall record the occurrence in an event

logger. The event logger shall include an event counter

(000 to 999), the date, and the time of at least the last

1000 occurrences.

(Added 2001)

N. Notes

N.1. Distance Tests.

N.1.1. Test Methods. - To determine compliance with

distance tolerances, a distance test of a taximeter shall be

conducted utilizing one or more of the following test

methods:

(a) Road Test. - A road test consists of driving the

vehicle over a precisely measured road course.

(b) Fifth-Wheel Test. - A fifth-wheel test consists of

driving the vehicle over any reasonable road course

and determining the distance actually traveled

through the use of a mechanism known as a "fifth

wheel" that is attached to the vehicle and that

independently measures and indicates the distance.

(c) Simulated-Road Test. - A simulated road test

consists of determining the distance traveled by use

of a roller device, or by computation from rolling

circumference and wheel-turn data.

(Amended 1977)

N.1.2. Test Procedures. - The distance test of a taximeter,

whether a road test, a simulated-road test, or a fifth-wheel

test, shall include at least duplicate runs of sufficient

length to cover at least the third money drop or 1 mile,

whichever is greater, and shall be at a speed approximating

the average speed traveled by the vehicle in normal

service. In the case of metric-calibrated taximeters, the test

should cover at least the third money drop or 2 km,

whichever is greater. (Amended 1977)
5.54. Taximeters



N.1.3. Test Conditions.

N.1.3.1. Vehicle Lading. - During the distance test of a

taximeter, the vehicle shall carry two persons, or in the case

of a simulated-road test, 70 kg or 150 lb of test weights may

be substituted in lieu of the second person.

N.1.3.2. Tire Pressure. - At the completion of test run or

runs, the tires of the vehicle under test shall be checked to

determine that the tire pressure is that operating tire pressure

posted in the vehicle. If not, the tire pressure should be

adjusted to the posted tire pressure and further tests may be

conducted to determine the operating characteristics of the

odometer.

(Amended 1977)

N.2. Time Test. - If a taximeter is equipped with a timing device

through which charges are made for time intervals, the timer shall

be tested at the initial interval, four separate subsequent intervals,

and an average time test of at least four consecutive subsequent

time intervals.

(Amended 1988)

N.3. Interference Test. - If a taximeter is equipped with a timing

device through which charges are made for time intervals, a test

shall be conducted to determine whether there is interference

between the time and distance elements. During the interference

test, the vehicle’s operating speed shall be 3 km/h or 4 km/h, or

2 mi/h or 3 mi/h faster than the speed at which the basic distance

rate equals the basic time rate. The basic rate per hour divided by

the basic rate per mile is the speed (km/h or mi/h) at which the

basic time rate and basic distance rate are equal.

(Amended 1988)

 T. Tolerances

T.1. Tolerance Values.

T.1.1. On Distance Tests. - Maintenance and acceptance

tolerances for taximeters shall be as follows:

(a) On Overregistration: 1 percent of the interval under test. which a taximeter is set, including the initial distance interval

(b) On Underregistration: 4 percent of the interval under schedule of extras when an extras indication is provided, shall

test, with an added tolerance of 30 m or 100 ft whenever be conspicuously displayed inside the front and rear pasthe initial interval is included in the interval under test. senger compartments. The words "Rate," "Rates," or "Rates

of Fare" shall precede the rate statement. The rate statement

T.1.2. On Time Tests.

T.1.2.1. On Individual Time Intervals. - Maintenance

and acceptance tolerances on individual time intervals shall

be as follows:

(a) On Overregistration: 3 seconds per minute

(5 percent).

(b) On Underregistration: 9 seconds per minute

(15 percent) on the initial interval, and 6 seconds

per minute (10 percent) on subsequent intervals.

T.1.2.2. On Average Time Interval Computed After

the Initial Interval. - Except for the initial interval,

maintenance and acceptance tolerances on the average time

interval shall be as follows:

(a) On Overregistration: 0.2 second per minute

(0.33 percent).

(b) On Underregistration: 3 seconds per minute

(5 percent).

(Amended 1991)

T.1.3. On Interference Tests.

T.1.3.1. The registration of a taximeter in the "time on"

position shall agree within one percent of its

performance in the "time off" position.

(Added 1988)

UR. User Requirements

UR.1. Inflation of Vehicle Tires. - The operational tire

pressure of passenger vehicles and truck tires shall be posted

in the vehicle and shall be maintained at the posted pressure.

(Amended 1977)

UR.2. Position and Illumination of Taximeter. - A

taximeter shall be so positioned and illuminated that its

indications, operational markings, and controls of passenger

interest can be conveniently read by a passenger seated in the

back seat of the vehicle.

(Amended 1985 and 1986)

UR.3. Statement of Rates. - The distance and time rates for

and the initial time interval, the local tax rate, and the

shall be fully informative, self-explanatory, and readily

understandable by the ordinary passenger, and shall either be

of a permanent character or be protected by glass or other

suitable transparent material.

(Amended 1977, 1988, 1990, and 1999)
5.55. Timing Devices



Sec. 5.55. Timing Devices

A. Application

A.1. - This code applies to devices used to measure time

during which services are being dispensed (such as vehicle

parking, laundry drying, and car washing).

A.2. - See also Sec. 1.10; General Code requirements.

 S. Specifications

S.1. Design of Indicating and Recording Elements and

of Recorded Representations.

S.1.1. Primary Elements.

S.1.1.1. General. - A timing device shall be equipped

with a primary indicating element, and may also be

equipped with a primary recording element. A readily

observable in-service light or other equally effective

means that automatically indicates when laundry

driers, vacuum cleaners, and car washes are in

operation shall be deemed an appropriate primary

indicating element.

(Amended 1979)

S.1.1.2. Units. - A timing device shall indicate and

record, if the device is equipped to record, the time in

terms of minutes for time intervals of 60 minutes or

less and in hours and minutes for time intervals

greater than 60 minutes.

S.1.1.3. Value of Smallest Unit. - The value of the

smallest unit of indicated time and recorded time, if

the device is equipped to record, shall not exceed the

equivalent of:

(a) one-half hour on parking meters indicating time

in excess of 2 hours;

(b) six minutes on parking meters indicating time in

excess of 1 but not greater than 2 hours; or

(c) five minutes on all other devices, except those

equipped with an in-service light.

(Amended 1975)

S.1.1.4. Advancement of Indicating and Recording

Elements. - Primary indicating and recording

elements shall be susceptible to advancement only

during the mechanical operation of the device, except

that clocks may be equipped to manually reset the

time.

S.1.1.5. Operation of In-Service Indicator Light. -

The in-service light indicator shall be operative only

during the time the device is in operation.

S.1.1.6. Discontinuous Indicating Parking Meters.

- An indication of the time purchased shall be

provided at the time the meter is activated in units of

no more than 1 minute for times less than 1 hour and

not more than 2 minutes for times of 1 hour or more.

Convenient means shall be provided to indicate to the

purchaser the unexpired time. 

(Added 1975 and amended 1976)

S.1.2. Graduations.

S.1.2.1. Length. - Graduations shall be so varied in

length that they may be conveniently read.

S.1.2.2. Width. - In any series of graduations, the

width of a graduation shall in no case be greater than

the width of the minimum clear interval between

graduations and the width of main graduations shall

be not more than 50 percent greater than the width of

subordinate graduations. Graduations shall in no case

be less than 0.2 mm (0.008 in) in width.

S.1.2.3. Clear Interval Between Graduations. - The

clear interval shall be not less than 0.75 mm

(0.03 in). If the graduations are not parallel, the measurement shall be made:

(a) along the line of relative movement between the

graduations at the end of the indicator, or

(b) if the indicator is continuous, at the point of

widest separation of the graduations.
5.55. Timing Devices



Table N.2.* Broadcast Times and Frequencies

Station Location,

Latitude,

Longitude

Frequency

(MHz)

Times of 

Transmission

(UTC)

WWV Fort Collins,

Colorado

40E41’ N

105E02’ W

2.5

5.0

10.0

15.0

Continuous

WWVH Kauai, Hawaii

21E59’ N

159E46’ W

2.5

5.0

15.0

Continuous

CHU Ottawa, Canada

45E18’ N

75E45’ W

3.330

7.335

14.670

Continuous

*From NIST Special Publication 559, "Time and Frequency Users’ Manual," 1979.

(Added 1988)

S.1.3. Indicators.

S.1.3.1. Symmetry. - The index of an indicator shall

be symmetrical with respect to the graduations, at least

throughout that portion of its length associated with

the graduations.

S.1.3.2. Length. - The index of an indicator shall

reach to the finest graduations with which it is used,

unless the indicator and the graduations are in the

same plane, in which case the distance between the

end of the indicator and the ends of the graduations,

measured along the line of the graduations, shall be

not more than 1.0 mm (0.04 in).

S.1.3.3. Width. - The width of the index of an

indicator in relation to the series of graduations with

which it is used shall be not greater than

(a) the width of the widest graduation, and

(b) the width of the minimum clear interval between

the graduations.

S.1.3.4. Parallax. - Parallax effect shall be reduced to

a practicable minimum.

S.1.4. Printed Tickets. - A printed ticket issued or

stamped by a timing device shall have printed clearly

thereon:

(a) the time and day when the service ends and the

time and day when the service begins, except that a

self-service money-operated device that clearly

displays the time of day need not record the time

and day when the service begins; or

(b) the time interval purchased, and the time and day

that the service either begins or ends.

(Amended 1983)

S.2. Marking Requirements, Operating Instructions. -

Operating instructions shall be clearly stated on the device.

 N. Notes

N.1. Test Method. - A timing device shall be tested with

a timepiece with an error of not greater than plus or minus

15 seconds per 24-hour period. In the test of timing devices

with a nominal capacity of 1 hour or less, stopwatches with

a minimum division of not greater than one-fifth second

shall be used. In the test of timing devices with a nominal

capacity of more than 1 hour, the value of the minimum

division on the timepiece shall be not greater than 1 second.

Time pieces and stopwatches shall be calibrated with

standard time signals as described in National Institute of

Standards and Technology Special Publication 432, NIST

Time and Frequency Dissemination Services, or any

superseding publication.

(Amended 1978)

N.2. Broadcast Times and Frequencies. - Time and

frequency standards are broadcast by the stations listed in

Table N.2.
5.55. Timing Devices



Table T.1.3.

Maintenance and Acceptance Tolerances

for Parking Meters

Maintenance and Acceptance Tolerances

Nominal time capacity On overregistration On underregistration

30 minutes or less

Over 30 minutes to and

 including 1 hour

Over 1 hour

No tolerance

No tolerance

No tolerance

10 seconds per minute, but not less

than 2 minutes

5 minutes plus 4 seconds per 

minute over 30 minutes

7 minutes plus 2 minutes per hour

over 1 hour

T. Tolerances UR. User Requirements

T.1. Tolerance Values. - Maintenance and acceptance UR.1. Statement of Rates. - The price in terms of money

tolerances for timing devices shall be as follows: per unit or units of time for the service dispensed and the

T.1.1. For Timing Devices Other Than Those

Specified in T.1.2. and T.1.3. - The maintenance and

acceptance tolerances shall be:

(a) On Overregistration: 5 seconds for any time UR.2. Time Representations. - Any time representation

interval of 1 minute or more; and shall be within plus or minus 2 minutes of the correct time

(b) On Underregistration: 6 seconds per indicated for "time out"; in addition, the time indication of the

minute. "time-out" clock shall be the same as or less than that of the

(Amended 1975 and 1986) "time-in" clock.

T.1.2. For Time Clocks and Time Recorders. - The

maintenance and acceptance tolerances on

overregistration and under registration shall be 3 seconds

per hour, but not to exceed 1 minute per day. 

 (Amended 1975) 

T.1.3. On Parking Meters. - The maintenance and

acceptance tolerances are shown in Table T.1.3.

T.2. Tests Involving Digital Indications or

Representations. - To the tolerances that would otherwise

be applied, there shall be added an amount equal to one-half

the minimum value that can be indicated or recorded.

number of coins the device will accept and be activated by

at one time, shall be clearly, prominently, and

conspicuously displayed.

(Amended 1976)

in effect in the area, except on an individual clock used only

(Amended 1975)
5.56.(a) Grain Moisture Meters



Sec. 5.56.(a) Grain Moisture Meters

Section 5.56. has been reorganized into two sections. This Section, 5.56.(a), applies to all NTEP grain moisture

meters. It also applies to any grain moisture meter manufactured or placed into service after January 1, 1998. (Code

reorganized and renumbered 1996.) 

A. Application

A.1. This code applies to grain moisture meters; that is,

devices used to indicate directly the moisture content of

cereal grain and oil seeds. The code consists of general

requirements applicable to all moisture meters and specific

requirements applicable only to certain types of moisture

meters.

A.2. This code does not apply to devices used for

in-motion measurement of grain moisture content or seed

moisture content.

A.3. Type Evaluation - The National Type Evaluation

Program will accept for type evaluation only those devices

that comply with this code. State enforcement will be based

upon the effective dates identified with each requirement

when specific dates are shown.

(Added 1993)

A.4. See also Sec. 1.10; General Code requirements.

 S. Specifications

S.1. Design of Indicating, Recording, and Measuring

Elements.

S.1.1. Digital Indications and Recording Elements.

(a) Meters shall be equipped with a digital indicating

element.

(b) The minimum height for the digits used to display

moisture content shall be 10 mm.

(c) Meters shall be equipped with a communication

interface that permits interfacing with a recording

element and transmitting the date, grain type, grain

moisture results, and calibration version

identification.

(d) A digital indicating element shall not display, and

a recording element shall not record, any moisture

content values before the end of the measurement

cycle.

(e) Moisture content results shall be displayed and

recorded as percent moisture content, wet basis.

Subdivisions of this unit shall be in terms of

decimal subdivisions (not fractions).

(f) A meter shall not display or record any moisture

content values when the moisture content of the

grain sample is beyond the operating range of the

device, unless the moisture representation includes

a clear error indication (and recorded error message

with the recorded representation).

(g) On multi-constituent meters (e.g., meters which

also measure grain protein), provision shall be

made for displaying and recording the constituent

label (such as moisture, protein, etc.) to make it

clear which constituent is associated with each of

the displayed and recorded values.

(Added 1995)

(Added 1993)(Amended 1994 and 1995)

S.1.2. Grain or Seed Kind and Class Selection and

Recording. - Provision shall be made for selecting and

recording the kind and class (as appropriate) of grain or

seed to be measured. The means to select the kind and

class of grain or seed shall be readily visible and the kind

and class of grain or seed selected shall be clearly and

definitely identified. Abbreviations for grain types

indicated on the meter must meet the minimum

acceptable abbreviations listed in Table S.1.2. Meters

shall have the capability (i.e., display capacity) of

indicating the grain type using a minimum of four

characters in order to accommodate the four-character

abbreviations listed in Table S.1.2. 

(Amended 1993 and 1995) 

S.1.3. Operating Range. - A meter shall automatically

and clearly indicate when the operating range of the

meter has been exceeded. The operating range shall

specify the following:

(a) Temperature Range of the Meter. The

temperature range over which the meter may be used and

still comply with the applicable requirements shall be

specified. The minimum temperature range shall be

10 EC to 30 EC. No moisture value may be displayed

when the temperature range is exceeded. An appropriate
5.56.(a) Grain Moisture Meters



6CDNG5)TCKP6[RGU%QPUKFGTGFHQT6[RG'XCNWCVKQPCPF%CNKDTCVKQP

CPF/KPKOWO#EEGRVCDNG#DDTGXKCVKQPU

)TCKP6[RG /KPKOWO

#EEGRVCDNG

#DDTGXKCVKQP

)TCKP6[RG /KPKOWO#EEGRVCDNG

#DDTGXKCVKQP

%QTP %140 5Q[DGCPU 51;$

&WTWO9JGCV

5QHV9JKVG9JGCV

*CTF4GF5RTKPI9JGCV

*CTF4GF9KPVGT9JGCV

5QHV4GF9KPVGT9JGCV

*CTF9JKVG9JGCV

&749

599

*459

*499

5499

*&99

6YQTQYGF$CTNG[

5KZTQYGF$CTNG[

1CVU

64$

54$

1#65

5WPHNQYGTUGGF1KN 570( .QPI)TCKP4QWIJ4KEG

/GFKWO)TCKP4QWIJ4KEG

.)44

/)44

)TCKP5QTIJWO 514)QT

/+.1

5OCNNQKNUGGFUWPFGT

EQPUKFGTCVKQP


6CDNG#FFGF
#OGPFGF=0QPTGVTQCEVKXGCUQH,CPWCT[?

error message shall be displayed when the temperature of the sample exceeds the specified difference.

the meter is outside its specified operating range. (Added 1993) (Amended 1995)

(b) Temperature Range of each Grain or Seed. The S.1.4. Value of Smallest Unit. - The display shall

temperature range for each grain or seed for which permit constituent value determination to both 0.01

the meter is to be used shall be specified. The percent and 0.1 percent resolution. The 0.1 percent

minimum temperature range for each grain shall be resolution is for commercial transactions; the 0.01

0 EC to 40 EC. No moisture value may be percent resolution is for type evaluation and calibration

displayed when the temperature range is exceeded. purposes only, not for commercial purposes.

An appropriate error message shall be displayed

when the temperature of the grain sample exceeds

the specified temperature range for the grain.

(c) Moisture Range of the Grain or Seed. The

moisture range for each grain or seed for which the

meter is to be used shall be specified. A moisture

value may be displayed when the moisture range is

exceeded if accompanied by a clear indication that

the moisture range has been exceeded.

(d) Maximum Allowable Meter/Grain Temperature

Difference. The maximum allowable difference in

temperature between the meter and the sample for

which an accurate moisture determination can be

made shall be specified. The minimum temperature

difference shall be 10 EC. No moisture value may

be displayed when the maximum allowable

temperature difference is exceeded. An appropriate

error message shall be displayed when the

difference in temperature between the meter and

S.1.5. Operating Temperature. -

(a) Warm up period: When a meter is turned on it

shall not display or record any usable values until

the operating temperature necessary for accurate

determination has been attained, or the meter shall

bear a conspicuous statement adjacent to the

indication stating that the meter shall be turned on

for a time period specified by the manufacturer

prior to use.

(b) A meter shall meet the requirements of T.2. -

Tolerance Values when operated in the temperature

range of 10 EC to 30 EC (50 EF to 86 EF) or within

the range specified by the meter manufacturer.

(c) If the manufacturer specifies a temperature range,

the range shall be at least 20 EC (36 EF).

(Added 1993) (Amended 1995 and 1996)
5.56.(a) Grain Moisture Meters



S.2. Design of Grain Moisture Meters

S.2.1. Minimum Sample Size. - Meters shall be

designed to measure the moisture content of

representative-size grain samples. The minimum S.2.4.2. Calibration Corruption. - If calibration

allowable sample size used in analysis shall be 100 g or constants are digitally stored in an electronically

400 kernels or seeds, whichever is smaller. alterable form, the meter shall be designed to make

(Added 1993) (Amended 1995) automatic checks to detect corruption of calibration

constants. An error message must be displayed if

S.2.2. Electric Power Supply.

S.2.2.1. Power Supply, Voltage and Frequency.

(a) A meter that operates using alternating current

must perform within the tolerances defined in

Section T.2. - Tolerance Values over the line

voltage range 100 V to 130 V, or 200 V to

250 V rms as designed, and over the frequency

range of 59.5 Hz to 60.5 Hz.

(b) Battery-operated instruments shall not indicate

or record values outside the applicable tolerance

limits when battery power output is excessive or

deficient.

S.2.2.2. Power Interruption. - A power interruption

shall not cause an indicating or recording element to

display or record any values outside the applicable

tolerance limits.

(Added 1988)

S.2.3 Level Indicating Means. - A meter shall be

equipped with a level indicator and leveling adjustments

if its performance is changed by an amount greater than

the applicable tolerance when the meter is moved from a

level position to a position that is out of level in any

upright direction by up to 5 percent (approximately 3

degrees). The level-indicating means shall be readable

without removing any meter parts requiring a tool. 

(Added 1988) (Amended 1994)

S.2.4. Calibration Integrity

S.2.4.1. Calibration Version. - A meter must be

capable of displaying either calibration constants, a

unique calibration name, or a unique calibration

version number for use in verifying that the latest 

version of the calibration is being used to make

moisture content determinations.

(Added 1993)(Amended 1995)

calibration constants have been electronically altered.

(Added 1993) (Amended 1995)

S.2.4.3. Calibration Transfer. - The instrument

hardware/software design and calibration procedures

shall permit calibration development and the transfer

of calibrations between instruments of like models

without requiring user slope or bias adjustments.

.

[Note: Only the manufacturer or the manufacturer’s

designated service agency may make standardization

adjustments on moisture meters This does not

preclude the possibility of the operator installing

manufacturer-specified calibration constants under

the instructions of the manufacturer or its designated

service agency.] Standardization adjustments (not to

be confused with grain calibrations) are those

physical adjustments or software parameters which

make meters of like type respond identically to the

grain(s) being measured.

 [Nonretroactive as of January 1, 1999.]

(Added 1994) (Amended 1998)

S.2.5. Provision for Sealing. - Provision shall be made

for applying a security seal in a manner that requires the

security seal to be broken, or for using other approved

means of providing security (e.g., audit trail available at

the time of inspection as defined in Table S.2.5.*) before

any change that affects the metrological integrity of the

device can be made to any mechanism.

[*Note: Table S.2.5. is Nonretroactive as of Jan, 1,

1999.]

[Note: Zero-setting and test point adjustments are

considered to affect metrological characteristics and must

be sealed.]

(Added 1993) (Amended 1995 and 1997)

S.2.6. Determination of Quantity and Temperature. -

The moisture meter system shall not require the operator

to judge the precise volume or weight and temperature

needed to make an accurate moisture determination.

External grinding, weighing, and temperature

measurement operations are not permitted.

(Added 1994)(Amended 1995)
5.56.(a) Grain Moisture Meters



S.3. Accessory Equipment - When the operating

instructions for a moisture meter require accessory PQVCFFGF

equipment separate from and external to the moisture meter,

such equipment shall be appropriate and complete for the

measurement.

S.4. Operating Instructions and Use Limitations. - The

manufacturer shall furnish operating instructions for the

device and accessories that include complete information

concerning the accuracy, sensitivity, and use of accessory

equipment necessary in obtaining a moisture content

Operating instructions shall include the following

information:


C PCOGCPFCFFTGUUQTVTCFGOCTMQHVJGOCPWHCEVWTGT


D VJGV[RGQTFGUKIPQHVJGFGXKEGYKVJYJKEJKVKU

KPVGPFGFVQDGWUGF


E FCVGQHKUUWG


F VJGMKPFQTENCUUGUQHITCKPQTUGGFHQTYJKEJVJG

FGXKEGKUFGUKIPGFVQOGCUWTGOQKUVWTGEQPVGPV


G VJGNKOKVCVKQPUQHWUGKPENWFKPIDWVPQVEQPHKPGFVQ

VJGOQKUVWTGOGCUWTGOGPVTCPIGITCKPQTUGGF

VGORGTCVWTGOCZKOWOCNNQYCDNGVGORGTCVWTG

FKHHGTGPEGDGVYGGPITCKPUCORNGCPFOGVGTMKPFQT

ENCUUQHITCKPQTUGGFOQKUVWTGOGVGTVGORGTCVWTG

XQNVCIGCPFHTGSWGPE[TCPIGUGNGEVTQOCIPGVKE

KPVGTHGTGPEGUCPFPGEGUUCT[CEEGUUQT[GSWKROGPV


#FFGF

00QVGU

06GUVKPI2TQEGFWTGU(KGNFGXCNWCVKQPQHITCKP

OQKUVWTGOGVGTUUJCNNDGRGTHQTOGFD[QPGQHVJG

HQNNQYKPIOGVJQFU

0#KT1XGP4GHGTGPEG/GVJQF6TCPUHGT

5VCPFCTFU1HHKEKCNITCKPUCORNGUUJCNNDGWUGFCUVJG

QHHKEKCNVTCPUHGTUVCPFCTFUYKVJOQKUVWTGEQPVGPVXCNWGU

CUUKIPGFD[VJGTGHGTGPEGOGVJQFU6JGTGHGTGPEG

OGVJQFUUJCNNDGVJGQXGPFT[KPIOGVJQFUCUURGEKHKGF

D[VJG75&#)+25#6QNGTCPEGUUJCNNDGCRRNKGFVQ

VJGCXGTCIGQHCVNGCUVVJTGGOGCUWTGOGPVUQPGCEJ

QHHKEKCNITCKPUCORNG1HHKEKCNITCKPUCORNGUUJCNNDG

ENGCPCPFPCVWTCNN[OQKUVDWVPQVVGORGTGFKGYCVGT


#OGPFGFCPF

0/KPKOWO6GUV #OKPKOWOVGUVQHCITCKP

OQKUVWTGOGVGTUJCNNEQPUKUVQHVGUVU


C YKVJUCORNGUPGGFPQVGZEGGFVJTGGQHGCEJITCKP

QTUGGFHQTYJKEJVJGFGXKEGKUWUGFCPF


D YKVJUCORNGUJCXKPICVNGCUVVYQFKHHGTGPV

OQKUVWTGEQPVGPVXCNWGUYKVJKPVJGQRGTCVKPITCPIG

QHVJGFGXKEG


#OGPFGFCPF

0/GVGTVQ.KMG6[RG/GVGT/GVJQF6TCPUHGT

5VCPFCTFU2TQRGTN[UVCPFCTFK\GFTGHGTGPEGOGVGTU

WUKPI0CVKQPCN6[RG'XCNWCVKQP2TQITCOCRRTQXGF

ECNKDTCVKQPUUJCNNDGWUGFCUVTCPUHGTUVCPFCTFU#

TGHGTGPEGOGVGTUJCNNDGQHVJGUCOGV[RGCUVJGOGVGT

WPFGTVGUV6GUVUUJCNNDGEQPFWEVGFUKFGD[UKFGWUKPI

CUCEQORCTKUQPOGFKWOITCKPUCORNGUVJCVCTGENGCP

CPFPCVWTCNN[OQKUVDWVPQVVGORGTGFKGYCVGTPQV

CFFGF


#FFGF

66QNGTCPEGU

66Q7PFGTTGIKUVTCVKQPCPFVQ1XGTTGIKUVTCVKQP

6JGVQNGTCPEGUJGTGKPCHVGTRTGUETKDGFUJCNNDGCRRNKGFVQ

GTTQTUQHWPFGTTGIKUVTCVKQPCPFGTTQTUQHQXGTTGIKUVTCVKQP

66QNGTCPEGU

6#KT1XGP4GHGTGPEG/GVJQF/CKPVGPCPEGCPF

CEEGRVCPEGVQNGTCPEGUUJCNNDGCUUJQYPKP6CDNG6

6QNGTCPEGUCTGGZRTGUUGFCUCHTCEVKQPQHVJGRGTEGPV

OQKUVWTGEQPVGPVQHVJGQHHKEKCNITCKPUCORNGVQIGVJGT

YKVJCOKPKOWOVQNGTCPEG


#OGPFGF
5.56.(a) Grain Moisture Meters



6CDNG6#EEGRVCPEGCPF/CKPVGPCPEG

6QNGTCPEGU#KT1XGP4GHGTGPEG/GVJQF

6[RGQH)TCKP

QT5GGF

6QNGTCPEG /KPKOWO

6QNGTCPEG

%QTPQCVU

TKEGUQTIJWO

UWPHNQYGT

QHVJGRGT

EGPVOQKUVWTG

EQPVGPV

RGTEGPV

KPOQKUVWTG

EQPVGPV

#NNQVJGTEGTGCN

ITCKPUCPFQKN

UGGFU

QHVJGRGT

EGPVOQKUVWTG

EQPVGPV

RGTEGPV

KPOQKUVWTG

EQPVGPV


#OGPFGF

6/GVGTVQ.KMG6[RG/GVGT/GVJQF

/CKPVGPCPEGCPFCEEGRVCPEGVQNGTCPEGUUJCNNDGCU

UJQYPKP6CDNG66JGVQNGTCPEGUUJCNNCRRN[VQCNN

V[RGUQHITCKPCPFUGGF


#FFGF

6CDNG6#EEGRVCPEGCPF/CKPVGPCPEG

6QNGTCPEGU

/GVGTVQ.KMG6[RG/GVGT/GVJQF

5CORNG4GHGTGPEG 6QNGTCPEG

/QKUVWTG

7RVQRGTEGPV RGTEGPVKPOQKUVWTG

EQPVGPV


6CDNG#FFGF

6(QT6GUV9GKIJV2GT$WUJGN+PFKECVKQPUQT

4GEQTFGF4GRTGUGPVCVKQPU6JGOCKPVGPCPEGCPF

CEEGRVCPEGVQNGTCPEGUQPVGUVYGKIJVRGTDWUJGNKPFKECVKQPU

QTTGEQTFGFTGRTGUGPVCVKQPUUJCNNDGMIJ.QT

NDDW6JGVGUVOGVJQFUWUGFUJCNNDGVJQUGURGEKHKGF

D[VJG75&#)+25#


#OGPFGF

747UGT4GSWKTGOGPVU

745GNGEVKQP4GSWKTGOGPVU

748CNWGQHVJG5OCNNGUV7PKVQP2TKOCT[

+PFKECVKPICPF4GEQTFKPI'NGOGPVU6JGTGUQNWVKQP

QHVJGOQKUVWTGOGVGTFKURNC[UJCNNDGRGTEGPV

OQKUVWTGFWTKPIEQOOGTEKCNWUG

745GG)74

74+PUVCNNCVKQP4GSWKTGOGPVU6JGITCKPOQKUVWTG

OGVGTUJCNNDGKPUVCNNGFKPCPGPXKTQPOGPVYKVJKPVJGTCPIG

QHVGORGTCVWTGCPFQTQVJGTGPXKTQPOGPVCNHCEVQTU

URGEKHKGFKPVJGQRGTCVKPIKPUVTWEVKQPU

747UG4GSWKTGOGPVU

741RGTCVKPI+PUVTWEVKQPU6JGQRGTCVKPI

KPUVTWEVKQPUHQTVJGWUGQHVJGITCKPOQKUVWTGOGVGT

UJCNNDGTGCFKN[CXCKNCDNGVQVJGWUGTUGTXKEGVGEJPKEKCP

CPFYGKIJVUCPFOGCUWTGUQHHKEKCNCVVJGRNCEGQH

KPUVCNNCVKQP+VUJCNNKPENWFGCNKUVQHCEEGUUQT[

GSWKROGPVCPFVJGMKPFUQHITCKPQTUGGFVQDG

OGCUWTGFYKVJVJGOQKUVWTGOGVGT


#OGPFGF

741VJGT&GXKEGUPQVWUGFHQT%QOOGTEKCN

/GCUWTGOGPV+HVJGTGCTGQVJGTOQKUVWTGOGVGTUQP

VJGRTGOKUGUPQVWUGFHQTVTCFGQTFGVGTOKPKPIQVJGT

EJCTIGUHQTUGTXKEGUVJGUGFGXKEGUUJCNNDGENGCTN[CPF

EQPURKEWQWUN[OCTMGF0QVHQT7UGKP6TCFGQT

%QOOGTEG

74/CKPVCKPKPI+PVGITKV[QH)TCKP5CORNGU

9JGPGXGTVJGTGKUCVKOGNCRUG
VGORGTCVWTGEJCPIG

DGVYGGPVCMKPIVJGUCORNGCPFVGUVKPIVJGUCORNG

OGCPUVQRTGXGPVEQPFGPUCVKQPQHOQKUVWTGQTNQUUQH

OQKUVWTGHTQOITCKPUCORNGUUJCNNDGWUGF(QT

GZCORNGCEQNFITCKPUCORNGOC[DGMGRVKPCENQUGF

EQPVCKPGTKPQTFGTVQRGTOKVVJGEQNFITCKPVQEQOGVQ

VJGQRGTCVKPIVGORGTCVWTGTCPIGQHVJGOGVGTDGHQTGVJG

ITCKPOQKUVWTGOGCUWTGOGPVUCTGOCFG

742TKPVGF6KEMGVU


C 2TKPVGFVKEMGVUUJCNNDGHTGGHTQOCP[RTGXKQWU

KPFKECVKQPQHOQKUVWTGEQPVGPVQTV[RGQHITCKPQT

UGGFUGNGEVGF


D 6JGEWUVQOGTUJCNNDGIKXGPCRTKPVGFVKEMGV

UJQYKPIVJGFCVGITCKPV[RGITCKPOQKUVWTG

TGUWNVUCPFECNKDTCVKQPXGTUKQPKFGPVKHKECVKQP6JG

VKEMGVUJCNNDGIGPGTCVGFD[VJGITCKPOQKUVWTG

OGVGTU[UVGO


#OGPFGFCPF

74#EEGUUQT[&GXKEGU#EEGUUQT[FGXKEGUKH

PGEGUUCT[KPVJGFGVGTOKPCVKQPQHCOQKUVWTGEQPVGPV

XCNWGUJCNNDGKPENQUGRTQZKOKV[VQVJGOQKUVWTGOGVGT

CPFCNNQYKOOGFKCVGWUG

745CORNKPI#ITCKPUCORNGUJCNNDGQDVCKPGF

D[HQNNQYKPICRRTQRTKCVGUCORNKPIOGVJQFUCPF

GSWKROGPV6JGUGKPENWFGDWVCTGPQVNKOKVGFVQITCKP

RTQDGUQHCRRTQRTKCVGNGPIVJWUGFCVTCPFQONQECVKQPU

KPVJGDWNMVJGWUGQHCRGNKECPUCORNGTQTQVJGT
5.56.(a) Grain Moisture Meters



VGEJPKSWGUCPFGSWKROGPVIKXKPIGSWKXCNGPVTGUWNVU

6JGITCKPUCORNGUJCNNDGVCMGPUWEJVJCVKVKU

TGRTGUGPVCVKXGQHVJGNQV

74.QECVKQP5GG)74

74.GXGN%QPFKVKQP+HGSWKRRGFYKVJCNGXGN

KPFKECVQTCOGVGTUJCNNDGOCKPVCKPGFKPCNGXGN

EQPFKVKQP


#FFGF

74%WTTGPV%CNKDTCVKQP&CVC)TCKPOQKUVWTG

FGVGTOKPCVKQPUUJCNNDGOCFGWUKPIQPN[VJGOQUV

TGEGPVN[RWDNKUJGFECNKDTCVKQPFCVC


#FFGF

742QUVKPIQH/GVGT1RGTCVKPI4CPIG6JG

QRGTCVKPITCPIGQHVJGITCKPOQKUVWTGOGVGTUJCNNDG

ENGCTN[CPFEQPURKEWQWUN[RQUVGFKPVJGRNCEGQH

DWUKPGUUUWEJVJCVVJGKPHQTOCVKQPKUTGCFKN[XKUKDNG

HTQOCTGCUQPCDNGEWUVQOGTRQUKVKQP6JGRQUVGF

KPHQTOCVKQPUJCNNKPENWFGVJGHQNNQYKPI


C 6JGVGORGTCVWTGTCPIGQXGTYJKEJVJGOGVGTOC[

DGWUGFCPFUVKNNEQORN[YKVJVJGCRRNKECDNG

TGSWKTGOGPVU+HVJGVGORGTCVWTGTCPIGXCTKGUHQT

FKHHGTGPVITCKPUQTUGGFVJGTCPIGUJCNNDG

URGEKHKGFHQTGCEJ


D 6JGOQKUVWTGTCPIGHQTGCEJITCKPQTUGGFHQT

YJKEJVJGOGVGTKUVQDGWUGF


E 6JGVGORGTCVWTGTCPIGHQTGCEJITCKPQTUGGFHQT

YJKEJVJGOGVGTKUVQDGWUGF


F 6JGOCZKOWOCNNQYCDNGFKHHGTGPEGKPVGORGTCVWTG

VJCVOC[GZKUVDGVYGGPVJGOGVGTCPFVJGUCORNG

HQTYJKEJCPCEEWTCVGOQKUVWTGFGVGTOKPCVKQPECP

DGOCFG#FFGF
5.56.(a) Grain Moisture Meters



Table S.2.5. Categories of Device and Methods of Sealing

Categories of Device Method of Sealing

Category 1: No remote configuration Seal by physical seal or two event counters: one for calibration

capability. parameters (000 to 999) and one for configuration parameters (000

to 999). If equipped with event counters, the device must be capable

of displaying, or printing through the device or through another onsite device, the contents of the counters.

Category 2: Remote configuration The hardware enabling access for remote communication must be at

capability, but access is controlled by the device and sealed using a physical seal or two event counters;

physical hardware. one for calibration parameters (000 to 999) and one for

A device shall clearly indicate that it is in the counters, the device must be capable of displaying, or printing

remote configuration mode and shall not be through the device or through another on-site device, the contents of

capable of operating in the measure mode the counters.

while enabled for remote configuration.

configuration parameters (000 to 999.) If equipped with event

Category 3: Remote configuration capability An event logger is required in the device; it must include an event

access may be unlimited or controlled counter (000 to 999), the parameter ID, the date and time of the

through a software switch (e.g., password). change , and the new value of the parameter (for calibration changes

When accessed remotely for the purpose of information must be available through the device or through another

modifying sealable parameters, the device on-site device. The event logger shall have a capacity to retain

shall clearly indicate that it is in the records equal to twenty-five (25) times the number of sealable

configuration mode and shall not be capable parameters in the device, but not more than 1000 records are

of operating in the measuring mode. required. (Note: Does not require 1000 changes to be stored for

consisting of multiple constants, the calibration version number may

be used rather than the calibration constants). A printed copy of the

each parameter.)

Category 3a: No remote capability, but Same as Category 3 

operator is able to make changes that affect

the metrological integrity of the device (e.g.,

slope, bias, etc.) in normal operation.

Category 3b: No remote capability, but Same as Category 3 

 access to metrological parameters is

controlled through a software switch (e.g.,

password).

[Nonretroactive as of January 1, 1999.] (Amended 1998)
5.56.(b) Grain Moisture Meters



Sec. 5.56.(b) Grain Moisture Meters

Section 5.56. has been reorganized into two Sections. This Section, 5.56.(b) is applies to all non-NTEP grain moisture

meters manufactured or placed into service before January 1, 1998.

(Code reorganized and renumbered 1996.)

A. Application

A.1. - This code applies to grain moisture meters; that is,

devices used to indicate directly or through conversion

and/or correction tables the moisture content of cereal grain

and oil seeds. The code consists of general requirements

applicable to all moisture meters and specific requirements

applicable only to certain types of moisture meters.

A.2. - This code does not apply to devices used for

in-motion measurement of grain moisture content or seed

moisture content.

A.3. - See also Sec. 1.10; General Code requirements.

 S. Specifications

S.1. Design of Indicating and Recording Elements and

of Recorded Representations.

S.1.1. Primary Elements, General. - A meter shall be

equipped with a primary indicating element and may also

be equipped with a primary recording element. If the

meter indicates directly and/or is equipped to record, the

meter shall indicate and/or record its measurements in

terms of percent moisture content, wet basis.

Subdivisions of this unit shall be in terms of decimal

subdivisions (not fractions). If the meter indicates in the

conventional scale and requires conversion or correction

tables, the resulting values after use of such tables shall

be in terms of percent moisture content, wet basis.

Subdivisions of this unit shall be in terms of decimal

subdivisions (not fractions).

S.1.2. Digital Indications.

S.1.2.1. Measurement Completion. - A digital

indicating element shall not display any values (either

moisture content or conventional scale) before the end

of the measurement cycle.

S.1.3. Graduations.

S.1.3.1. Length. - Graduations shall be so varied in

length that they may be conveniently read.

S.1.3.2. Width. - In any series of graduations, the

width of a graduation shall in no case be greater than

the width of the minimum clear interval between

graduations, and the width of the main graduations

shall be not more than 50 percent greater than the

width of subordinate graduations. Graduations shall in

no case be less than 0.2 mm (0.008 in) in width.

S.1.3.3. Clear Interval Between Graduations. - The

clear interval shall be not less than 0.75 mm (0.03 in)

between graduations. If the graduations are not

parallel, the measurement shall be made:

(a) along the line of relative movement between the

graduations at the end of the indicator, or

(b) if the indicator is continuous, at the point of

widest separation of the graduations.

S.1.4. Indicators.

S.1.4.1. Symmetry. - The index of an indicator shall

be symmetrical with respect to the graduations, at least

throughout that portion of its length associated with

the graduations.

S.1.4.2. Length. - The index of an indicator shall

reach to the finest graduations with which it is used,

unless the indicator and the graduations are in the

same plane, in which case the distance between the

end of the indicator and the ends of the graduations,

measured along the line of the graduations, shall be

not more than 1.0 mm (0.04 in).

S.1.4.3. Width. - The width of the index of an indicator in relation to the series of graduations with

which it is used shall be not greater than:

(a) the width of the widest graduation, nor

(b) the width of the minimum clear interval between

graduations.

When the index of an indicator extends along the

entire length of a graduation, that portion of the index

of the indicator that may be brought into coincidence

with the graduation shall be of the same width as the
5.56.(b) Grain Moisture Meters



graduation throughout the length of the index that coincides be used and still comply with the applicable

with the graduation. requirements;

S.1.4.4. Clearance. - The clearance between the

index of an indicator and the graduations shall in no

case be more than 1.5 mm (0.06 in).

S.1.4.5. Parallax. - Parallax effects shall be reduced

to the practicable minimum.

S.1.5. Recording Elements.

S.1.5.1. General. - If a meter is equipped with a

recording element, it shall record in terms of percent

moisture content, wet basis only, and not in terms of

conventional scale.

S.1.5.2. Measurement Completion. - A recording

element shall not record any values before the end of

the measurement cycle.

S.1.5.3. Range of Moisture Content. - A recording

element shall not record any values when the moisture

content of the grain sample is beyond the operating

range of the device.

S.1.6. Design of Direct Reading Grain Moisture Meters.

S.1.6.1. Grain or Seed Kind and Class Selection

and Recording. - Provision shall be made for

selecting and recording, if equipped to record, the

kind and class (as appropriate) of grain or seed to be

measured. The means to select the kind and class of

grain or seed shall be readily visible and the kind and

class of grain or seed selected shall be clearly and

definitely identified in letters (such as Wheat or WHT,

HRWW, etc.).

S.1.6.2. Operating Range. - A meter shall

automatically and clearly indicate when the operating

range of the meter has been exceeded or the manufacturer shall:

(a) clearly and conspicuously mark the operating

ranges on the meter; or

(b) furnish the operating ranges of the meter and

the means to clearly and conspicuously display

this information on or immediately adjacent to

the device.

The operating range shall specify the following:

(a) the temperature range over which the meter may

(b) the moisture range for each grain or seed for

which the meter is to be used;

(c) the temperature range for each grain or seed for

which the meter is to be used; and

(d) the maximum allowable difference in

temperature between the meter and the sample

for which an accurate moisture determination

can be made.

Examples of clearly indicating these conditions

include an error indication, flashing the displayed

moisture value, or blanking the display.

[Nonretroactive as of January 1, 1989.]

(Amended 1986 and 1988)

S.1.6.3. Value of Smallest Unit.

(a) The value of the minimum indicated or recorded

moisture indication shall not be greater than

0.1 percent.

S.1.7. Electric Power Supply.

S.1.7.1. Power Supply, Voltage and Frequency.

(a) A meter that operates using alternating current

must perform within the tolerances defined in

Section T.2. - Tolerance Values over the line

voltage range 100 V to 130 V, or 200 V to 250 V

rms as designed, and over the frequency range

of 59.5 Hz to 60.5 Hz.

(b) Battery-operated instruments shall not indicate

or record values outside the applicable

tolerance limits when battery power output is

excessive or deficient.

[Nonretroactive as of January 1, 1989.]

S.1.7.2. Power Interruption. - A power interruption

shall not cause an indicating or recording element to

display or record any values outside the applicable

tolerance limits.

[Nonretroactive as of January 1, 1989.]

(Added 1988)

S.1.8. Level Indicating Means. A meter shall be

equipped with a level indicator and leveling adjustments

if its performance is changed by an amount greater than

the applicable tolerance when the meter is moved from

a level position to a position that is out of level in any
5.56.(b) Grain Moisture Meters



upright direction by up to 5 percent (approximately 3

degrees).

The level-indicating means shall be readable without

removing any meter parts requiring a tool.

[Nonretroactive as of January 1, 1989.]

(Added 1988) (Amended 1994)

S.1.9. Operating Temperature.

(a) A meter shall not display or record any usable

values until the operating temperature necessary

for accurate determination has been attained, or

the meter shall bear a conspicuous statement

adjacent to the indication stating that the meter

shall be turned on for a time period specified by the

manufacturer prior to use.

(b) A meter shall meet the requirements of T.2. -

Tolerance Values when operated in the

temperature range of 2 EC to 40 EC (35 EF to

104 EF) or within the range specified by the meter

manufacturer.

(c) If the manufacturer specifies a temperature range,

the range shall be at least 10 EC (20 EF) and shall

be marked on the device.

[Nonretroactive as of January 1, 1989.]

(Added 1988)

S.2. Design of Measuring Elements.

S.2.1. Design of Zero-Setting and Test Point

Mechanisms. - If a grain moisture meter is equipped

with a zero setting and/or test point mechanism(s), this

(these) mechanism(s) shall be adjustable only with a tool

outside and entirely separate from this mechanism or

enclosed in a cabinet. This requirement shall not apply

to manual operations that the operator must make

(following operating instructions) in order to obtain a

meter reading on a grain sample.

S.2.2. Provision for Sealing. - Provision shall be made

for applying a security seal in a manner that requires the

security seal to be broken before an adjustment can be

made to any component of the grain moisture meter that

is set by the manufacturer or authorized service

representative and not intended to be adjusted by the

user.

S.3. Accessory Equipment. - When the operating

instructions for a moisture meter require accessory

equipment separate from and external to the moisture meter,

such equipment shall be appropriate and complete for the

measurement.

S.3.1. Grain-Test Scale. - If the moisture meter requires

the weighing of the grain sample, the weighing device

shall meet the requirements of the General Code and

those applicable portions of the Scales Code.

S.3.2. Thermometers or Other Temperature Sensing

Equipment. -

(a) The temperature sensing equipment or thermometer

shall be designed to be in direct contact with a

grain sample in a closed container. It is acceptable

to insert thermometer through a small hole in the lid

of the container used to hold the grain sample.

(b) A separate thermometer or other temperature

sensing equipment shall have temperature divisions

not greater than the temperature increments used by

the manufacturer in the correction table.

(Amended 1988)

S.3.3. Conversion and Correction Tables. -

Conversion and correction tables, charts, graphs, slide

rules, or other apparatus to convert the conventional

scale values read from a moisture meter to moisture

content values, if such apparatus is required, shall be

appropriate and correct for the moisture meter being used

and shall be marked with the following information:

(a) name and address or trademark of the

manufacturer;

(b) the type or design of the device with which it is

intended to be used;

(c) date of issue;

(d) the kind or classes of grain or seed for which the

device is designed to measure moisture content;

(e) the limitations of use, including but not confined to

the moisture measurement range, grain or seed

temperature, kind or class of grain or seed,

moisture meter temperature, voltage and frequency

ranges, electromagnetic interferences, and

necessary accessory equipment; but

(f) values exceeding any measurement range shall not

be included. 

(Added 1984)

S.3.4. Operating Instructions and Use Limitations. -

Operating instructions shall be furnished by the

manufacturer with each device with all of the

information required by paragraph S.3.3. Complete

information concerning the accuracy, sensitivity, and use
5.56.(b) Grain Moisture Meters



6JGUGVQNGTCPEGUFQPQVCRRN[VQVGUVUKPYJKEJITCKP 

OQKUVWTGOGVGTUCTGVJGVTCPUHGTUVCPFCTFU

of accessory equipment (e.g., test weight per bushel expressed as a fraction of the percent moisture content of

equipment, thermometer, etc.) necessary in obtaining a the official grain sample, together with a minimum

moisture content shall be included. tolerance.

N. Notes

N.1. Testing Procedures.

N.1.1. Transfer Standards. - Official grain samples

shall be used as the official transfer standards with

moisture content values assigned by the reference

methods. The reference methods shall be the oven

drying methods as specified by the USDA GIPSA.

Tolerances shall be applied to the average of at least

three measurements on each official grain sample.

Official grain samples shall be clean and naturally moist,

but not tempered (i.e., water not added).

(Amended 1992)

N.1.2. Minimum Test.- A minimum test of a grain

moisture meter shall consist of tests:

(a) with samples (need not exceed three) of each grain

or seed for which the device is used, and

(b) with samples having at least two different moisture

content values within the operating range of the

device.

(Amended 1986, 1989)

N.1.3. Temperature Measuring Equipment. - The than 0.1 percent.

accuracy of accessory temperature measuring equipment

shall be determined by comparison with a calibrated

temperature sensor, such as a total immersion

thermometer with 0.1 EC (0.2 EF) subdivisions,

indicating over a range of from 0 EC to 40 EC (32 EF to

104 EF) with a maximum error of ±0.1 EC (0.2 EF).

Tests shall be conducted at two temperatures using liquid

baths (e.g., ice water and room temperature water). The

two temperatures selected shall not exceed the range of

temperatures identified in the moisture meter operating

instructions. (Amended 1988)

T. Tolerances1

T.1. To Underregistration and to Overregistration. -

The tolerances hereinafter prescribed shall be applied to

errors of under registration and errors of overregistration.

T.2. Tolerance Values. - Maintenance and acceptance

tolerances shall be as shown in Table T.2. Tolerances are

T.3. For Test Weight Per Bushel Indications or

Recorded Representations. - The maintenance and

acceptance tolerances on test weight per bushel indications

or recorded representations shall be 0.193 kg/hL or

0.15 lb/bu. The test methods used shall be those specified

by the USDA GIPSA.

(Amended 1992)

T.4. Thermometers or Other Temperature Sensing

Equipment. - The tolerance for a separate thermometer or

temperature sensing equipment used to determine the

temperature of grain samples for the purpose of making

temperature corrections in moisture determinations shall be

±0.5 EC (1 EF). (Added 1988)

UR. User Requirements

UR.1. Selection Requirements.

UR.1.1. Value of the Smallest Unit on Primary

Indicating and Recording Elements. - The value of the

smallest unit on a moisture meter, whether the moisture

meter reads directly in terms of moisture content, or

when the conventional scale unit is converted or

corrected to moisture content, shall be equal to or less

Table T.2. Acceptance and Maintenance Tolerances for

Grain Moisture Meters
5.56.(b) Grain Moisture Meters



Corn, oats, 0.05 of the percent 0.8 percent in

rice, sorg- moisture content moisture

hum, content

sunflower

All other 0.04 of the percent 0.7 percent in

cereal grains moisture content moisture

and oil seeds content

UR.1.2. See G-UR.1.2.

UR.2. Installation Requirements. - The grain moisture

meter shall be installed in an environment within the range

of temperature and/or other environmental factors specified

(a) in the operating manual, and (b) on the conversion or

correction tables if such tables are necessary for the

operation of the device. 

UR.3. Use Requirements.

UR.3.1. Operating Instructions. - The operating

instructions for the use of the grain moisture meter shall

be readily available to the user, service technician, and

weights and measures official at the place of installation.

It shall include a list of accessory equipment, conversion

and correction charts if any are required to obtain

moisture content values, and the kinds of grain or seed to

be measured with the moisture meter.

(Amended 1988)

UR.3.2. Other Devices not used for Commercial

Measurement. - If there are other moisture meters on

the premises not used for trade or determining other

charges for services, these devices shall be clearly and

conspicuously marked "Not for Use in Trade or

Commerce."

UR.3.3. Maintaining Integrity of Grain Samples. -

Whenever there is a time lapse (temperature change)

between taking the sample and testing the sample, means

to prevent condensation of moisture or loss of moisture

from grain samples shall be used. For example, a cold

grain sample may be kept in a closed container in order

to permit the cold grain to come to the operating

temperature range of the meter before the grain moisture

measurements are made.

UR.3.4. Printed Tickets. - Printed tickets shall be free

from any previous indication of moisture content or type

of grain or seed selected.

UR.3.5. Accessory Devices. - Accessory devices, if

necessary in the determination of a moisture content

value, shall be in close proximity to the moisture meter

and allow immediate use.

UR.3.6. Sampling. - A grain sample shall be obtained

by following appropriate sampling methods and

equipment. These include, but are not limited to, grain

probes of appropriate length used at random locations in

the bulk, the use of a pelican sampler, or other

techniques and equipment giving equivalent results. The

grain sample shall be taken such that it is representative

of the lot.

UR.3.7. Location. - See G-UR.3.3. - Position of

Equipment.

UR.3.8. Level Condition. - If equipped with a level

indicator, a meter shall be maintained in a level

condition.

(Added 1988)

UR.3.9. Operating Limitation. - Unless otherwise

specified by the meter manufacturer, moisture

determinations shall not be made when the difference in

temperatures between the grain sample and the meter

exceeds 10 EC (20 EF).

(Added 1988)

UR.3.10. Current Calibration Chart or Data. - Grain

moisture determinations shall be made using only the

most recently published calibration charts or calibration

data.

(Added 1988)

UR.3.11. Posting of Meter Operating Range. - The

operating range of the grain moisture meter shall be

clearly and conspicuously posted in the place of business

such that the information is readily visible from a reasonable customer position. The posted information shall

include the following:

(a) The temperature range over which the meter may

be used and still comply with the applicable

requirements. If the temperature range varies for

different grains or seed, the range shall be specified

for each.

(b) The moisture range for each grain or seed for

which the meter is to be used.

(c) The temperature range for each grain or seed for

which the meter is to be used.

(d) The maximum allowable difference in temperature

that may exist between the meter and the sample for

which an accurate moisture determination can be

made.

(Added 1988)
5.56.(b) Grain Moisture Meters


5.57. Near-Infrared Grain Analyzers



Sec. 5.57. Near-Infrared Grain Analyzers - Tentative Code

This tentative code has only a trial or experimental status and is not intended to be enforced. The requirements are

designed for study prior to the development and adoption of a final Code for Near-Infrared Grain Analyzers. All

nonretroactive and effective dates have been changed to January 1, 2002. 

 (Tentative Code Added 1994.)

A. Application

A.1. - This code applies to near-infrared (NIR) grain

analyzers; that is, devices used to indicate the constituent

values (other than moisture content) of grain using

near-infrared reflectance or transmittance technology.

These instruments may analyze either whole grain or

ground grain samples. The code consists of general

requirements applicable to all NIR analyzers and specific

requirements applicable only to certain types of NIR

analyzers, grain types, or grain constituents.

A.2. - This code does not apply to devices used for

in-motion measurement of grain constituent values.

A.3. Type Evaluation. - The National Type Evaluation

Program will accept for type evaluation only those devices

that comply with the nonretroactive requirements scheduled

to take effect on January 1, 2002. State enforcement will be

based upon the effective dates identified with each

requirement when specific dates are shown.

A.3.1. Dual Type Evaluations. - In addition to meeting

the requirements of this code, a whole grain NIR

analyzer that displays a measured moisture value must

also comply with the requirements of the Grain Moisture

Meters Code and be type evaluated as a grain moisture

meter by NTEP. 

(Added 2001)

A.3.2 Calibrations. - The National Type Evaluation

Program Certificate of Conformance (CC) shall indicate

the native moisture basis of each calibration. The

"native" moisture basis is the default moisture basis of

the sealable constituent calibration (or constituent

calibration pair when a non-displayed moisture

calibration is also involved). If an NIR analyzer uses a

self-generated moisture measurement internally but does

NOT display or record a moisture value, the moisture

calibration shall be considered to be a part of the

constituent calibration. For such calibrations, the CC

shall state: "Includes non-displayed moisture

calibration." Changes to any part of such calibrations

shall require changes to the CC.

(Added 2001)

A.4. - See also Sec. 1.10; General Code requirements.

S. Specifications

S.1. Design of Indicating, Recording, and Measuring

Elements.

S.1.1. Digital Indications and Recording Elements.

(a) Analyzers shall be equipped with a digital

indicating element.

(b) The minimum height for the digits used to display

constituent values shall be 10 mm.

(c) Analyzers shall be equipped with a communication

interface that permits interfacing with a recording

element and transmitting the date, grain type or

class, constituent values, the moisture basis for

each constituant value (except moisture), and

calibration version identification.

(Amended 2001)

(d) A digital indicating element shall not display, and

a recording element shall not record, any

constituent value before the end of the

measurement cycle.

(e) Constituent content shall be recorded and

displayed as percent of total mass. The moisture

basis shall also be recorded and displayed for each

constituent content result (except moisture). If a

whole grain analyzer that is calibrated to display

results on an "as is" moisture basis does NOT

display or record a moisture value, it must clearly

indicate that results are expressed on an "as is"

moisture basis. Ground grain analyzers must

ALWAYS display and record a moisture

measurement for "as is" content results (except

moisture).

(Amended 2001)

(f) An analyzer shall not display or record any

constituent value that is beyond the operating

range of the device unless the constituent value

representation includes a clear error indication
5.57. Near-Infrared Grain Analyzers



(and recorded error message with the recorded range of the device has been exceeded. The statement of

representation). the operating range shall be specified in the operator’s

(g) If an NIR analyzer is used to determine a moisture

value, either to determine the moisture of an "as is" (a) The ambient temperature range over which the

constituent content measurement or to convert from analyzer may be used and still comply with the

one moisture basis to another, the moisture applicable requirements shall be specified. The

measurement must be concurrent with the minimum temperature range shall be 10 EC to

measurement of other constituents. 30 EC. No constituent value may be displayed

(Added 2001) when the temperature range is exceeded. An

[Nonretroactive as of January 1, 2002] appropriate error message shall be displayed when

the temperature of the analyzer is outside its

S.1.2. Selecting Grain Class and Constituent. -

Provision shall be made for selecting, and recording

the type or class of grain and the constituent(s) to be

measured. The means to select the grain type or

class and constituent(s) shall be readily visible and

the type or class of grain and constituent(s) selected

shall be clearly and definitely identified in letters

(such as HRWW, HRSW, etc. or PROT, etc.). A

symbol to identify the display of the type or class of

grain and constituent(s) selected is permitted provided

that it is clearly defined adjacent to the display.

Minimum acceptable abbreviations are listed in Table

S.1.2. Meters shall have the capability (i.e., display

capacity) of indicating the grain type using a

minimum of four characters in order to accommodate

the abbreviations listed in Table S.1.2.

[Nonretroactive as of January 1, 2002]

Table S.1.2. Grain Types Considered for Type Evaluation

and Calibration

and Minimum Acceptable Abbreviations

Grain Type Minimum Acceptable Abbreviation

Durum Wheat

Hard Red Spring Wheat

Hard Red Winter Wheat

Hard White Wheat

Soft Red Winter Wheat

Soft White Wheat

Soybeans

Two-rowed Barley

Six-rowed Barley

Corn

DURW

HRSW

HRWW

HDWW

SRWW

SWW

SOYB

TRB

SRB

CORN

[Nonretroactive as of January 1, 2002]

(Table Amended 2001)

S.1.3. Operating Range. - An analyzer shall

automatically and clearly indicate when the operating

manual and shall operate as follows:

specified operating range.

(b) The constituent range at the moisture basis

specified in Table N.1.1. shall be specified for each

grain or seed for which the analyzer is to be used.

A constituent value may be displayed when the

constituent range is exceeded if accompanied by a

clear indication that the constituent range has been

exceeded.

(Amended 2001)

(c) For whole grain analyzers only, the temperature

range shall be specified for each grain or seed for

which an analyzer is to be used. The minimum

temperature range for each grain shall be 10 EC to

30 EC. No constituent value may be displayed

when the temperature range is exceeded. An

appropriate error message shall be displayed when

the temperature of the grain sample exceeds the

temperature range for the grain. The requirements

of this subsection (c) are not applicable to ground

grain analyzers.

(d) For whole grain analyzers, the maximum allowable

difference in temperature between the instrument

environment (ambient temperature) and the sample

for which an accurate constituent determination

can be made shall be specified. The minimum

temperature range shall cover at least 10 EC. No

constituent value may be displayed when the

maximum allowable temperature difference is

exceeded. An appropriate error message shall be

displayed when the difference between the ambient

temperature and the sample temperature exceeds

the specified difference. The requirements of this

subsection (d) are not applicable to ground grain

analyzers.

[Nonretroactive and effective as of January 1, 2002]
5.57. Near-Infrared Grain Analyzers



S.1.4. Operating Temperature:

(a) An analyzer shall not display or record any usable

values until the internal operating temperature

necessary to meet tolerance requirements has been

attained, or the analyzer shall bear a conspicuous

statement adjacent to the indication stating that the

analyzer shall be turned on for a time period

specified by the manufacturer prior to use.

(b) If an instrument does not meet tolerance

requirements because there is an upper internal

operating temperature limit that could be exceeded

when operating within the ambient temperature

range specified by the manufacturer, then a means

of sensing and indicating an over-temperature

condition must be provided.

[Nonretroactive as of January 1, 2002]

S.1.5. Value of Smallest Unit. - The display shall

permit constituent value determination to both 0.01

percent and 0.1 percent resolution. The 0.1 percent

resolution is for commercial transactions; the 0.01

percent resolution is for calibration purposes only, not

for commercial purposes.

[Nonretroactive as of January 1, 2002]

S.2. Design of NIR Analyzers

S.2.1. Minimum Sample Size. - Analyzers shall be

designed to measure constituent values of representative

size grain samples. The minimum allowable sample size

used in analysis shall be 20 g.

[Nonretroactive and effective as of January 1, 2002]

S.2.2. Electric Power Supply. (Note added 1995) (Amended 1995)

S.2.2.1. Power Supply, Voltage and Frequency. - An

analyzer that operates using alternating current must

perform within tolerance requirements over the line

voltage range 100 V to 130 V and over the frequency

range of 59.5 Hz to 60.5 Hz.

[Nonretroactive as of January 1, 2002]

S.2.2.2. Power Interruption. - A power interruption

shall not cause an indicating or recording element to

display or record any values outside the applicable

tolerance limits.

[Nonretroactive and effective as of January 1, 2002]

S.2.3. Level Indicating Means. - Analyzers shall be

equipped with a level indicator and leveling adjustments

if its performance is changed by an amount greater than

the tolerance requirement when the instrument is moved

from a level position into a position that is out of level in

any upright direction by up to 5 percent (approximately

3 degrees). The level-indicating means shall be readable

without removing any instrument parts requiring a tool.

[Nonretroactive and effective as of January 1, 2002]

S.2.4. Environmental Conditions. - Instrument optics

and electronics must be protected from exposure to dust

by either sealing these areas or by protecting them with

a dust filtration system suitable for the removal of

air-borne grain dust.

[Nonretroactive as of January 1, 2002]

S.2.5. Calibration Transfer and Verification.

S.2.5.1. Calibration Transfer. - The instrument

hardware/software design and calibration procedures

shall permit calibration development and the

mathematical transfer of calibrations between

instruments of like models.

Note: Only the manufacturer or the manufacturer’s

designated service agency may make calibration

transfer or slope adjustments on near-infrared grain

analyzers and, except for instrument failure and

repair, only during a prescribed period of time during

the year. This does not preclude the possibility of the

operator installing the manufacturer-specified

calibration constants or standardization parameters

under the instructions of the manufacturer or the

manufacturer’s designated service agency. Nor does

it preclude operator bias adjustments when made

under the conditions specified in UR.2.8.

[Nonretroactive and effective as of January 1,

2002.]

S.2.5.2. Calibration Version. - An instrument must be

capable of displaying either calibration constants, a

unique calibration name, or a unique calibration

version number for use in verifying that the latest

version of the calibration is being used to make

constituent determinations, and that the appropriate

instrument settings have been made for the calibration

being used.

[Nonretroactive and effective as of January 1, 2002].

(Amended 2001)

S.2.5.3. Calibration Corruption. - If calibration

constants are digitally stored in an electronically

alterable form, the analyzer shall be designed to make

automatic checks to detect corruption of calibration
5.57. Near-Infrared Grain Analyzers



constants. An error message must be displayed if e. the limitations of use, including but not limited to

calibration constants have been electronically altered. constituent range, grain or seed temperature, kind or

[Nonretroactive as of January 1, 2002]. class of grain or seed, instrument temperature, voltage

S.2.6. Provision for Sealing. - An event logger is

required in the device; it must include an event counter

(000 to 999), the parameter ID, the date and time of the

change, and the new value of the parameter (for

calibration changes consisting of multiple constants, the

calibration version number may be used rather than the

calibration constants.)

A printed copy of the information must be available

through the device or through another on-site device.

The event logger shall have a capacity to retain records

equal to twenty-five (25) times the number of sealable

parameters in the device, but not more than 1000

records are required. (Note: Does not require 1000

changes to be stored for each parameter.)

[Nonretroactive as of January 1, 2002]

(Amended 1997)

S.3. Accessory Equipment. When the operating

instructions for a NIR analyzer require accessory

equipment separate from and external to the analyzer, such

equipment shall be appropriate and complete for the

measurement.

[Nonretroactive and effective as of January 1, 2002]

S.3.1. Grinders. - The make and model of grinder used

for ground grain NIR analyzers must be specified by the

manufacturer and required as auxiliary equipment in the

determination of constituent values for applicable grain

types.

[Nonretroactive as of January 1, 2002]

S.4. Operating Instructions and Use Limitations. - The

manufacturer shall furnish operating instructions for the

device and accessories that include complete information

concerning the accuracy, sensitivity, and use of accessory

equipment necessary in obtaining a constituent value.

Operating instructions shall include the following

information:

a. name and address or trademark of the manufacturer;

b. the type or design of the device for which the operating

instructions are intended to be used;

c. date of issue;

d. the kind or classes of grain or seed for which the device

is designed to measure constituent values; and

and frequency ranges, electromagnetic interferences,

and necessary accessory equipment.

[Nonretroactive as of January 1, 2002]

N. Notes

N.1. Testing Procedures.

N.1.1. Field Inspection. - Whole grain samples shall be

used as the official field inspection standards. Five

samples per grain type or class shall be used to check

instrument performance. Each sample will be analyzed

once. One of the samples will be analyzed an additional

four times to test instrument repeatability. For ground

grain instruments, the ground sample will be repacked

four times. A new grind is not required. Test results

must be converted to the standard moisture bases shown

in Table N.1.1. before applying the tolerances of Table

T.2. Test results on whole grain analyzers that produce

results on an "as is" basis without displaying or

recording a moisture value shall be converted to the

standard moisture bases shown in Table N.1.1. using

sample moisture values determined with the facility’s

moisture meter (which must be certified for commercial

use).

(Amended 2001)

Constituent values shall be assigned to test samples by

the Grain Inspection, Packers and Stockyards

Administration (GIPSA). Tolerances shall be applied to

individual sample measurements, the average of

individual measurements on each of the five test

samples, and the maximum difference (range) in results

for five analyses on one of the test samples.

(Amended 2001)
5.57. Near-Infrared Grain Analyzers



Table T.2. Acceptance and Maintenance Tolerances

for NIR Grain Analyzers

Type of Grain Constituent

Individual

Samples

(percent)

Average

for Five

Samples

(percent)

Range for

Five

Retests

(percent)

Durum Wheat, Hard

Red Spring Wheat,

Hard Red Winter

Wheat, Hard White

Wheat, Soft Red

Winter Wheat, Soft

White Wheat

protein 0.60 0.40 0.40

Soybeans protein 0.80 0.60 0.60

oil 0.70 0.50 0.50

Two-rowed Barley

Six-rowed Barley protein 0.70 0.50 0.50

Corn protein 0.80 0.60 0.60

oil 0.70 0.50 0.50

starch 1.00 0.80 0.80

(Amended 2001)

Table N.1.1. Constant Moisture Basis for Type

Evaluation and Field Inspection

)TCKP6[RGQT %QPUVKVWGPVU
U /QKUVWTG$CUKU

%NCUU

&WTWO9JGCV 2TQVGKP RGTEGPV

*CTF4GF

5RTKPI9JGCV

*CTF4GF

9KPVGT9JGCV

*CTF9JKVG

9JGCV5QHV

4GF9KPVGT

9JGCV5QHV

9JKVG9JGCV

5Q[DGCPU 2TQVGKP RGTEGPV

1KN

6YQTQYGF 2TQVGKP RGTEGPVFT[

$CTNG[ DCUKU

5KZTQYGF

$CTNG[

%QTP 2TQVGKP RGTEGPVFT[

1KN DCUKU

5VCTEJ

=0QPTGVTQCEVKXGCUQH,CPWCT[?


6CDNG#FFGF

N.1.2. Standard Reference Samples. - Reference

samples used for field inspection purposes shall be clean

and selected to reasonably represent the constituent

range. These samples shall be selected such that the

difference between constituent values obtained using the

GIPSA standard reference method and an official GIPSA

NIR grain analyzer does not exceed one-half of the

acceptance tolerance shown in Table T.2. for individual

test samples or 0.375 times the acceptance tolerance

shown for the average of five samples.

(Amended 2001)

T. Tolerances

T.1. To Underregistration and to Overregistration. -

The tolerances hereinafter prescribed shall be applied to

errors of underregistration and errors of overregistration

and shall be based on constituent values expressed at the

moisture bases shown in Table N.1.1.

(Amended 2001)

T.2. Tolerance Values. - Acceptance and maintenance

tolerances shall be equal. Tolerances for individual samples

and the average for five samples are as shown in Table T.2.

UR. User Requirements

UR.1. Installation Requirements. - The NIR analyzer

shall be installed in an environment within the range of

temperature and/or other environmental factors specified in

the operating manual.

UR.2. User Requirements.

UR.2.1. Operating Instructions. - The operating

instructions for the NIR analyzer shall be readily

available to the user, service technician, and weights and

measures official at the place of installation. It shall

include a list of accessory equipment if any are required

to obtain constituent values, and the type or class of grain

to be measured with the NIR analyzer. If an NIR

analyzer has the capability, the user is permitted to select

the moisture basis to be used on any measurement.

(Amended 2001)

UR.2.2. Other Devices not used for Commercial

Measurement. - If there are other NIR analyzers on the

premises not used for trade or determining other charges

for services, these devices shall be clearly and
5.57. Near-Infrared Grain Analyzers



conspicuously marked "Not for Use in Trade or have been measured on a moisture meter certified for

Commerce." commercial use.

UR.2.3. Printed Tickets. -

(a) Printed tickets shall be free from any previous

indication of constituent or grain type selected.

The printed ticket shall indicate constituent values

and the moisture basis associated with each

constituent value (except moisture). If the analyzer

is calibrated to display results on an "as is"

moisture basis and does NOT display or record a

moisture value, the ticket must clearly indicate that

results are expressed on an "as is" moisture basis.

(Amended 2001)

(b) The customer shall be given a printed ticket

showing the date, grain type or class, constituent

results, and calibration version identification. If

the analyzer converts constituent results to a

manually entered moisture basis, the "native"

concentration and the "native" moisture basis must

appear on the printed ticket in addition to the

converted results and the manually entered

moisture basis. If the manually entered moisture

basis is intended to be the moisture value for an

"as is" constituent concentration measurement, that

moisture value must have been obtained on the

same sample and must have been measured on a

moisture meter certified for commercial use. The

information presented on the ticket shall be

arranged in a consistent and unambiguous manner.

The ticket shall be generated by the near-infrared

grain analyzer system.

[Nonretroactive as of January 1, 2002]

(Amended 2001)

UR.2.4. Grinders. - Place grinders in a separate room

from the NIR analyzer to avoid instrument

contamination. If a separate room is not available, the

grinder may be in the same room with the NIR analyzer

provided the grinder is not placed within 1 meter of the

air intake on the NIR.

UR.2.5. Sampling. - Samples shall be obtained by

following appropriate sampling methods and equipment.

These include, but are not limited to grain probes of

appropriate length used at random locations in the bulk,

the use of a pelican sampler, or other techniques and

equipment giving equivalent results. The sample shall be

taken such that it is representative of the lot. If an NIR

analyzer permits user entry of the moisture value for an

"as is" constituent measurement, that moisture value

must have been obtained on the same sample and must

(Amended 2001)

UR.2.6. Level Condition. - If equipped with a level

indicator, an analyzer shall be maintained in a level

condition.

UR.2.7. Operating Limitation. - Constituent

determinations shall not be made when the difference in

temperatures between the grain sample and the

instrument environment (ambient temperature) exceeds

manufacturer recommendations.

UR.2.8. Slope and Bias Adjustments. - Bias changes

shall be made only on the basis of tests run on a current

set of Standard Reference Samples (SRS) traceable to

GIPSA Master Instruments. A written explanation and 1

record of all calibration changes, including those changes

made by a manufacturer or the manufacturer’s designated

service agency, shall be maintained. The log shall

indicate the date and magnitude of changes in bias and

slope constants and the instrument serial number. A

Calibration Adjustment Data Sheet for each log entry

shall be available for inspection upon request by the field

inspector. Data Sheets shall be retained by the user for

a period of no less than 18 months following any

calibration adjustment. The Data Sheet must show:

Date of test and adjustment, serial number of the

instrument, calibration identification, the nature of the

adjustment, the unique identification number and source

of sample sets used, and, for each sample in the set,

reference values, initial instrument results (except in the

cases of instrument failure and repair), and instrument

results after calibration adjustment or instrument repair.

 Established error must be known. 1

(Amended 1995)








Fundamental Considerations

A-1

Appendix A

 Fundamental Considerations

Associated with the

 Enforcement of Handbook 44 Codes

1. Uniformity of Requirements

1.1. National Conference Codes. - Weights and measures

jurisdictions are urged to promulgate and adhere to the

National Conference codes, to the end that uniform

requirements may be in force throughout the country. This

action is recommended even though a particular jurisdiction

does not wholly agree with every detail of the National

Conference codes. Uniformity of specifications and

tolerances is an important factor in the manufacture of

commercial equipment. Deviations from standard designs

to meet the special demands of individual weights and

measures jurisdictions are expensive, and any increase in

costs of manufacture is, of course, passed on to the

purchaser of equipment. On the other hand, if designs can

be standardized by the manufacturer to conform to a single

set of technical requirements, production costs can be kept

down, to the ultimate advantage of the general public.

Moreover, it seems entirely logical that equipment that is

suitable for commercial use in the "specification" States

should be equally suitable for such use in other States.

Another consideration supporting the recommendation for

uniformity of requirements among weights and measures

jurisdictions is the cumulative and regenerative effect of the

widespread enforcement of a single standard of design and

performance. The enforcement effort in each jurisdiction

can then reinforce the enforcement effort in all other

jurisdictions. More effective regulatory control can be

realized with less individual effort under a system of

uniform requirements than under a system in which even

minor deviations from standard practice are introduced by

independent State action.

Since the National Conference codes represent the majority

opinion of a large and representative group of experienced

regulatory officials, and since these codes are recognized by

equipment manufacturers as their basic guide in the design

and construction of commercial weighing and measuring

equipment, the acceptance and promulgation of these codes

by each State are strongly recommended.

1.2. Form of Promulgation. - A convenient and very

effective form of promulgation already successfully used in

a considerable number of States is promulgation by citation

of National Institute of Standards and Technology

Handbook 44. It is especially helpful when the citation is

so made that, as amendments are adopted from time to time

by the National Conference on Weights and Measures,

these automatically go into effect in the State regulatory

authority. For example, the following form of promulgation

has been used successfully and is recommended for

consideration:

The specifications, tolerances, and other technical

requirements for weighing and measuring devices as

recommended by the National Conference on Weights

and Measures and published in the National Institute of

Standards and Technology Handbook 44, Specifications, Tolerances, and Other Technical

Requirements for Weighing and Measuring Devices,

and supplements thereto or revisions thereof, shall

apply to commercial weighing and measuring devices

in the State.

In some States, it is preferred to base technical requirements

upon specific action of the State legislature rather than upon

an act of promulgation by a State officer. The advantages

cited above may be obtained and may yet be surrounded by

adequate safeguards to insure proper freedom of action by

the State enforcing officer if the legislature adopts the

National Conference requirements by language somewhat

as follows:

The specifications, tolerances, and other technical

requirements for weighing and measuring devices as

recommended by the National Conference on Weights

and Measures shall be the specifications, tolerances,

and other technical requirements for weighing and

measuring devices of the State except insofar as

specifically modified, amended, or rejected by a

regulation issued by the State (insert title of enforcing

officer).
Fundamental Considerations

A-2

 See General Code, Section 1.10.; User Requirement G- 1

UR.4.3.

 The numerical values of the tolerances recommended by 2

the National Institute of Standards and Technology, for the

standards of length, mass, and capacity used by weights and

measures officials, may be obtained upon request from the

Office of Weights and Measures of the National Institute of

Standards and Technology.

2. Tolerances for Commercial Equipment

2.1. Acceptance and Maintenance Tolerances. - The

official tolerances prescribed by a weights and measures

jurisdiction for commercial equipment are the limits of

inaccuracy officially permissible within that jurisdiction. It

is recognized that errorless value or performance of 3.1. Adequacy. - Tests can be made properly only if,

mechanical equipment is unattainable. Tolerances are among other things, adequate testing apparatus is available.

established, therefore, to fix the range of inaccuracy within Testing apparatus may be considered adequate only when it

which equipment will be officially approved for commercial is properly designed for its intended use, when it is so

use. In the case of classes of equipment on which the constructed that it will retain its characteristics for a

magnitude of the errors of value or performance may be reasonable period under conditions of normal use, when it

expected to change as a result of use, two sets of tolerances is available in denominations appropriate for a proper

are established: acceptance tolerances and maintenance determination of the value or performance of the

tolerances. commercial equipment under test, and when it is accurately

Acceptance tolerances are applied to new or newly

reconditioned or adjusted equipment, and are smaller than

(usually one-half of) the maintenance tolerances.

Maintenance tolerances thus provide an additional range of

inaccuracy within which equipment will be approved on

subsequent tests, permitting a limited amount of

deterioration before the equipment will be officially

rejected for inaccuracy and before reconditioning or

adjustment will be required. In effect, there is assured a

reasonable period of use for equipment after it is placed in

service before reconditioning will be officially required.

The foregoing comments do not apply, of course, when

only a single set of tolerance values is established, as is the

case with equipment such as glass milk bottles and

graduates, which maintain their original accuracy regardless

of use, and measure-containers, which are used only once.

2.2. Theory of Tolerances. - Tolerance values are so fixed and measuring equipment be so established and maintained

that the permissible errors are sufficiently small that there is that the use of corrections is not necessary. Also, whenever

no serious injury to either the buyer or the seller of it can readily be done, it will be desirable to reduce the error

commodities, yet not so small as to make manufacturing or on a standard below the one-third point previously

maintenance costs of equipment disproportionately high. mentioned.

Obviously, the manufacturer must know what tolerances his

equipment is required to meet, so that he can manufacture

economically. His equipment must be good enough to

satisfy commercial needs, but should not be subject to such

stringent tolerance values as to make it unreasonably costly,

complicated, or delicate.

2.3. Tolerances and Adjustments. - Tolerances are

primarily accuracy criteria for use by the regulatory official.

However, when equipment is being adjusted for accuracy,

either initially or following repair or official rejection, the

objective should be to adjust as closely as practicable to

zero error. Equipment owners should not take advantage of

tolerances by deliberately adjusting their equipment to have

a value, or to give performance, at or close to the tolerance

limit. Nor should the repair or service personnel bring

equipment merely within tolerance range when it is possible

to adjust closer to zero error.1

3. Testing Apparatus

calibrated.

3.2. Tolerances for Standards. - The error in a standard 2

used by a weights and measures official should be known

and corrected for when the standard is used; or if the

standard is to be used without correction, its error should be

not greater than one-third of the smallest tolerance to be

applied when the standard is used. The reason for this is to

keep at a minimum the proportion of the tolerance on the

item tested that will be used up by the error of the standard.

Expressed differently, the reason is to give the item being

tested as nearly as practicable the full benefit of its own

tolerance.

Field testing operations are complicated to some degree

when corrections to standards are applied. Except for work

of relatively high precision, it is recommended that the

accuracy of standards used in testing commercial weighing

3.3. Accuracy of Standards. - Prior to the official use of

testing apparatus, its accuracy should invariably be verified.

Standards should be re-verified as often as circumstances
Fundamental Considerations

A-3

require. By their nature, metal volumetric standards are This is the way the official learns whether or not the design

more susceptible to damage in handling than are standards and construction of the device conform to the specification

of some other types. A standard should be re-calibrated requirements. But even a device of a type with which the

whenever damage is known or suspected to have occurred official is thoroughly familiar and that he has previously

or significant repairs have been made. In addition, found to meet specification requirements should not be

standards, particularly volumetric standards, should be re- accepted entirely on faith. Some part may have become

calibrated with sufficient frequency to affirm their damaged, or some detail of design may have been changed

continued accuracy, so that the official may always be in an by the manufacturer, or the owner or operator may have

unassailable position with respect to the accuracy of his removed an essential element or made an objectionable

testing apparatus. Secondary standards, such as special addition. Such conditions may be learned only by

fabric testing tapes, should be verified much more inspection. Some degree of inspection is therefore an

frequently than such basic standards as steel tapes or essential part of the official examination of every piece of

volumetric provers to demonstrate their constancy of value weighing or measuring equipment.

or performance.

Accurate and dependable results cannot be obtained with

faulty or inadequate standards. If either the service person

or official is poorly equipped, their results cannot be

expected to check consistently. Disagreements can be

avoided and the servicing of commercial equipment can be

expedited and improved if service persons and officials give

equal attention to the adequacy and maintenance of their

testing apparatus.

4. Inspection of Commercial Equipment

4.1. Inspection Versus Testing. - A distinction may be

made between the inspection and the testing of commercial

equipment that should be useful in differentiating between

the two principal groups of official requirements; i.e.,

specifications and performance requirements. Although the

term inspection is frequently loosely used to include

everything that the official has to do in connection with

commercial equipment, it is useful to limit the scope of that

term primarily to examinations made to determine

compliance with design, maintenance, and user

requirements. The term testing may then be limited to those

operations carried out to determine the accuracy of value or

performance of the equipment under examination by

comparison with the actual physical standards of the

official. These two terms will be used herein in the limited

senses defined.

4.2. Necessity for Inspection. - It is not enough merely to

determine that the errors of equipment do not exceed the

appropriate tolerances. Specification and user requirements

are as important as tolerance requirements and should be

enforced. Inspection is particularly important, and should

be carried out with unusual thoroughness whenever the

official examines a type of equipment not previously

encountered.

4.3. Specification Requirements. - A thorough knowledge

by the official of the specification requirements is a

prerequisite to competent inspection of equipment. The

inexperienced official should have his specifications before

him when making an inspection, and should check the

requirements one by one against the equipment itself.

Otherwise some important requirement may be overlooked.

As experience is gained, the official will become

progressively less dependent on the Handbook, until finally

observance of faulty conditions becomes almost automatic

and the time and effort required to do the inspecting are

reduced to a minimum. The printed specifications,

however, should always be available for reference to

refresh the official’s memory or to be displayed to support

his decisions, and they are an essential item of his kit.

Specification requirements for a particular class of

equipment are not all to be found in the separate code for

that class. The requirements of the General Code apply, in

general, to all classes of equipment, and these must always

be considered in combination with the requirements of the

appropriate separate code to arrive at the total of the

requirements applicable to a piece of commercial

equipment.

4.4. General Considerations. - The simpler the commercial device, the fewer are the specification requirements

affecting it, and the more easily and quickly can adequate

inspection be made. As mechanical complexity increases,

however, inspection becomes increasingly important and

more time consuming, because the opportunities for the

existence of faulty conditions are multiplied. It is on the

relatively complex device, too, that the official must be on

the alert to discover any modification that may have been

made by an operator that might adversely affect the proper

functioning of the device.

It is essential for the officials to familiarize themselves with

the design and operating characteristics of the devices that
Fundamental Considerations

A-4

See Sec. 1.10.; General Code and Appendix D. 3

Definitions.

he inspects and tests. Such knowledge can be obtained from any of the requirements for correct equipment is incorrect.

the catalogs and advertising literature of device Only equipment that is correct should be sealed and

manufacturers, from trained service persons and plant approved for commercial use.

engineers, from observation of the operations performed by

service persons when reconditioning equipment in the field,

and from a study of the devices themselves.

Inspection should include any auxiliary equipment and

general conditions external to the device that may affect its

performance characteristics. In order to prolong the life of

the equipment and forestall rejection, inspection should also

include observation of the general maintenance of the

device and of the proper functioning of all required

elements. The official should look for worn or weakened

mechanical parts, leaks in volumetric equipment, or

elements in need of cleaning.

4.5. Misuse of Equipment. - Inspection, coupled with

judicious inquiry, will sometimes disclose that equipment is

being improperly used, either through ignorance of the

proper method of operation or because some other method

is preferred by the operator. Equipment should be operated

only in the manner that is obviously indicated by its

construction or that is indicated by instructions on the

equipment, and operation in any other manner should be

prohibited.

4.6. Recommendations. - A comprehensive knowledge of

each installation will enable the official to make

constructive recommendations to the equipment owner

regarding proper maintenance of his weighing and measuring devices and the suitability of his equipment for the

purposes for which it is being used or for which it is

proposed that it be used. Such recommendations are always

in order and may be very helpful to an owner. The official

will, of course, carefully avoid partiality toward or against

equipment of specific makes, and will confine his

recommendations to points upon which he is qualified, by

knowledge and experience, to make suggestions of practical

merit.

4.7. Accurate and Correct Equipment. - Finally, the

weights and measures official is reminded that commercial

equipment may be accurate without being correct. A piece

of equipment is accurate when its performance or value

(that is, its indications, its deliveries, its recorded

representations, or its capacity or actual value, etc., as deter

mined by tests made with suitable standards) conforms to

the standard within the applicable tolerances and other

performance requirements. Equipment that fails so to

conform is inaccurate. A piece of equipment is correct

when in addition to being accurate, it meets all applicable

specification requirements. Equipment that fails to meet

3

5. Correction of Commercial Equipment

5.1. Adjustable Elements. - Many types of weighing and

measuring instruments are not susceptible of adjustment for

accuracy by means of adjustable elements. Linear

measures, liquid measures, graduates, measure-containers,

milk and lubricating-oil bottles, farm milk tanks, dry

measures, and some of the more simple types of scales are

in this category. Other types (for example, taximeters and

odometers and some metering devices) may be adjusted in

the field, but only by changing certain parts such as gears in

gear trains.

Some types, of which fabric-measuring devices and

cordage-measuring devices are examples, are not intended

to be adjusted in the field and require reconditioning in

shop or factory if inaccurate. Liquid-measuring devices and

most scales are equipped with adjustable elements, and

some vehicle-tank compartments have adjustable indicators.

Field adjustments may readily be made on such equipment.

In the discussion that follows, the principles pointed out

and the recommendations made apply to adjustments on any

commercial equipment, by whatever means accomplished.

5.2. When Corrections Should be Made. - One of the

primary duties of a weights and measures official is to

determine whether equipment is suitable for commercial

use. If a device conforms to all legal requirements, the

official "marks" or "seals" it to indicate approval. If it does

not conform to all official requirements, the official is

required to take action to ensure that the device is corrected

within a reasonable period of time. Devices with

performance errors that could result in serious economic

injury to either party in a transaction should be prohibited

from use immediately and not allowed to be returned to

service until necessary corrections have been made. The

official should consider the most appropriate action, based

on all available information and economic factors.

Some officials contend that it is justifiable for the official to

make minor corrections and adjustments if there is no

service agency nearby or if the owner or operator depends

on this single device and would be "out of business" if the

use of the device were prohibited until repairs could be

made.
Fundamental Considerations

A-5

Before adjustments are made at the request of the owner or situations, but seizure and destruction should be resorted to

the owner’s representative, the official should be confident only when clearly justified.

that the problem is not due to faulty installation or a

defective part, and that the adjustment will correct the On the other hand, rejection is clearly inappropriate for

problem. The official should never undertake major many items of measuring equipment. This is true for most

repairs, or even minor corrections, if services of commercial linear measures, many liquid and dry measures, and

agencies are readily available. The official should always graduates, measure-containers, milk bottles, lubricating-oil

be mindful of conflicts of interest before attempting to bottles, and some scales. When such equipment is

perform any services other than normal device examination "incorrect," it is either impractical or impossible to adjust or

and testing duties. repair it, and the official has no alternative to outright con-

(Amended 1995) demnation. When only a few such items are involved,

5.3. Gauging. - In the majority of cases, when the weights

and measures official tests commercial equipment, he is

verifying the accuracy of a value or the accuracy of the

performance as previously established either by himself or

by someone else. There are times, however, when the test

of the official is the initial test on the basis of which the

calibration of the device is first determined or its

performance first established. The most common example

of such gauging is in connection with vehicle tanks the

compartments of which are used as measures. Frequently

the official makes the first determination on the capacities

of the compartments of a vehicle tank, and his test results

are used to determine the proper settings of the compartment indicators for the exact compartment capacities

desired. Adjustments of the position of an indicator under

these circumstances are clearly not the kind of adjustments

discussed in the preceding paragraph.

6. Rejection of Commercial Equipment

6.1. Rejection and Condemnation. - The uniform

Weights and Measures Law contains a provision stating that

the director shall reject and order to be corrected such

physical weights and measures or devices found to be

incorrect. Weights and measures and devices that have

been rejected may be seized if not corrected within a

reasonable time or if used or disposed of in a manner not

specifically authorized. The director shall remove from

service and may seize weights and measures found to be

incorrect that are not capable of being made correct.

These broad powers should be used by the official with

discretion. The director should always keep in mind the

property rights of an equipment owner, and cooperate in

working out arrangements whereby an owner can realize at

least something from equipment that has been rejected. In

cases of doubt, the official should initially reject rather than

condemn outright. Destruction and confiscation of

equipment are harsh procedures. Power to seize and

destroy is necessary for adequate control of extreme

immediate destruction or confiscation is probably the best

procedure. If a considerable number of items are involved

(as, for example, a stock of measures in the hands of a

dealer or a large shipment of bottles), return of these to the

manufacturer for credit or replacement should ordinarily be

permitted provided that the official is assured that they will

not get into commercial use. In rare instances, confiscation

and destruction are justified as a method of control when

less harsh methods have failed.

In the case of incorrect mechanisms such as

fabric-measuring devices, taximeters, liquid-measuring

devices, and most scales, repair of the equipment is usually

possible, so rejection is the customary procedure. Seizure

may occasionally be justified, but in the large majority of

instances this should be unnecessary. Even in the case of

worn-out equipment, some salvage is usually possible, and

this should be permitted under proper controls.

(Amended 1995)

7. Tagging of Equipment

7.1. Rejected and Condemned. - It will ordinarily be

practicable to tag or mark as rejected each item of

equipment found to be incorrect and considered susceptible

of proper reconditioning. However, it can be considered

justifiable not to mark as rejected incorrect devices capable

of meeting acceptable performance requirements if they are

to be allowed to remain in service for a reasonable time

until minor problems are corrected since marks of rejection

may tend to be misleading about a device’s ability to

produce accurate measurements during the correction

period. The tagging of equipment as condemned, or with a

similar label to indicate that it is permanently out of service,

is not recommended if there is any other way in which the

equipment can definitely be put out of service. Equipment

that cannot successfully be repaired should be dismantled,

removed from the premises, or confiscated by the official

rather than merely being tagged as "condemned."

(Amended 1995)
Fundamental Considerations

A-6

7.2. Nonsealed and Noncommercial. - Rejection is not

appropriate if measuring equipment cannot be tested by the

official at the time of his regular visit–for example, when

there is no gasoline in the supply tank of a

gasoline-dispensing device. Some officials affix to such

equipment a nonsealed tag stating that the device has not

been tested and sealed and that it must not be used

commercially until it has been officially tested and approved. This is recommended whenever considerable time

will elapse before the device can be tested.

Where the official finds in the same establishment

equipment that is in commercial use and also equipment

suitable for commercial use that is not presently in service,

but which may be put into service at some future time, he

may treat the latter equipment in any of the following ways:

a. Test and approve the same as commercial equipment

in use.

b. Refrain from testing it and remove it from the

premises to preclude its use for commercial

purposes.

c. Mark the equipment nonsealed.

Where the official finds commercial equipment and

noncommercial equipment installed or used in close

proximity, he may treat the noncommercial equipment in

any of the following ways:

a. Test and approve the same as commercial equipment.

b. Physically separate the two groups of equipment so

that misuse of the noncommercial equipment will be

prevented.

c. Tag it to show that it has not been officially tested

and is not to be used commercially.

8. Records of Equipment

8.1. The official will be well advised to keep careful samples selected at random from new stocks.

records of equipment that is rejected, so that he may follow

up to insure that the necessary repairs have been made. As Another exception to the general rule for sealing approved

soon as practicable following completion of repairs, the equipment is found in certain very small weights whose size

equipment should be retested. Complete records should precludes satisfactory stamping with a steel die.

also be kept of equipment that has been tagged as nonsealed

or noncommercial. Such records may be invaluable should

it subsequently become necessary to take disciplinary steps

because of improper use of such equipment.

9. Sealing of Equipment

9.1. Types of Seals and Their Locations. - Most weights

and measures jurisdictions require that all equipment

officially approved for commercial use (with certain

exceptions to be pointed out later) be suitably marked or

sealed to show approval. This is done primarily for the

benefit of the public to show that such equipment has been

officially examined and approved. The seal of approval

should be as conspicuous as circumstances permit and

should be of such a character and so applied that it will be

reasonably permanent. Uniformity of position of the seal

on similar types of equipment is also desirable as a further

aid to the public.

The official will need more than one form of seal to meet

the requirements of different kinds of equipment. Good

quality, weather-resistant, water-adhesive, or

pressure-sensitive seals or decalcomania seals are

recommended for fabric-measuring devices,

liquid-measuring devices, taximeters, and most scales,

because of their permanence and good appearance. Steel

stamps are most suitable for liquid and dry measures, for

some types of linear measures, and for weights. An etched

seal, applied with suitable etching ink, is excellent for steel

tapes, and greatly preferable to a seal applied with a steel

stamp. The only practicable seal for a graduate is one

marked with a diamond or carbide pencil, or one etched

with glass-marking ink. For a vehicle tank, the official may

wish to devise a relatively large seal, perhaps of metal, with

provision for stamping data relative to compartment

capacities, the whole to be welded or otherwise permanently

attached to the shell of the tank. In general, the

lead-and-wire seal is not suitable as an approval seal.

9.2. Exceptions. - Commercial equipment such as

measure-containers, milk bottles, and lubricating-oil bottles

are not tested individually because of the time element

involved. Because manufacturing processes for these items

are closely controlled, an essentially uniform product is

produced by each manufacturer. The official normally tests

samples of these items prior to their sale within his

jurisdiction and subsequently makes spot checks by testing

 10. Rounding Off Numerical Values

10.1. Definition. - To round off or round a numerical value

is to change the value of recorded digits to some other value
Fundamental Considerations

A-7

considered more desirable for the purpose at hand by significant figure of the final result is to be, the entire series

dropping or changing certain figures. For example, if a of such figures must be rounded off in one step and not in

computed, observed, or accumulated value is 4738, this can two or more successive rounding steps. [Expressed

be rounded off to the nearest thousand, hundred, or ten, as differently, when two or more such figures are involved,

desired. Such rounded-off values would be, respectively, these are not to be rounded off individually, but are to be

5000, 4700, 4740. Similarly, a value such as 47.382 can be rounded off as a group.] Thus, when rounding off 47.3499

rounded off to two decimal places, to one decimal place, or to the first decimal place, the result becomes 47.3. In

to the units place. The rounded-off figures in this example arriving at this result, the figures "499" are treated as a

would be, respectively, 47.38, 47.4, 47. group. Since the 4 next beyond the last figure to be retained

10.2. General Rules. - The general rules for rounding off

may be stated briefly as follows:

a. When the figure next beyond the last figure or place

to be retained is less than 5, the figure in the last 10.3. Rules for Reading of Indications. - An important

place retained is to be kept unchanged. When aspect of rounding off values is the application of these

rounding off 4738 to the nearest hundred, it is noted rules to the reading of indications of an inthat the figure 3 (next beyond the last figure to be dicator-and-graduated-scale combination (where the

retained) is less than 5. Thus the rounded-off value majority of the indications may be expected to lie somewould be 4700. Likewise, 47.382 rounded to two where between two graduations) if it is desired to read or

decimal places becomes 47.38. record values only to the nearest graduation. Consider a

b. When the figure next beyond the last figure or place indicator is between two graduations but is closer to one

to be retained is greater than 5, the figure in the last graduation than it is to the other adjacent graduation, the

place retained is to be increased by 1. When value of the closer graduation is the one to be read or

rounding off 4738 to the nearest thousand, it is recorded.

noted that the figure 7 (next beyond the last figure to

be retained) is greater than 5. Thus the rounded-off In the case where, as nearly as can be determined, the

value would be 5000. Likewise, 47.382 rounded to indicator is midway between two graduations, the

one decimal place becomes 47.4. odd-and-even rule is invoked, and the value to be read or rec. When the figure next beyond the last figure to be example, if the indicator lies exactly midway between two

retained is 5 followed by any figures other than graduations having values of 471 and 472, respectively, the

zero(s), treat as in (b) above; that is, the figure in indication should be read or recorded as 472, this being an

the last place retained is to be increased by 1. When even value. If midway between graduations having values

rounding off 4501 to the nearest thousand, 1 is of 474 and 475, the even value 474 should be read or

added to the thousands figure and the result recorded. Similarly, if the two graduations involved had

becomes 5000. values of 470 and 475, the even value of 470 should be read

d. When the figure next beyond the last figure to be

retained is 5 and there are no figures, or only zeros, A special case not covered by the foregoing paragraph is

beyond this 5, the figure in the last place to be that of a graduated scale in which successive graduations

retained is to be left unchanged if it is even (0, 2, 4, are numbered by two’s, all graduations thus having even

6, or 8) and is to be increased by 1 if it is odd (1, 3, values; for example, 470, 472, 474, etc. When, in this case,

5, 7, or 9). This is the odd and even rule, and may an indication lies midway between two graduations, the

be stated as follows: "If odd, then add." Thus, recommended procedure is to depart from the practice of

rounding off to the first decimal place, 47.25 would reading or recording only to the value of the nearest

become 47.2 and 47.15 would become 47.2. Also, graduation and to read or record the intermediate odd value.

rounded to the nearest thousand, 4500 would For example, an indication midway between 470 and 472

become 4000 and 1500 would become 2000. should be read as 471.

It is important to remember that, when there are two or

more figures to the right of the place where the last

is less than 5, the "499" is dropped (see subparagraph (a)

above). It would be incorrect to round off these figures

successively to the left so that 47.3499 would become

47.350 and then 47.35 and then 47.4.

vertical graduated scale and an indicator. Obviously, if the

corded is that of the graduation whose value is even. For

or recorded.

10.4. Rules for Common Fractions. - When applying the

rounding-off rules to common fractions, the principles are
Fundamental Considerations

A-8

to be applied to the numerators of the fractions that have, if

necessary, been reduced to a common denominator. The

principle of "5s" is changed to the one-half principle; that

is, add if more than one-half, drop if less than one-half, and

apply the odd-and even rule if exactly one-half.

For example, a series of values might be 1-1/32, 1-2/32,

1-3/32, 1-4/32, 1-5/32, 1-6/32, 1-7/32, 1-8/32, 1-9/32.

Assume that these values are to be rounded off to the

nearest eighth (4/32). Then,

1-1/32 becomes 1. (1/32 is less than half of 4/32 and

accordingly is dropped.)

1-2/32 becomes 1. (2/32 is exactly one-half of 4/32; it is

dropped because it is round (down) to the "even" eighth,

which in this instance is 0/8.)

1-3/32 becomes 1-4/32 or 1-1/8. (3/32 is more than half

of 4/32,and accordingly is rounded "up" to 4/32 or 1/8).

1-4/32 remains unchanged, being an exact eighth (1-1/8).

1-5/32 becomes 1-4/32 or 1-1/8. (5/32 is 1/32 more than

an exact 1/8; 1/32 is less than half of 4/32 and

accordingly is dropped.)

1-6/32 becomes 1-2/8 or 1-1/4. (6/32 is 2/32 more than

an exact 1/8; 2/32 is exactly one-half of 4/32, and the

final fraction is rounded (up) to the "even" eighth, which

in this instance is 2/8.)

1-7/32 becomes 1-2/8 or 1-1/4. (7/32 is 3/32 more than

an exact 1/8; 3/32 is more than one-half of 4/32 and

accordingly the final fraction is rounded (up) to 2/8 or

1/4.)

1-8/32 remains unchanged, being an exact eighth (1-2/8

or 1-1/4.)

1-9/32 becomes 1-2/8 or 1-1/4. (9/32 is 1/32 more than

an exact 1/8; 1/32 is less than half of 4/32 and

accordingly is dropped.)
Units and Systems

B-1

Appendix B

Units and Systems of Measurement 

Their Origin, Development, and Present Status

1. Introduction

The National Institute of Standards and Technology (NIST)

(formerly the National Bureau of Standards) was established

by Act of Congress in 1901 to serve as a national scientific

laboratory in the physical sciences, and to provide

fundamental measurement standards for science and

industry. In carrying out these related functions the Institute

conducts research and development in many fields of

physics, mathematics, chemistry, and engineering. At the

time of its founding, the Institute had custody of two

primary standards–the meter bar for length and the kilogram

cylinder for mass. With the phenomenal growth of science

and technology over the past half century, the Institute has

become a major research institution concerned not only with

everyday weights and measures, but also with hundreds of

other scientific and engineering standards that are necessary

to the industrial progress of the Nation. Nevertheless, the

country still looks to NIST for information on the units of

measurement, particularly their definitions and equivalents.

The subject of measurement systems and units can be treated

from several different standpoints. Scientists and engineers

are interested in the methods by which precision

measurements are made. State weights and measures

officials are concerned with laws and regulations that assure

equity in the marketplace, protect public health and safety,

and with methods for verifying commercial weighing and

measuring devices. But a vastly larger group of people is

interested in some general knowledge of the origin and

development of measurement systems, of the present status

of units and standards, and of miscellaneous facts that will

be useful in everyday life. This material has been prepared

to supply that information on measurement systems and

units that experience has shown to be the common subject of

inquiry.

2. Units and Systems of Measurement

The expression "weights and measures" is often used to

refer to measurements of length, mass, and capacity or

volume, thus excluding such quantities as electrical and time

measurements and thermometry. This section on units and

measurement systems presents some fundamental

information to clarify the concepts of this subject and to

eliminate erroneous and misleading use of terms.

It is essential that the distinction between the terms "units"

and "standards" be established and kept in mind.

A unit is a special quantity in terms of which other

quantities are expressed. In general, a unit is fixed by

definition and is independent of such physical conditions as

temperature. Examples: the meter, the liter, the gram, the

yard, the pound, the gallon.

A standard is a physical realization or representation of a

unit. In general, it is not entirely independent of physical

conditions, and it is a representation of the unit only under

specified conditions. For example, a meter standard has a

length of one meter when at some definite temperature and

supported in a certain manner. If supported in a different

manner, it might have to be at a different temperature to

have a length of one meter.

2.1. Origin and Early History of Units and Standards.

2.1.1. General Survey of Early History of

Measurement Systems. - Weights and measures were

among the earliest tools invented by man. Primitive

societies needed rudimentary measures for many tasks:

constructing dwellings of an appropriate size and shape,

fashioning clothing, or bartering food or raw materials.

Man understandably turned first to parts of the body and

the natural surroundings for measuring instruments. Early

Babylonian and Egyptian records and the Bible indicate

that length was first measured with the forearm, hand, or

finger and that time was measured by the periods of the

sun, moon, and other heavenly bodies. When it was

necessary to compare the capacities of containers such as

gourds or clay or metal vessels, they were filled with

plant seeds which were then counted to measure the

volumes. When means for weighing were invented, seeds

and stones served as standards. For instance, the "carat,"

still used as a unit for gems, was derived from the carob

seed.
Units and Systems

B-2

Our present knowledge of early weights and measures cubit. In any case, the Greeks and Romans inherited the

comes from many sources. Archaeologists have recovered foot from the Egyptians. The Roman foot was divided

some rather early standards and preserved in museums. into both 12 unciae (inches) and 16 digits. The Romans

The comparison of the dimensions of buildings with the also introduced the mile of 1000 paces or double steps,

descriptions of contemporary writers is another source of the pace being equal to five Roman feet. The Roman

information. An interesting example of this is the mile of 5000 feet was introduced into England during the

comparison of the dimensions of the Greek Parthenon occupation. Queen Elizabeth, who reigned from 1558 to

with the description given by Plutarch from which a fairly 1603, changed, by statute, the mile to 5280 feet or 8

accurate idea of the size of the Attic foot is obtained. In furlongs, a furlong being 40 rods of 5-1/2 yards each.

some cases, we have only plausible theories and we must

sometimes select the interpretation to be given to the The introduction of the yard as a unit of length came

evidence. later, but its origin is not definitely known. Some believe

For example, does the fact that the length of the originated from cubic measure. Whatever its origin, the

double-cubit of early Babylonia was equal (within two early yard was divided by the binary method into 2, 4, 8,

parts per thousand) to the length of the seconds pendulum and 16 parts called the half-yard, span, finger, and nail.

at Babylon suggest a scientific knowledge of the The association of the yard with the "gird" or

pendulum at a very early date, or do we merely have a circumference of a person’s waist or with the distance

curious coincidence? By studying the evidence given by from the tip of the nose to the end of the thumb of Henry

all available sources, and by correlating the relevant facts, I are probably standardizing actions, since several yards

we obtain some idea of the origin and development of the were in use in Great Britain.

units. We find that they have changed more or less

gradually with the passing of time in a complex manner The point, which is a unit for measuring print type, is

because of a great variety of modifying influences. We recent. It originated with Pierre Simon Fournier in 1737.

find the units modified and grouped into measurement It was modified and developed by the Didot brothers,

systems: The Babylonian system, the Egyptian system, Francois Ambroise and Pierre Francois, in 1755. The

the Phileterian system of the Ptolemaic age, the Olympic point was first used in the United States in 1878 by a

system of Greece, the Roman system, and the British Chicago type foundry (Marder, Luse, and Company). 

system, to mention only a few. Since 1886, a point has been exactly 0.3514598

2.1.2. Origin and Development of Some Common

Customary Units. - The origin and development of units

of measurement has been investigated in considerable

detail and a number of books have been written on the

subject. It is only possible to give here, somewhat

sketchily, the story about a few units.

Units of length: The cubit was the first recorded unit

used by ancient peoples to measure length. There were

several cubits of different magnitudes that were used.

The common cubit was the length of the forearm from the

elbow to the tip of the middle finger. It was divided into

the span of the hand (one-half cubit), the palm or width of

the hand (one sixth), and the digit or width of a finger

(one twenty-fourth). The Royal or Sacred Cubit, which

was 7 palms or 28 digits long, was used in constructing

buildings and monuments and in surveying. The inch,

foot, and yard evolved from these units through a

complicated transformation not yet fully understood.

Some believe they evolved from cubic measures; others

believe they were simple proportions or multiples of the

the origin was the double cubit, others believe that it

millimeters, or about 1/72 inch. 

Units of mass: The grain was the earliest unit of mass

and is the smallest unit in the apothecary, avoirdupois,

Tower, and Troy systems. The early unit was a grain of

wheat or barleycorn used to weigh the precious metals

silver and gold. Larger units preserved in stone standards

were developed that were used as both units of mass and

of monetary currency. The pound was derived from the

mina used by ancient civilizations. A smaller unit was

the shekel, and a larger unit was the talent. The

magnitude of these units varied from place to place. The

Babylonians and Sumerians had a system in which there

were 60 shekels in a mina and 60 minas in a talent. The

Roman talent consisted of 100 libra (pound) which were

smaller in magnitude than the mina. The Troy pound

used in England and the United States for monetary

purposes, like the Roman pound, was divided into

12 ounces, but the Roman uncia (ounce) was smaller.

The carat is a unit for measuring gemstones that had its

origin in the carob seed, which later was standardized at

1/144 ounce and then 0.2 gram.
Units and Systems

B-3

Goods of commerce were originally traded by number or to as the modern metric system.

volume. When weighing of goods began, units of mass

based on a volume of grain or water were developed. For The adoption of the metric system in France was slow,

example, the talent in some places was approximately but its desirability as an international system was

equal to the mass of one cubic foot of water. Was this a recognized by geodesists and others. On May 20, 1875,

coincidence or by design? The diverse magnitudes of an international treaty known as the International Metric

units having the same name, which still appear today in Convention or the Treaty of the Meter was signed by 17

our dry and liquid measures, could have arisen from the countries including the United States. This treaty

various commodities traded. The larger avoirdupois established the following organizations to conduct

pound for goods of commerce might have been based on international activities relating to a uniform system for

volume of water which has a higher bulk density than measurements:

grain. For example, the Egyptian hon was a volume unit

about 11 percent larger than a cubic palm and 1) The General Conference on Weights and Measures

corresponded to one mina of water. It was almost (French initials: CGPM), an intergovernmental

identical in volume to the present U.S. pint. conference of official delegates of member nations

The stone, quarter, hundredweight, and ton were larger

units of mass used in Great Britain. Today only the stone 2) The International Committee of Weights and

continues in customary use for measuring personal body Measures (French initials: CIPM), consisting of

weight. The present stone is 14 pounds, but an earlier selected scientists and metrologists, which prepares

unit appears to have been 16 pounds. The other units and executes the decisions of the CGPM and is

were multiples of 2, 8, and 160 times the stone, or 28, responsible for the supervision of the International

112, and 2240 pounds, respectively. The hundredweight Bureau of Weights and Measures;

was approximately equal to two talents. In the United

States the ton of 2240 pounds is called the “long ton.” 3) The International Bureau of Weights and Measures

The “short ton” is equal to 2000 pounds. (French initials: BIPM), a permanent laboratory and

Units of time and angle: We can trace the division of the which include the establishment of the basic

circle into 360 degrees and the day into hours, minutes, standards and scales of the principal physical

and seconds to the Babylonians who had a sexagesimal quantities and maintenance of the international

system of numbers. The 360 degrees may have been prototype standards.

related to a year of 360 days.

2.2. The Metric System.

2.2.1. Definition, Origin, and Development. - Metric

systems of units have evolved since the adoption of the

first well defined system in France in 1791. During this

evolution the use of these systems spread throughout the

world, first to the non-English speaking countries, and

more recently to the English speaking countries. The first

metric system was based on the centimeter, gram, and

second (cgs) and these units were particularly convenient

in science and technology. Later metric systems were

based on the meter, kilogram, and second (mks) to 2.2.2. International System of Units. - At the end of

improve the value of the units for practical applications. World War II, a number of different systems of

The present metric system is the International System of measurement still existed throughout the world. Some of

Units (SI). It is also based on the meter, kilogram and these systems were variations of the metric system, and

second as well as additional base units for temperature, others were based on the customary inch-pound system

electric current, luminous intensity, and amount of of the English-speaking countries. It was recognized that

substance. the International System of Units is referred

and the supreme authority for all actions;

world center of scientific metrology, the activities of

The National Institute of Standards and Technology

provides official United States representation in these

organizations. The CGPM, the CIPM, and the BIPM

have been major factors in the continuing refinement of

the metric system on a scientific basis and in the

evolution of the International System of Units. 

Multiples and submultiples of metric units are related by

powers of ten. This relationship is compatible with the

decimal system of numbers and it contributes greatly to

the convenience of metric units.
Units and Systems

B-4

additional steps were needed to promote a worldwide by light in a vacuum during an interval of 1/299 792 458

measurement system. As a result the 9th GCPM, in 1948, of a second. 

asked the ICPM to conduct an international study of the

measurement needs of the scientific, technical, and The kilogram, originally defined as the mass of one cubic

educational communities. Based on the findings of this decimeter of water at the temperature of maximum

study, the 10th General Conference in 1954 decided that density, was known as the Kilogram of the Archives. It

an international system should be derived from six base was replaced after the International Metric Convention in

units to provide for the measurement of temperature and 1875 by the International Prototype Kilogram which

optical radiation in addition to mechanical and became the unit of mass without reference to the mass of

electromagnetic quantities. The six base units a cubic decimeter of water or to the Kilogram of the

recommended were the meter, kilogram, second, ampere, Archives. Each country that subscribed to the

Kelvin degree (later renamed the kelvin), and the candela. International Metric Convention was assigned one or

In 1960, the 11th General Conference of Weights and known as National Prototype Meters and Kilograms.

Measures named the system based on the six base

quantities the International System of Units, abbreviated The liter is a unit of capacity or volume. In 1964, the

SI from the French name: Le Système International 12th GCPM redefined the liter as being one cubic

d'Unités. The SI metric system is now either obligatory decimeter. By its previous definition -- the volume

or permissible throughout the world. occupied, under standard conditions, by a quantity of

2.2.3. Units and Standards of the Metric System. - In

the early metric system there were two fundamental or

base units, the meter and the kilogram, for length and

mass. The other units of length and mass, and all units of

area, volume, and compound units such as density were

derived from these two fundamental units.

The meter was originally intended to be one ten-millionth

part of a meridional quadrant of the earth. The Meter of

the Archives, the platinum length standard which was the

standard for most of the 19th century, at first was

supposed to be exactly this fractional part of the quadrant.

 More refined measurements over the earth's surface

showed that this supposition was not correct. In 1889, a

new international metric standard of length, the

International Prototype Meter, a graduated line standard

of platinum-iridium, was selected from a group of bars

because precise measurements found it to have the same

length as the Meter of the Archives. The meter was then

defined as the distance, under specified conditions,

between the lines on the International Prototype Meter 2.2.4. International Bureau of Weights and

without reference to any measurements of the earth or to Measures. - The International Bureau of Weights and

the Meter of the Archives, which it superseded. Measures (BIPM) was established at Sèvres, a suburb of

Advances in science and technology have made it Paris, France, by the International Metric Convention of

possible to improve the definition of the meter and reduce May 20, 1875. The BIPM maintains the International

the uncertainties associated with artifacts. From 1960 to Prototype Kilogram, many secondary standards, and

1983, the meter was defined as the length equal to equipment for comparing standards and making precision

1 650 763.73 wavelengths in a vacuum of the radiation measurements. The Bureau, funded by assessment of the

corresponding to the transition between the specified signatory governments, is truly international. In recent

energy levels of the krypton 86 atom. Since 1983 the years the scope of the work at the Bureau has been

meter has been defined as the length of the path traveled considerably broadened. It now carries on researches in

more copies of the international standards; these are

pure water having a mass of one kilogram -- the liter was

larger than the cubic decimeter by 28 parts per 1 000 000.

Except for determinations of high precision, this

difference is so small as to be of no consequence.

The modern metric system (SI) includes two classes of

units:

base units for length, mass, time, temperature,

electric current, luminous intensity, and amount

of substance; and

derived units for all other quantities (e.g., work,

force, power) expressed in terms of the seven

base units.

For details, see NIST Special Publication 330 (2001),

The International System of Units (SI) and NIST Special

Publication 811(1995), Guide for the Use of the

International System of Units.
Units and Systems

B-5

the fields of electricity, photometry and radiometry, interpreted or modified for the United States by the

ionizing radiations, and time and frequency besides its Secretary of Commerce (sec. 4(4)- Pub. L. 94-168; sec.

work in mass, length, and thermometry. 403(a)(3)- Pub. L. 93-380). The Secretary has delegated

2.2.5. Status of the Metric System in the United

States. - The use of the metric system in this country was

legalized by Act of Congress in 1866, but was not made

obligatory then or since.

Following the signing of the Convention of the Meter in

1875, the United States acquired national prototype

standards for the meter and the kilogram. U.S. Prototype

Kilogram No. 20 continues to be the primary standard for

mass in the United States. It is recalibrated from time to

time at the BIPM. The prototype meter has been replaced

by modern stabilized lasers following the most recent

definition of the meter.

From 1893 until 1959, the yard was defined as equal

exactly to 3600/3937 meter. In 1959, a small change was

made in the definition of the yard to resolve discrepancies

both in this country and abroad. Since 1959, we define

the yard as equal exactly to 0.9144 meter; the new yard is

shorter than the old yard by exactly two parts in a

million. At the same time, it was decided that any data

expressed in feet derived from geodetic surveys within

the United States would continue to bear the relationship

as defined in 1893 (one foot equals 1200/3937 meter).

We call this foot the U.S. Survey Foot, while the foot

defined in 1959 is called the International Foot.

Measurements expressed in U.S. statute miles, survey

feet, rods, chains, links, or the squares thereof, and acres

should be converted to the corresponding metric values

by using pre-1959 conversion factors if more than five

significant figure accuracy is required.

Since 1970, actions have been taken to encourage the use

of metric units of measurement in the United States. A

brief summary of actions by Congress is provided below

as reported in the Federal Register Notice dated July 28,

1998.

Section 403 of Public Law 93-380, the Education Measurement. - In the past, the customary system of

Amendment of 1974, states that it is the policy of the weights and measures in the British Commonwealth

United States to encourage educational agencies and countries and that in the United States were very similar;

institutions to prepare students to use the metric system of however, the SI metric system is now the official system of

measurement as part of the regular education program. units in the United Kingdom, while the customary units are

Under both this act and the Metric Conversion Act of still predominantly used in the United States. Because

1975, the "metric system of measurement" is defined as references to the units of the old British customary system

the International System of Units as established in 1960 are still found, the following discussion describes the

by the General Conference on Weights and Measures and

authority under these subsections to the Director of the

National Institute of Standards and Technology.

Section 5164 of Public Law 100-418, the Omnibus Trade

and Competitiveness Act of 1988, amends Public Law

94-168, The Metric Conversion Act of 1975. In

particular, section 3 Metric Conversion Act is amended

to read as follows:

"Sec. 3. It is therefore the declared policy of the

United States–

(1) to designate the metric system of measurement as

the preferred system of weights and measures for

United States trade and commerce;

(2) to require that each Federal agency, by a date

certain and to the extent economically feasible by

the end of the fiscal year 1992, use the metric system

of measurement in its procurements, grants, and

other business-related activities, except to the extent

that such use is impractical or is likely to cause

significant inefficiencies or loss of markets to U.S.

firms, such as when foreign competitors are

producing competing products in non-metric units;

(3) to seek ways to increase understanding of the

metric system of measurement through educational

information and guidance and in Government

publications; and

(4) to permit the continued use of traditional systems

of weights and measures in nonbusiness activities."

The Code of Federal Regulations makes the use of metric

units mandatory for agencies of the Federal Government.

(Federal Register, Vol. 56, No. 23, page 160, January 2,

1991.)

2.3. British and United States Systems of
Units and Systems

B-6

differences between the U.S. and British customary systems 1 U.S. fluid ounce = 1.041 British fluid ounces

of units. 1 British fluid ounce = 0.961 U.S. fluid ounce

After 1959, the U.S. and the British inch were defined 1 British Imperial gallon = 1.201 U.S. gallons

identically for scientific work and were identical in

commercial usage. A similar situation existed for the U.S. Among other differences between the customary British and

and the British pound, and many relationships, such as 12 the United States measurement systems, we should note that

inches = 1 foot, 3 feet = 1 yard, and 1760 yards = 1 they abolished the use of the troy pound in England January

international mile, were the same in both countries; but there 6, 1879, they retained only the troy ounce and its

were some very important differences. subdivisions, whereas the troy pound is still legal in the

In the first place, the U.S. customary bushel and the U.S. mention again the common use, for body weight, in

gallon, and their subdivisions differed from the England of the stone of 14 pounds, this being a unit now

corresponding British Imperial units. Also the British ton is unused in the United States, although its influence was

2240 pounds, whereas the ton generally used in the United shown in the practice until World War II of selling flour by

States is the short ton of 2000 pounds. The American the barrel of 196 pounds (14 stone). In the apothecary

colonists adopted the English wine gallon of 231 cubic system of liquid measure the British add a unit, the fluid

inches. The English of that period used this wine gallon and scruple, equal to one third of a fluid drachm (spelled dram

they also had another gallon, the ale gallon of 282 cubic in the United States) between their minim and their fluid

inches. In 1824, the British abandoned these two gallons drachm. In the United States, the general practice now is to

when they adopted the British Imperial gallon, which they sell dry commodities, such as fruits and vegetables, by their

defined as the volume of 10 pounds of water, at a mass.

temperature of 62 EF, which, by calculation, is equivalent to

277.42 cubic inches. At the same time, they redefined the

bushel as 8 gallons.

In the customary British system the units of dry measure are

the same as those of liquid measure. In the United States

these two are not the same, the gallon and its subdivisions

are used in the measurement of liquids; the bushel, with its

subdivisions, is used in the measurement of certain dry

commodities. The U.S. gallon is divided into four liquid

quarts and the U.S. bushel into 32 dry quarts. All the units

of capacity or volume mentioned thus far are larger in the

customary British system than in the U.S. system. But the

British fluid ounce is smaller than the U.S. fluid ounce,

because the British quart is divided into 40 fluid ounces

whereas the U.S. quart is divided into 32 fluid ounces.

From this we see that in the customary British system an

avoirdupois ounce of water at 62 EF has a volume of one

fluid ounce, because 10 pounds is equivalent to 160

avoirdupois ounces, and 1 gallon is equivalent to 4 quarts,

or 160 fluid ounces. This convenient relation does not exist

in the U.S. system because a U.S. gallon of water at 62 EF

weighs about 8-1/3 pounds, or 133-1/3 avoirdupois ounces,

and the U.S. gallon is equivalent to 4 x 32, or 128 fluid

ounces.

1 U.S. gallon = 0.833 British Imperial gallon

United States, although it is not now greatly used. We can

2.4. Subdivision of Units. - In general, units are

subdivided by one of three methods: (a) decimal, into

tenths; (b) duodecimal, into twelfths; or (c) binary, into

halves (twos). Usually the subdivision is continued by

using the same method. Each method has its advantages for

certain purposes, and it cannot properly be said that any one

method is "best" unless the use to which the unit and its

subdivisions are to be put is known.

For example, if we are concerned only with measurements

of length to moderate precision, it is convenient to measure

and to express these lengths in feet, inches, and binary

fractions of an inch, thus 9 feet 4-3/8 inches. However, if

these lengths are to be subsequently used to calculate area

or volume, that method of subdivision at once becomes

extremely inconvenient. For that reason, civil engineers,

who are concerned with areas of land, volumes of cuts, fills,

excavations, etc., instead of dividing the foot into inches

and binary subdivisions of the inch, divide it decimally; that

is, into tenths, hundredths, and thousandths of a foot.

The method of subdivision of a unit is thus largely made

based on convenience to the user. The fact that units have

commonly been subdivided into certain subunits for

centuries does not preclude their also having another mode

of subdivision in some frequently used cases where

convenience indicates the value of such other method.

Thus, while we usually subdivide the gallon into quarts and
Units and Systems

B-7

pints, most gasoline-measuring pumps, of the system of fractions from the Romans, (c) the decimal

price-computing type, are graduated to show tenths of a system from the Chinese and Egyptians, some developments

gallon. having been made by the Hindus, and (d) the sexagesimal

Although the mile has for centuries been divided into rods, units of angle and of time, from the ancient Babylonians.

yards, feet, and inches, the odometer part of an automobile The use of decimal numbers in measurements is becoming

speedometer shows tenths of a mile. Although we divide the standard practice.

our dollar into 100 parts, we habitually use and speak of

halves and quarters. An illustration of rather complex

subdividing is found on the scales used by draftsmen. These

scales are of two types: (a) architects, which are commonly

graduated with scales in which 3/32, 3/16, 1/8, 1/4, 3/8, 1⁄2,

3/4, 1, 1-1/2, and 3 inches, respectively, represent 1 foot full

scale, and also having a scale graduated in the usual manner

to 1/16 inch; and (b) engineers, which are commonly

subdivided to 10, 20, 30, 40, 50, and 60 parts to the inch.

The dictum of convenience applies not only to subdivisions

of a unit but also to multiples of a unit. Land elevations

above sea level are given in feet although the height may be

several miles; the height of aircraft above sea level as given

by an altimeter is likewise given in feet, no matter how high

it may be.

On the other hand, machinists, toolmakers, gauge makers,

scientists, and others who are engaged in precision

measurements of relatively small distances, even though

concerned with measurements of length only, find it

convenient to use the inch, instead of the tenth of a foot, but

to divide the inch decimally to tenths, hundredths,

thousandths, etc., even down to millionths of an inch.

Verniers, micrometers, and other precision measuring

instruments are usually graduated in this manner. Machinist

scales are commonly graduated decimally along one edge

and are also graduated along another edge to binary fractions

as small as 1/64 inch. The scales with binary fractions are

used only for relatively rough measurements.

It is seldom convenient or advisable to use binary

subdivisions of the inch that are smaller than 1/64. In fact,

1/32-, 1/16-, or 1/8-inch subdivisions are usually preferable

for use on a scale to be read with the unaided eye.

2.5. Arithmetical Systems of Numbers. - The subdivision

of units of measurement is closely associated with

arithmetical systems of numbers. The systems of units used

in this country for commercial and scientific work, having

many origins as has already been shown, naturally show

traces of the various number systems associated with their

origins and developments. Thus, (a) the binary subdivision

has come down to us from the Hindus, (b) the duodecimal

system (division by 60) now illustrated in the subdivision of

3. Standards of Length, Mass, and Capacity

 or Volume

3.1. Standards of Length. - The meter, which is defined

in terms of the speed of light in a vacuum, is the unit on

which all length measurements are based. 
Units and Systems

B-8

 See Federal Register for July 1, 1959. See also next to last 1

paragraph of 2.2.5.

 See Federal Register for July 1, 1959. 2

The yard is defined as follows: The avoirdupois pound is defined in terms of the kilogram 1

1 yard = 0.914 4 meter

and the inch is exactly equal to 25.4 millimeters.

3.1.1. Calibration of Length Standards. - NIST

calibrates standards of length including meter bars, yard

bars, miscellaneous precision line standards, steel tapes,

invar geodetic tapes, precision gauge blocks,

micrometers, and limit gauges. It also measures the linear

dimensions of miscellaneous apparatus such as

penetration needles, cement sieves, and hemacytometer

chambers. In general, NIST accepts for calibration only

apparatus of such material, design, and construction as to 3.2.1. Mass and Weight. - The mass of a body is a

ensure accuracy and permanence sufficient to justify measure of its inertial property or how much matter it

calibration by the Institute. NIST performs calibrations contains. The weight of a body is a measure of the force

in accordance with fee schedules, copies of which may be exerted on it by gravity or the force needed to support it.

obtained from NIST. Gravity on earth gives a body a downward acceleration

NIST does not calibrate carpenters’ rules, machinist used as a synonym for mass as in weights and measures.)

scales, draftsman scales, and the like. Such apparatus, if The incorrect use of weight in place of mass should be

they require calibration, should be submitted to State or phased out, and the term mass used when mass is meant.

local weights and measures officials.

3.2. Standards of Mass. - The primary standard of mass

for this country is United States Prototype Kilogram 20,

which is a platinum-iridium cylinder kept at NIST. We

know the value of this mass standard in terms of the

International Prototype Kilogram, a platinum-iridium

standard which is kept at the International Bureau of

Weights and Measures.

In Colonial Times the British standards were considered the

primary standards of the United States. Later, the U.S.

avoirdupois pound was defined in terms of the Troy Pound

of the Mint, which is a brass standard kept at the United

States Mint in Philadelphia. In 1911, the Troy Pound of the

Mint was superseded, for coinage purposes, by the Troy

Pound of the Institute.

by the relation:

1 avoirdupois pound = 0.453 592 37 kilogram.2

These changes in definition have not made any appreciable

change in the value of the pound.

The grain is 1/7000 of the avoirdupois pound and is

identical in the avoirdupois, troy, and apothecary systems.

The troy ounce and the apothecary ounce differ from the

avoirdupois ounce but are equal to each other, and equal to

480 grains. The avoirdupois ounce is equal to 437.5 grains.

of about 9.8 m/s . (In common parlance, weight is often 2

Standards of mass are ordinarily calibrated by

comparison to a reference standard of mass. If two

objects are compared on a balance and give the same

balance indication, they have the same "mass" (excluding

the effect of air buoyancy). The forces of gravity on the

two objects are balanced. Even though the value of the

acceleration of gravity, g, is different from location to

location, because the two objects of equal mass in the

same location (where both masses are acted upon by the

same g) will be affected in the same manner and by the

same amount by any change in the value of g, the two

objects will balance each other under any value of g.

However, on a spring balance the mass of a body is not

balanced against the mass of another body. Instead, the

gravitational force on the body is balanced by the

restoring force of a spring. Therefore, if a very sensitive

spring balance is used, the indicated mass of the body

would be found to change if the spring balance and the 
Units and Systems

B-9

body were moved from one locality to another locality with calibrates mass standards submitted, but it does not

a different acceleration of gravity. But a spring balance is manufacture or sell them. Information regarding the

usually used in one locality and is adjusted or calibrated to mass calibration service of NIST and the regulations

indicate mass at that locality. governing the submission of standards of mass to NIST

3.2.2. Effect of Air Buoyancy. - Another point that must

be taken into account in the calibration and use of

standards of mass is the buoyancy or lifting effect of the

air. A body immersed in any fluid is buoyed up by a 3.3. Standards of Capacity. - Units of capacity or volume,

force equal to the force of gravity on the displaced fluid. being derived units, are in this country defined in terms of

Two bodies of equal mass, if placed one on each pan of linear units. Laboratory standards have been constructed

an equal-arm balance, will balance each other in a and are maintained at NIST. These have validity only by

vacuum. A comparison in a vacuum against a known calibration with reference either directly or indirectly to the

mass standard gives "true mass." If compared in air, linear standards. Similarly, NIST has made and distributed

however, they will not balance each other unless they are standards of capacity to the several States. Other standards

of equal volume. If of unequal volume, the larger body of capacity have been verified by calibration for a variety of

will displace the greater volume of air and will be buoyed uses in science, technology, and commerce.

up by a greater force than will the smaller body, and the

larger body will appear to be of less mass than the smaller

body.

The greater the difference in volume, and the greater the

density of the air in which we make the comparison

weighing, the greater will be the apparent difference in

mass. For that reason, in assigning a precise numerical

value of mass to a standard, it is necessary to base this

value on definite values for the air density and the density

of the mass standard of reference.

The apparent mass of an object is equal to the mass of

just enough reference material of a specified density (at

20 EC) that will produce a balance reading equal to that

produced by the object if the measurements are done in

air with a density of 1.2 mg/cm at 20 EC. The original 3

basis for reporting apparent mass is apparent mass versus

brass. The apparent mass versus a density of 8.0 g/cm is 3

the more recent definition, and is used extensively

throughout the world. The use of apparent mass versus

8.0 g/cm is encouraged over apparent mass versus brass. 3

The difference in these apparent mass systems is

insignificant in most commercial weighing applications.

A full discussion of this topic is given in NIST

Monograph 133, Mass and Mass Values, by Paul E.

Pontius [for sale by the National Technical Information

Service, 5285 Port Royal Road, Springfield, VA 22161

(COM 7450309).]

3.2.3. Calibrations of Standards of Mass. - Standards

of mass regularly used in ordinary trade should be tested

by State or local weights and measures officials. NIST

for calibration are contained in NIST Special Publication

250, Calibration and Related Measurement Services of

NIST, latest edition.

3.3.1. Calibrations of Standards of Capacity. - NIST

makes calibrations on capacity or volume standards that

are in the customary units of trade; that is, the gallon, its

multiples, and submultiples, or in metric units. Further,

NIST calibrates precision-grade volumetric glassware

which is normally in metric units. NIST makes

calibrations in accordance with fee schedules, copies of

which may be obtained from NIST.

3.4. Maintenance and Preservation of Fundamental

Standard of Mass. - It is a statutory responsibility of NIST

to maintain and preserve the national standard of mass at

NIST and to realize all the other base units. The U.S.

Prototype Kilogram maintained at NIST is fully protected

by an alarm system. All measurements made with this

standard are conducted in special air-conditioned

laboratories to which the standard is taken a sufficiently

long time before the observations to ensure that the standard

will be in a state of equilibrium under standard conditions

when the measurements or comparisons are made. Hence,

it is not necessary to maintain the standard at standard

conditions, but care is taken to prevent large changes of

temperature. More important is the care to prevent any

damage to the standard because of careless handling.
Units and Systems

B-10

4. Specialized use of the Terms

"Ton" and "Tonnage" 

As weighing and measuring are important factors in our cubical capacity or volume of a ship expressed in register

everyday lives, it is quite natural that questions arise about tons of 100 cubic feet, or 2.83 cubic meters, less such

the use of various units and terms and about the magnitude space as hatchways, bakeries, galleys, etc., as are

of quantities involved. For example, the words "ton" and exempted from measurement by different governments.

"tonnage" are used in widely different senses, and a great There is some lack of uniformity in the gross tonnages as

deal of confusion has arisen regarding the application of given by different nations due to lack of agreement on

these terms. the spaces that are to be exempted. Official merchant

The ton is used as a unit of measure in two distinct senses: of the gross register tonnage. Press references to ship

(1) as a unit of mass, and (2) as a unit of capacity or volume. tonnage are usually to the gross tonnage.

In the first sense, the term has the following meanings: The net tonnage, or net register tonnage, is the gross

a. The short, or net ton of 2000 pounds. nations in their measurement rules and laws. The spaces

b. The long, gross, or shipper’s ton of 2240 pounds. such as the engine room, coal bunkers, crew quarters,

c. The metric ton of 1000 kilograms, or 2204.6 pounds. in computing how much cargo that can be loaded on a

In the second sense (capacity), it is usually restricted to uses similar charges.

relating to ships and has the following meaning:

a. The register ton of 100 cubic feet. of a ship under her tonnage deck expressed in register

b. The measurement ton of 40 cubic feet. deck is the second from the keel.

c. The English water ton of 224 British Imperial There are several variations of displacement tonnage.

gallons.

In the United States and Canada the ton (mass) most "loaded" and "light" displacement tonnages of a vessel.

commonly used is the short ton. In Great Britain, it is the It is expressed in terms of the long ton of 2240 pounds,

long ton, and in countries using the metric system, it is the or the metric ton of 2204.6 pounds, and is the weight of

metric ton. The register ton and the measurement ton are fuel, passengers, and cargo that a vessel can carry when

capacity or volume units used in expressing the tonnage of loaded to its maximum draft.

ships. The English water ton is used, chiefly in Great

Britain, in statistics dealing with petroleum products. The second variety of tonnage, cargo tonnage, refers to

There have been many other uses of the term ton such as the In overseas traffic it is usually expressed in long tons of

timber ton of 40 cubic feet and the wheat ton of 20 bushels, 2240 pounds or metric tons of 2204.6 pounds. The short

but their use has been local and the meanings have not been ton is only occasionally used. Therefore, the cargo

consistent from one place to another. tonnage is very distinct from vessel tonnage.

Properly, the word "tonnage" is used as a noun only in

respect to the capacity or volume and dimensions of ships,

and to the amount of the ship’s cargo. There are two distinct

kinds of tonnage; namely, vessel tonnage and cargo tonnage

and each of these is used in various meanings.

The several kinds of vessel tonnage are as follows:

Gross tonnage, or gross register tonnage, is the total

marine statistics of most countries are published in terms

tonnage less the different spaces specified by maritime

deducted are those totally unavailable for carrying cargo,

chart and instrument room, etc. The net tonnage is used

ship. It is used as the basis for wharfage and other

The register under-deck tonnage is the cubical capacity

tons. In a vessel having more than one deck, the tonnage

The dead weight tonnage is the difference between the

the weight of the particular items making up the cargo.
General Tables of Units of Measurement

C-1

Appendix C

General Tables of Units of Measurement

These tables have been prepared for the benefit of those requiring tables of units for occasional ready reference. In

Section 4 of this Appendix, the tables are carried out to a large number of decimal places and exact values are indicated

by underlining. In most of the other tables, only a limited number of decimal places are given, therefore making the

tables better adopted to the average user.

1. TABLES OF METRIC UNITS OF MEASUREMENT

In the metric system of measurement, designations of multiples and subdivisions of any unit may be arrived at by

combining with the name of the unit the prefixes deka, hecto, and kilo meaning, respectively, 10, 100, and 1 000, and

deci, centi, and milli, meaning, respectively, one-tenth, one-hundredth, and one- thousandth. In some of the following

metric tables, some such multiples and subdivisions have not been included for the reason that these have little, if any

currency in actual usage.

In certain cases, particularly in scientific usage, it becomes convenient to provide for multiples larger than 1 000 and

for subdivisions smaller than one-thousandth. Accordingly, the following prefixes have been introduced and these are

now generally recognized:

yotta, (Y) meaning 10 deci, (d), meaning 10 24 -1

zetta, (Z), meaning 10 centi, (c), meaning 10 21 -2

exa, (E), meaning 10 milli, (m), meaning 10 18 -3

peta, (P), meaning 10 micro, (μ), meaning 10 15 -6

tera, (T), meaning 10 nano, (n), meaning 10 12 -9

giga, (G), meaning 10 pico, (p), meaning 10 9 -12

mega, (M), meaning 10 femto, (f), meaning 10 6 -15

kilo, (k), meaning 10 atto, (a), meaning 10 3 -18

hecto, (h), meaning 10 zepto, (z), meaning 10 2 -21

deka, (da), meaning 10 yocto, (y), meaning 10 1 -24

 Thus a kilometer is 1000 meters and a millimeter is 0.001 meter.

Units of Length

10 millimeters (mm) = 1 centimeter (cm)

10 centimeters = 1 decimeter (dm) = 100 millimeters

10 decimeters = 1 meter (m) = 1000 millimeters

10 meters = 1 dekameter (dam)

10 dekameters = 1 hectometer (hm) = 100 meters

10 hectometers = 1 kilometer (km) = 1000 meters
General Tables of Units of Measurement

 By action of the 12th General Conference on Weights and Measures (1964) the liter is a special name for the cubic 1

decimeter. 

C-2

Units of Area

100 square millimeters (mm ) = 1 square centimeter (cm ) 2 2

100 square centimeters = 1 square decimeter (dm )2

100 square decimeters = 1 square meter (m )2

100 square meters = 1 square dekameter (dam ) = 1 are 2

100 square dekameters = 1 square hectometer (hm ) = 1 hectare (ha) 2

100 square hectometers = 1 square kilometer (km )2

Units of Liquid Volume

10 milliliters (mL) = 1 centiliter (cL)

10 centiliters = 1 deciliter (dL) = 100 milliliters

10 deciliters = 1 liter = 1000 milliliters 1

10 liters = 1 dekaliter (daL)

10 dekaliters = 1 hectoliter (hL) = 100 liters

10 hectoliters = 1 kiloliter (kL) = 1000 liters

Units of Volume

1000 cubic millimeters (mm ) = 1 cubic centimeter (cm ) 3 3

1000 cubic centimeters = 1 cubic decimeter (dm )3

= 1 000 000 cubic millimeters

1000 cubic decimeters = 1 cubic meter (m )3

= 1 000 000 cubic centimeters

= 1 000 000 000 cubic millimeters

Units of Mass

10 milligrams (mg) = 1 centigram (cg)

10 centigrams = 1 decigram (dg) = 100 milligrams

10 decigrams = 1 gram (g) = 1000 milligrams

10 grams = 1 dekagram (dag)

10 dekagrams = 1 hectogram (hg) = 100 grams

10 hectograms = 1 kilogram (kg) = 1000 grams

1000 kilograms = 1 megagram (Mg) or 1 metric ton(t)
General Tables of Units of Measurement

 This section lists units of measurement that have traditionally been used in the United States. In keeping with the 2

Omnibus Trade and Competitiveness Act of 1988, the ultimate objective is to make the International System of Units

the primary measurement system used in the United States.

 Squares and cubes of customary but not of metric units are sometimes expressed by the use of abbreviations rather than 3

symbols. For example, sq ft means square foot, and cu ft means cubic foot.

C-3

2. TABLES OF U.S. UNITS OF MEASUREMENT2

In these tables where foot or mile is underlined, it is survey foot or U.S. statute mile rather than international foot or mile

that is meant.

Units of Length

12 inches (in) = 1 foot (ft)

3 feet = 1 yard (yd)

16-1/2 feet = 1 rod (rd), pole, or perch

40 rods = 1 furlong (fur) = 660 feet

8 furlongs = 1 U.S. statute mile (mi) = 5280 feet

1852 meters = 6076.115 49 feet (approximately)

= 1 international nautical mile

Units of Area3

144 square inches (in ) = 1 square foot (ft ) 2 2

9 square feet = 1 square yard (yd )2

= 1296 square inches

272-1/4 square feet = 1 square rod (sq rd)

160 square rods = 1 acre = 43 560 square feet

640 acres = 1 square mile (mi )2

1 mile square = 1 section of land

6 miles square = 1 township

= 36 sections = 36 square miles

Units of Volume3

1728 cubic inches (in ) = 1 cubic foot (ft ) 3 3

27 cubic feet = 1 cubic yard (yd )3

Gunter’s or Surveyors Chain Units of Measurement

0.66 foot (ft) = 1 link (li)

100 links = 1 chain (ch)

= 4 rods = 66 feet

80 chains = 1 U.S. statute mile (mi)

= 320 rods = 5280 feet
General Tables of Units of Measurement

 When necessary to distinguish the liquid pint or quart from the dry pint or quart, the word "liquid" or the abbreviation 4

"liq" should be used in combination with the name or abbreviation of the liquid unit.

 When necessary to distinguish dry pint or quart from the liquid pint or quart, the word "dry" should be used in 5

combination with the name or abbreviation of the dry unit.

C-4

Units of Liquid Volume4

4 gills (gi) = 1 pint (pt)= 28.875 cubic inches

2 pints = 1 quart (qt) = 57.75 cubic inches

4 quarts = 1 gallon (gal) = 231 cubic inches 

= 8 pints = 32 gills

Apothecaries Units of Liquid Volume

60 minims (min or ) = 1 fluid dram (fl dr or ƒ )

= 0.225 6 cubic inch

8 fluid drams = 1 fluid ounce (fl oz or ƒ )

 = 1.804 7 cubic inches

16 fluid ounces = 1 pint (pt or )

= 28.875 cubic inches 

= 128 fluid drams

2 pints = 1 quart (qt) = 57.75 cubic inches 

= 32 fluid ounces = 256 fluid drams

4 quarts = 1 gallon (gal) = 231 cubic inches 

= 128 fluid ounces = 1024 fluid drams

Units of Dry Volume5

2 pints (pt) = 1 quart (qt) = 67.200 6 cubic inches

8 quarts = 1 peck (pk) = 537.605 cubic inches 

= 16 pints

4 pecks = 1 bushel (bu) = 2150.42 cubic inches

= 32 quarts
General Tables of Units of Measurement

 When necessary to distinguish the avoirdupois dram from the apothecaries dram, or to distinguish the avoirdupois dram 6

or ounce from the fluid dram or ounce, or to distinguish the avoirdupois ounce or pound from the troy or apothecaries

ounce or pound, the word "avoirdupois" or the abbreviation "avdp" should be used in combination with the name or

abbreviation of the avoirdupois unit.

 When the terms "hundredweight" and "ton" are used unmodified, they are commonly understood to mean the 100- 7

pound hundredweight and the 2000-pound ton, respectively; these units may be designated "net" or "short" when

necessary to distinguish them from the corresponding units in gross or long measure.

C-5

Avoirdupois Units of Mass6

[The "grain" is the same in avoirdupois, troy, and apothecaries units of mass.

27-11/32 grains = 1 dram (dr)

16 drams = 1 ounce (oz)

= 437-1/2 grains

16 ounces = 1 pound (lb)

= 256 drams

= 7000 grains

100 pounds = 1 hundredweight (cwt)7

20 hundredweights = 1 ton

= 2000 pounds7

In "gross" or "long" measure, the following values are recognized:

112 pounds = 1 gross or long hundredweight 7

20 gross or long hundredweights = 1 gross or long ton

= 2240 pounds7

Troy Units of Mass

[The "grain" is the same in avoirdupois, troy, and apothecaries units of mass.

24 grains = 1 pennyweight (dwt)

20 pennyweights = 1 ounce troy (oz t)= 480 grains

12 ounces troy = 1 pound troy (lb t)

= 240 pennyweights = 5760 grains

Apothecaries Units of Mass

[The "grain" is the same in avoirdupois, troy, and apothecaries units of mass.

20 grains = 1 scruple (s ap or )

3 scruples = 1 dram apothecaries (dr ap or )

= 60 grains

8 drams apothecaries = 1 ounce apothecaries (oz ap or )

= 24 scruples = 480 grains 

12 ounces apothecaries = 1 pound apothecaries (lb ap)

= 96 drams apothecaries 

= 288 scruples = 5760 grains
General Tables of Units of Measurement

C-6

3. NOTES ON BRITISH UNITS OF MEASUREMENT

In Great Britain, the yard, the avoirdupois pound, the troy pound, and the apothecaries pound are identical with the units

of the same names used in the United States. The tables of British linear measure, troy mass, and apothecaries mass

are the same as the corresponding United States tables, except for the British spelling "drachm" in the table of

apothecaries mass. The table of British avoirdupois mass is the same as the United States table up to 1 pound; above

that point the table reads:

14 pounds = 1 stone

2 stones = 1 quarter = 28 pounds

4 quarters = 1 hundredweight = 112 pounds

 20 hundredweight = 1 ton = 2240 pounds

The present British gallon and bushel--known as the "Imperial gallon" and "Imperial bushel"--are, respectively, about

20 percent and 3 percent larger than the United States gallon and bushel. The Imperial gallon is defined as the volume

of 10 avoirdupois pounds of water under specified conditions, and the Imperial bushel is defined as 8 Imperial gallons.

Also, the subdivision of the Imperial gallon as presented in the table of British apothecaries fluid measure differs in two

important respects from the corresponding United States subdivision, in that the Imperial gallon is divided into 160 fluid

ounces (whereas the United States gallon is divided into 128 fluid ounces), and a "fluid scruple" is included. The full

table of British measures of capacity (which are used alike for liquid and for dry commodities) is as follows:

 4 gills = 1 pint

 2 pints = 1 quart

 4 quarts = 1 gallon

 2 gallons = 1 peck

 8 gallons (4 pecks) = 1 bushel

 8 bushels = 1 quarter

The full table of British apothecaries measure is as follows:

20 minims = 1 fluid scruple

3 fluid scruples = 1 fluid drachm

= 60 minims

8 fluid drachms = 1 fluid ounce

 20 fluid ounces = 1 pint

8 pints = 1 gallon (160 fluid ounces)
General Tables of Units of Measurement

 One international foot = 0.999 998 survey foot (exactly) 8

 One international mile = 0.999 998 survey mile (exactly)

Note: 1 survey foot = 1200/3937 meter (exactly)

 1 international foot = 12 x 0.0254 meter (exactly)

 1 international foot = 0.0254 x 39.37 survey foot (exactly)

C-7

4. TABLES OF UNITS OF MEASUREMENT

(all underlined figures are exact)

Units of Length - International Measure8

Units Inches Feet Yards Miles Centimeters Meters

1 inch = 1 0.083 333 33 0.027 777 78 0.000 015 782 83 2.54 0.025 4

1 foot = 12 1 0.333 333 3 0.000 189 393 9 30.48 0.304 8

1 yard = 36 3 1 0.000 568 181 8 91.44 0.914 4

1 mile = 63 360 5 280 1 760 1 160 934.4 1609.344

1 centimeter = 0.393 700 8 0.032 808 40 0.010 936 13 0.000 006 213 712 1 0.01

1 meter = 39.370 08 3.280 840 1.093 613 0.000 621 371 2 100 1

 Units of Length - Survey Measure8

Units Links Feet Rods Chains Miles Meters

1 link = 1 0.66 0.04 0.01 0.000 125 0.201 168 4

1 foot = 1.515 152 1 0.060 606 06 0.015 151 52 0.000 189 393 9 0.304 800 6

1 rod = 25 16.5 1 0.25 0.003 125 5.029 210

1 chain = 100 66 4 1 0.0125 20.116 84

1 mile = 8 000 5 280 320 80 1 1609.347

1 meter = 4.970 960 3.280 833 0.198 838 4 0.049 709 60 0.000 621 369 9 1
General Tables of Units of Measurement

 One square survey foot = 1.000 004 square international feet 9

 One square survey mile = 1.000 004 square international miles

C-8

 Units of Area - International Measure 9

 (all underlined figures are exact)

Units Square Inches Square Feet Square Yards

1 square inch = 1 0.006 944 444 0.000 771 604 9

1 square foot = 144 1 0.111 111 1

1 square yard = 1296 9 1

1 square mile = 4 014 489 600 27 878 400 3 097 600

1 square centimeter = 0.155 000 3 0.001 076 391 0.000 119 599 0

1 square meter = 1550.003 10.763 91 1.195 990

Units Square Miles Square Centimeters Square Meters

1 square inch = 0.000 000 000 249 097 7 6.451 6 0.000 645 16

1 square foot = 0.000 000 035 870 06 929.030 4 0.092 903 04

1 square yard = 0.000 000 322 830 6 8361.273 6 0.836 127 36

1 square mile = 1 25 899 881 103.36 2 589 988.110 336

1 square centimeter = 0.000 000 000 038 610 22 1 0.0001

1 square meter = 0.000 000 386 102 2 10 000 1

 Units of Area - Survey Measure9

Units Square Feet Square Rods Square Chains Acres

1 square foot = 1 0.003 673 095 0.000 229 568 4 0.000 022 956 84

1 square rod = 272.25 1 0.062 5 0.006 25

1 square chain = 4356 16 1 0.1

1 acre = 43 560 160 10 1

1 square mile = 27 878 400 102 400 6400 640

1 square meter = 10.763 87 0.039 536 70 0.002 471 044 0.000 247 104 4

1 hectare = 107 638.7 395.367 0 24.710 44 2.471 044
General Tables of Units of Measurement

C-9

Units Square Miles Square Meters Hectares

1 square foot = 0.000 000 035 870 06 0.092 903 41 0.000 009 290 341

1 square rod = 0.000 009 765 625 25.292 95 0.002 529 295

1 square chain = 0.000 156 25 404.687 3 0.040 468 73

1 acre = 0.001 562 5 4 046.873 0.404 687 3

1 square mile = 1 2 589 998 258.999 8

1 square meter = 0.000 000 386 100 6 1 0.000 1

1 hectare = 0.003 861 006 10 000 1

 Units of Volume

 (all underlined figures are exact)

Units Cubic Inches Cubic Feet Cubic Yards

1 cubic inch = 1 0.000 578 703 7 0.000 021 433 47

1 cubic foot = 1728 1 0.037 037 04

1 cubic yard = 46 656 27 1

1 cubic centimeter = 0.061 023 74 0.000 035 314 67 0.000 001 307 951

1 cubic decimeter = 61.023 74 0.035 314 67 0.001 307 951

1 cubic meter = 61 023.74 35.314 67 1.307 951

Units Milliliters Liters Cubic Meters

(Cubic Centimeters) (Cubic Decimeters)

1 cubic inch = 16.387 064 0.016 387 064 0.000 016 387 064

1 cubic foot = 28 316.846 592 28.316 846 592 0.028 316 846 592

1 cubic yard = 764 554.857 984 764.554 857 984 0.764 554 857 984

1 cubic centimeter = 1 0.001 0.000 001

1 cubic decimeter = 1000 1 0.001

1 cubic meter = 1 000 000 1000 1
General Tables of Units of Measurement

C-10

Units of Capacity or Volume - Dry Volume Measure

Units Dry Pints Dry Quarts Pecks Bushels

1 dry pint = 1 0.5 0.062 5 0.015 625

1 dry quart = 2 1 0.125 0.031 25

1 peck = 16 8 1 0.25

1 bushel = 64 32 4 1

1 cubic inch = 0.029 761 6 0.014 880 8 0.001 860 10 0.000 465 025

1 cubic foot = 51.428 09 25.714 05 3.214 256 0.803 563 95

1 liter = 1.816 166 0.908 083 0 0.113 510 4 0.028 377 59

1 cubic meter = 1 816.166 908.083 0 113.510 4 28.377 59

Units Cubic Inches Cubic Feet Liters Cubic Meters

1 dry pint = 33.600 312 5 0.019 444 63 0.550 610 5 0.000 550 610 5

1 dry quart = 67.200 625 0.038 889 25 1.101 221 0.001 101 221

1 peck = 537.605 0.311 114 8.809 768 0.008 809 768

1 bushel = 2 150.42 1.244 456 35.239 07 0.035 239 07

1 cubic inch = 1 0.000 578 703 7 0.016 387 06 0.000 016 387 06

1 cubic foot = 1728 1 28.316 85 0.028 316 85

1 liter = 61.023 74 0.035 314 67 1 0.001

1 cubic meter = 61 023.74 35.314 67 1000 1
General Tables of Units of Measurement

C-11

Units of Capacity or Volume - Liquid Volume Measure

(All underlined figures are exact)

Units Minims Fluid Drams Fluid Ounces Gills

1 minim = 1 0.016 666 67 0.002 083 333 0.000 520 833 3

1 fluid dram = 60 1 0.125 0.031 25

1 fluid ounce = 480 8 1 0.25

1 gill = 1920 32 4 1

1 liquid pint = 7680 128 16 4

1 liquid quart = 15 360 256 32 8

1 gallon = 61 440 1024 128 32

1 cubic inch = 265.974 0 4.432 900 0.554 112 6 0.138 528 1

1 cubic foot = 459 603.1 7660.052 957.506 5 239.376 6

1 milliliter = 16.230 73 0.270 512 2 0.033 814 02 0.008 453 506

1 liter = 16 230.73 270.512 2 33.814 02 8.453 506

Units Liquid Pints Liquid Quarts Gallons Cubic Inches

1 minim = 0.000 130 208 3 0.000 065 104 17 0.000 016 276 04 0.003 759 766

1 fluid dram = 0.007 812 5 0.003 906 25 0.000 976 562 5 0.225 585 94

1 fluid ounce = 0.062 5 0.031 25 0.007 812 5 1.804 687 5

1 gill = 0.25 0.125 0.031 25 7.218 75

1 liquid pint = 1 0.5 0.125 28.875

1 liquid quart = 2 1 0.25 57.75

1 gallon = 8 4 1 231

1 cubic inch = 0.034 632 03 0.017 316 02 0.004 329 004 1

1 cubic foot = 59.844 16 29.922 08 7.480 519 1728

1 milliliter = 0.002 113 376 0.001 056 688 0.000 264 172 1 0.061 023 74

1 liter = 2.113 376 1.056 688 0.264 172 1 61.023 74
General Tables of Units of Measurement

C-12

Units Cubic Feet Milliliters Liters

1 minim = 0.000 002 175 790 0.061 611 52 0.000 061 611 52

1 fluid dram = 0.000 130 547 4 3.696 691 0.003 696 691

1 fluid ounce = 0.001 044 379 29.573 53 0.029 573 53

1 gill = 0.004 177 517 118.294 1 0.118 294 1

1 liquid pint = 0.016 710 07 473.176 5 0.473 176 5

1 liquid quart = 0.033 420 14 946.352 9 0.946 352 9

1 gallon = 0.133 680 6 3785.412 3.785 412

1 cubic inch = 0.000 578 703 7 16.387 06 0.016 387 06

1 cubic foot = 1 28 316.85 28.316 85

1 milliliter = 0.000 035 314 67 1 0.001

1 liter = 0.035 314 67 1000 1

 Units of Mass Not Less Than Avoirdupois Ounces

 (all underlined figures are exact)

Units Avoirdupois Avoirdupois Short Hundred- Short tons

Ounces Pounds weights

1 avoirdupois ounce = 1 0.0625 0.000 625 0.000 031 25

1 avoirdupois pound = 16 1 0.01 0.000 5

1 short 1600 100 1 0.05

hundredweight =

1 short ton = 32 000 2000 20 1

1 long ton = 35 840 2240 22.4 1.12

1 kilogram = 35.273 96 2.204 623 0.022 046 23 0.001 102 311

1 metric ton = 35 273.96 2204.623 22.046 23 1.102 311

Units Long Tons Kilograms Metric Tons

1 avoirdupois ounce = 0.000 027 901 79 0.028 349 523 125 0.000 028 349 523 125

1 avoirdupois pound = 0.000 446 428 6 0.453 592 37 0.000 453 592 37

1 short hundredweight = 0.044 642 86 45.359 237 0.045 359 237

1 short ton = 0.892 857 1 907.184 74 0.907 184 74

1 long ton = 1 1016.046 908 8 1.016 046 908 8

1 kilogram = 0.000 984 206 5 1 0.001

1 metric ton = 0.984 206 5 1000 1
General Tables of Units of Measurement

C-13

Units of Mass Not Greater Than Pounds and Kilograms

 (all underlined figures are exact)

Units Grains Apothecaries Scruples Pennyweights Avoirdupois Drams

1 grain = 1 0.05 0.041 666 67 0.036 571 43

1 apoth. scruple = 20 1 0.833 333 3 0.731 428 6

1 pennyweight = 24 1.2 1 0.877 714 3

1 avdp. dram = 27.343 75 1.367 187 5 1.139 323 1

1 apoth. dram = 60 3 2.5 2.194 286

1 avdp. ounce = 437.5 21.875 18.229 17 16

1 apoth. or troy oz. = 480 24 20 17.554 29

1 apoth. or troy pound= 5760 288 240 210.651 4

1 avdp. pound = 7000 350 291.666 7 256

1 milligram = 0.015 432 36 0.000 771 617 9 0.000 643 014 9 0.000 564 383 4

1 gram = 15.432 36 0.771 617 9 0.643 014 9 0.564 383 4

1 kilogram = 15432.36 771.617 9 643.014 9 564.383 4

Units Apothecaries Drams Avoirdupois Ounces Apothecaries or Troy Apothecaries or Troy

Ounces Pounds

1 grain = 0.016 666 67 0.002 285 714 0.002 083 333 0.000 173 611 1

1 apoth. scruple = 0.333 333 3 0.045 714 29 0.041 666 67 0.003 472 222

1 pennyweight = 0.4 0.054 857 14 0.05 0.004 166 667

1 avdp. dram = 0.455 729 2 0.062 5 0.56 966 15 0.004 747 179

1 apoth. dram = 1 0.137 142 9 0.125 0.010 416 67

1 avdp. ounce = 7.291 667 1 0.911 458 3 0.075 954 86

1 apoth. or troy ounce= 8 1.097 143 1 0.083 333 333

1 apoth. or troy pound= 96 13.165 71 12 1

1 avdp. pound = 116.666 7 16 14.583 33 1.215 278

1 milligram = 0.000 257 206 0 0.000 035 273 96 0.000 032 150 75 0.000 002 679 229

1 gram = 0.257 206 0 0.035 273 96 0.032 150 75 0.002 679 229

1 kilogram = 257.206 0 35.273 96 32.150 75 2.679 229
General Tables of Units of Measurement

 The angstrom is basically defined as 10 meter. 10 -10

C-14

Units Avoirdupois Pounds Milligrams Grams Kilograms

1 grain = 0.000 142 857 1 64.798 91 0.064 798 91 0.000 064 798 91

1 apoth. scruple = 0.002 857 143 1295.978 2 1.295 978 2 0.001 295 978 2

1 pennyweight = 0.003 428 571 1555.173 84 1.555 173 84 0.001 555 173 84

1 avdp. dram = 0.003 906 25 1771.845 195 312 5 1.771 845 195 312 5 0.001 771 845 195 312 5

1 apoth. dram = 0.008 571 429 3887.934 6 3.887 934 6 0.003 887 934 6

1 avdp. ounce = 0.062 5 28 349.523 125 28.349 523 125 0.028 349 523 125

1 apoth. or troy 0.068 571 43 31 103.476 8 31.103 476 8 0.031 103 476 8

ounce =

1 apoth. or troy 0.822 857 1 373 241.721 6 373.241 721 6 0.373 241 721 6

pound =

1 avdp. pound = 1 453 592.37 453.592 37 0.453 592 37

1 milligram = 0.000 002 204 623 1 0.001 0.000 001

1 gram = 0.002 204 623 1000 1 0.001

1 kilogram = 2.204 623 1 000 000 1000 1

5. TABLES OF EQUIVALENTS

In these tables it is necessary to differentiate between the "international foot" and the "survey foot" the survey foot is

underlined.

When the name of a unit is enclosed in brackets (thus, [1 hand] . . . ), this indicates (1) that the unit is not in general

current use in the United States, or (2) that the unit is believed to be based on "custom and usage" rather than on formal

authoritative definition.

Equivalents involving decimals are, in most instances, rounded off to the third decimal place except where they are

exact, in which cases these exact equivalents are so designated. The equivalents of the imprecise units "tablespoon"

and "teaspoon" are rounded to the nearest milliliter.

UNITS OF LENGTH

0.1 nanometer (exactly).

0.000 1 micrometer (exactly).

angstrom (D) ----------------------- 0.000 000 1 millimeter (exactly). 10

0.000 000 004 inch.

120 fathoms (exactly).

1 cable’s length----------------------- 720 feet (exactly).

219 meters.

1 centimeter (cm)---------------------- 0.393 7 inch.

1 chain (ch)--------------------------- 66 feet (exactly). 
General Tables of Units of Measurement

 The term "statute mile" originated with Queen Elizabeth I who changed the definition of the mile from the Roman 11

mile of 5000 feet to the statute mile of 5280 feet. The international mile and the U.S. statute mile differ by about 3

millimeters although both are defined as being equal to 5280 feet. The international mile is based on the international

foot (0.3048 meter) whereas the U.S. statute mile is based on the survey foot (1200/3937 meter).

C-15

 (Gunter’s or surveyors) 

20.116 8 meters.

1 decimeter (dm)----------------------- 3.937 inches.

1 dekameter (dam)---------------------- 32.808 feet.

6 feet (exactly).

1 fathom------------------------------- 1.828 8 meters.

1 foot (ft)---------------------------- 0.304 8 meter (exactly).

 

10 chains (surveyors) (exactly).

1 furlong (fur)------------------------ 660 feet (exactly).

1/8 U.S. statute mile (exactly).

201.168 meters.

[1 hand]------------------------------- 4 inches.

1 inch (in)---------------------------- 2.54 centimeters (exactly).

1 kilometer (km)----------------------- 0.621 mile.

1 league (land)------------------------ 3 U.S. statute miles (exactly).

4.828 kilometers.

0.66 foot (exactly).

1 link (li) (Gunter’s or surveyors)---- 0.201 168 meter.

1 meter (m)---------------------------- 39.37 inches.

1.094 yards.

1 micrometer--------------------------- 0.001 millimeter (exactly).

0.000 039 37 inch.

0.001 inch (exactly).

1 mil---------------------------------- 0.025 4 millimeter (exactly).

1 mile (mi) (U.S. statute) ---------- 5280 feet survey (exactly). 11

1.609 kilometers.

1 mile (mi) (international)------------ 5280 feet international (exactly).
43560 ' 208.710%

General Tables of Units of Measurement

 The international nautical mile of 1 852 meters (6 076.115 49...feet) was adopted effective July 1, 1954, for use in 12

the United States. The value formerly used in the United States was 6 080.20 feet = 1 nautical (geographical or sea)

mile.

 The question is often asked as to the length of a side of an acre of ground. An acre is a unit of area containing 43 560 13

square feet. It is not necessarily square, or even rectangular. But, if it is square, then the length of a side is equal to

 feet.

C-16

1 mile (mi) (international nautical) 1.852 kilometers (exactly). 12

1.151 survey miles.

1 millimeter (mm)---------------------- 0.039 37 inch.

 0.001 micrometer (exactly).

1 nanometer (nm)----------------------- 0.000 000 039 37 inch.

 

0.013 837 inch (exactly).

1 Point (typography)------------------- 1/72 inch (approximately).

0.351 millimeter.

1 rod (rd), pole, or perch------------- 16 1/2 feet (exactly).

5.029 2 meters.

1 yard (yd)---------------------------- 0.914 4 meter (exactly).

UNITS OF AREA

43 560 square feet (exactly).

1 acre ---------------------------- 0.405 hectare. 13

1 are------------------------------ 119.599 square yards. 

 0.025 acre.

1 hectare-------------------------- 2.471 acres.

[1 square (building)]--------------- 100 square feet.

1 square centimeter (cm2)----------- 0.155 square inch. 

1 square decimeter (dm )------------ 15.500 square inches. 2

1 square foot (ft )----------------- 929.030 square centimeters. 2

1 square inch (in )----------------- 6.451 6 square centimeters (exactly). 2

1 square kilometer (km )------------ 247.104 acres. 2

 0.386 square mile.

1 square meter (m )----------------- 1.196 square yards. 2

 10.764 square feet.
General Tables of Units of Measurement

 There are a variety of "barrels" established by law or usage. For example, Federal taxes on fermented liquors are 14

based on a barrel of 31 gallons; many State laws fix the "barrel for liquids" as 31-1/2 gallons; one State fixes a 36-gallon

barrel for cistern measurement; Federal law recognizes a 40-gallon barrel for "proof spirits"; by custom, 42 gallons

comprise a barrel of crude oil or petroleum products for statistical purposes, and this equivalent is recognized "for

liquids" by four States. 

 Frequently recognized as 1-1/4 bushels, struck measure. 15

C-17

1 square mile (mi )----------------- 258.999 hectares. 2

1 square millimeter (mm )----------- 0.002 square inch. 2

1 square rod (rd ), sq pole, 2

 or sq perch----------------------- 25.293 square meters. 

1 square yard (yd )----------------- 0.836 square meter. 2

UNITS OF CAPACITY OR VOLUME

1 barrel (bbl), liquid----------------- 31 to 42 gallons.14

1 barrel (bbl), standard for fruits, 7056 cubic inches.

 vegetables, and other dry------------ 105 dry quarts.

 commodities, except cranberries 3.281 bushels, struck measure.

5 826 cubic inches.

1 barrel (bbl), standard, cranberry---- 86 45/64 dry quarts.

 2.709 bushels, struck measure.

2150.42 cubic inches (exactly).

1 bushel (bu) (U.S.) struck measure---- 35.239 liters.

2 747.715 cubic inches.

[1 bushel, heaped (U.S.)]-------------- 1.278 bushels, struck measure.15

[1 bushel (bu) (British Imperial)

1.032 U.S. bushels, struck measure 

(struck measure)]---------------------- 2219.36 cubic inches. 

1 cord (cd) (firewood)----------------- 128 cubic feet (exactly).

1 cubic centimeter (cm )--------------- 0.061 cubic inch. 3

1 cubic decimeter (dm )---------------- 61.024 cubic inches. 3

1 cubic foot (ft )--------------------- 7.481 gallons. 3

28.316 cubic decimeters. 

 0.554 fluid ounce.

1 cubic inch (in )------------------------ 4.433 fluid drams. 3

 16.387 cubic centimeters.
General Tables of Units of Measurement

C-18

1 cubic meter (m )--------------------- 1.308 cubic yards. 3

1 cubic yard (yd )--------------------- 0.765 cubic meter. 3

8 fluid ounces (exactly).

1 cup, measuring----------------------- 237 milliliters.

 1/2 liquid pint (exactly).

1 dekaliter (daL)---------------------- 2.642 gallons.

 1.135 pecks.

 1/8 fluid ounce (exactly).

1 dram, fluid (or liquid) (fl dr 0.226 cubic inch.

or ƒ ) (U.S.)------------------------ 3.697 milliliters.

 1.041 British fluid drachms.

 0.961 U.S. fluid dram.

[1 drachm, fluid (fl dr) (British)]---- 0.217 cubic inch. 

 3.552 milliliters.

 231 cubic inches (exactly). 

3.785 liters.

1 gallon (gal) (U.S.)------------------ 0.833 British gallon. 

 128 U.S. fluid ounces (exactly).

 277.42 cubic inches. 

 1.201 U.S. gallons.

[1 gallon (gal) (British Imperial)]---- 4.546 liters. 

 160 British fluid ounces (exactly).

 7.219 cubic inches.

1 gill (gi)---------------------------- 4 fluid ounces (exactly). 

0.118 liter.

1 hectoliter (hL)---------------------- 26.418 gallons.

 2.838 bushels.

1.057 liquid quarts.

1 liter (1 cubic decimeter exactly)---- 0.908 dry quart. 

 61.025 cubic inches 

 0.271 fluid dram.

1 milliliter (mL)---------------------- 16.231 minims. 

 0.061 cubic inch.

1 ounce, fluid (or liquid) (fl oz 1.805 cubic inches.

or ƒ )(U.S.)------------------------- 29.573 milliliters.

 1.041 British fluid ounces.

0.961 U.S. fluid ounce.

[1 ounce, fluid (fl oz) (British)]----- 1.734 cubic inches. 

 28.412 milliliters.

1 peck (pk)---------------------------- 8.810 liters. 
General Tables of Units of Measurement

 The equivalent "1 teaspoon = 1-1/3 fluid drams" has been found by the Bureau to correspond more closely with the 16

actual capacities of "measuring" and silver teaspoons than the equivalent "1 teaspoon = 1 fluid dram," which is given

by a number of dictionaries.

 Used in assaying. The assay ton bears the same relation to the milligram that a ton of 2000 pounds avoirdupois bears 17

to the ounce troy; hence the mass in milligrams of precious metal obtained from one assay ton of ore gives directly the

number of troy ounces to the net ton.

C-19

33.600 cubic inches.

1 pint (pt), dry----------------------- 0.551 liter.

 28.875 cubic inches exactly.

1 pint (pt), liquid-------------------- 0.473 liter.

 67.201 cubic inches.

1 quart (qt), dry (U.S.)--------------- 1.101 liters.

 0.969 British quart.

57.75 cubic inches (exactly).

1 quart (qt), liquid (U.S.)------------ 0.946 liter.

 0.833 British quart.

 69.354 cubic inches.

[1 quart (qt) (British)]--------------- 1.032 U.S. dry quarts. 

 1.201 U.S. liquid quarts. 

 3 teaspoons (exactly).

1 tablespoon, measuring---------------- 15 milliliters.

 4 fluid drams.

 1/2 fluid ounce (exactly).

1/3 tablespoon (exactly).

1 teaspoon, measuring------------------ 5 milliliters.

1-1/3 fluid drams.16

270.91 U.S. gallons.

1 water ton (English)------------------ 224 British Imperial gallons.

(exactly).

UNITS OF MASS

1 assay ton (AT)--------------------- 29.167 grams. 17

200 milligrams (exactly).

1 carat (c)---------------------------- 3.086 grains.

1 dram apothecaries (dr ap or )----- 60 grains (exactly).

3.888 grams.

1 dram avoirdupois (dr avdp)----------- 27-11/32 (= 27.344) grains.

1.777 grams.
General Tables of Units of Measurement

 The gross or long ton and hundredweight are used commercially in the United States to only a very limited extent, 18

usually in restricted industrial fields. The units are the same as the British "ton" and "hundredweight."

C-20

1 gamma ( )--------------------------- 1 microgram (exactly).

1 grain-------------------------------- 64.798 91 milligrams (exactly).

1 gram (g)----------------------------- 15.432 grains.

0.035 ounce, avoirdupois.

1 hundredweight, gross or long 112 pounds (exactly). 18

 (gross cwt)-------------------------- 50.802 kilograms.

1 hundredweight, gross or short 100 pounds (exactly).

 (cwt or net cwt)--------------------- 45.359 kilograms.

1 kilogram (kg)------------------------ 2.205 pounds.

1 microgram (μg) [the Greek letter mu

 in combination with the letter g]--- 0.000 001 gram (exactly).

1 milligram (mg)----------------------- 0.015 grain.

437.5 grains (exactly).

1 ounce, avoirdupois (oz avdp)--------- 0.911 troy or apothecaries ounce.

28.350 grams.

1 ounce, troy or apothecaries 480 grains (exactly).

(oz t or oz ap or )---------------- 1.097 avoirdupois ounces.

31.103 grams.

1 pennyweight (dwt)-------------------- 1.555 grams.

1 point-------------------------------- 0.01 carat.

2 milligrams.

7000 grains (exactly).

1 pound, avoirdupois (lb avdp)--------- 1.215 troy or apothecaries pounds.

453.592 37 grams (exactly).

1 pound, troy or apothecaries 5760 grains (exactly).

 (lb t or lb ap)---------------------- 0.823 avoirdupois pound.

373.242 grams.

1 scruple (s ap or )---------------- 20 grains (exactly).

1.296 grams.
General Tables of Units of Measurement

 The gross or long ton and hundredweight are used commercially in the United States to a limited extent only, usually 19

in restricted industrial fields. These units are the same as the British "ton" and "hundredweight."

C-21

2240 pounds (exactly).

1 ton, gross or long ----------------- 1.12 net tons (exactly). 19

1.016 metric tons.

2204.623 pounds.

1 ton, metric (t)---------------------- 0.984 gross ton.

1.102 net tons.

2000 pounds (exactly).

1 ton, net or short-------------------- 0.893 gross ton.

0.907 metric ton.
Definitions

D-1

Appendix D

Definitions

The specific code to which the definition applies is shown in [brackets] at the end of the definition. Definitions for the

General Code [1.10] apply to all codes in Handbook 44.

A

absolute value. The absolute value of a number is the

magnitude of that number without considering the

positive or negative sign.[2.20]

acceptance test. The first official test of a farm milk

tank, at a particular location, in which the tank is

accepted as correct. This test applies to newly

constructed tanks, relocated used tanks, and recalibrated tanks.[4.43]

accurate. A piece of equipment is "accurate" when its

performance or value–that is, its indications, its

deliveries, its recorded representations, or its capacity

or actual value, etc., as determined by tests made with

suitable standards–conforms to the standard within the

applicable tolerances and other performance requirements. Equipment that fails so to conform is "inaccurate." (Also see "correct.")[1.10]

analog type. A system of indication or recording in

which values are presented as a series of graduations in

combination with an indicator, or in which the most

sensitive element of an indicating system moves

continuously during the operation of the device.[1.10]

animal scale. A scale designed for weighing single

heads of livestock.[2.20]

(Amended 1987)

apparent mass versus 8.0 g/cm . The apparent mass 3

of an object versus 8.0 g/cm is the mass of material of 3

density 8.0 g/cm that produces exactly the same 3

balance reading as the object when the comparison is

made in air with a density of 1.2 mg/cm at 3

20 EC.[3.30, 3.32]

approval seal. A label, tag, stamped or etched

impression, or the like, indicating official approval of

a device. (Also see "security seal.")[1.10]

atmospheric pressure. The average atmospheric

pressure agreed to exist at the meter at various ranges

of elevation, irrespective of variations in atmospheric

pressure from time to time.[3.33]

audit trail. An electronic count and/or information

record of the changes to the values of the calibration or

configuration parameters of a device.

[1.10, 2.20, 3.30] (Added 1993)

automatic zero-setting mechanism. Automatic

means provided to maintain zero balance indication

without the intervention of an operator.[2.20]

automatic bulk weighing system. A weighing system

adapted to the automatic weighing of bulk commodities

in successive drafts of predetermined amounts,

automatically recording the no-load and loaded weight

values and accumulating the net weight of each

draft.[2.20]

automatic hopper scale. One adapted to the

automatic weighing of bulk commodity in successive

drafts of predetermined amounts. (This is not an "automatic-indicating scale" defined below.)[2.20]

automatic temperature or density compensation.

The use of integrated or ancillary equipment to obtain

from the output of a volumetric meter an equivalent

mass, or an equivalent liquid volume at a normal

temperature of 70 EF and absolute pressure of

14.696 lb/in absolute.[3.34] 2

automatic-indicating scale. One on which the

weights of applied loads of various magnitudes are

automatically indicated throughout all or a portion of

the weighing range of the scale. (A scale that

automatically weighs out commodity in predetermined

drafts, such as an automatic hopper scale, a packaging

scale, and the like, is not an "automatic-indicating"

scale.)[2.20]

auxiliary indicator. Any indicator other than the

master weight totalizer that indicates the weight of

material determined by the scale.[2.21]

axle-load scale. A scale permanently installed in a

fixed location, having a load-receiving element

specially adapted to determine the combined load of all

wheels (1) on a single axle or (2) on a tandem axle of

a highway vehicle.[2.20]
Definitions

D-2

B

badge. A metal plate affixed to the meter by the

manufacturer showing the manufacturer’s name, serial

number and model number of the meter, and its rated

capacity.[3.33]

balance, zero-load. See "zero-load balance."[2.20]

balance indicator. A combination of elements, one or

both of which will oscillate with respect to the other,

for indicating the balance condition of a nonautomatic

indicating scale. The combination may consist of two

indicating edges, lines, or points, or a single edge, line,

or point and a graduated scale.[2.20]

balancing mechanism. A mechanism (including a

balance ball) that is designed for adjusting a scale to an

accurate zero-load balance condition.[2.20]

base pressure. The absolute pressure used in defining

the gas measurement unit to be used, and is the gauge

pressure at the meter plus an agreed atmospheric

pressure.[3.33]

basic time rate. The charge for time for all intervals

except the initial interval.[5.54]

basic tolerances. Basic tolerances are those tolerances

on underregistration and on overregistration, or in

excess and in deficiency, that are established by a particular code for a particular device under all normal

tests, whether maintenance or acceptance. Basic

tolerances include minimum tolerance values when

these are specified. Special tolerances, identified as

such and pertaining to special tests, are not basic

tolerances.[1.10]

basic distance rate. The charge for distance for all

intervals except the initial interval.[5.54]

batching meter. A device used for the purpose of

measuring quantities of water to be used in a batching

operation.[3.36]

beam scale. One on which the weights of loads of

various magnitudes are indicated solely by means of

one or more weighbeam bars either alone or in combination with counterpoise weights.[2.20]

beam. See "weighbeam."[2.20]

bell prover. A calibrated cylindrical metal tank of the

annular type with a scale thereon that, in the downward

travel in a surrounding tank containing a sealing

medium, displaces air through the meter being proved

or calibrated.[3.33]

belt-conveyor. An endless moving belt for

transporting material from place to place.[2.21]

belt-conveyor scale. A device that employs a

weighing element in contact with a belt to sense the

weight of the material being conveyed and the speed

(travel) of the material, and integrates these values to

produce total delivered weight.[2.21]

belt-conveyor scale systems area. The scale area

refers to the scale suspension, weigh idlers attached to

the scale suspension, 5 approach (-) idlers, and 5 retreat

(+) idlers. [2.21] (Added 2001)

bench scale. See "counter scale."[2.20]

binary submultiples. Fractional parts obtained by

successively dividing by the number 2. Thus, one-half,

one-fourth, one-eighth, one-sixteenth, and so on, are

binary submultiples.[1.10]

C

calibration parameter. Any adjustable parameter that

can affect measurement or performance accuracy and,

due to its nature, needs to be updated on an ongoing

basis to maintain device accuracy, e.g., span

adjustments, linearization factors, and coarse zero

adjustments. [2.20, 3.30] (Added 1993)

car-wash timer. A timer used in conjunction with a

coin-operated device to measure the time during which

car-wash water, cleaning solutions, or waxing solutions

are dispensed.[5.55]

center-reading tank. One so designed that the gauge

rod or surface gauge, when properly positioned for use,

will be approximately in the vertical axis of the tank,

centrally positioned with respect to the tank

walls.][4.43]

cereal grain and oil seeds. Agricultural commodities

including, but not limited to, corn, wheat, oats, barley,

flax, rice, sorghum, soybeans, peanuts, dry beans,

safflower, sunflower, fescue seed, etc.[5.56]

chart recorder. An element used with a belt-conveyor

scale that continuously records the rate-of-flow of bulk

material over the scale.[2.21]
Definitions

D-3

check rate. A rate of flow usually 20 percent of the

capacity rate.[3.33]

checkweighing scale. One used to verify

predetermined weight within prescribed limits.[2.20]

class of grain. Hard Red Winter Wheat as

distinguished from Hard Red Spring Wheat as

distinguished from Soft Red Winter Wheat, etc.[5.56]

clear interval between graduations. The distance

between adjacent edges of successive graduations in a

series of graduations. If the graduations are "staggered," the interval shall be measured, if necessary,

between a graduation and an extension of the adjacent

graduation. (Also see "minimum clear interval.")[1.10]

cleared. A taximeter is "cleared" when it is inoperative

with respect to all fare indication, when no indication

of fare or extras is shown and when all parts are in

those positions in which they are designed to be when

the vehicle on which the taximeter is installed is not

engaged by a passenger.[5.54]

coal-mine scale. One used at a coal mining operation

for determining the basic wages of miners on a

production basis.[2.20]

cold-tire pressure. The pressure in a tire at ambient

temperature.[5.53, 5.54]

computing type or computing type device. A device

designed to indicate, in addition to weight or measure,

the total money value of product weighed or measured,

for one of a series of unit prices.[1.10]

computing scale. One that indicates the money values

of amounts of commodity weighed, at predetermined

unit prices, throughout all or part of the weighing range

of the scale.[2.20]

concave curve. A change in the angle of inclination of

a belt conveyor where the center of the curve is above

the conveyor.[2.21]

concentrated load capacity (CLC). A capacity rating

of a vehicle, axle-load, or livestock scale, specified by

the manufacturer, defining the maximum load

concentration for which the weighbridge is designed.

In the case of vehicle and axle-load scales, it is the

maximum axle-load concentration (for a group of two

axles with a centerline spaced 4 feet apart and an axle

width of 8 feet) for which the weighbridge is designed

as specified by the manufacturer. The concentrated

load capacity rating is for both test and use.[2.20]

(Added 1988)(Amended 1991, 1994)

configuration parameter. Any adjustable or

selectable parameter for a device feature that can affect

the accuracy of a transaction or can significantly

increase the potential for fraudulent use of the device

and, due to its nature, needs to be updated only during

device installation or upon replacement of a

component, e.g., division value (increment), sensor

range, and units of measurement. [2.20, 3.30] (Added

1993)

consecutive-car test train. A train consisting of cars

weighed on a reference scale, then coupled

consecutively and run over the coupled-in-motion

railway track scale under test.[2.20]

(Added 1990)

construction-material hopper scale. A scale adapted

to weighing construction materials such as sand, gravel,

cement, and hot oil.[2.20]

contract sale. A sale where there is a written

agreement stating the price as either a fixed price, a

price above cost, or an adjustment from the posted

price. [3.30] (Added 1993)

conventional scale. If the use of conversion tables is

necessary to obtain a moisture content value, the

moisture meter indicating scale is called "conventional

scale." The values indicated by the scale are

dimensionless.[5.56]

conversion table. Any table, graph, slide rule, or other

external device used to determine the moisture content

from the value indicated by the moisture meter.[5.56]

correction table. Any table, graph, slide rule, or other

external device used to determine the moisture content

from the value indicated by the moisture meter when

the indicated value is altered by a parameter not

automatically corrected for in the moisture meter (for

example, temperature or test weight).[5.56]

convex curve. A change in the angle of inclination of

a belt conveyor where the center of the curve is below

the conveyor.[2.21]

conveyor stringers. Support members for the

conveyor on which the scale and idlers are

mounted.[2.21]

correct. A piece of equipment is "correct" when, in

addition to being accurate, it meets all applicable

specification requirements. Equipment that fails to
Definitions

D-4

meet any of the requirements for correct equipment is level of the sealing medium properly adjusted, the

"incorrect." (Also see "accurate.")[1.10] bottle, when lowered, will displace exactly one cubic

counter scale. One that, by reason of its size,

arrangement of parts, and moderate nominal capacity,

is adapted for use on a counter or bench. Sometimes

called "bench scale."[2.20]

counterbalance weight. One intended for application

near the butt of a weighbeam for zero-load balancing

purposes.[2.20]

counterpoise weight. A slotted or "hanger" weight

intended for application near the tip of the weighbeam

of a scale having a multiple greater than 1.[2.20]

coupled-in-motion railroad weighing system. A

device and related installation characteristics consisting

of (1) the associated approach trackage, (2) the scale

(i.e., the weighing element, the load-receiving element,

and the indicating element with its software), and (3)

the exit trackage which permit the weighing of railroad

cars coupled in motion.[2.20]

(Added 1992)

crane scale. One with a nominal capacity of 5000

pounds or more designed to weigh loads while they are

suspended freely from an overhead, trackmounted

crane.[2.20]

cryogenic liquid-measuring device. A system

including a mechanism or machine of (a) the meter or

mass flow type, or (b) a weighing type of device

mounted on a vehicle, designed to measure and deliver

cryogenic liquids in the liquid state. Means may be

provided to indicate automatically, for one of a series

of unit prices, the total money value of the liquid

measured.[3.34]

(Amended 1986)

cryogenic liquids. Fluids whose normal boiling point

is below 120 kelvin (-243 EF).[3.34]

cubic foot, standard. That quantity of gas that

occupies a volume of one cubic foot when under a

pressure of 14.73 lb/in absolute and at a temperature 2

of 60 EF.[3.33]

cubic foot, metered. That quantity of gas that

occupies one cubic foot when under pressure and

temperature conditions existing in the meter.[3.33]

cubic-foot bottle. A metal bottle open at the lower end

and so supported that it may be easily raised or lowered

in a tank that contains a sealing medium. With the

foot of air upon coming to rest on the bottom of the

tank. The marks on the bottle defining the cubic foot

are the bottom of the lower neck and the gauge mark

that partially surrounds the gauge glass in the upper

neck.[3.33]

cubic foot, gas. The amount of a cryogenic liquid in

the gaseous state at a temperature of 70 EF and under

a pressure of 14.696 lb/in absolute that occupies one 2

cubic foot. (See NTP.)[3.34]

D

d, value scale division. See "scale division, value of

(d)." [2.20.]

dairy-product-test scale. A scale used in determining

the moisture content of butter and/or cheese or in

determining the butterfat content of milk, cream, or

butter.[2.20]

decreasing-load test. A test for automatic-indicating

scales only, wherein the performance of the scale is

tested as the load is reduced.[2.20]

(Amended 1987)

deficiency. See "excess and deficiency."[1.10]

digital type. A system of indication or recording of

the selector type or one that advances intermittently in

which all values are presented digitally, or in numbers.

In a digital indicating or recording element, or in digital

representation, there are no graduations.[1.10]

direct sale. A sale in which both parties in the

transaction are present when the quantity is being

determined. An unattended automated or customeroperated weighing or measuring system is considered

to represent the device/business owner in transactions

involving an unattended device. [1.10] (Amended

1993)

discharge hose. A flexible hose connected to the

discharge outlet of a measuring device or its discharge

line. [3.30]

(Added 1987)

discharge line. A rigid pipe connected to the outlet of

a measuring device. [3.30]

(Added 1987)
Definitions

D-5

discrimination (of an automatic-indicating scale).

The value of the test load on the load-receiving element

of the scale that will produce a specified minimum

change of the indicated or recorded value on the

scale.[2.20] equal-arm scale. A scale having only a single lever

dispenser. See motor-fuel device.[3.30]

distributed-car test train. A train consisting of cars

weighed first on a reference scale, cars coupled

consecutively in groups at different locations within the

train, then run over the coupled-in-motion railway track

scale under test. The groups are typically placed at the

front, middle, and rear of the train.[2.20]

(Added 1990)

dry-hose type. A type of device in which it is

intended that the discharge hose be completely drained

following the mechanical operations involved in each

delivery. [See "dry hose."] [3.30, 3.34]

dry hose. A discharge hose intended to be completely

drained at the end of each delivery of liquid. [See

"dry-hose type."][3.30]

dynamic monorail weighing system. A weighing

system which employs hardware or software to

compensate for dynamic effects from the load or the

system that do not exist in a static weighing, in order to

provide a stable indication. Dynamic factors may

include shock or impact loading, system vibrations,

oscillations, etc. and can occur even when the load is

not moving across the load receiving element. [2.20]

(Added 1999)

E

e, value of verification scale division. See

"verification scale division, value of (e)." [2.20]

electronic link. An electronic connection between the

weighing/load receiving or other sensing element and

indicating element where one recognizes the other and

neither can be replaced without calibration. [2.20]

(Added 2001)

with equal arms (that is, with a multiple of 1), equipped

with two similar or dissimilar load-receiving elements

(pan, plate, platter, scoop, or the like), one intended to

receive material being weighed and the other intended

to receive weights. There may or may not be a

weighbeam.[2.20]

e (minimum verification scale division). min The

smallest scale division for which a weighing element

complies with the applicable requirements.[2.20, 2.21,

2.24] 

(Added 1997)

event counter. A nonresettable counter that

increments once each time the mode that permits

changes to sealable parameters is entered and one or

more changes are made to sealable calibration or

configuration parameters of a device. [2.20, 3.30]

(Added 1993)

event logger. A form of audit trail containing a series

of records where each record contains the number from

the event counter corresponding to the change to a

sealable parameter, the identification of the parameter

that was changed, the time and date when the parameter

was changed, and the new value of the parameter.

[2.20, 3.30]

(Added 1993)

excess and deficiency. When an instrument or device

is of such a character that it has a value of its own that

can be determined, its error is said to be "in excess" or

"in deficiency," depending upon whether its actual

value is, respectively, greater or less than its nominal

value. (See "nominal.") Examples of instruments

having errors "in excess" are: a linear measure that is

too long; a liquid measure that is too large; and a

weight that is "heavy." Examples of instruments

having errors "in deficiency" are: a lubricating-oil

bottle that is too small; a vehicle tank compartment that

is too small; and a weight that is "light."[1.10]

extras. Charges to be paid by a passenger in addition

to the fare, including any charge at a flat rate for the

transportation of passengers in excess of a stated

number and any charge for the transportation of

baggage. [5.54]

F
Definitions

D-6

face. That side of a taximeter on which passenger gasoline gallon equivalent (GGE). Gasoline gallon

charges are indicated. [5.54] equivalent (GGE) means 5.660 pounds of natural gas.

face. That portion of a computing-type pump or

dispenser which displays the actual computation of

price per unit, delivered quantity, and total sale price. gasoline liter equivalent (GLE). Gasoline liter

In the case of some electronic displays, this may not be equivalent (GLE) means 0.678 kilograms of natural

an integral part of the pump or dispenser.[3.30] gas. [3.37]

(Added 1987) (Added 1994)

fare. That portion of the charge for the hire of a gauge pressure. The difference between the pressure

vehicle that is automatically calculated by a taximeter at the meter and the atmospheric pressure (psi).[3.33]

through the operation of the distance and/or time

mechanism.[5.54]

farm milk tank. A unit for measuring milk or other to be read at the point where the liquid surface crosses

fluid dairy product, comprising a combination of (1) a the rod.[4.43]

stationary or portable tank, whether or not equipped

with means for cooling its contents, (2) means for

reading the level of liquid in the tank, such as a

removable gauge rod or a surface gauge, and (3) a chart

for converting level-of-liquid readings to volume; or

such a unit in which readings are made on gauge rod or

surface gauge directly in terms of volume. Each compartment of a subdivided tank shall, for purposes of

this code, be construed to be a "farm milk tank."][4.43]

feeding mechanism. The means for depositing

material to be weighed on the belt conveyor. [2.21]

fifth wheel. A commercially-available distancemeasuring device which, after calibration, is

recommended for use as a field transfer standard for

testing the accuracy of taximeters and odometers on

rented vehicles.[5.53, 5.54]

fifth-wheel test. A distance test similar to a road test,

except that the distance traveled by the vehicle under

test is determined by a mechanism known as a "fifth-

-wheel" that is attached to the vehicle and that independently measures and indicates the distance.[5.53, 5.54]

flag. A plate at the end of the lever arm or similar part

by which the operating condition of a taximeter is

controlled and indicated.[5.54]

fractional bar. A weighbeam bar of relatively small

capacity for obtaining indications intermediate between

notches or graduations on a main or tare bar.[2.20]

ft /h. Cubic feet per hour.[3.33] 3

G

[3.37]

(Added 1994)

gauge rod. A graduated, "dip-stick" type of measuring

rod designed to be partially immersed in the liquid and

gauging. The process of determining and assigning

volumetric values to specific graduations on the gauge

or gauge rod that serve as the basis for the tank volume

chart.[4.43]

graduated interval. The distance from the center of

one graduation to the center of the next graduation in a

series of graduations. (Also see "value of minimum

graduated interval.")[1.10]

graduation. A defining line, or one of the lines

defining the subdivisions of a graduated series. The

term includes such special forms as raised or indented

or scored reference "lines" and special characters such

as dots. (Also see "main graduation" and "subordinate

graduation.")[1.10]

grain hopper scale. One adapted to the weighing of

individual loads of varying amounts of grain.[2.20]

grain moisture meter. Any device indicating either

directly or through conversion tables and/or correction

tables the moisture content of cereal grains and oil

seeds. Also termed "moisture meter."[5.56]

grain sample. That portion of grain or seed taken

from a bulk of grain or seed to be bought or sold and

used to determine the moisture content of the

bulk.[5.56]

grain-test scale. A scale adapted to weighing grain

samples used in determining moisture content, dockage,

weight per unit volume, etc.[2.20, 5.56]

gravity type. A type of device designed for discharge

by gravity.[3.30]
Definitions

D-7

H

head pulley. The pulley at the discharge end of the

belt conveyor. The power drive to drive the belt is

generally applied to the head pulley.[2.21]

hired. A taximeter is "hired" when it is operative with

respect to all applicable indications of fare or extras.

The indications of fare include time and distance where

applicable unless qualified by another indication of

"Time Not Recording" or an equivalent expression.-

[5.54]

hopper scale. A scale designed for weighing bulk

commodities whose load-receiving element is a tank,

box, or hopper mounted on a weighing element. (Also,

see "automatic hopper scale," "grain hopper scale," and

"construction-material hopper scale."[2.20]

I

idler space. The center-to-center distance between

idler rollers measured parallel to the belt.[2.21]

idlers or idler rollers. Freely turning cylinders

mounted on a frame to support the conveyor belt. For

a flat belt the idlers consist of one or more horizontal

cylinders transverse to the direction of belt travel. For

a troughed belt, the idlers consist of one or more

horizontal cylinders and one or more cylinders at an

angle to the horizontal to lift the sides of the belt to

form a trough.[2.21]

in-service light indicator. A light used to indicate that

a timing device is in operation.[5.55]

increasing-load test. The normal basic performance

test for a scale in which observations are made as

increments of test load are successively added to the

load-receiving element of the scale.[2.20]

increment. The value of the smallest change in value

that can be indicated or recorded by a digital device in

normal operation.[1.10]

index of an indicator. The particular portion of an

indicator that is directly utilized in making a

reading.[1.10]

indicating element. An element incorporated in a

weighing or measuring device by means of which its

performance relative to quantity or money value is

"read" from the device itself as, for example, an

index-and-graduated-scale combination, a weighbeam-and-poise combination, a digital indicator, and

the like. (Also see "primary indicating or recording

element.")[1.10]

indicator, balance. See "balance indicator."[2.20]

initial zero-setting mechanism. Automatic means

provided to set the indication to zero at the time the

instrument is switched on and before it is ready for

use.[2.20]

(Added 1990)

initial distance or time interval. The interval

corresponding to the initial money drop.[5.54]

interval, graduated. See "graduated interval."[1.10]

interval, clear, between graduations. See "clear

interval between graduations."[1.10]

J

jewelers’ scale. One adapted to weighing gems and

precious metals.[2.20]

K

kind of grain. Corn as distinguished from soybeans as

distinguished from wheat, etc.[5.56]

L

label. A printed ticket, to be attached to a package,

produced by a printer that is a part of a prepackaging

scale or that is an auxiliary device.[2.20]

large-delivery device. Devices used primarily for

single deliveries greater than 200 gallons, 2000

pounds, 20 000 cubic feet, 2000 liters, or 2000 kilograms.[3.34]

laundry-drier timer. A timer used in conjunction

with a coin-operated device to measure the period of

time that a laundry drier is in operation.[5.55]

liquefied petroleum gas vapor-measuring device. A

system including a mechanism or device of the meter

type, equipped with a totalizing index, designed to

measure and deliver liquefied petroleum gas in the

vapor state by definite volumes, and generally installed

in a permanent location. The meters are similar in

construction and operation to the conventional naturaland manufactured-gas meters.[3.32]
Definitions

D-8

liquefied petroleum gas. A petroleum product loading point. The location at which material to be

composed predominantly of any of the following conveyed is applied to the conveyor.[2.21]

hydrocarbons or mixtures thereof: propane, propylene,

butanes (normal butane or isobutane), and

butylenes.[3.32, 3.33]

liquefied petroleum gas liquid-measuring device. A

system including a mechanism or machine of the meter

type designed to measure and deliver liquefied petroleum gas in the liquid state by a definite quantity,

whether installed in a permanent location or mounted

on a vehicle. Means may or may not be provided to

indicate automatically, for one of a series of unit prices,

the total money value of the liquid measured.[3.33]

(Amended 1987)

liquid volume correction factor. A correction factor

used to adjust the liquid volume of a cryogenic product

at the time of measurement to the liquid volume at

NBP.[3.34]

liquid-fuel device. A device designed for the measurement and delivery of liquid fuels.[3.30]

liquid-measuring device. A mechanism or machine

designed to measure and deliver liquid by definite

volume. Means may or may not be provided to indicate

automatically, for one of a series of unit prices, the

total money value of the liquid measured, or to make

deliveries corresponding to specific money values at a

definite unit price.[3.30]

liquid fuel. Any liquid used for fuel purposes, that is,

as a fuel, including motor fuel.[3.30]

livestock scale. A scale equipped with stock racks

and gates and adapted to weighing livestock standing

on the scale platform.[2.20]

(Amended 1989)

load-receiving element. That element of a scale that

is designed to receive the load to be weighed; for

example, platform, deck, rail, hopper, platter, plate,

scoop.[2.20]

load cell. A device, whether electric, hydraulic, or

pneumatic, that produces a signal proportional to the

load applied.[2.20]

load cell verification interval (v). The load cell

interval, expressed in units of mass, used in the test of

the load cell for accuracy classification.[2.20, 2.21]

(Added 1996)

low-flame test. A test simulating extremely low-flow

rates such as caused by pilot lights.[3.33]

lubricant device. A device designed for the

measurement and delivery of liquid lubricants,

including, but not limited to, heavy gear lubricants and

automatic transmission fluids (automotive).[3.30]

M

m /h. Cubic meters per hour.[3.33] 3

main-weighbeam elements. The combination of a

main bar and its fractional bar, or a main bar alone if no

fractional bar is associated with it.[2.20]

main bar. A principal weighbeam bar, usually of

relatively large capacity as compared with other bars of

the same weighbeam. (On an automatic-indicating

scale equipped with a weighbeam, the main weighbeam

bar is frequently called the "capacity bar.")[2.20]

main graduation. A graduation defining the primary

or principal subdivisions of a graduated series. [Also

see "graduation."][1.10]

manual zero-setting mechanism. Nonautomatic

means provided to attain a zero balance indication by

the direct operation of a control.[2.20]

manufactured device. Any commercial weighing or

measuring device shipped as new from the original

equipment manufacturer.[1.10]

(Amended 2001)

mass flow meter. A device that measures the mass of

a product flowing through the system. The mass

measurement may be determined directly from the

effects of mass on the sensing unit or may be inferred

by measuring the properties of the product, such as the

volume, density, temperature, or pressure, and

displaying the quantity in mass units.[3.30, 3.32]

master meter test method. A method of testing milk

tanks that utilizes an approved master meter system for

measuring test liquid removed from or introduced into

the tank.[4.43]

master weight totalizer. An indicating element used

with a belt conveyor scale to indicate the weight of

material that was passed over the scale. The master
Definitions

D-9

weight totalizer is a primary indicating element of the

belt-conveyor scale.[2.21]

material test. The test of a belt-conveyor scale using

material (preferably that for which the device is

normally used) that has been weighed to an accuracy of

0.1 percent.[2.21]

(Amended 1989)

maximum capacity. The largest load that may be

accurately weighed. [2.24]

(Added 1999)

maximum cargo load. The maximum cargo load for

trucks is the difference between the manufacturer’s

rated gross vehicle weight and the actual weight of the

vehicle having no cargo load.[5.53]

meter register. An observation index for the

cumulative reading of the gas flow through the meter.

In addition there are one or two proving circles in

which one revolution of the test hand represents 1/2, 1,

2, 5, or 10 cubic feet, or 0.025, 0.05, 0.1, 0.2, or 0.25

cubic meter, depending on meter size. If two proving

circles are present, the circle representing the smallest

volume per revolution is referred to as the "leak-test

circle."[3.33]

metrological integrity (of a device). The design,

features, operation, installation, or use of a device that

facilitates (1) the accuracy and validity of a

measurement or transaction, (2) compliance of the

device with weights and measures requirements, or (3)

the suitability of the device for a given application.

[1.10, 2.20] (Added 1993)

minimum capacity. The smallest load that may be

accurately weighed. The weighing results may be

subject to excessive error if used below this value.

[2.24]

(Added 1999)

minimum totalized load. The least amount of weight

for which the scale is considered to be performing

accurately.[2.21]

minimum tolerances. Minimum tolerances are the

smallest tolerance values that can be applied to a scale.

Minimum tolerances are determined on the basis of the

value of the minimum graduated interval or the nominal

or reading face capacity of the scale. (See also

definition for basic tolerances.)[2.20]

minimum clear interval. The shortest distance

between adjacent graduations when the graduations are

not parallel. (Also see "clear interval.")[3.30]

minimum delivery. The least amount of weight that

is to be delivered as a single weighment by a

belt-conveyor scale system in normal use.[2.21]

moisture content (wet basis). The mass of water in a

grain or seed sample (determined by the reference

method) divided by the mass of the grain or seed

sample expressed as a percentage (%).[5.56]

money-operated type. A device designed to be

released for service by the insertion of money, or to be

actuated by the insertion of money to make deliveries

of product.[1.10]

money drop. An increment of fare indication. The

"initial money drop" is the first increment of fare

indication following activation of the taximeter.[5.54]

motor-fuel device or motor-fuel dispenser or retail

motor-fuel device. A device designed for the

measurement and delivery of liquids used as fuel for

internal-combustion engines. The term "motor-fuel

dispenser" means the same as "motor-fuel device"; the

term "retail motor-fuel device" applies to a unique

category of device (see definition of "retail

device").[3.30]

motor fuel. Liquid used as fuel for internalcombustion engines.[3.30]

multi-interval scale. A scale having one weighing

range which is divided into partial weighing ranges,

each with different scale intervals, with the weighing

range determined automatically according to the load

applied, both on increasing and decreasing loads.

(Added 1995) [2.20]

multi-revolution scale. An automatic-indicating scale

having a nominal capacity that is a multiple of the

reading-face capacity and that is achieved by more than

one complete revolution of the indicator.[2.20]

multiple cell application load cell. A load cell

intended for use in a weighing system which

incorporates more than one load cell. A multiple cell

application load cell is designated with the letter "M"

or the term "Multiple." (See also “single cell

application load cell”) [2.20]

(Added 1999)
Definitions

D-10

multiple of a scale. In general, the multiplying power

of the entire system of levers or other basic weighing

elements. (On a beam scale, the multiple of the scale is

the number of pounds on the load-receiving element

that will be counterpoised by 1 pound applied to the tip

pivot of the weighbeam.)[2.20]

multiple range scale. A scale having two or more

weighing ranges with different maximum capacities

and different scale intervals for the same load receptor,

each range extending from zero to its maximum

capacity. [2.20]

(Added 1995)

multiple-tariff taximeter. One that may be set to

calculate fares at any one of two or more rates.[5.54]

multiple. An integral multiple; that is, a result

obtained by multiplying by a whole number. (Also see

"multiple of a scale.")[1.10]

N

natural gas. A gaseous fuel, composed primarily of

methane, that is suitable for compression and

dispensing into a fuel storage container(s) for use as an

engine fuel. [3.37]

(Added 1994)

NBP. Normal boiling point of a cryogenic liquid at

14.696 lb/in absolute.[3.34] 2

n (maximum number of scale divisions). max The

maximum number of scale divisions for which a main

element or load cell complies with the applicable

requirements. The maximum number of scale divisions

permitted for an installation is limited to the lowest nmax

marked on the scale indicating element, weighing

element, or load cell.[2.20, 2.21, 2.24]

(Added 1997)

no-load reference value. A positive weight value

indication with no load in the load-receiving element

(hopper) of the scale. (Used with automatic

bulk-weighing systems and certain single-draft,

manually-operated receiving hopper scales installed

below grade and used to receive grain.)[2.20]

nominal capacity. The nominal capacity of a scale is

(a) the largest weight indication that can be obtained by

the use of all of the reading or recording elements in

combination, including the amount represented by any

removable weights furnished or ordinarily furnished

with the scale, but excluding the amount represented by

any extra removable weights not ordinarily furnished

with the scale, and excluding also the capacity of any

auxiliary weighing attachment not contemplated by the

original design of the scale, and excluding any

fractional bar with a capacity less than 2-1/2 percent of

the sum of the capacities of the remaining reading

elements, or (b) the capacity marked on the scale by the

manufacturer, whichever is less. (Also see "nominal

capacity, batching scale"; "nominal capacity, hopper

scale.")[2.20]

nominal capacity, batching scale. The nominal

capacity of a batching scale is the capacity as marked

on the scale by the scale manufacturer, or the sum of

the products of the volume of each of the individual

hoppers, in terms of cubic feet, times the weight per

cubic foot of the heaviest material weighed in each

hopper, whichever is less.[2.20]

nominal capacity, hopper scale. The nominal

capacity of a hopper scale is the capacity as marked on

the scale by the scale manufacturer, or the product of

the volume of the hopper in bushels or cubic feet times

the maximum weight per bushel or cubic foot, as the

case may be, of the commodity normally weighed,

whichever is less.[2.20]

nominal. Refers to "intended" or "named" or "stated,"

as opposed to "actual." For example, the "nominal"

value of something is the value that it is supposed or

intended to have, the value that it is claimed or stated to

have, or the value by which it is commonly known.

Thus, "1-pound weight," "1-gallon measure," "1-yard

indication," and "500-pound scale" are statements of

nominal values; corresponding actual values may be

greater or lesser. (See nominal capacity of a

scale)[1.10]

nonretroactive. "Nonretroactive" requirements are

enforceable after the effective date for:

1. devices manufactured within a State after the

effective date;

2. both new and used devices brought into a State after

the effective date; and

3. devices used in noncommercial applications which

are placed into commercial use after the effective date.

Nonretroactive requirements are not enforceable with

respect to devices that are in commercial service in the

State as of the effective date or to new equipment in the

stock of a manufacturer or a dealer in the State as of the

effective date. (Nonretroactive requirements are

printed in italic type.)[1.10]

(Amended 1989)
Definitions

D-11

nose-iron. A slide-mounted, manually-adjustable pivot official with statutory authority. The representative

assembly for changing the multiple of a lever.[2.20] of the jurisdiction(s) responsible for certifying the

notes. A section included in each of a number of

codes, containing instructions, pertinent directives, and

other specific information pertaining to the testing of operating tire pressure. The pressure in a tire

devices. Notes are primarily directed to weights and immediately after a vehicle has been driven for at least

measures officials.[1.10] 5 miles or 8 kilometers.[5.53, 5.54]

NTP density and volume correction factor. A over-and-under indicator. An automatic-indicating

correction factor used to adjust the liquid volume of a element incorporated in or attached to a scale and

cryogenic product at the time of measurement to the comprising an indicator and a graduated scale with a

gas equivalent at NTP.[3.34] central or intermediate "zero" graduation and a limited

NTP. Normal temperature of 21 EC (70 EF) and

pressure of 101.325 kPa (14.696 lb/in absolute) 2

respectively.[3.34]

O

odometer. A device that automatically indicates the

total distance traveled by a vehicle. For the purpose of

this code, this definition includes hub odometers, cable-

-driven odometers, and the distance-indicating or

odometer portions of "speedometer" assemblies for

automotive vehicles.[5.53]

official grain samples. Grain or seed used by the

official as the official transfer standard from the

reference standard method to test the accuracy and

precision of grain moisture meters.[5.56]

accuracy of the device.[2.20, 2.21, 2.22]

(Added 1991)

range of weight graduations on either side of the zero

graduation, for indicating weights greater than and less

than the predetermined values for which other elements

of the scale may be set. (A scale having an

over-and-under indicator is classed as an automatic-indicating scale.)[2.20]

overregistration and underregistration. When an

instrument or device is of such a character that it

indicates or records values as a result of its operation,

its error is said to be in the direction of over

registration or underregistration, depending upon

whether the indications are, respectively, greater or less

than they should be. Examples of devices having errors

of "overregistration" are: a fabric-measuring device

that indicates more than the true length of material

passed through it; and a liquid-measuring device that

indicates more than the true amount of the liquid

delivered by the device. Examples of devices having

errors of "underregistration" are: a meter that indicates

less than the true amount of product that it delivers; and

a weighing scale that indicates or records less than the

true weight of the applied load.[1.10]

P

parallax. The apparent displacement, or apparent

difference in height or width, of a graduation or other

object with respect to a fixed reference, as viewed from

different points.[1.10]

parking meter. A coin-operated device for measuring

parking time for vehicles.[5.55]

passenger vehicles. Vehicles such as automobiles,

recreational vehicles, limousines, ambulances, and

hearses.[5.53]

performance requirements. Performance

requirements include all tolerance requirements and, in

the case of nonautomatic-indicating scales, sensitivity
Definitions

D-12

requirements (SR). (See definitions for "tolerance" and meters or scales so equipped.) The term "primary" is

"sensitivity requirement.")[1.10] not applied to such auxiliary elements as, for example,

point-of-sale system. An assembly of elements

including a weighing or measuring element, an

indicating element, and a recording element (and may

also be equipped with a “scanner”) used to complete a

direct sales transaction. [2.20, 3.30, 3.32, 3.37]

(Added 1986) (Amended 1997)

poise. A movable weight mounted upon or suspended

from a weighbeam bar and used in combination with

graduations, and frequently with notches, on the bar to

indicate weight values. (A suspended poise is proving indicator. The test hand or pointer of the

commonly called a "hanging poise.")[2.20] proving or leak-test circle on the meter register or

portable cubic-foot standard. A gasometer of the

annular type, the bell being sealed with a light oil, the prover test method. A method of testing milk tanks

amount of its rise (and consequently of the volume of that utilizes approved volumetric prover(s) for

air or gas being measured) being under absolute control measuring the test liquid removed from or introduced

so that an exact cubic foot can be delivered.[3.33] into the tank.[4.43]

postal scale. A scale (usually a computing scale)

designed for use to determine shipping weight or

delivery charges for letters or parcels delivered by the

U.S. Postal Service or private shipping companies. A

weight classifier may be used as a postal scale.[2.20]

(Added 1987)

prepackaging scale. A computing scale specially

designed for putting up packages of random weights in

advance of sale.[2.20]

prescription scale. A scale or balance adapted to

weighing the ingredients of medicinal and other

formulas prescribed by physicians and others and used

or intended to be used in the ordinary trade of

pharmacists.[2.20]

pressure type (device). A type of device designed for

operation with the liquid under artificially produced

pressure.[3.30]

primary indicating or recording elements. The term

"primary" is applied to those principal indicating

(visual) elements and recording elements that are

designed to, or may, be used by the operator in the

normal commercial use of a device. The term

"primary" is applied to any element or elements that

may be the determining factor in arriving at the sale

representation when the device is used commercially.

(Examples of primary elements are the visual indicators

for meters or scales not equipped with ticket printers or

other recording elements and both the visual indicators

and the ticket printers or other recording elements for

the totalizing register or predetermined-stop mechanism

on a meter or the means for producing a running record

of successive weighing operations, these elements

being supplementary to those that are the determining

factors in sales representations of individual deliveries

or weights. (See "indicating element" and "recording

element.")[1.10]

prover oil. A light oil of low vapor pressure used as a

sealing medium in bell provers, cubic-foot bottles, and

portable cubic-foot standards.[3.33]

index.[3.33]

R

T HCEVQT#EQORWVCVKQPHQTFGVGTOKPKPIVJG

UWKVCDKNKV[QHCXGJKENGUECNGHQTYGKIJKPIXGJKENGU

YKVJXCT[KPICZNGEQPHKIWTCVKQPU6JGHCEVQTYCU

FGTKXGFD[FKXKFKPIVJGYGKIJVUKP(*9#(GFGTCN

*KIJYC[$TKFIG)TQUU9GKIJV6CDNG$D[

NDU
6JGTGUWNVCPVHCEVQTUEQPVCKPGFKP6CDNG

74=?

radio frequency interference (RFI). Radio frequency

interference is a type of electrical disturbance that,

when introduced into electronic and electrical circuits,

may cause deviations from the normally expected

performance.[1.10]

ranges, weight. See "weight ranges."[2.20]

rated scale capacity. That value representing the

weight that can be delivered by the device in one hour.-

[2.21]

rated capacity. The rate of flow in cubic meters per

hour of a hydrocarbon gas vapor-measuring device as

recommended by the manufacturer. This rate of flow

should cause a pressure drop across the meter not

exceeding 1/2-inch water column.[3.33]

ratio test. A test to determine the accuracy with which

the actual multiple of a scale agrees with its designed

multiple. This test is used for scales employing

counterpoise weights and is made with standard test

weights substituted in all cases for the weights
Definitions

D-13

commercially used on the scale. (It is appropriate to

use this test for some scales not employing

counterpoise weights.)[2.20]

reading-face capacity. The largest value that may be

indicated on the reading face, exclusive of the

application or addition of any supplemental or

accessory elements.[1.10]

reading face. That portion of an automatic-indicating

weighing or measuring device that gives a visible

indication of the quantity weighed or measured. A

reading face may include an indicator and a series of

graduations or may present values digitally, and may

also provide money-value indications.[1.10]

recorded representation. The printed, embossed, or (Added 1993)

other representation that is recorded as a quantity by a

weighing or measuring device.[1.10]

recording element. An element incorporated in a single deliveries of less than 378 L (100 gal),

weighing or measuring device by means of which its

performance relative to quantity or money value is per- retail deliveries of motor fuels to individual

manently recorded on a tape, ticket, card, or the like, in highway vehicles, or

the form of a printed, stamped, punched, or perforated

representation.[1.10, 2.21] single deliveries of liquefied petroleum gas for

recording scale. One on which the weights of applied

loads may be permanently recorded on a tape, ticket,

card, or the like in the form of a printed, stamped,

punched, or perforated representation.[2.20] retroactive. "Retroactive" requirements are

reference weight car. A railroad car weighed on a

scale for temporary use as a mass standard over a short

period of time (typically, the time required to test one

scale) as part of a test train. road test. A distance test, over a measured course, of

Note: A test weight car that is representative of the vehicle, the mechanism being actuated as a result of

types of cars typically weighed on the scale under test vehicle travel. [5.53]

may be used wherever reference weight cars are

specified. [2.20]

(Added 1991)

remanufactured device. A device that is odometer. If more than one wheel actuates the

disassembled, checked for wear, parts replaced or taximeter or odometer, the rolling circumference is the

fixed, reassembled and made to operate like a new average distance traveled per revolution of the

device of the same type. [1.10] actuating wheels.[5.53, 5.54]

(Added 2001)

remanufactured element. An element that is

disassembled, checked for wear, parts replaced or

fixed, reassembled and made to operate like a new

element of the same type.

(Added 2001) [1.10]

repaired device. A device to which work is performed

that brings the device back into proper operating

condition.

(Added 2001) [1.10]

repaired element. An element to which work is

performed that brings the element back into proper

operating condition. 

(Added 2001) [1.10]

remote configuration capability. The ability to

adjust a weighing or measuring device or change its

sealable parameters from or through some other device

that is not itself necessary to the operation of the

weighing or measuring device or is not a permanent

part of that device. [2.20, 3.30]

retail device. A device used for:

domestic use and liquified petroleum gas or liquid

anhydrous ammonia for nonresale use.

(Amended 1987) [3.32]

enforceable with respect to all equipment. Retroactive

requirements are printed herein in upright roman type.

(Also see "nonretroactive.")[1.10]

a complete taximeter assembly when installed on a

rolling circumference. The rolling circumference is

the straight line distance traveled per revolution of the

wheel (or wheels) that actuates the taximeter or
n ' Capacity

e

Definitions

D-14

S

scale area, belt-conveyor. See belt-conveyor scale

systems area. [2.21]

(Added 2001)

scale division, number of (n). Quotient of the

capacity divided by the value of the verification scale

division. [2.20]

scale division, value of (d). The value of the scale

division, expressed in units of mass, is the smallest

subdivision of the scale for analog indication or the

difference between two consecutively indicated or

printed values for digital indication or printing. (Also

see "verification scale division.")[2.20]

scale section. A part of a vehicle, axle-load, livestock,

or railway track scale consisting of two main load

supports, usually transverse to the direction in which

the load is applied.[2.20]

scale. See specific type of scale.[2.20]

seal. See "approval seal," "security seal."[1.10]

section capacity. The section capacity of a scale is the

maximum live load that may be divided equally on the

load pivots or load cells of a section.

(Added 2001)

section test. A shift test in which the test load is

applied over individual sections of the scale. This test

is conducted to disclose the weighing performance of

individual sections, since scale capacity test loads are

not always available and loads weighed are not always

distributed evenly over all main load supports.[2.20]

 

security means. A method used to prevent access by

other than qualified personnel, or to indicate that access

has been made to certain parts of a scale that affect the

performance of the device.[2.21]

security seal. A uniquely identifiable physical seal,

such as a lead-and-wire seal or other type of locking

seal, a pressure-sensitive seal sufficiently permanent to

reveal its removal, or similar apparatus attached to a

weighing or measuring device for protection against or

indication of access to adjustment. (Also see "approval

seal.")[1.10]

(Amended 1994)

selector-type. A system of indication or recording in

which the mechanism selects, by means of a

ratchet-and-pawl combination or by other means, one

or the other of any two successive values that can be

indicated or recorded.[1.10]

semi-automatic zero-setting mechanism. Automatic

means provided to attain a direct zero balance

indication requiring a single initiation by an

operator.[2.20]

sensitivity requirement (SR). A performance

requirement for a non automatic-indicating scale;

specifically, the minimum change in the position of rest

of the indicating element or elements of the scale in

response to the increase or decrease, by a specified

amount, of the test load on the load-receiving element

of the scale.[2.20]

sensitivity (of a nonautomatic-indicating scale). The

value of the test load on the load-receiving element of

the scale that will produce a specified minimum change

in the position of rest of the indicating element or

elements of the scale.[2.20]

shift test. A test intended to disclose the weighing

performance of a scale under off-center loading.[2.20]

side. That portion of a pump or dispenser which faces

the consumer during the normal delivery of

product.[3.30]

(Added 1987)

simulated-road test. A distance test during which the

taximeter or odometer may be actuated by some means

other than road travel. The distance traveled is either

measured by a properly calibrated roller device or

computed from rolling circumference and wheel-turn

data.[5.53, 5.54]

simulated test. A test using artificial means of loading

the scale to determine the performance of a

belt-conveyor scale.[2.21]

single cell application load cell. A load cell intended

for use in a weighing system which incorporates one or

more load cells. A single cell application load cell is 
Definitions

D-15

designated with the letter "S" or the term "Single."

(See also “multiple cell application load cell”) [2.20]

(Added 1999)

single-tariff taximeter. One that calculates fares at a

single rate only.[5.54]

skirting. Stationary side boards or sections of belt

conveyor attached to the conveyor support frame or

other stationary support to prevent the bulk material

from falling off the side of the belt.[2.21]

slow-flow meter. A retail device designed for the

measurement, at very slow rates (less than 40 L (10 gal)

per hour), of liquid fuels at individual domestic installations.[3.30]

small-delivery device. Any device other than a

large-delivery device.[3.34]

span (structural). The distance between adjoining

sections of a scale.[2.20]

(Added 1988)

specification. A requirement usually dealing with the

design, construction, or marking of a weighing or

measuring device. Specifications are directed primarily

to the manufacturers of devices.[1.10]

static monorail weighing system. A weighing system

in which the load being applied is stationary during the

weighing operation. [2.20]

(Added 1999)

strain-load test. The test of a scale beginning with the

scale under load and applying known test weights to

determine accuracy over a portion of the weighing

range. The scale errors for a strain-load test are the

errors observed for the known test loads only. The

tolerances to be applied are based on the known test

load used for each error that is determined.[2.20, 2.22]

subordinate graduation. Any graduation other than

a main graduation. (Also see "graduation.")[1.10]

subsequent distance or time intervals. The intervals

corresponding to money drops following the initial

money drop.[5.54]

surface gauge. A combination of (1) a stationary

indicator, and (2) a movable, graduated element

designed to be brought into contact with the surface of

the liquid from above.[4.43]

T

tail pulley. The pulley at the opposite end of the

conveyor from the head pulley.[2.21]

take-up. A device to provide sufficient tension in a

conveyor belt so that the belt will be positively driven

by the drive pulley. A counter-weighted take-up

consists of a pulley free to move in either the vertical or

horizontal direction with dead weights applied to the

pulley shaft to provide the tension required.[2.21]

tare-weighbeam elements. The combination of a tare

bar and its fractional bar, or a tare bar alone if no

fractional bar is associated with it.[2.20]

tare mechanism. A mechanism (including a tare bar)

designed for determining or balancing out the weight of

packaging material, containers, vehicles, or other materials that are not intended to be included in netweight

determinations.[2.20]

taximeter. A device that automatically calculates, at a

predetermined rate or rates, and indicates the charge for

hire of a vehicle.[5.54]

testing. An operation consisting of a series of

volumetric determinations made to verify the accuracy

of the volume chart that was developed by

gauging.[4.43]

test liquid. The liquid used during the test of a

device.[3.30]

test chain. A device used for simulated tests

consisting of a series of rollers or wheels linked

together in such a manner as to assure uniformity of

weight and freedom of motion to reduce wear, with

consequent loss of weight, to a minimum.[2.21]

test train. A train consisting of or including reference

weight cars and used to test coupled-in-motion railway

track scales. The reference weight cars may be placed

consecutively or distributed in different places within

a train.[2.20]

(Added 1990)(Amended 1991)

test weight car. A railroad car designed to be a stable

mass standard to test railway track scales. The test

weight car may be one of the following types: a selfcontained composite car, a self-propelled car, or a

standard rail car.[2.20]

(Added 1991)
Definitions

D-16

time recorder. A clock-operated mechanism designed unit train. A unit train is defined as a number of

to record the time of day. Examples of time recorders contiguous cars carrying a single commodity from one

are those used in parking garages to record the "in" and consignor to one consignee. The number of cars is

"out" time of day for parked vehicles.[5.55] determined by agreement among the consignor,

timing device. A device used to measure the time

during which a particular paid-for service is dispensed. user requirement. A requirement dealing with the

Examples of timing devices are laundry driers, selection, installation, use, or maintenance of a

car-wash timers, parking meters, and parking-garage weighing or measuring device. User requirements are

clocks and recorders.[5.55] directed primarily to the users of devices.[1.10]

tolerance. A value fixing the limit of allowable error usual and customary. Commonly or ordinarily found

or departure from true performance or value. (See also in practice or in the normal course of events and in

"basic tolerances.")[1.10] accordance with established practices.[1.10]

training idlers. Idlers of special design or mounting

intended to shift the belt sideways on the conveyor to

assure the belt is centered on the conveying

idlers.[2.21]

transfer standard. A measurement system designed

for use in proving and testing cryogenic

liquid-measuring devices.[3.34]

tripper. A device for unloading a belt conveyor at a

point between the loading point and the head

pulley.[2.21]

U

uncoupled-in-motion railroad weighing system. A

device and related installation characteristics consisting

of (1) the associated approach trackage, (2) the scale

(i.e., the weighing element, the load-receiving element, verification scale division, value of (e). A value,

and the indicating element with its software), and (3) expressed in units of weight (mass) and specified by

the exit trackage which permit the weighing of railroad the manufacturer of a device, by which the tolerance

cars uncoupled in motion. [2.20] (Added 1993) values and the accuracy class applicable to the device

underregistration. See "overregistration" and

"underregistration."[1.10]

unit price. The price at which the product is being

sold and expressed in whole units of

measurement.[3.30]

(Added 1992)

unit weight. One contained within the housing of an

automatic-indicating scale and mechanically applied to

and removed from the mechanism. The application of

a unit weight will increase the range of automatic

indication, normally in increments equal to the

reading-face capacity.[2.20]

consignee, and the operating railroad.[2.20]

V

value of minimum graduated interval. The value

represented by the interval from the center of one

graduation to the center of the succeeding graduation.

Also, the increment between successive recorded

values. (Also see "graduated interval.")[1.10]

vehicle on-board weighing system. A weighing

system designed as an integral part of or attached to the

frame, chassis, lifting mechanism, or bed of a vehicle,

trailer, industrial truck, industrial tractor, or forklift

truck. [2.20] (Amended 1993)

vehicle scale. A scale adapted to weighing highway,

farm, or other large industrial vehicles (except railroad

freight cars), loaded or unloaded.[2.20]

are determined. The verification scale division is

applied to all scales, in particular to ungraduated

devices since they have no graduations. The

verification scale division, e, may be different from the

displayed scale division, d, for certain other devices

used for weight classifying or weighing in

pre-determined amounts, and certain other Class I and

II scales.[2.20]

visible type. A type of device in which the measurement takes place in a see-through glass measuring

chamber.[3.30]

v (minimum load cell verification interval). min The

smallest load cell verification interval, expressed in

units of mass* into which the load cell measuring range

can be divided. [2.20, 2.21, 2.24]
Definitions

D-17

[*Nonretroactive as of January 1, 2001] material off the underside (return side) of the

(Added 1996) (Amended 1999) belt.[2.21]

W Z

weighbeam. An element comprising one or more bars,

equipped with movable poises or means for applying

counterpoise weights or both.[2.20]

weighing element. That portion of a scale that supports

the load-receiving element and transmits to the

indicating element a signal or force resulting from the

load applied to the load-receiving element.[2.20]

(Added 1988)

weighment. A single complete weighing

operation.[2.20, 2.21]

(Added 1986)

weight, unit. See "unit weight."[2.20]

weight classifier. A digital scale that rounds weight

values up to the next scale division. These scales

usually have a verification scale division, e, that is

smaller than the displayed scale division.[2.20]

(Added 1987)

weight ranges. Electrical or electro-mechanical

elements incorporated in an automatic indicating scale

through the application of which the range of automatic

indication of the scale is increased, normally in

increments equal to the reading-face capacity.[2.20]

wet basis. See "moisture content (wet basis)."[5.56]

wet hose. A discharge hose intended to be full of

liquid at all times. [See "wet-hose type."][3.30]

wet-hose type. A type of device designed to be

operated with the discharge hose full of liquid at all

times. [See "wet hose."][3.30]

wheel-load weighers. Compact, self-contained,

portable weighing elements specially adapted to

determining the wheel loads or axle loads of vehicles

on highways for the enforcement of highway weight

laws only.[2.20]

wholesale device. Any device other than a retail

device. [See "retail device."][3.30, 3.32] 

wing pulley. A pulley made of widely spaced metal

bars in order to set up a vibration to shake loose

zero-load balance. A correct weight indication or

representation of zero when there is no load on the

load-receiving element. (See also "zero-load balance

for an automatic-indicating scale," "zero-load balance

for a nonautomatic-indicating scale," "zero-load

balance for a recording scale.")[2.20]

zero-load balance, automatic-indicating scale. A

condition in which the indicator is at rest at, or

oscillates through approximately equal arcs on either

side of, the zero graduation.[2.20]

zero-load balance, nonautomatic-indicating scale.

A condition in which (a) the weighbeam is at rest at, or

oscillates through approximately equal arcs above and

below, the center of a trig loop; (b) the weighbeam or

lever system is at rest at, or oscillates through

approximately equal arcs above and below, a horizontal

position or a position midway between limiting stops;

or (c) the indicator of a balance indicator is at rest at, or

oscillates through approximately equal arcs on either

side of, the zero graduation.[2.20]

zero-load balance for a recording scale. A condition

in which the scale will record a representation of zero

load.[2.20]

zero-setting mechanism. Means provided to attain a

zero balance indication with no load on the

load-receiving element. Three types of these

mechanisms are:[2.20]

 

manual zero-setting mechanism.

Nonautomatic means provided to attain a zero

balance indication by the direct operation of a

control.[2.20]

semiautomatic zero-setting mechanism.

Automatic means provided to attain a direct zero

balance indication requiring a single initiation by

an operator.[2.20]

automatic zero-setting mechanism. Automatic

means provided to maintain zero balance

indication without the intervention of an

operator.[2.20]

zone of uncertainty. The zone between adjacent

increments on a digital device in which the value of
Definitions

D-18

either of the adjacent increments may be

displayed.[2.20]
United States

Department of

Commerce

SPECIFICATIONS,

Technology

Administration

National

Institute of

Standards and

Technology

TOLERANCES, AND

OTHER TECHNICAL

REQUIREMENTS FOR

WEIGHING AND

MEASURING DEVICES

as adopted by the 86th

National Conference on

Weights and Measures 2001

NIST

Handbook44

2002