Operating Instructions

Bedienungsanleitung

Mode d‘emploi

Instrucciones de manejo

Istruzioni d‘uso

SevenEasyTM

pH Meter S20

 Italiano Español Français Deutsch English

METTLER TOLEDO SevenEasy 1

Contents

Contents

1. Introduction 3

2. Safety measures 3

Measures for your protection 3

Measures for operational safety 3

3. Description of the instrument 4

3.1 Display 4

3.2 Keypad 5

4. Installation 6

5. Sample Measurement 8

5.1 pH measurement 8

5.2 mV measurement 8

5.3 Settings 8

5.3.1 ATC 8

5.3.2 MTC 8

5.3.3 Data output 8

6. Calibration 9

6.1 Settings 9

6.2 Select a fixed buffer group 9

6.3 Set a user-defined buffer group 9

6.4 Calibration 10

6.4.1 1-Point calibration 10

6.4.2 2-Point calibration 10

6.4.3 3-Point calibration 10

7. Self-Diagnosis 11

8. Optional battery operation 11

9. Error messages 12

10. Maintenance 13

10.1 Meter maintenance 13

10.2 Electrode maintenance 13

10.3 Disposal 13

11. Accessories 14

12. Specifications 15

13. Quick Guide 16

14. Buffer sets 17
2 METTLER TOLEDO SevenEasy
METTLER TOLEDO SevenEasy 3

Introduction / Safety measures

1. Introduction

SevenEasy – an instrument which offers far more than simple pH measurement, without breaking your 

budget. An instrument with so many advantages: 

- SevenEasy saves you time. The user interface is designed in such a logical way that you will no 

longer need to consult your user manual. 

- SevenEasy can be battery operated. Thanks to this option you can now easily move your instrument 

from one working area to another even when no power supply is available. 

- SevenEasy has added value. Our Service Option allows for regular Equipment Qualification which 

will improve the reliability and accuracy of your instrument. 

2. Safety measures

Measures for your protection

- Never work in an environment subject to explosion hazards! The housing of the instrument is not gas tight (explosion hazard due to spark formation, corrosion caused by 

the ingress of gases).

- When using chemicals and solvents, comply with the instructions of the producer and 

the general lab safety rules!

Measures for operational safety

- Have the instrument serviced only by METTLER TOLEDO Service!

- Always wipe off splashed liquids immediately! The instrument is not waterproof.

- Use batteries of the specified type only. Otherwise, proper operation cannot be guaranteed.

- Exclude the following environmental influences:

- powerful vibrations,

- direct sunlight,

- atmospheric humidity greater than 80%,

- corrosive gases present,

- temperatures below 5 °C and above 40 °C,

- powerful electric or magnetic fields!
4 METTLER TOLEDO SevenEasy

Description of the instrument

1 2 3 4 5 6

8

11

13

7

10

12

9

3. Description of the instrument

3.1 Display

1 Battery status 

2 Auto-off override during battery operation

3 Data transfer to PC/Printer 

4 Electrode condition

 slope: 95-105 % slope: 90-94 % slope: 85-89 %

 offset: ± (0-15) mV offset: ± (15-35) mV offset: ± (>35) mV

 Electrode is in Electrode needs Electrode is faulty

 good condition cleaning

5 Meter self-diagnosis

 Self-diagnosis Indication to Self-diagnosis 

 indicator press key passed

6 Error index 

7 pH/mV-reading

8 Auto/Manual Temperature compensation

9 Endpoint stability/autom endpoint

 Endpoint stability A Auto endpoint

10 Temperature 

11 Calibration point

12 Buffer groups

13 Electrode offset and slope
METTLER TOLEDO SevenEasy 5

Description of the instrument

3.2 Keypad

Press & release Press & hold for 2 seconds 

Meter on/off Auto-off override during battery 

operation

- Start or endpoint measurement

- Return to measurement mode

- Store entered value

Turn auto endpoint on/off / 

- Start calibration

- Confirm buffer group selected

Displaying the buffers used for the 

calibration

Switch between pH- and mV measurement 

mode

Data transfer to PC or printer

- Select calibration buffer groups

- Increase value during setting

- Set MTC-temperature

- Decrease value during setting

Start meter self-diagnosis 
6 METTLER TOLEDO SevenEasy

4. Installation

1. Unpack the meter, power adapter, electrode, electrode arm and other accessories. Keep the calibration certificate in a safe place.

2. Make sure the power adapter matches your local power supply. If not, please contact your vendor.

3. Install the electrode stand

- The electrode stand can be used stand alone or it can be attached 

to the instrument. The electrode arm is placed in one of the three 

positions on the base. If the electrode stand is used stand alone it is 

recommended to use the middle position. 

 The appropriate lid needs to be removed. Use the fixing screw supplied to tighten the connection. Installation is finished if the electrode 

stand is used stand alone.

- If the electrode stand is attached to the instrument, place the base 

in such a way that the free wing of the base faces away from you. 

The appropriate lid needs to be removed. The electrode stand can 

be used on the left or right hand side.

- Adjust the tension knob as required. Loosen

Tighten

Installation
METTLER TOLEDO SevenEasy 7

Installation

DC power supply unit socket

ATC (Automatic temperature compensation) 

probe socket

pH combination electrode socket

RS232 data output socket

4. Disconnect the shorting clip from the pH socket.

5. Connect the electrode. If you are using an electrode with a built-in temperature probe, connect the 

other lead to the ATC socket.

6. If you are using a separate temperature probe, connect it to the ATC socket.

7. Connect the power supply unit to the DC socket.
8 METTLER TOLEDO SevenEasy

5. Sample Measurement

5.1 pH measurement

Place electrode in sample and press to start the measurement: the decimal point flashes.

The display shows the pH of the sample. The automatic endpoint A is the meter‘s default setting. When 

the sensor output has stabilized, the display freezes automatically, and appears.

The automatic endpoint algorithm is 0.1 mV per 5 seconds.

By pressing and holding the key, you can switch between auto and manual endpoint mode. To 

manually endpoint a measurement, press , the display freezes and appears.

5.2 mV measurement

To perform a mV measurement, follow the same procedure as for pH measurement. To see the mV 

value during pH measurement, simply press .

5.3 Settings

5.3.1 ATC

For better accuracy, we recommend the use of either a built-in or a separate temperature probe. When 

a temperature probe is used, the symbol ATC and the sample temperature are displayed.

5.3.2 MTC

When the meter does not detect a temperature probe, it automatically switches to manual temperature 

compensation mode, and MTC appears.

To set the MTC temperature, press to start, use and to increase or decrease the value to the 

temperature of your sample. Press the key to confirm your setting. The default setting is 25 °C.

5.3.3 Data output 

If a PC or printer is connected, every endpointed reading is sent to PC or printed through the RS232 

interface.

By pressing and holding the key appears. The meter sends out a reading every second until 

it has endpointed.

Measurement
METTLER TOLEDO SevenEasy 9

6. Calibration

6.1 Settings

The SevenEasy pH meter allows you to perform 1-, 2- and 3-point calibrations. If you select your calibration buffer group from the 4 fixed groups defined in the meter, the buffers are automatically recognized 

during calibration (auto buffer recognition).

The 4 fixed buffer groups are:

B1: (25 °C) 7.00 4.00 10.01 1.68

B2: (25 °C) 7.00 4.01 9.21 2.00 11.00

B3: (20 °C) 7.00 4.00 9.00 2.00 12.00

B4: (25 °C) 6.86 4.01 9.18 1.68 

You can also follow the buffer setting procedure below to define your own buffer group, but in this case 

the auto buffer recognition will not work during calibration.

6.2 Select a fixed buffer group

Press the key, the current buffer group starts blinking. If the current setting is the user-defined group, 

the blank frame blinks.

Use the or key to select other groups in the upper or lower position. When the desired buffer 

group blinks, press to confirm your selection.

6.3 Set a user-defined buffer group

In step 6.2, when the blank frame blinks, press to start setting. The meter displays your current 

temperature value setting and the dot and frame blink (default temperature is 25 °C). Use or 

key to change the value. Press to store the value and continue.

After setting the temperature value, the meter displays the current setting of the first calibration buffer (default 

value 4.00). Use or key to change the value. Press to store the value and continue.

After setting the first calibration buffer, press to start setting the next point. 

The procedure is the same as for the 1st point.You can set up to 3 user-defined calibration buffers. When 

you have completed your settings, press to exit.

Note

When using a user-defined buffer group in the calibration, the screen will display the buffer value you 

set. Make sure that you are using the correct buffer. You should also keep the buffer temperature at the 

set value. When a temperature probe is used, if the temperature measured differs by more than 1 degree 

from the set value, Err 5 appears.

Calibration
10 METTLER TOLEDO SevenEasy

6.4 Calibration

6.4.1 1-Point calibration

Place the electrode in a calibration buffer and press .

The SevenEasy pH meter automatically endpoints when calibrating. To manually endpoint, press . 

The meter displays and freezes the relevant buffer value, and shows the electrode offset.

To return to sample measurement, press .

6.4.2 2-Point calibration

Step 1 Perform the first point calibration as described in “1-Point calibration”.

Step 2 Use distilled water to rinse the electrode.

Step 3 Place the electrode in the next calibration buffer and press .

The SevenEasy pH meter automatically endpoints when calibrating. To manually endpoint, press . The 

meter displays and freezes the relevant buffer value, updates the electrode offset and shows its slope.

To return to sample measurement, press .

6.4.3 3-Point calibration 

Perform the same steps as in the “2-Point calibration”, then repeat steps 2 and 3 for the third point 

calibration.

Note

The use of a temperature probe or electrode with a built-in temperature probe is recommended. If you 

use the MTC mode, you should keep all buffer and sample solutions at the same set temperature.

To ensure the most accurate pH readings, you should perform a calibration regularly.

By pressing and holding the key, the number and type of buffers used for the calibration are displayed. The buffer values appear in alternating order.

Calibration
METTLER TOLEDO SevenEasy 11

7. Self-Diagnosis

Press and hold and simultaneously until the meter self-diagnosis icon appears.

The meter displays the full screen first, then each icon blinks one after the other. The final step is to check 

that the keys function. This requires the user‘s cooperation. 

When the icon blinks, depending on the flashing content, press the corresponding key within 

5 seconds.

a. When flashes, press the key;

b. When flashes, press the key;

c. When flashes, press the pH/mV key;

d. When flashes, press the key;

e. When flashes, press the key.

When self-diagnosis is completed, a tick icon appears. If the self-diagnosis fails, turn to “9. Error 

Messages” in these operating instructions for the proper action(s).

8. Optional battery operation

The SevenEasy pH meter offers optional battery operation. Install 4 AA batteries in the rear of the meter. 

If the power adapter is disconnected, the meter is operated by battery and the icon appears.

When the batteries lose power, the meter displays .

During battery operation, the meter has an auto-off function - if no key is pressed during the next 10 

minutes, the meter will automatically switch off to save battery power. To override the auto-off function, 

press and hold key for 2 seconds until appears.

Self-Diagnosis/Battery operation
12 METTLER TOLEDO SevenEasy

9. Error messages

Error 1 - Offset out of range

Make sure you have the correct buffer and that it is fresh.

Clean or replace the electrode.

Error 2 - Slope out of range

Make sure you have the correct buffer and that it is fresh.

Clean or replace the electrode.

Error 3 - Meter cannot recognize the buffer

Make sure you have the correct buffer and that it is fresh.

Check whether the buffer is used more than once in the calibration.

Error 4 - Data entry error in setting the user-defined buffer

When entering the user-defined buffer value, the meter doesn‘t accept a value whose pH differs by less 

than 1 pH unit from other pre-set values. Re-enter a value.

Error 5 - ATC measured temperature is different from the user-defined value.

Keep the buffer or sample at the set temperature or change the temperature setting.

Error 6 - the measured buffer temperature is out of the range (5... 50°C)

Keep the buffer temperature within the range.

Error 7, (----) - mV value out of range

Make sure the electrode is connected.

If no electrode is connected, place the shorting clip in the socket.

Error 8 - pH measuring value out of range

Check if the electrode wetting cap is removed and if the electrode is properly connected and placed in 

a sample solution.

Error 9 - Self-diagnosis failed

Repeat self-diagnosis process and make sure that you press the correct keys while the icon is 

blinking. If Err 9 still appears, call METTLER TOLEDO service.

Error messages
METTLER TOLEDO SevenEasy 13

10. Maintenance

10.1 Meter maintenance

There are no user-replaceable parts in the meter or power supply unit. Do not remove the covers. 

The SevenEasy needs no maintenance except for an occasional wipe with a damp cloth. The housing 

is made of ABS/PC which is attacked by some organic solvents, such as toluene, xylene and methyl 

ethyl ketone. It is good laboratory practice to wipe away any spillage immediately.

10.2 Electrode maintenance

Make sure the electrode is always kept filled with the appropriate filling solution. For maximum accuracy, any filling solution that may have “crept” and encrusted the outside of the electrode should be 

removed with distilled water.

Always store the electrode properly and do not allow it to dry out.

If the electrode slope value falls rapidly, or if the response becomes sluggish or inaccurate, the following 

procedures may help. Try them one by one in the order given.

1. Degrease the membrane with cotton wool soaked in either acetone or soap solution.

2. Soak the tip of the electrode in 0.1 M HCl overnight.

3. If a protein build-up has occurred, remove deposits by soaking electrode in a HCl/pepsin solution 

(Order Nr. 51340068).

4. If a silver sulfide contamination has occurred, remove deposits by soaking electrode in a thiourea 

solution (Order Nr. 51340070).

Note

Cleaning and filling solutions should be handled with the same care given to toxic or corrosive substances.

10.3 Disposal

In conformance with the European Directive 2002/96/ EC on Waste Electrical and Electronic 

Equipment (WEEE) this device may not be disposed of in domestic waste. This also applies 

to countries outside the EU, per their specific requirements.

Please dispose of this product in accordance with local regulations at the collecting point 

specified for electrical and electronic equipment.

If you have any questions, please contact the responsible authority or the distributor from which you 

purchased this device.

Should this device be passed on to other parties (for private or professional use), the content of this 

regulation must also be related.

Thank you for your contribution to environmental protection.

Maintenance
14 METTLER TOLEDO SevenEasy

11. Accessories

Order Nr.

Electrode arm complete 51302820

Guide to pH measurement 51300047

Guide to conductivity and dissolved oxygen 51724716

Guide to ion selective measurement 51300075

pH 4.01 buffer sachets, 30 x 20mL 51302069

pH 4.01 buffer solution, 6 x 250mL 51340058

pH 7.00 buffer sachets, 30 x 20mL 51302047

pH 7.00 buffer solution, 6 x 250mL 51340060

pH 9.21 buffer sachets, 30 x 20mL 51302070

pH 9.21 buffer solution, 6 x 250mL 51300194

pH 10.01 buffer sachets, 30 x 20mL 51302079

pH 10.01 buffer solution, 6 x 250mL 51340231

InLab®410, three-in-one pH sensor, glass, refillable 52000118

InLab®412, laboratory pH sensor, glass, refillable 52000112

InLab®413, rugged 3-in-1 pH sensor, PEEK shaft, ATC 52000100

InLab®420, special pH sensor with PTFE junction 52000113

HCl/Pepsin solution (removes protein contamination) 51340068

Thiourea solution (removes silver sulfide contamination) 51340070

Reactivation solution for pH electrodes 51340073

Accessories
METTLER TOLEDO SevenEasy 15

Specifications

12. Specifications

pH mV Temperature

Measurement Range 0.00 - 14.00 ± 1999 –5 ... 105 °C

Resolution 0.01 1 0.1 °C

Relative Accuracy ± 0.01 ± 1 ± 0.5 °C

pH-calibration 1, 2, or 3 points

Isopotential Point 7.00 pH

Calibration Buffer B1: (25 °C) 7.00 4.00 10.01 1.68

B2: (25 °C) 7.00 4.01 9.21 2.00 11.00

B3: (20 °C) 7.00 4.00 9.00 2.00 12.00

B4: (25 °C) 6.86 4.01 9.18 1.68

B5: User-defined

Temperature Compensation –5 ... 105 °C automatic or manual

Display Liquid crystal

Outputs RS232 serial, 

Baud rate: 1200

Data bit: 8

Stop bit: 1

Parity: none

pH input Impedance > 1012 Ohm

Ambient Conditions Ambient temperature: 5 °C ... 40 °C

Relative humidity: 5 % ... 80 % (non condensing)

Installation category: II

Pollution degree: 2

Size/Weight 180 x 180 x 65 mm / 0.61 kg

Materials Housing: ABS, PC enforced

Electrode stand: ABS, PC enforced

Membrane keypad: Polyester

Power Requirements The SevenEasy pH meter is supplied with an appropriate 

power supply unit:

- USA: 120 V / 60 Hz, 10 VA, 9 V DC

- Europe: 230 V / 50 Hz, 10 VA, 9 V DC

- UK: 240 V / 50 Hz, 10 VA, 9 V DC

- Japan: 110 V / 50 Hz, 10 VA, 9 V DC

- Australia: 240 V / 50 Hz, 10 VA, 9 V DC

- China: 220 V / 50 Hz, 10 VA, 9 V DC

- Battery (optional): 4 x AA (LR6)

Note

The SevenEasy pH meter should only be used with the power supply unit supplied, or with batteries.
16 METTLER TOLEDO SevenEasy

13. Quick Guide 1. Prepare Electrode

Remove cap

2. 1-Point Calibration

pH Buffer 7

Stir, then 

leave

auto enpoint or

3. Rinse Electrode 4. 2 or 3-Point Calibration

Repeat step 2 when carrying out a 2 or 3-point 

calibration, then rinse

5. Measure sample

Sample

Stir, then 

leave

auto enpoint or

6. Rinse Electrode 7. Store Electrode

Electrolyte

Quick Guide
METTLER TOLEDO SevenEasy 17

14. Buffer sets

SevenEasy pH Meter automatically corrects for setting temperature using the values shown in the 

tables:

Buffer Set 1 (ref. 25 °C) MT US

 5 7.09 4.00 10.25 1.67

 10 7.06 4.00 10.18 1.67

 15 7.04 4.00 10.12 1.67

 20 7.02 4.00 10.06 1.68

25 7.00 4.00 10.01 1.68

 30 6.99 4.01 9.97 1.68

 35 6.98 4.02 9.93 1.69

 40 6.97 4.03 9.89 1.69

 45 6.97 4.04 9.86 1.70

 50 6.97 4.06 9.83 1.71

Buffer Set 2 (ref. 25 °C) MT Europa

 5 7.09 4.01 9.45 2.02 11.72

 10 7.06 4.00 9.38 2.01 11.54

 15 7.04 4.00 9.32 2.00 11.36

 20 7.02 4.00 9.26 2.00 11.18

 25 7.00 4.01 9.21 2.00 11.00

 30 6.99 4.01 9.16 1.99 10.82

 35 6.98 4.02 9.11 1.99 10.64

 40 6.97 4.03 9.06 1.98 10.46

 45 6.97 4.04 9.03 1.98 10.28

 50 6.97 4.06 8.99 1.98 10.10

Buffer Set 3 (ref. 20 °C) Merck

 5 7.07 4.04 9.16 2.01 12.41

 10 7.05 4.02 9.11 2.01 12.26

 15 7.02 4.01 9.05 2.00 12.10

20 7.00 4.00 9.00 2.00 12.00

 25 6.98 4.01 8.95 2.00 11.88

 30 6.98 4.01 8.91 2.00 11.72

 35 6.96 4.01 8.88 2.00 11.67

 40 6.95 4.01 8.85 2.00 11.54

 45 6.95 4.01 8.82 2.00 11.44

 50 6.95 4.00 8.79 2.00 11.33

Buffer Set 4 (ref. 25 °C) Japan JIS

 5 6.95 4.00 9.40 1.67 

 10 6.92 4.00 9.33 1.67 

 15 6.90 4.00 9.28 1.67 

 20 6.88 4.00 9.22 1.68 

 25 6.86 4.01 9.18 1.68 

 30 6.85 4.02 9.14 1.68 

 35 6.84 4.02 9.10 1.69

 40 6.84 4.04 9.07 1.69

 45 6.83 4.05 9.04 1.70

 50 6.83 4.06 9.01 1.71 

Buffer sets
18 METTLER TOLEDO SevenEasy
METTLER TOLEDO SevenEasy 1

Inhalt

Inhalt

1. Einleitung 3

2. Sicherheitsmassnahmen 3

Massnahmen zu Ihrem Schutz 3

Massnahmen zur Betriebssicherheit 3

3. Beschreibung des Gerätes 4

3.1 Anzeige 4

3.2 Tastatur 5

4. Installation 6

5. Probenmessung 8

5.1 pH-Messung 8

5.2 mV-Messung 8

5.3 Einstellungen 8

5.3.1 ATC 8

5.3.2 MTC 8

5.3.3 Datenausgang 8

6. Kalibrierung 9

6.1 Einstellungen 9

6.2 Auswahl einer vordefinierten Puffergruppe 9

6.3 Einstellen einer vom Anwender definierten Puffergruppe 9

6.4 Kalibrierung 10

6.4.1 Einpunkt-Kalibrierung 10

6.4.2 Zweipunkt-Kalibrierung 10

6.4.3 Dreipunkt-Kalibrierung 10

7. Selbsttest 11

8. Alternativer Batteriebetrieb 11

9. Fehlermeldungen 12

10. Wartung 13

10.1 Wartung des Gerätes 13

10.2 Wartung der Elektrode 13

10.3 Entsorgung 13

11. Zubehör 14

12. Technische Daten 15

13. Quick Guide 16

14. Puffergruppen 17
2 METTLER TOLEDO SevenEasy
METTLER TOLEDO SevenEasy 3

Einleitung / Sicherheitsmassnahmen

1. Einleitung

SevenEasy – ein Gerät, welches Ihre Ansprüche weit über die reine pH-Messung erfüllt ohne Ihr Budget 

unnötig zu beanspruchen. Ein Gerät, das viele Vorteile bringt:

- Die Benutzeroberfläche ist so gestaltet, dass sämtliche Bedienungsschritte selbsterklärend ausgeführt 

werden können. Somit sparen Sie viel Zeit.

- Dank der Batteriespeisung kann das Gerät schnell auf einem anderen Labortisch oder sogar in einem 

anderen Labor installiert und eingesetzt werden. Auch an Orten, wo der Netzanschluss nicht verwendet 

werden kann.

- Servicedienstleistungen ermöglichen die regelmässige Qualifizierung Ihrer Geräte und erhöhen die 

Verfügbarkeit und die Genauigkeit Ihres Gerätes.

2. Sicherheitsmassnahmen

Massnahmen zu Ihrem Schutz

- Nicht in explosionsgefährdeter Umgebung arbeiten! Das Gehäuse des Geräts ist nicht 

gasdicht. Sonst besteht Explosionsgefahr durch Funkenbildung und/oder Korrosionsgefahr durch eindringende Gase.

- Beachten Sie bei der Verwendung von Chemikalien und Lösemitteln die Vorschriften des 

Herstellers und die allgemeinen Sicherheitsregeln im Labor!

Massnahmen zur Betriebssicherheit

- Lassen Sie das Gerät nur vom METTLER TOLEDO Service warten!

- Trocknen Sie Flüssigkeits-Spritzer jeweils ab! Das Gerät ist nicht wasserdicht.

- Nur Batterien des spezifizierten Typs verwenden. Ein einwandfreies Funktionieren ist 

sonst nicht gewährleistet.

- Schliessen Sie folgende Umwelteinflüsse aus:

- starke Vibrationen,

- direkte Sonneneinstrahlung,

- Luftfeuchtigkeit über 80%,

- korrosive Gasatmosphäre,

- Temperaturen unter 5 °C und über 40 °C,

- starke elektrische oder magnetische Felder!
4 METTLER TOLEDO SevenEasy

Beschreibung des Gerätes

1 2 3 4 5 6

8

11

13

7

10

12

9

3. Beschreibung des Gerätes

3.1 Anzeige

1 Batteriezustand 

2 Ausserkraftsetzung des automatischen Ausschaltens während des Batteriebetriebs

3 Datenübertragung zum PC/Drucker 

4 Zustand der Elektrode

 Steilheit: 95-105 % Steilheit: 90-94 % Steilheit: 85-89 %

 Offset: ± (0-15) mV Offset: ± (15-35) mV Offset: ± (>35) mV

 Guter Zustand Elektrode benötigt Elektrode ist defekt

 der Elektrode Reinigung

5 Gerät Selbsttest

 Selbsttest- Aufforderung zum Selbsttest 

 Anzeige Tasten drücken bestanden

6 Fehlernummer 

7 pH/mV-Messung

8 Autom./Manueller Temperaturausgleich

9 Endpunkt Stabilität/autom. Endpunkt

 Endpunkt-Stabilität A Autom. Endpunkt

10 Temperatur 

11 Kalibrierungspunkt

12 Puffergruppen

13 Offset und Steigung der Elektrode
METTLER TOLEDO SevenEasy 5

Beschreibung des Gerätes

3.2 Tastatur

Drücken & loslassen Drücken und während 2 Sek. 

gedrückt halten 

Meter ein/aus Deaktivieren des automatischen

Ausschaltens mit Batteriebetrieb

- Anfangs- oder Endpunktmessung

- Zurück zum Messmodus

- Eingegebenen Wert speichern

Autom. Endpunkt ein-/ausschalten

 / 

- Kalibrierung beginnen

- Ausgewählte Puffergruppe bestätigen

Anzeige der für die Kalibrierung 

verwendeten Puffer

Zwischen pH- und mV-Messmodus hin und 

her schalten

Datenübertragung zum PC oder 

Drucker

- Auswahl der Kalibrierungspuffergruppen

- Wert während der Einstellung heraufsetzen

- MTC-Temperatur einstellen

- Wert während der Einstellung herabsetzen

Autotest beginnen 
6 METTLER TOLEDO SevenEasy

4. Installation

1. Packen Sie das Gerät, den Netzadapter, die Elektrode, den Elektrodenarm und das andere Zubehör 

aus. Bewahren Sie das Kalibrierungszertifikat an einem sichern Ort auf. 

2. Vergewissern Sie sich, dass der Netzadapter mit ihrer Netzspeisung übereinstimmt. Wenn nicht, setzen 

Sie sich mit Ihrer METTLER TOLEDO Vertretung in Verbindung.

3. Installieren sie den Elektrodenarm.

- Der Elektrodenhalter kann alleine stehend verwendet werden oder am 

Gerät montiert werden. Der Elektrodenarm wird in eine der drei Positionen auf dem Sockel platziert. Es wird empfohlen, das mittlere Loch 

zu benutzen, falls der Elektrodenhalter alleine stehend verwendet wird. 

Zuerst muss die entsprechende Abdeckung entfernt werden. Fixieren 

Sie die Verbindung mittels der mitgelieferten Befestigungsschraube. 

Die Installation ist beendet, wenn der Elektrodenhalter alleine stehend 

verwendet werden soll.

- Falls der Elektrodenhalter am Gerät befestigt werden soll, setzen Sie 

das Gerät so auf den Sockel, dass der frei stehende Flügel des Sockels 

nach hinten schaut. Zuerst muss die entsprechende Abdeckung entfernt werden. Der Elektrodenhalter kann links oder rechts des Geräts 

befestigt werden.

- Verstellen sie den Spannungsdrehknopf nach Bedarf.

Installation

Lösen

Anziehen
METTLER TOLEDO SevenEasy 7

Installation

Anschluss für Steckernetzteil

Anschluss für ATC Temperatursensor (automatische 

Temperaturkompensation)

Anschluss für kombinierte pH Elektrode

RS232 Datenausgang

4. Entfernen Sie die Kurzschlussklammer vom pH-Anschluss.

5. Verbinden Sie die Elektrode. Wenn Sie eine Elektrode mit integrierter Temperatursonde verwenden, 

verbinden Sie das andere Kabel mit dem ATC-Anschluss.

6. Wenn Sie eine separate Temperatursonde verwenden, verbinden Sie ihn mit dem ATC-Anschluss. 

7. Verbinden Sie die Netzeinheit mit dem Gleichstromanschluss (DC). 
8 METTLER TOLEDO SevenEasy

5. Probenmessung

5.1 pH-Messung

Stellen Sie die Elektrode in die Probe und drücken Sie um die Messung zu beginnen – der Dezimalpunkt blinkt. 

Auf der Anzeige steht der aktuelle pH-Wert der Probe. Der automatische Endpunkt A ist die Grundeinstellung des Geräts. Wenn sich das Signal des Sensors stabilisiert hat, bleibt die Anzeige stehen und 

 erscheint.

Der Algorithmus der automatischen Endpunkterkennung ist 0,1 mV pro 5 Sekunden.

Durch anhaltendes Drücken der Taste können Sie zwischen automatischem oder manuellem Endpunktformat hin und her schalten. Um den Endpunkt für eine Messung manuell einzugeben, drücken 

Sie die Taste . Die Anzeige erstarrt und erscheint.

5.2 mV-Messung

Um eine mV-Messung durchzuführen, gehen Sie wie bei einer pH-Messung vor. Um den mV-Wert zu 

sehen müssen Sie die Taste drücken.

5.3 Einstellungen

5.3.1 ATC

Für eine höhere Genauigkeit empfehlen wir die Verwendung einer integrierten oder separaten Temperatursonde. Wenn eine Temperatursonde verwendet wird, erscheint das Symbol ATC und die aktuelle 

Messtemperatur auf der Anzeige.

5.3.2 MTC

Wenn das Gerät feststellt, dass keine Temperatursonde angeschlossen ist, wechselt es automatisch in 

den manuellen Temperaturkompensationsmodus über, und MTC erscheint auf der Anzeige. 

Um die MTC-Temperatur einzustellen, drücken Sie . Benützen Sie und , um den Temperaturwert nach unten oder oben zu verschieben. Bestätigen Sie Ihre Eingabe durch Drücken der Taste 

. Die Grundeinstellung ist 25 °C.

 

5.3.3 Datenausgang

Wenn ein PC oder Drucker angeschlossen ist, wird jede Endpunktmessung über den RS232-Anschluss 

an den PC oder Drucker übermittelt. 

Durch Drücken und Halten der Taste erscheint in der Anzeige. Das Gerät übermittelt eine 

Messung pro Sekunde, bis der Endpunkt erreicht ist.

Messungen
METTLER TOLEDO SevenEasy 9

6. Kalibrierung

6.1 Einstellungen

Das SevenEasy pH-Meter ermöglicht Ein-, Zwei- und Dreipunkt-Kalibrierungen. Wenn Sie Ihre Kalibrierungspuffergruppe aus den vier im Gerät vorgegebenen Gruppen auswählen, so wird die automatische 

Puffergruppenerkennung während der Kalibrierung durchgeführt.

Die 4 vordefinierten Puffergruppen sind:

B1: (25 °C) 7,00 4,00 10,01 1,68

B2: (25 °C) 7,00 4,01 9,21 2,00 11,00

B3: (20 °C) 7,00 4,00 9,00 2,00 12,00 

B4: (25 °C) 6,86 4,01 9,18 1,68

Mit der unten beschriebenen Puffereinstellung können Sie Ihre eigene Puffergruppe definieren. Während 

der Kalibrierung ist die automatische Puffererkennung nicht aktiv. 

6.2 Auswahl einer vordefinierten Puffergruppe

Drücken Sie die Taste und die aktuelle Puffergruppe beginnt zu blinken. Ist die aktuelle Gruppe eine 

vom Anwender definierte Gruppe, so blinkt der leere Rahmen.

Benützen Sie die Tasten oder , um andere Gruppen in der oberen oder unteren Position auszuwählen. Wenn die gewünschte Gruppe blinkt, drücken Sie , um ihre Auswahl zu bestätigen.

6.3 Einstellen einer vom Anwender definierten Puffergruppe

Wenn unter Schritt 6.2 der leere Rahmen blinkt, dann drücken sie die Taste, um die Einstellung 

zu beginnen. Das Gerät zeigt den aktuell eingestellten Temperaturwert an, und der Punkt und Rahmen 

blinken (Grundeinstellung 25 °C). Benützen Sie die Tasten oder , um den Wert zu verändern. 

Drücken sie die Taste um den Wert abzuspeichern und fortzufahren.

Nachdem Sie den Temperaturwert eingegeben haben, zeigt das Gerät die aktuelle Einstellung des ersten 

Kalibrierungspuffers an (Grundeinstellung 4.00). Verwenden Sie die Tasten oder , um den Wert 

zu verändern. Drücken sie die Taste , um den Wert abzuspeichern und fortzufahren.

Nachdem Sie den ersten Kalibrierungspuffer eingestellt haben, drücken Sie , um den nächsten 

Punkt einzustellen.

Das Verfahren ist dasselbe wie für den ersten Punkt. Sie können bis zu drei Anwender-definierte Kalibrationspuffer einrichten. Wenn Sie Ihre Einstellungen vorgenommen haben, drücken Sie die Taste , 

um das Menü zu verlassen.

Hinweis

Wenn Sie eine vom Anwender definierte Puffergruppe für die Kalibrierung verwenden, erscheint der 

von Ihnen eingegeben Pufferwert. Vergewissern Sie sich, dass Sie den richtigen Puffer verwenden. Die 

Puffertemperatur sollte den eingegeben Wert einhalten. Wird eine Temperatursonde eingesetzt und die 

gemessene Temperatur unterscheidet sich um mehr als 1 °C vom eingegebenen Wert, so erscheint Err 

5 auf der Anzeige.

Kalibrierung
10 METTLER TOLEDO SevenEasy

6.4 Kalibrierung

6.4.1 Einpunkt-Kalibrierung

Stellen Sie die Elektrode in einen Kalibrierungspuffer und drücken Sie .

Ihr SevenEasy pH-Meter erkennt bei der Kalibrierung den Endpunkt automatisch. Um den Endpunkt 

manuell zu bestimmen, drücken Sie . Das Gerät zeigt den relevanten Pufferwert an. Der Offset-Wert 

der Elektrode wird angezeigt. 

Um zur Probenmessung zurückzukehren, drücken Sie .

6.4.2 Zweipunkt-Kalibrierung

1. Schritt Führen Sie eine “Einpunkt-Kalibrierung” wie oben beschrieben durch.

2. Schritt Spülen Sie die Elektrode mit destilliertem Wasser.

3. Schritt Stellen Sie die Elektrode in den nächsten Puffer und drücken Sie .

Ihr SevenEasy pH-Meter erkennt bei der Kalibrierung den Endpunkt automatisch. Um den Endpunkt 

manuell zu bestimmen, drücken Sie . Das Gerät zeigt den relevanten Pufferwert an. Der Offset-Wert 

der Elektrode wird aktualisiert und die Steigung wird angezeigt. 

Um zur Probenmessung zurückzukehren, drücken Sie .

6.4.3 Dreipunkt-Kalibrierung 

Führen Sie eine “Zweipunkt-Kalibrierung” gemäss obigem Beschrieb durch. Wiederholen Sie die Schritte 

2 und 3 für die Dreipunkt-Kalibrierung. 

Hinweis

Die Verwendung einer Temperatursonde oder einer Elektrode mit integrierter Temperatursonde wird empfohlen. Wenn Sie den MTC-Modus benützen, sollten alle Puffer und Probelösungen bei der gleichen, 

eingestellten Temperatur gehalten werden.

Um eine möglichst genaue pH-Messung zu erhalten, sollten Sie das Gerät regelmässig kalibrieren.

Um die Anzahl und Art der Puffer anzuzeigen, die für die Kalibrierung verwendet wurden, drücken und 

halten Sie die Taste. Die Puffer-Werte werden abwechselnd angezeigt.

Kalibrierung
METTLER TOLEDO SevenEasy 11

7. Selbsttest

Drücken und halten Sie die beiden Tasten und gleichzeitig bis das Symbol für den Selbsttest 

 erscheint.

Auf dem Gerät erscheinen zuerst alle Anzeigeelemente. Dann blinken die einzelnen Symbol nacheinander. 

Danach müssen die Tasten durch den Anwender geprüft werden.

Wenn das Symbol blinkt, drücken Sie die entsprechende Taste innerhalb 5 Sekunden gemäss der 

nachfolgenden Liste:

a. Wenn blinkt, drücken Sie die Taste ;

b. Wenn blinkt, drücken Sie die Taste ;

c. Wenn blinkt, drücken Sie die Taste ;

d. Wenn blinkt, drücken Sie die Taste ;

e. Wenn blinkt, drücken Sie die Taste .

Ist der Selbsttest abgeschlossen, erscheint ein Häkchen in der Anzeige. Schlägt der Selbsttest fehl, 

sehen Sie unter “9. Fehlermeldungen” nach und leiten die notwendigen Massnahmen ein.

8. Alternativer Batteriebetrieb

Das SevenEasy pH-Meter kann auch mit Batterien betrieben werden. Setzen Sie 4 AA (LR6) Batterien 

auf der Unterseite des Geräts ein. Wird das Netzadapterkabel am Gerät ausgezogen, wechselt das 

pH-Meter automatisch auf Batteriebetrieb und das Symbol erscheint.

Wenn die Batterien leer sind, erscheint das Symbol für die Batterie in der Anzeige.

Im Batteriebetrieb verfügt das Gerät über eine automatische Ausschaltfunktion. Wird während 10 Minuten keine Taste gedrückt, so schaltet das Gerät automatisch aus, um die Batterie zu schonen. Um 

diese automatische Ausschaltfunktion zu umgehen, drücken und halten Sie die Taste während 

2 Sekunden, bis das Symbol erscheint.

Selbsttest/Batteriebetrieb
12 METTLER TOLEDO SevenEasy

9. Fehlermeldungen

Error 1 - Offset liegt ausserhalb des erlaubten Bereiches

Vergewissern Sie sich, dass Sie den richtigen Puffer verwenden und dass er frisch ist. Elektrode reinigen 

oder auswechseln.

Error 2 - Steigung liegt ausserhalb des erlaubten Bereiches

Vergewissern Sie sich, dass Sie den richtigen Puffer verwenden und dass er frisch ist. Elektrode reinigen 

oder auswechseln.

Error 3 - Instrument erkennt den Puffer nicht

Vergewissern Sie sich, dass Sie den richtigen Puffer verwenden und dass er frisch ist. Überprüfen Sie, 

ob der Puffer bei der Kalibrierung mehrfach verwendet wurde. 

Error 4 - Eingabefehler des anwenderdefinierten Pufferwertes

Wenn Sie die anwenderdefinierten Pufferwerte eingeben, akzeptiert das Instrument keine pH-Werte, 

welche weniger als 1 pH-Einheit von anderen definierten Werten entfernt sind. Geben Sie einen neuen 

Wert ein.

Error 5 - ATC Temperatur weicht vom anwenderdefinierten Wert ab.

Die Temperatur des Puffers oder der Probe sollte der eingegebenen Temperatur entsprechen. Wenn nötig, 

ändern Sie die Temperatureinstellung. 

Error 6 - Gemessene Puffertemperatur liegt ausserhalb des erlaubten Bereiches (5... 50°C)

Die Temperatur des Puffers muss innerhalb der Bandbreite gehalten werden.

Error 7, (----) - mV Wert liegt ausserhalb des Messbereiches

Vergewissern Sie sich, dass die Elektrode angeschlossen ist.

Ist die Elektrode nicht angeschlossen, setzen Sie bitte die Kurzschlussklammer ein. 

Error 8 - pH Wert liegt ausserhalb des Messbereiches

Überprüfen Sie, ob der Deckel der Elektrode entfernt wurde und ob die Elektrode richtig angeschlossen 

und in die Probe eingetaucht wurde. 

Error 9 - Selbsttest fehlgeschlagen

Wiederholen Sie den Selbsttest und vergewissern Sie sich, dass Sie die richtigen Tasten gedrückt haben, 

wenn das Symbol blinkt. Erscheint Err 9 immer noch, so kontaktieren Sie den METTLER TOLEDO 

Service.

Fehlermeldungen
METTLER TOLEDO SevenEasy 13

10. Wartung

10.1 Wartung des Gerätes

Bei diesem Gerät und dem Netzadapter gibt es keine vom Anwender zu ersetzende Teile. Bitte öffnen 

Sie die Gehäuse nicht. 

Das SevenEasy benötigt ausser einer gelegentlichen Reinigung mit einem feuchten Lappen keinen 

Unterhalt. Das Gehäuse besteht aus ABS/PC, das von einigen organischen Lösungen wie Toluol, Xylol 

und Methylethylketon angegriffen werden kann. Gutes Laborverhalten erfordert sofortiges Wegwischen 

ausgeleerter Flüssigkeiten. 

10.2 Wartung der Elektrode

Vergewissern Sie sich, dass die Elektrode immer mit der angemessenen Lösung gefüllt ist. Für eine 

maximale Messgenauigkeit sollte jede eingefüllte Lösung, die nach aussen “gekrochen” und die Elektrode 

verkrustet haben könnte, mit destilliertem Wasser entfernt werden. 

Lagern Sie die Elektrode immer richtig und lassen Sie sie nicht austrocknen.

Fällt der Steigungswert der Elektrode schnell ab, wird die Reaktionszeit träge oder werden die Messungen 

ungenau, können folgende Verfahren helfen. Wenden Sie sie in der nachstehenden Reihenfolge an.

1. Entfetten Sie die Membran mit in Aceton- oder Seifenlösung getauchter Watte. 

2. Tauchen Sie die Elektrode über Nacht in 0.1 M HCl.

3. Entfernen Sie Proteinrückstände indem Sie die Elektrode in eine HCl /Pepsin-Lösung einlegen (Best.-

Nr: 51340068).

4. Im Falle einer Silbersulfidkontamination, tauchen Sie die Elektrode in eine Thioharnstofflösung (Best.-

Nr: 51340070).

Hinweis

Reinigungs- und Fülllösungen sollten mit der gleichen Vorsicht gehandhabt werden wie giftige oder 

ätzende Substanzen.

10.3 Entsorgung

In Übereinstimmung mit den Anforderungen der Europäischen Richtlinie 2002/96 EG über 

Elektro- und Elektronik-Altgeräte (WEEE) darf dieses Gerät nicht mit dem Hausmüll entsorgt 

werden. Sinngemäss gilt dies auch für Länder ausserhalb der EU entsprechend den geltenden 

nationalen Regelungen.

Bitte entsorgen Sie dieses Produkt gemäss den örtlichen Bestimmungen in einer getrennten 

Sammlung für Elektro- und Elektronikgeräte.

Bei allfälligen Fragen wenden Sie sich bitte an die zuständige Behörde oder den Händler, bei dem Sie 

dieses Gerät erworben haben.

Bei Weitergabe dieses Gerätes (z.B. für private oder gewerbliche/industrielle Weiternutzung) ist diese 

Bestimmung sinngemäss weiterzugeben.

Vielen Dank für Ihren Beitrag zum Schutz der Umwelt.

Unterhalt
14 METTLER TOLEDO SevenEasy

Zubehör

11. Zubehör

Best.-Nr

Elektrodenarm komplett 51302820

Anleitung zur pH-Bestimmung 51300058

Anleitung zur Leitfähigkeits- und Sauerstoffmessung 51724715

Anleitung zur ionenselektiven Messung 51300201

pH 4,01 Puffer, 30 Beutel zu 20mL 51302069

pH 4,01 Puffer, 6 Flaschen zu 250mL 51340058

pH 7,00 Puffer, 30 Beutel zu 20mL 51302047

pH 7,00 Puffer, 6 Flaschen zu 250mL 51340060

pH 9,21 Puffer, 30 Beutel zu 20mL 51302070

pH 9,21 Puffer, 6 Flaschen zu 250mL 51300194

pH 10,01 Puffer, 30 Beutel zu 20mL 51302079

pH 10,01 Puffer, 6 Flaschen zu 250mL 51340231

InLab®410, pH-Elektrode, Temp.-Fühler, nachfüllbar 52000118

InLab®412, Labor pH-Elektrode, Glas, nachfüllbar 52000112

InLab®413, robuste pH-Elektrode, PEEK-Schaft, ATC 52000100

InLab®420, pH-Elektrode mit PTFE-Schliffdiaphragma 52000113

HCl/Pepsin-Lösung (entfernt Proteinverschmutzungen) 51340068

Thioharnstofflösung (entfernt Silbersulfidverschmutzungen) 51340070

Reaktivierungslösung für pH-Elektroden 51340073
METTLER TOLEDO SevenEasy 15

Technische Daten

12. Technische Daten

pH mV Temperatur

Messbereich 0,00 -14,00 ± 1999 –5 ... 105 °C

Auflösung 0,01 1 0,1 °C

Relative Genauigkeit ± 0,01 ± 1 ± 0,5 °C

pH-Kalibrierung 1, 2, oder 3 Punkte

Isopotentialpunkt 7,00 pH

Kalibrierungspuffer B1: (25 °C) 7,00 4,00 10,01 1,68

B2: (25 °C) 7,00 4,01 9,21 2,00 11,00

B3: (20 °C) 7,00 4,00 9,00 2,00 12,00

B4: (25 °C) 6,86 4,01 9,18 1,68

B5: vom Anwender definiert

Temperaturkompensation –5 ... 105 °C automatisch oder manuell

Anzeige Flüssigkristallanzeige

Ausgänge RS232 seriell, 

Baudrate: 1200

Databit: 8

Stoppbit: 1

Parität: keine

pH-Eingang Impedanz > 1012 Ohm

Umgebungsbedingungen Raumtemperatur: 5°C ... 40 °C

Relative Luftfeuchtigkeit: 5 % ... 80 %, (nicht kondensierend)

Überspannungskategorie: II

Verschmutzungsgrad: 2

Abmessung/Gewicht 180 x 180 x 65 (mm) / 0,61 kg

Materialien Gehäuse: ABS, PC verstärkt

Elektrodenhalter: ABS, PC verstärkt

Tastaturfolie: Polyester

Stromversorgung Das SevenEasy pH-Meter ist mit einem Ihrer Region 

entsprechenden Netzadapter ausgerüstet:

- USA: 120 V / 60 Hz, 10 VA, 9 V DC

- Europa: 230 V / 50 Hz, 10 VA, 9 V DC

- GB: 240 V / 50 Hz, 10 VA, 9 V DC

- Japan: 110 V / 50 Hz, 10 VA, 9 V DC

- Australien: 240 V / 50 Hz, 10 VA, 9 V DC

- China: 220 V / 50 Hz, 10 VA, 9 V DC

- Batterie (fakultativ): 4 x AA (LR6)

Hinweis

Das SevenEasy pH-Meter sollte nur mit dem gelieferten Netzadapter oder mit Batterien betrieben werden.
16 METTLER TOLEDO SevenEasy

13. Quick Guide 1. Elektrode vorbereiten

Wässerungskappe entfernen

2. Einpunkt-Kalibrierung

pH Puffer 7

Rühren, dann 

belassen

automatischer Endpunkt oder

3. Elektrode spülen 4. Zwei- oder Dreipunkt-Kalibrierung

Wiederholen Sie Schritt 2 bei einer Zwei- oder Dreipunkt-Kalibrierung, dann spülen

5. Probe messen

Probe

Rühren, dann 

belassen

automatische Endpunkt oder

6. Elektrode spülen 7. Elektrode lagern

Elektrolyt

Quick Guide
METTLER TOLEDO SevenEasy 17

14. Puffergruppen

Das SevenEasy pH-Meter kompensiert die eingestellte Temperatur automatisch gemäss den Werten in 

den Tabellen:

Puffergruppe 1 (ref. 25 °C) MT USA

 5 7,09 4,00 10,25 1,67

 10 7,06 4,00 10,18 1,67

 15 7,04 4,00 10,12 1,67

 20 7,02 4,00 10,06 1,68

25 7,00 4,00 10,01 1,68

 30 6,99 4,01 9,97 1,68

 35 6,98 4,02 9,93 1,69

 40 6,97 4,03 9,89 1,69

 45 6,97 4,04 9,86 1,70

 50 6,97 4,06 9,83 1,71

Puffergruppe 2 (ref. 25 °C) MT Europa

 5 7,09 4,01 9,45 2,02 11,72

 10 7,06 4,00 9,38 2,01 11,54

 15 7,04 4,00 9,32 2,00 11,36

 20 7,02 4,00 9,26 2,00 11,18

 25 7,00 4,01 9,21 2,00 11,00

 30 6,99 4,01 9,16 1,99 10,82

 35 6,98 4,02 9,11 1,99 10,64

 40 6,97 4,03 9,06 1,98 10,46

 45 6,97 4,04 9,03 1,98 10,28

 50 6,97 4,06 8,99 1,98 10,10

Puffergruppe 3 (ref. 20 °C) Merck

 5 7,07 4,04 9,16 2,01 12,41

 10 7,05 4,02 9,11 2,01 12,26

 15 7,02 4,01 9,05 2,00 12,10

20 7,00 4,00 9,00 2,00 12,00

 25 6,98 4,01 8,95 2,00 11,88

 30 6,98 4,01 8,91 2,00 11,72

 35 6,96 4,01 8,88 2,00 11,67

 40 6,95 4,01 8,85 2,00 11,54

 45 6,95 4,01 8,82 2,00 11,44

 50 6,95 4,00 8,79 2,00 11,33

Puffergruppe 4 (ref. 25 °C) Japan JIS

 5 6,95 4,00 9,40 1,67 

 10 6,92 4,00 9,33 1,67 

 15 6,90 4,00 9,28 1,67 

 20 6,88 4,00 9,22 1,68 

 25 6,86 4,01 9,18 1,68 

 30 6,85 4,02 9,14 1,68 

 35 6,84 4,02 9,10 1,69

 40 6,84 4,04 9,07 1,69

 45 6,83 4,05 9,04 1,70

 50 6,83 4,06 9,01 1,71 

Puffergruppen
18 METTLER TOLEDO SevenEasy
METTLER TOLEDO SevenEasy 1

Sommaire

Sommaire

1. Introduction 3

2. Consignes de sécurité 3

Pour votre propre sécurité 3

Pour la sécurité de fonctionnement 3

3. Description de l’appareil 4

3.1 Affichage 4

3.2 Clavier 5

4. Installation 6

5. Mesure d’un échantillon 8

5.1 Mesure du pH 8

5.2 Mesure en mV 8

5.3 Réglages 8

5.3.1 ATC 8

5.3.2 MTC 8

5.3.3 Sortie de données 8

6. Etalonnage 9

6.1 Réglages 9

6.2 Sélection d’un ensemble de solutions tampon prédéfini 9

6.3 Réglage d’un ensemble de tampons défini par l’utilisateur 9

6.4 Etalonnage 10

6.4.1 Etalonnage à 1 point 10

6.4.2 Etalonnage à 2 points 10

6.4.3 Etalonnage à 3 points 10

7. Test d’autocontrôle 11

8. Fonctionnement facultatif sur pile 11

9. Messages d’erreur 12

10. Entretien 13

10.1 Entretien de l’appareil 13

10.2 Entretien de l’électrode 13

10.3 Elimination 13

11. Accessoires 14

12. Caractéristiques techniques 15

13. Guide rapide 16

14. Ensembles de solutions tampon d’ étalonnage 17
2 METTLER TOLEDO SevenEasy
METTLER TOLEDO SevenEasy 3

Introduction / consignes de sécurité

1. Introduction

SevenEasy – un appareil qui répond à vos exigences bien au-delà de la simple mesure du pH et sans 

pour autant obérer votre budget. Un appareil aux nombreux avantages:

- L’interface utilisateur est conçue de manière à ce que toutes les opérations de commande s’expliquent 

d’elles-mêmes. Vous gagnez ainsi beaucoup de temps.

- Grâce à l’alimentation piles, l’appareil peut être rapidement installé et utilisé sur une autre table du 

laboratoire, voire même dans un autre laboratoire, et aussi à des endroits où un raccordement au 

secteur n’est pas possible.

- Les prestations de service permettent la qualification périodique de vos appareils et augmentent la 

disponibilité et l’exactitude de votre appareil.

2. Consignes de sécurité

Pour votre propre sécurité

- Travailler en atmosphère non explosible! Le boîtier de l’appareil n’est pas étanche aux 

gaz. La pénétration de gaz pourrait créer un danger de corrosion et/ou d’explosion en 

cas d’étincelles.

- Respecter les instructions du fabricant pour les produits chimiques et solvants employés, 

ainsi que les règles de sécurité courantes du laboratoire!

Pour la sécurité de fonctionnement

- Confier la maintenance de l’appareil exclusivement à METTLER TOLEDO!

- Essuyer régulièrement les projections de liquides! L’appareil n’est pas étanche à 

l’eau.

- N’utiliser que des piles du modèle spécifié. Sinon le parfait fonctionnement ne serait 

pas garanti.

- Eviter les facteurs ambiants suivants:

- fortes vibrations,

- exposition directe au soleil,

- humidité relative de l’air supérieure à 80%,

- atmosphère de gaz corrosifs,

- température inférieure à 5 °C et supérieure à 40 °C,

- champs électriques ou magnétiques puissants!
4 METTLER TOLEDO SevenEasy

Description de l’appareil

1 2 3 4 5 6

8

11

13

7

10

12

9

3. Description de l’appareil

3.1 Affichage

1 Etat de la pile 

2 Désactivation de l’arrêt automatique pendant le fonctionnement piles

3 Transfert des données vers PC/imprimante 

4 Etat de l’électrode

 pente: 95-105 % pente: 90-94 % pente: 85-89 %

 zéro: ± (0-15) mV zéro: ± (15-35) mV zéro: ± (>35) mV

 électrode en bon l’électrode a besoin l’électrode est 

 état d’un nettoyage défectueuse

5 Test d’autocontrôle de l’appareil

 affichage du test injonction de presser test d’autocontrôle

 d’autocontrôle des touches réussi

6 Numéro de l’erreur 

7 Mesure pH/mV

8 Compensation de température automatique/manuelle

9 Stabilité du point final/point final automatique

 Stabilité du point final A Point final automatique 

10 Température 

11 Point d’étalonnage

12 Ensembles de solutions tampon d‘étalonnage

13 Valeur de décalage et pente de l’électrode
METTLER TOLEDO SevenEasy 5

Description de l’appareil

3.2 Clavier

presser et relâcher presser pendant 

2 secondes 

appareil marche/arrêt désactivation de l’arrêt auto 

matique peandant le fonction 

nement piles

- mesure du point de départ ou point final

- retour au mode mesure

- enregistrer la valeur entrée

point final automatique marche/arrêt 

 / 

- commencer l’étalonnage

- confirmer le groupe de tampons 

sélectionné

Indication des tampons utilisés 

pour la calibration

commuter entre les modes de mesure transfert des données pH et mVtransfert des données vers le PC 

ou l’imprimante

- séquence des solutions tampon 

d‘étalonnage

- augmenter la valeur pendant le réglage

- régler la température MTC,

- diminuer la valeur pendant le réglage

commencer le test d’autocontrôle
6 METTLER TOLEDO SevenEasy

4. Installation

1. Déballer l’appareil, l’adaptateur secteur, l’électrode, le bras porte-électrode et les autres accessoires. 

Conserver le certificat d’étalonnage en un endroit sûr. 

2. S’assurer que l’adaptateur secteur correspond à la tension secteur locale. Sinon, contactez votre 

représentant METTLER TOLEDO.

3. Installer le bras porte-électrode.

- Le porte-électrode peut être employé seul ou être monté sur l’appareil. 

Il est placé sur le socle dans une des trois positions prévues. Si le 

porte-électrode doit être empoyé seul, il est recommandé d’utiliser le 

trou médian. D’abord enlever le cache. Fixer le support à l’aide de la 

vis fournie. Si le porte-électrode doit être employé seul, l’installation 

est alors terminée.

- Si le porte-électrode doit être monté sur l’appareil, placer l’appareil 

sur le socle de manière à ce que l’aile libre du socle regarde vers 

l’arrière. D’abord enlever le cache. Le porte-électrode peut être fixé 

sur la gauche ou la droite de l’appareil.

- Tourner le bouton de serrage en fonction des besoins.

Installation

desserrer

serrer
METTLER TOLEDO SevenEasy 7

Installation

connecteur pour l’adaptateur secteur

connecteur pour capteur de température ATC (compensation de température automatique)

connecteur pour électrode de pH combinée

sortie de données RS232 

4. Enlever la pince de court-circuitage du connecteur pH.

5. Raccorder l’électrode. En cas d’utilisation d’une électrode avec capteur de température intégré, relier 

l’autre câble au connecteur ATC.

6. En cas d’utilisation d’un capteur de température séparé, le relier au connecteur ATC. 

7. Relier l’adaptateur secteur au connecteur de courant continu (DC). 
8 METTLER TOLEDO SevenEasy

5. Mesure d’un échantillon

5.1 Mesure du pH

Plonger l’électrode dans l’échantillon et presser pour commencer la mesure – le point décimal 

clignote. 

L’afficheur indique le pH de l’échantillon et le réglage de base de l’appareil est point final automatique A. Lorsque les indications du capteur se sont stabilisées, l’afficheur se fige automatiquement et affiche .

Pour ce point final automatique, la valeur doit changer moins de 0,1 mV pendant 5 seconds.

On peut passer du format point final automatique à manuel et inversement en maintenant la pression 

sur la touche . Pour entrer manuellement le format point final d’une mesure, presser la touche . 

L’afficheur se fige et affiche .

5.2 Mesure en mV

Pour mesurer en mV, procéder comme pour une mesure du pH. Pour voir la valeur mV, il faut presser 

la touche .

5.3 Réglages

5.3.1 ATC

En vue d’une meilleure exactitude nous recommandons l’utilisation d’un capteur de température intégré 

ou séparé. Lorsqu’un capteur de température est utilisé, le symbole ATC et la température de l’échantillon sont affichés.

5.3.2 MTC

Quand l’appareil constate l’absence d’un capteur de température, il passe automatiquement en mode 

manuel de compensation de la température et affiche MTC. 

Pour régler la température MTC, presser . Les touches et servent à augmenter ou diminuer la 

valeur de la température. Confirmer l’entrée en pressant la touche . Le réglage de base est 25 °C.

5.3.3 Sortie de données

En cas de raccordement d’un PC ou d’une imprimante, chaque mesure à point final est transmise au 

PC ou à l’imprimante à travers le connecteur RS232. 

En maintenant la pression sur la touche , l’afficheur indique . L’appareil transmet une mesure 

par seconde jusqu’à ce que le point final soit atteint.

Mesures
METTLER TOLEDO SevenEasy 9

6. Etalonnage

6.1 Réglages

Le pH-mètre SevenEasy permet des étalonnages à 1, 2 et 3 points. En sélectionnant votre ensemble de 

solutions tampon d‘étalonnage parmi les quatre groupes prédéfinis dans l’appareil, la reconnaissance 

automatique du groupe se fait en cours d’étalonnage.

Les 4 ensembles de solutions tampons d‘étalonnage prédéfinis sont:

B1: (25 °C) 7,00 4,00 10,01 1,68

B2: (25 °C) 7,00 4,01 9,21 2,00 11,00

B3: (20 °C) 7,00 4,00 9,00 2,00 12,00

B4: (25 °C) 6,86 4,01 9,18 1,68 

Vous pouvez définir votre propre groupe de tampons grâce aux réglages décrits ci-après. Pendant 

l’étalonnage, la reconnaissance automatique des tampons est inactive. 

6.2 Sélection d’un ensemble de solutions tampon prédéfini

Presser la touche et l’ensemble actuel clignote. Si l’ensemble actuel est un ensemble défini par 

l’utilisateur, le cadre vide clignote.

Utiliser les touches ou , pour sélectionner un autre ensemble en position supérieure ou inférieure. 

Lorsque l’ensemble voulu clignote, presser pour confirmer la sélection.

6.3 Réglage d’un ensemble de tampons défini par l’utilisateur

A l’étape 6.2, lorsque le cadre vide clignote, presser la touche pour commencer le réglage. L’appareil indique la valeur actuelle de réglage de la température, et le point et le cadre clignotent (réglage 

de base 25 °C). Utiliser les touches ou pour modifier la valeur. Presser la touche pour 

enregistrer la valeur et poursuivre.

Après entrée de la température, l’appareil indique le réglage actuel du premier tampon d’étalonnage 

(réglage de base 4.00). Utiliser les touches ou pour modifier la valeur. Presser la touche 

 pour enregistrer la valeur et poursuivre.

Après avoir réglé le premier tampon d’étalonnage, presser pour passer à l’étape suivante.

Le procédé est le même que pour la première étape. Vous pouvez installer jusqu’à trois tampons 

d’étalonnage définis par l’utilisateur. Lorsque les réglages sont terminés, presser la touche pour 

quitter le menu.

Remarque

Si vous utilisez, pour l’étalonnage, un ensemble de tampons défini par l’utilisateur, l’appareil indique la 

valeur tampon que vous avez entrée. Vérifiez que vous utilisez bien le bon tampon. La température du 

tampon doit correspondre à la valeur entrée. Lorsqu’un capteur de température est employé et que la 

température mesurée s’écarte de plus de 1 °C de la valeur entrée, l’appareil affiche Err 5.

Etalonnage
10 METTLER TOLEDO SevenEasy

6.4 Etalonnage

6.4.1 Etalonnage à 1 point

Plonger l’électrode dans une solution tampon d’étalonnage et presser .

Le pH-mètre SevenEasy atteint automatiquement le point final. Pour atteindre le point final manuellement, presser . L’appareil indique la valeur tampon et la maintient. Il affiche la valeur du zéro de 

l’électrode. 

Pour revenir à la mesure d’échantillons, presser .

6.4.2 Etalonnage à 2 points

Etape 1 Effectuer un “étalonnage à 1 point” comme décrit ci-dessus.

Etape 2 Rincer l’électrode avec de l’eau distillée.

Etape 3 Plonger l’électrode dans le tampon suivant et presser .

Le pH-mètre SevenEasy atteint automatiquement le point final. Pour atteindre le point final manuellement, presser . L’appareil indique la valeur tampon et la maintient. Il actualise la valeur du zéro 

de l’électrode et affiche la pente. 

Pour revenir à la mesure d’échantillons, presser .

6.4.3 Etalonnage à 3 points

Effectuer un “étalonnage à 2 points” comme décrit ci-dessus. Répéter les étapes 2 et 3 pour l’étalonnage 

à 3 points. 

Remarque

Il est recommandé d’utiliser un capteur de température ou une électrode avec capteur de température 

intégré. Si vous travaillez en mode MTC, tous les tampons et solutions échantillons devraient être 

maintenus à la même température.

Pour obtenir une mesure du pH aussi exacte que possible, l’appareil doit être étalonné régulièrement.

Pour afficher le nombre et le type de tampon utilisés pour la calibration, presser la touche plus de 

2 secondes. Les tampons apparaîtront tour à tour.

Etalonnage
METTLER TOLEDO SevenEasy 11

7. Test d’autocontrôle

Presser simultanément et maintenir les deux touches et jusqu’à l’affichage du symbole 

d’autocontrôle .

L’afficheur présente d’abord tous les éléments d’affichage. Les différents symboles clignotent ensuite 

successivement. Finalement, il faut encore contrôler les touches, ce qui nécessite l’intervention de 

l’utilisateur. 

Lorsque le symbole clignote, presser la touche correspondante en l’espace de 5 secondes, conformément à la liste suivante:

a. lorsque clignote, presser la touche ;

b. lorsque clignote, presser la touche ;

c. lorsque clignote, presser la touche ;

d. lorsque clignote, presser la touche ;

e. lorsque clignote, presser la touche .

Lorsque le test d’autocontrôle est terminé, un crochet est affiché. Si le test d’autocontrôle échoue, 

consulter “9. Messages d’erreur” et prendre les mesures qui s’imposent.

8. Fonctionnement facultatif sur pile

Le pH-mètre SevenEasy peut également fonctionner sur pile. Insérer 4 piles AA (LR6) à la base de 

l’appareil. Lorsque le câble de l’adaptateur secteur est retiré de l’appareil, ce dernier fonctionne sur pile 

et le symbole est affiché.

Si les piles sont déchargées, le symbole est affiché.

En fonctionnement sur pile, l’appareil dispose d’une fonction d’arrêt automatique. Au bout de 10 minutes 

sans action sur une touche, l’appareil s’arrête automatiquement pour ménager la pile. Pour contourner 

cette fonction d’arrêt automatique, presser et maintenir la touche pendant 2 secondes jusqu’à 

l’affichage du symbole .

Autocontrôle/Fonctionnement sur pile
12 METTLER TOLEDO SevenEasy

9. Messages d’erreur

Error 1 - Le décalage tombe hors de la plage permise

S’assurer que vous utilisez le bon tampon et qu’il est frais. Nettoyer ou remplacer l’électrode.

Error 2 - La pente tombe hors de la plage permise

S’assurer que vous utilisez le bon tampon et qu’il est frais. Nettoyer ou remplacer l’électrode.

Error 3 - L’instrument ne reconnaît pas le tampon

S’assurer que vous utilisez le bon tampon et qu’il est frais. Vérifier que le même tampon n’a pas été 

utilisé deux fois pour l’étalonnage. 

Error 4 - Erreur d’entrée de la valeur du tampon définie par l’utilisateur

Lors de l’entrée des valeurs tampons définies par l’utilisateur, l’appareil refuse les valeurs de pH qui diffèrent de moins d’une unité de pH par rapport aux autres valeurs définies. Entrer une nouvelle valeur.

Error 5 - La température ATC s’écarte de la valeur définie par l’utilisateur.

Le tampon ou l’échantillon devrait être à la température définie. Si nécessaire, changer le paramètre 

température. 

Error 6 - La température mesurée du tampon tombe hors de la plage permise (5... 50°C)

Il faut maintenir la température du tampon à l’intérieur de la plage permise.

Error 7, La valeur (----) - mV tombe hors de la plage de mesure

S’assurer que l’électrode est bien raccordée.

Si l’électrode n’est pas raccordée, insérer la pince de court-circuitage. 

Error 8 - Le pH tombe hors de la plage de mesure

Vérifier que le capuchon de l’électrode a été enlevé et que l’électrode est bien raccordée et plonge 

effectivement dans l’échantillon. 

Error 9 - Echec du test d’autocontrôle

Répéter le test d’autocontrôle et s’assurer que vous avez bien pressé les bonnes touches lorsque le 

symbole clignote. Si Err 9 se répète, contacter le service après-vente METTLER TOLEDO.

Messages d’erreur
METTLER TOLEDO SevenEasy 13

10. Entretien

10.1 Entretien de l’appareil

Cet appareil et l’adaptateur secteur ne comportent pas de pièces pouvant être remplacées par l’utilisateur. 

Prière de ne pas ouvrir le boîtier. 

A part un nettoyage occasionnel à l’aide d’un chiffon humide, le SevenEasy ne requiert pas d’entretien. 

Le boîtier est en ABS/PC qui peut être attaqué par des solvants organiques tels le toluène, le xylène et 

la butanone. La bonne pratique du laboratoire veut que les liquides renversés soient immédiatement 

essuyés. 

10.2 Entretien de l’électrode

S’assurer que l’électrode est toujours remplie de la bonne solution. Pour un maximum d’exactitude 

de mesure, il convient de laver à l’eau distillée toute trace de solution de remplissage pouvant avoir 

“grimpé” sur l’électrode et l’avoir encroûtée. 

Toujours conserver correctement l’électrode et ne jamais la laisser se dessécher.

Les procédés suivant peuvent aider lorsque la valeur de la pente de l’électrode tombe rapidement, que 

le temps de réaction devient long ou que les mesures deviennent inexactes. Les utiliser dans l’ordre 

suivant.

1. Dégraisser la membrane avec de l’ouate trempée d’acétone ou de solution savonneuse. 

2. Laisser tremper la pointe de l’électrode pendant la nuit dans 0.1 M HCl.

3. En cas de dépôt de protéines, l’enlever en faisant tremper l’électrode dans une solution de HCl /pepsine 

(référence pour la commande: 51340068).

4. En cas de contamination par du sulfure d’argent, tremper l’électrode dans une solution de thiourée 

(référence pour la commande: 51340070).

Remarque

Manipuler les solutions de nettoyage et de remplissage avec les mêmes précautions que celles prises 

pour les substances toxiques ou corrosives.

10.3 Elimination

En conformité avec les exigences de la directive européenne 2002/96 CE relative aux déchets 

d‘équipements électriques et électroniques (DEEE), cet appareil ne doit pas être éliminé avec 

les déchets ménagers. Logiquement, ceci est aussi valable pour les pays en dehors de l’UE 

conformément aux règlementations nationales en vigueur.

Veuillez éliminer cet appareil conformément aux prescriptions locales dans un conteneur 

séparé pour appareils électriques et électroniques.

Pour toute question, adressez-vous aux autorités compétentes ou au revendeur chez qui vous avez 

acheté cet appareil.

En cas de remise de cet appareil (p. ex. pour une utilisation privée ou artisanale/industrielle), cette 

prescription doit être transmise en substance.

Merci pour votre contribution à la protection de l’environnement.

Entretien
14 METTLER TOLEDO SevenEasy

Accessoires

11. Accessoires

référence

Bras porte-électrode complet 51302820

Guide pour les mesures de pH 51300185

Guide de mesure de la conductivité et de l’oxygène dissous 51724717

Guide pour les mesures ionometriques 51300200

Tampon pH 4,01, 30 sachets de 20mL 51302069

Tampon pH 4,01, 6 flacons de 250mL 51340058

Tampon pH 7,00, 30 sachets de 20mL 51302047

Tampon pH 7,00, 6 flacons de 250mL 51340060

Tampon pH 9,21, 30 sachets de 20mL 51302070

Tampon pH 9,21, 6 flacons de 250mL 51300194

Tampon pH 10,01, 30 sachets de 20mL 51302079

Tampon pH 10,01, 6 flacons de 250mL 51340231

InLab®410, électrode de pH, capteur de température, remplissable 52000118

InLab®412, électrode de pH de laboratoire, verre, remplissable 52000112

InLab®413, électrode de pH robuste, tige en PEEK, ATC 52000100

InLab®420, électrode de pH avec diaphragme PTFE à rodage 52000113

Solution de HCl/pepsine (enlève les dépôts de protéines) 51340068

Solution de thiourée (enlève les dépôts de sulfure d’argent) 51340070

Solution de réactivation pour électrodes de pH 51340073
METTLER TOLEDO SevenEasy 15

Caractéristiques techniques

12. Caractéristiques techniques

pH mV température

Plage de mesure 0,00 -14,00 ± 1999 –5 ... 105 °C

Résolution 0,01 1 0,1 °C

Exactitude relative ± 0,01 ± 1 ± 0,5 °C

Etalonnage pH 1, 2, ou 3 points

Point isopotentiel 7,00 pH

Tampons d’étalonnage B1: (25 °C) 7,00 4,00 10,01 1,68

B2: (25 °C) 7,00 4,01 9,21 2,00 11,00

B3: (20 °C) 7,00 4,00 9,00 2,00 12,00

B4: (25 °C) 6,86 4,01 9,18 1,68

B5: défini par l’utilisateur

Compensation de température –5 ... 105 °C, automatique ou manuelle

Affichage à cristaux liquides

Sorties série RS232, 

vitesse de transmission: 1200 baud

bits de données: 8

bit d’arrêt: 1

parité: néant

Entrée pH impédance > 1012 Ohm

Conditions ambiantes température: 5 °C ... 40 °C

humidité relative: 5 % ... 80 %, (non condensante)

catégorie de surtension: II

degré d’encrassement: 2

Dimensions/poids 180 x 180 x 65 (mm) / 0,61 kg

Matériaux Boîtier: ABS, fortifié avec PC 

Porte-électrode: ABS, fortifié avec PC 

Feuille du clavier: Polyester

Alimentation le pH-mètre SevenEasy est équipé d’un adaptateur secteur 

pour votre région:

- Etats-Unis: 120 V / 60 Hz, 10 VA, 9 V DC

- Europe: 230 V / 50 Hz, 10 VA, 9 V DC

- Royaume Uni: 240 V / 50 Hz, 10 VA, 9 V DC

- Japon: 110 V / 50 Hz, 10 VA, 9 V DC

- Australie: 240 V / 50 Hz, 10 VA, 9 V DC

- Chine: 220 V / 50 Hz, 10 VA, 9 V DC

- piles (facultatives): 4 x AA (LR6)

Remarque

N’alimenter le pH-mètre SevenEasy qu’à l’aide de l’adaptateur secteur fourni avec l’appareil ou par 

des piles.
16 METTLER TOLEDO SevenEasy

13. Guide rapide 1. Préparer l’électrode

enlever le capuchon humidificateur

2. Etalonnage à 1point

tampon pH 7

agiter puis laisser reposer

point final automatique ou

3. Rincer l’électrode 4. Etalonnage à 2 ou 3 points

répéter l’étape 2 pour l’étalonnage à 2 ou 3 points, 

puis rincer

5. Mesurer l’échantillon

échantillon

agiter puis laisser reposer

point final automatique ou

6. Rincer l’électrode 7. Conserver l’électrode

électrolyte

Guide rapide
METTLER TOLEDO SevenEasy 17

14. Ensembles de solutions tampon d’ étalonnage

Le pH-mètre SevenEasy compense automatiquement la température réglée en fonction des valeurs 

figurant dans les tableaux:

Groupe de tampons 1 (réf. 25 °C) MT US

 5 7,09 4,00 10,25 1,67

 10 7,06 4,00 10,18 1,67

 15 7,04 4,00 10,12 1,67

 20 7,02 4,00 10,06 1,68

25 7,00 4,00 10,01 1,68

 30 6,99 4,01 9,97 1,68

 35 6,98 4,02 9,93 1,69

 40 6,97 4,03 9,89 1,69

 45 6,97 4,04 9,86 1,70

 50 6,97 4,06 9,83 1,71

Groupe de tampons 2 (réf. 25 °C) MT Europe

 5 7,09 4,01 9,45 2,02 11,72

 10 7,06 4,00 9,38 2,01 11,54

 15 7,04 4,00 9,32 2,00 11,36

 20 7,02 4,00 9,26 2,00 11,18

 25 7,00 4,01 9,21 2,00 11,00

 30 6,99 4,01 9,16 1,99 10,82

 35 6,98 4,02 9,11 1,99 10,64

 40 6,97 4,03 9,06 1,98 10,46

 45 6,97 4,04 9,03 1,98 10,28

 50 6,97 4,06 8,99 1,98 10,10

Groupe de tampons 3 (réf. 20 °C) Merck étalon

 5 7,07 4,04 9,16 2,01 12,41

 10 7,05 4,02 9,11 2,01 12,26

 15 7,02 4,01 9,05 2,00 12,10

20 7,00 4,00 9,00 2,00 12,00

 25 6,98 4,01 8,95 2,00 11,88

 30 6,98 4,01 8,91 2,00 11,72

 35 6,96 4,01 8,88 2,00 11,67

 40 6,95 4,01 8,85 2,00 11,54

 45 6,95 4,01 8,82 2,00 11,44

 50 6,95 4,00 8,79 2,00 11,33

Groupe de tampons 4 (ref. 25 °C) Japon JIS

 5 6,95 4,00 9,40 1,67 

 10 6,92 4,00 9,33 1,67 

 15 6,90 4,00 9,28 1,67 

 20 6,88 4,00 9,22 1,68 

 25 6,86 4,01 9,18 1,68 

 30 6,85 4,02 9,14 1,68 

 35 6,84 4,02 9,10 1,69

 40 6,84 4,04 9,07 1,69

 45 6,83 4,05 9,04 1,70

 50 6,83 4,06 9,01 1,71 

Ensembles de solutions tampon d‘étalonnage
18 METTLER TOLEDO SevenEasy
METTLER TOLEDO SevenEasy 1

Sumario

Sumario

1. Introducción 3

2. Medidas de seguridad 3

Medidas para su protección 3

Medidas para la seguridad funcional 3

3. Descripción del instrumento 4

3.1 Indicador 4

3.2 Teclado 5

4. Instalación 6

5. Medición de las muestras 8

5.1 Medición de pH 8

5.2 Medición de mV 8

5.3 Ajustes 8

5.3.1 ATC 8

5.3.2 MTC 8

5.3.3 Salida de datos 8

6. Calibración 9

6.1 Ajustes 9

6.2 Selección de un grupo de tampones predefinidos 9

6.3 Ajuste de un grupo de tampones definido por el usuario 9

6.4 Calibración 10

6.4.1 Calibración a 1 punto 10

6.4.2 Calibración de 2 puntos 10

6.4.3 Calibración de 3 puntos 10

7. Test de autocontrol 11

8. Operación con pilas opcional 11

9. Mensajes de error 12

10. Mantenimiento 13

10.1 Mantenimiento del instrumento 13

10.2 Mantenimiento del electrodo 13

10.3 Eliminación de residuos 13

11. Accesorios 14

12. Características técnicas 15

13. Quick Guide 16

14. Grupo de tampones 17
2 METTLER TOLEDO SevenEasy
METTLER TOLEDO SevenEasy 3

Introducción / Medidas de seguridad

1. Introducción

SevenEasy – un instrumento que satisface sus exigencias, mucho más allá de la simple medición del 

pH, con un costo reducido y numerosas ventajas:

- El entorno del usuario está ideado de forma que la ejecución de cada fase operativa se explica por 

sí sola, lo que le permite ahorrar mucho tiempo.

- Gracias a la alimentación por pilas, el instrumento se puede instalar y utilizar rápidamente en otra 

mesa de laboratorio, o incluso en un laboratorio distinto, así como en lugares donde no se pueda 

usar alimentación eléctrica.

- Las prestaciones de servicio permiten la cualificación periódica de su instrumento, aumentando así 

la disponibilidad y exactitud del mismo.

2. Medidas de seguridad

Medidas para su protección

- ¡No trabajar en entorno expuesto a explosiones! La carcasa del instrumento no es 

hermética a los gases y existe el peligro de explosión por formación de chispas y/o 

peligro de corrosión por los gases introducidos.

- ¡Si utiliza productos químicos y disolventes siga las normas del fabricante y las normas 

de seguridad generales del laboratorio!

Medidas para la seguridad funcional

- ¡Encargue el mantenimiento del instrumento sólo al servicio METTLER TOLEDO!

- ¡Seque las salpicaduras de líquidos! El instrumento no es hermético al agua.

- No utilice pilas que no sean del tipo especificado. En otro caso no se garantiza un 

funcionamiento correcto.

- Evite los efectos ambientales siguientes:

- vibraciones fuertes,

- radiación solar directa,

- humedad del aire superior al 80%,

- atmósfera de gas corrosivo,

- temperaturas inferiores a 5 °C y superiores a 40 °C,

- ¡campos eléctricos o magnéticos fuertes!
4 METTLER TOLEDO SevenEasy

Descripción del instrumento

1 2 3 4 5 6

8

11

13

7

10

12

9

3. Descripción del instrumento

3.1 Indicador

1 Estado de las pilas

2 Anulación de la desactivación automática durante la operación con pilas

3 Transferencia de datos a PC/impresora 

4 Estado del electrodo

 Pendiente: 95-105 % Pendiente: 90-94 % Pendiente: 85-89 %

 Offset: ± (0-15) mV Offset: ± (15-35) mV Offset: ± (>35) mV

 Buen estado El electrodo necesita Electrodo defectuoso

 del electrodo limpieza

5 Autotest del instrumento

 Autotest del Indicación de Autotest 

 indicador pulsar tecla positivo 

6 Número de error 

7 Medición de pH/mV

8 Compensación de temperatura autom./manual

9 Estabilidad del punto final/punto final autom.

 Estabilidad del punto final A Punto final autom.

10 Temperatura 

11 Punto de calibración

12 Grupos de tampones

13 Offset y pendiente del electrodo
METTLER TOLEDO SevenEasy 5

Descripción del instrumento

3.2 Teclado

Pulsar y soltar Pulsar y mantener 

pulsada durante 2 s

Medidor on/off Anulación de la desactivación 

autom. en operación con pilas

- Medición de punto inicial o final

- Retorno al modo medida

- Almacenar valor introducido

Activar/desactivar punto final 

autom.

 / 

- Comenzar calibración

- Confirmar grupo de tampones seleccionados

Visualización del tampón usado 

en la calibración.

Alternar entre modo de medida pH y mV Transmisión de datos al PC o 

impresora

- Selección de los grupos de tampones de 

calibración

- Aumenta el valor durante el ajuste

- Ajustar temperatura MTC

- Reduce el valor durante el ajuste

Iniciar el autotest
6 METTLER TOLEDO SevenEasy

4. Instalación

1. Desembale el instrumento, el alimentador, el electrodo, el brazo portaelectrodo y los demás accesorios. 

Conserve el certificado de calibración en lugar seguro. 

2. Asegurese de que el alimentador corresponde a su red eléctrica. En caso contrario, póngase en 

contacto con su distribuidor METTLER TOLEDO.

3. Instale el brazo portaelectrodo.

- Se puede usar el portaelectrodo suelto de pie, o montado en el instrumento. Se pone sobre el soporte en una de las tres posiciones. En 

caso de que se use suelto de pie, se recomienda utilizar el orificio 

central. Primero hay que retirar la tapa correspondiente. Afiance la 

unión mediante el tornillo de fijación entregado. La instalación termina 

aquí si el portaelectrodo se va a usar de pie suelto.

- Si se va a sujetar el portaelectrodo en el instrumento, ponga éste 

sobre el soporte, de modo que la aleta libre del soporte quede orientada hacia atrás. Primero hay que retirar la tapa correspondiente. 

El portaelectrodo se puede fijar en la parte izquierda o derecha del 

instrumento.

- Regule el botón giratorio de tensión según sea necesario.

Instalación

aflojar

apretar
METTLER TOLEDO SevenEasy 7

Instalación

Conexión para alimentador

Conexión para sonda de temperatura ATC (compensación automática de temperatura)

Conexión para electrodo de pH combinado

Salida de datos RS232

4. Retire la horquilla cortacircuito de la conexión pH.

5. Conecte el electrodo. Si utiliza un electrodo con sonda de temperatura integrada, conecte el otro cable 

a la conexión ATC.

6. Si utiliza una sonda de temperatura separada, conéctela a la conexión ATC.

7. Conecte la unidad de alimentación al conector de corriente continua (DC).
8 METTLER TOLEDO SevenEasy

5. Medición de las muestras

5.1 Medición de pH

Ponga el electrodo en la muestra y pulse para empezar la medición – el punto decimal parpadea. 

El indicador presenta el valor de pH actual. El adjuste básico del instrumento es el punto final A (automático). Cuando la señal del sensor se han estabilizado, el indicador se para automáticamente y 

aparece .

El algoritmo del punto final automático es 0,1 mV por 5 segundos. 

Manteniendo pulsada la tecla puede Vd. cambiar entre formato de punto final automático o 

manual. Para introducir a mano el punto final para una medición, pulse la tecla . El indicador se 

detiene y aparece .

5.2 Medición de mV 

Para efectuar una medición de mV, proceda del mismo modo que con una medición de pH. Pulsando 

la tecla verá el valor en mV.

5.3 Ajustes

5.3.1 ATC

Para una mayor exactitud recomendamos usar una sonda de temperatura integrada o separada. Al utilizar 

una sonda de temperatura aparecen en el indicador el símbolo ATC y la temperatura de la muestra.

5.3.2 MTC

Cuando el instrumento detecta que no hay conectada sonda de temperatura, cambia automáticamente 

al modo de compensación de temperatura manual, y el indicador señala MTC . 

Para ajustar el temperatura MTC, pulse . Utilice y para desplazar el valor de temperatura 

hacia abajo o hacia arriba. Confirme su entrada pulsando la tecla El ajuste básico es 25 °C.

 

5.3.3 Salida de datos

Si hay conectado un PC o impresora, cualquier medición de punto final es transmitida al PC o impresora 

a través de la conexión RS232.

Manteniendo pulsada la tecla , aparece en el indicador. El instrumento transmite una medición 

por segundo hasta llegar al punto final.

Mediciones
METTLER TOLEDO SevenEasy 9

6. Calibración

6.1 Ajustes

El pH-metro SevenEasy permite calibraciones a 1, 2 y 3 puntos. Si Vd. selecciona su grupo de tampones 

de calibración entre los cuatro grupos predefinidos en el instrumento, durante la calibración se lleva a 

cabo la detección automática del grupo de tampones.

Los 4 grupos de tampones predefinidos son:

B1: (25 °C) 7,00 4,00 10,01 1,68

B2: (25 °C) 7,00 4,01 9,21 2,00 11,00

B3: (20 °C) 7,00 4,00 9,00 2,00 12,00

B4: (25 °C) 6,86 4,01 9,18 1,68 

Con el ajuste de tampón abajo descrito, puede Vd. definir su propio grupo de tampones. Durante la 

calibración, la detección automática de tampón no está activada. 

6.2 Selección de un grupo de tampones predefinidos

Pulse la tecla y el grupo de tampones actuales empezará a parpadear. Si el grupo actual está 

definido por el usuario, el cuadro vacío parpadea.

Utilice la tecla o la para seleccionar otros grupos en la posición superior o inferior. Cuando 

parpadee el grupo deseado, pulse para confirmar su selección.

6.3 Ajuste de un grupo de tampones definido por el usuario

Si en el paso 6.2 parpadea el cuadro vacío, pulse la tecla para empezar el ajuste. El instrumento 

señala el valor de temperatura ajustado en ese momento, y el punto y el cuadro parpadean (ajuste 

básico 25 °C). Utilice la tecla o la para cambiar el valor. Pulse la tecla para almacenar 

el valor y continuar

Después de introducir el valor de temperatura, el instrumento muestra el ajuste actual del primer tampón 

de calibración (ajuste básico 4.00). Utilice la tecla ó para cambiar el valor. Pulse la tecla 

 para guardar el valor y continuar.

Después de ajustar el primer tampón de calibración, pulse la tecla para ajustar el punto siguiente.

El procedimiento es el mismo que para el primer punto. Puede introducir hasta tres tampones de 

calibración definidos por el usuario. Cuando haya efectuado sus ajustes, pulse la tecla para salir 

del menú.

Observación

Si utiliza un grupo de tampones definidos por el usuario para la calibración, aparece el valor tampón 

introducido por Vd. Asegúrese de utilizar el tampón adecuado. La temperatura del tampón debe atenerse 

al valor introducido. Cuando se trabaja con sonda y la diferencia entre temperatura medida y valor 

introducido supera 1 °C, aparece Err 5 en el indicador.

Calibración
10 METTLER TOLEDO SevenEasy

6.4 Calibración

6.4.1 Calibración a 1 punto

Introduzca el electrodo en un tampón de calibración y pulse .

El pH-metro SevenEasy alcanza automáticamente durante la calibración al punto final. Para alcanzar el 

punto final manualmente, pulse la tecla . El instrumento señala el valor correspondiente del tampón 

y se detiene en él. Aparece indicado el valor de offset del electrodo.

Para volver a la medición de muestra, pulse .

6.4.2 Calibración de 2 puntos

Paso 1o Efectúe una “calibración de 1 punto”, como se ha explicado antes.

Paso 2o Lave el electrodo con agua destilada.

Paso 3o Introduzca el electrodo en el tampón siguiente y pulse .

El pH-metro SevenEasy alcanza automáticamente durante la calibración al punto final. Para alcanzar 

manualmente al instrumento al punto final, pulse . El instrumento señala el valor tampón correspondiente y se detiene en él. Se actualiza al valor offset del electrodo y se indica la pendiente.

Para volver a la medición de la muestra, pulse .

6.4.3 Calibración de 3 puntos

Realice una “calibración de 2 puntos” como anteriormente. Repita los pasos 2 y 3 para la calibración 

de 3 puntos.

Observación

Se recomienda utilizar una sonda de temperatura o un electrodo con sonda de temperatura integrada. 

Si recurre al modo MTC, todos los tampones y soluciones de la muestra deben mantenerse a la misma 

temperatura regulada.

A fin de obtener una medida de pH lo más exacta posible, se debe calibrar periódicamente el instrumento.

Para visualizar el número y el tipo de tampón usado en la calibración, pulsar y mantener pulsada la 

tecla durante 2 segundos. Aparecen sucesivamente los valores de los tampones.

Calibración
METTLER TOLEDO SevenEasy 11

7. Test de autocontrol

Mantenga pulsadas las dos teclas y al mismo tiempo, hasta que aparezca el símbolo del 

autotest .

En el instrumento aparece primero la indicación completa. A continuación parpadean sucesivamente 

los distintos símbolos. Luego hay que comprobar las teclas con la intervención del usuario.

Cuando parpadee el símbolo , pulse la tecla correspondiente antes de 5 segundos, de acuerdo 

con la lista siguiente

a. Si parpadea , pulse la tecla ;

b. Si parpadea , pulse la tecla ;

c. Si parpadea , pulse la tecla ;

d. Si parpadea , pulse la tecla ;

e. Si parpadea , pulse la tecla .

Cuando el test de autocontrol está terminado aparece la señal en el indicador. Si el autocontrol es 

negativo, consulte bajo “9. Mensajes de error” y adopte las medidas necesarias.

8. Operación con pilas opcional

El pH-metro SevenEasy permite el funcionamiento opcional con pilas. Introduzca 4 pilas AA (LR6) en la 

parte inferior del instrumento. Al retirar el cable de alimentador del instrumento, éste pasa a la operación 

con pilas y aparece el símbolo .

Cuando las pilas están descargadas aparece el símbolo en el indicador.

En la operación con pilas el instrumento dispone de una función de desactivación automática. Si durante 

10 minutos no se pulsa ninguna tecla, el instrumento se desactiva automáticamente para proteger las 

pilas. Para salir de esta función de desactivación automática, mantenga pulsada la tecla durante 

2 segundos hasta que aparezca el símbolo .

Autocontrol/Operación con pilas
12 METTLER TOLEDO SevenEasy

9. Mensajes de error

Error 1 - Offset excede el intervalo permitido

Asegúrese de que el tampón es el adecuado y reciente. Limpie o sustituya el electrodo.

Error 2 - La pendiente excede el intervalo permitido

Asegúrese de que el tampón es el adecuado y reciente. Limpie o sustituya el electrodo.

Error 3 - El instrumento no reconoce el tampón

Asegúrese de que el tampón es el adecuado y reciente. Verifique que no se ha usado dos veces para 

la calibración.

Error 4 - Error de entrada del valor tampón definido por el usuario

Cuando Vd. introduce valores tampones definidos por el usuario, el instrumento no acepta valores de 

pH inferiores en 1 unidad en los valores definidos por el usuario. Introduzca un valor nuevo.

Error 5 - La temperatura ATC se desvía del valor definido por el usuario.

La temperatura del tampón o de la muestra deben corresponder a la temperatura introducida. Si hiciera 

falta, cambie el ajuste de temperatura.

Error 6 - La temperatura de tampón medida excede el intervalo permitido (5...50 °C)

La temperatura del tampón debe mantenerse dentro del rango permitido.

Error 7, (——) - El valor mV excede el intervalo de medida

Asegúrese de que el electrodo está conectado.

Si el electrodo no está conectado, introduzca la horquilla cortacircuito.

Error 8 - El valor pH excede el intervalo de medida

Compruebe que el capuchón protector del electrodo está quitado y que el electrodo se encuentra correctamente conectado y sumergido en la muestra.

Error 9 - Test de autocontrol negativo

Repita el test de autocontrol y cerciórese de haber pulsado las teclas adecuadas cuando parpadee el 

símbolo . Si sigue apareciendo Err 9, póngase en contacto con el servicio METTLER TOLEDO.

Mensajes de error
METTLER TOLEDO SevenEasy 13

10. Mantenimiento

10.1 Mantenimiento del instrumento

Ni las piezas de este instrumento ni las del alimentador son sustituibles por el usuario. Por favor, no 

abra la carcasa.

El SevenEasy no necesita mantenimiento alguno, a excepción de una limpieza ocasional con paño 

húmedo. La carcasa es de ABS/PC, que resistente a algunos disolventes orgánicos como tolueno, 

xileno y metiletilcetona. La buena práctica de laboratorio requiere la inmediata eliminación de los 

líquidos derramados.

10.2 Mantenimiento del electrodo

Asegúrese de que el electrodo está siempre lleno de electrolito. Para obtener máxima exactitud, cualquier solución añadida que “resbale” hacia fuera y pueda formar costra sobre el electrodo, debe ser 

eliminada con agua destilada.

Guarde el electrodo siempre en condiciones adecuadas, y no lo deje secar.

Si el valor de pendiente del electrodo disminuye rápidamente, la reacción se hace lenta, o las mediciones 

imprecisas, las operaciones siguientes pueden resultar útiles. Aplíquelas en este mismo orden:

1. Desengrasar la membrana con algodón empapado en acetona o solución jabonosa.

2. Sumerja la punta del electrodo durante la noche en 0.1 M HCl.

3. Si se origina formación proteínica, eliminar los residuos metiendo el electrodo en una solución de 

HCl/pepsina (no de pedido: 51340068).

4. En caso de contaminación con sulfuro de plata, sumergir el electrodo en una solución de tiourea (no 

de pedido: 51340070).

Observación

Las soluciones de limpieza y de relleno han de manipularse con las mismas precauciones que las 

sustancias tóxicas o corrosivas.

10.3 Eliminación de residuos

De conformidad con las exigencias de la directiva europea 2002/96/CE sobre residuos de 

aparatos eléctricos y electrónicos (RAEE), este equipo no puede eliminarse como basura 

doméstica. Esta prohibición es asimismo válida para los países que no pertenecen a la UE 

cuyas normativas nacionales en vigor así lo reflejan.

Elimine este producto, según las disposiciones locales, mediante el sistema de recogida 

selectiva de aparatos eléctricos y electrónicos.

Si tiene alguna pregunta al respecto, diríjase a las autoridades responsables o al distribuidor que le 

proporcionó el equipo.

Si transfiere este equipo (por ejemplo, para la continuación de su uso con fines privados, comerciales 

o industriales), deberá transferir con él esta disposición.

Muchas gracias por su contribución a la conservación medioambiental.

Mantenimiento
14 METTLER TOLEDO SevenEasy

Accesorios

11. Accesorios

No de ped.

Brazo portaelectrodos completo 51302820

Guide to pH measurement (English) 51300047

Guide to conductivity and dissolved oxygen (English) 51724716

Guide to ion selective measurement (English) 51300075

Tampón pH 4,01, 30 bolsas de 20mL 51302069

Tampón pH 4,01, 6 frascos de 250mL 51340058

Tampón pH 7,00, 30 bolsas de 20mL 51302047

Tampón pH 7,00, 6 frascos de 250mL 51340060

Tampón pH 9,21, 30 bolsas de 20mL 51302070

Tampón pH 9,21, 6 frascos de 250mL 51300194

Tampón pH 10,01, 30 bolsas de 20mL 51302079

Tampón pH 10,01, 6 frascos de 250mL 51340231

InLab®410, electrodo de pH, sonda de temperatura, rellenable 52000118

InLab®412, electrodo de pH de laboratorio, vidrio, rellenable 52000112

InLab®413, electrodo de pH robusto, cuerpo de PEEK, ATC 52000100

InLab®420, electrodo de pH con diafragma esmerilado de PTFE 52000113

Solución de HCl/pepsina (elimina restos proteínicos) 51340068

Solución de tiourea (elimina restos de sulfuro de plata) 51340070

Solución de reactivación para electrodos de pH 51340073
METTLER TOLEDO SevenEasy 15

Características técnicas

12. Características técnicas

pH mV Temperatura

Intervalo de medida 0,00 -14,00 ± 1999 –5 ... 105 °C

Resolución 0,01 1 0,1 °C

Exactitud relativa ± 0,01 ± 1 ± 0,5 °C

Calibración de pH 1, 2, ó 3 puntos

Punto isopotencial 7,00 pH

Tampónes de calibración B1: (25 °C) 7,00 4,00 10,01 1,68

B2: (25 °C) 7,00 4,01 9,21 2,00 11,00

B3: (20 °C) 7,00 4,00 9,00 2,00 12,00

B4: (25 °C) 6,86 4,01 9,18 1,68

B5: definido por el usuario

Compensación de temperat. –5 ... 105 °C automática o manual

Indicador de cristal líquido

Salidas RS232 serie, 

Vel. de transmisión: 1200 baudios

Bits de datos: 8

Bits de parada: 1

Paridad: no

Entrada pH Impedancia > 1012 ohmios

Condiciones ambientales Temperatura ambiente: 5°C ... 40 °C

Humedad rel. del aire: 5 % ... 80 %, (sin condensación)

Categ.de sobrevoltaje: II

Grado de contaminación: 2

Dimensiones/peso 180 x 180 x 65 (mm) / 0,61 kg

Material Carcasa: ABS, reforzado con PC 

Portaelectrodo: ABS, reforzado con PC 

Tarjeta del teclado: Polyester

Alimentación eléctrica El pH-metro SevenEasy va provisto de un alimentador 

apropiado para su región:

- EEUU: 120 V / 60 Hz, 10 VA, 9 V DC

- Europa: 230 V / 50 Hz, 10 VA, 9 V DC

- Reino Unido: 240 V / 50 Hz, 10 VA, 9 V DC

- Japón: 110 V / 50 Hz, 10 VA, 9 V DC

- Australia: 240 V / 50 Hz, 10 VA, 9 V DC

- China: 220 V / 50 Hz, 10 VA, 9 V DC

- Pilas (opcionales): 4 x AA (LR6)

Observación

El pH-metro SevenEasy sólo debe operar con el alimentador suministrado, o con pilas.
16 METTLER TOLEDO SevenEasy

13. Quick Guide 1. Preparar electrodo

retirar capuchón protector

2. Calibración a 1 punto

tampón pH 7

agitar, dejar 

en reposo

punto final automático 

, o bien

3. Lavar electrodo 4. Calibración de 2 ó 3 puntos

Repita el paso 2 para una calibración a 2 ó 3 puntos, luego lave

5. Medir muestra

muestra

agitar, dejar 

en reposo

6. Lavar electrodo 7. Guardar electrodo

electrolito

punto final automático, 

o bien

Quick Guide
METTLER TOLEDO SevenEasy 17

14. Grupo de tampones

El pH-metro SevenEasy compensa automáticamente la temperatura ajustada con arreglo a los valores 

de las tablas:

Grupo tampones 1 (ref. 25 °C) MT USA

 5 7,09 4,00 10,25 1,67

 10 7,06 4,00 10,18 1,67

 15 7,04 4,00 10,12 1,67

 20 7,02 4,00 10,06 1,68

25 7,00 4,00 10,01 1,68

 30 6,99 4,01 9,97 1,68

 35 6,98 4,02 9,93 1,69

 40 6,97 4,03 9,89 1,69

 45 6,97 4,04 9,86 1,70

 50 6,97 4,06 9,83 1,71

Grupo tampones 2 (ref. 25 °C) MT EU

 5 7,09 4,01 9,45 2,02 11,72

 10 7,06 4,00 9,38 2,01 11,54

 15 7,04 4,00 9,32 2,00 11,36

 20 7,02 4,00 9,26 2,00 11,18

 25 7,00 4,01 9,21 2,00 11,00

 30 6,99 4,01 9,16 1,99 10,82

 35 6,98 4,02 9,11 1,99 10,64

 40 6,97 4,03 9,06 1,98 10,46

 45 6,97 4,04 9,03 1,98 10,28

 50 6,97 4,06 8,99 1,98 10,10

Grupo tampones 3 (ref. 20 °C) Tampones Merck

 5 7,07 4,04 9,16 2,01 12,41

 10 7,05 4,02 9,11 2,01 12,26

 15 7,02 4,01 9,05 2,00 12,10

20 7,00 4,00 9,00 2,00 12,00

 25 6,98 4,01 8,95 2,00 11,88

 30 6,98 4,01 8,91 2,00 11,72

 35 6,96 4,01 8,88 2,00 11,67

 40 6,95 4,01 8,85 2,00 11,54

 45 6,95 4,01 8,82 2,00 11,44

 50 6,95 4,00 8,79 2,00 11,33

Grupo tampones 4 (ref. 25 °C) Japón JIS

 5 6,95 4,00 9,40 1,67 

 10 6,92 4,00 9,33 1,67 

 15 6,90 4,00 9,28 1,67 

 20 6,88 4,00 9,22 1,68 

 25 6,86 4,01 9,18 1,68 

 30 6,85 4,02 9,14 1,68 

 35 6,84 4,02 9,10 1,69

 40 6,84 4,04 9,07 1,69

 45 6,83 4,05 9,04 1,70

 50 6,83 4,06 9,01 1,71 

Grupo de tampones
18 METTLER TOLEDO SevenEasy
METTLER TOLEDO SevenEasy 1

Indice del contenuto

Indice del contenuto

1. Introduzione 3

2. Misure di sicurezza 3

Misure per la protezione dell’utente 3

Misure per la sicurezza di funzionamento 3

3. Descrizione dello strumento 4

3.1 Indicatore 4

3.2 Tastiera 5

4. Installazione 6

5. Misurazione campioni 8

5.1 Misurazione di pH 8

5.2 Misurazione di mV 8

5.3 Regolazioni 8

5.3.1 ATC 8

5.3.2 MTC 8

5.3.3 Emissione dati 8

6. Calibrazione 9

6.1 Regolazioni 9

6.2 Selezione di un gruppo di tamponi predefinito 9

6.3 Impostazione di un gruppo di tamponi definito dall’utente 9

6.3 Calibrazione 10

6.3.1 Calibrazione a 1 punto 10

6.3.2 Calibrazione a 2 punti 10

6.3.3 Calibrazione a 3 punti 10

7. Autodiagnosi 11

8. Funzionamento opzionale con alimentazione a batterie 11

9. Messaggi d’errore 12

10. Manutenzione 13

10.1 Manutenzione dello strumeno 13

10.2 Manutenzione dell’elettrodo 13

10.3 Smaltimento 13

11. Accessori 14

12. Specifiche tecniche 15

13. Guida Rapida 16

14. Gruppi di tamponi 17
2 METTLER TOLEDO SevenEasy
METTLER TOLEDO SevenEasy 3

Introduzione / Misure di sicurezza

1. Introduzione

SevenEasy – uno strumento che soddisfa le vostre esigenze ben oltre la pura misurazione del pH, senza 

sovraccaricare inutilmente il vostro budget. Uno strumento che offre molti vantaggi:

- L’interfaccia utente è concepita in maniera tale da renere non più necessaria la consultazione del 

manuale di utilizzo. In questo modo risparmiate molto tempo.

- Grazie all’alimentazione a batteria, lo strumento può essere rapidamente installato e montato su un 

altro tavolo da laboratorio, o persino in un altro laboratorio, anche in luoghi in cui non è possibile 

utilizzare l’alimentazione da rete.

- Le prestazioni di servizio consentono una regolare qualificazione del vostro strumento e ne migliorano 

l’affidabilità e la precisione.

2. Misure di sicurezza

Misure per la protezione dell’utente

- Mai lavorare in un ambiente a rischio d’esplosione. Lo chassis dello strumento non 

è a tenuta di gas. Vi è un rischio di esplosione in caso di scintille e/o un rischio di 

corrosione in presenza di gas).

- In caso di utilizzo di agenti chimici e solventi, attenersi alle istruzioni del produttore ed 

alle normative a carattere generale in tema di sicurezza in laboratorio.

Misure per la sicurezza di funzionamento

- Fare eseguire la manutenzione solo dal personale dell’Assistenza METTLER TOLEDO.

- Asciugare subito eventuali spruzzi di liquido. Lo strumento non è a tenuta d’acqua.

- Usare solo batterie del tipo specificato. In caso contrario, non può essere garantito il 

corretto funzionamento dello strumento.

- Eliminare le seguenti influenze ambientali:

- forti vibrazioni

- esposizione diretta alla luce solare,

- umidità atmosferica superiore all’80%,

- presenza di gas corrosivi nell’atmosfera,

- temperature inferiori a 5 °C e superiori a 40 °C,

- campi elettrici o magnetici di forte intensità.
4 METTLER TOLEDO SevenEasy

Descrizione dello strumento

1 2 3 4 5 6

8

11

13

7

10

12

9

3. Descrizione dello strumento

3.1 Indicatore

1 Stato della batteria

2 Disattivazione autospegnimento automatico durante il funzionamento a batterie

3 Trasferimento dati al PC/Stampante 

4 Stato dell’elettrodo

 Pendenza: 95-105 % Pendenza: 90-94 % Pendenza: 85-89 %

 Offset: ± (0-15) mV Offset: ± (15-35) mV Offset: ± (>35) mV

 L’elettrodo è in L’elettrodo deve L’elettrodo è difettoso

 buono stato L’elettrodo deve 

5 Autodiagnosi dello strumento 

 Indicatore Richiesta di Autodiagnosi 

 autodiagnosi premere un tasto superata

6 Codice errore 

7 Misurazione pH/mV

8 Compensazione temperatura in automatico/manuale

9 Stabilità punto finale/punto finale in automatico

 Stabilità punto finale A Punto finale automatico

10 Temperatura 

11 Punto di calibrazione

12 Gruppi di tamponi

13 Punto zero e pendenza dell’elettrodo
METTLER TOLEDO SevenEasy 5

Descrizione dello strumento 

3.2 Tastiera

Premere e rilasciare Premere e tenere premuto per 2 secondi

Misuratore on/off Disattivazione autospegnimento 

con funzionamento a batterie

- Misurazione punto iniziale o finale

- Ritorno al modo misurazione

- Memorizzazione valore introdotto

Attivaz./disattivaz. punto finale 

autom.

 / 

- Inizio calibrazione

- Conferma gruppo di tamponi selezionato

Indicatore delle soluzioni tampone impiegate per la calibrazione.

Commutazione tra modo misurazione 

pH e mV 

Trasferimento dati al PC o alla 

stampante

- Selezione gruppi di tamponi calibrazione

- Aumento valore durante l’impostazione

- Impostazione temperatura MTC

- Riduzione valore durante l’impostazione

Avvio autodiagn. 
6 METTLER TOLEDO SevenEasy

4. Installazione 

1. Disimballare lo strumento, l’alimentatore, l’elettrodo, il braccio dell’elettrodo e gli altri accessori. 

Conservare il certificato di calibrazione in un luogo sicuro. 

2. Accertarsi che la tensione indicata sull’alimentatore corrisponda a quella della rete di alimentazione. 

Altrimenti, contattare il rivenditore METTLER TOLEDO.

3. Installare il braccio dell’elettrodo.

- Il portaelettrodo può essere utilizzato da solo oppure può essere 

montato sullo strumento. Il braccio dell’elettrodo deve essere posizionato in una delle tre posizioni sul supporto. Se il portaelettrodo viene 

usato da solo, si raccomanda di usare la posizione intermedia. Per 

prima cosa, rimuovere il coperchio corrispondente. Utilizzare la vite 

di fissaggio fornita per bloccare il braccio in posizione. Se si utilizza 

il portaelettrodo da solo, l’installazione è terminata.

- Per fissare il portaelettrodo allo strumento, posizionare il supporto 

in modo tale che l’ala del supporto non occupata risulti rivolta verso 

il basso dello strumento. Per prima cosa, rimuovere il coperchio 

corrispondente. Il portaelettrodo può essere fissato sul lato sinistro 

oppure sul lato destro.

- Regolare la vite di fissaggio secondo necessità. 

Installazione 

Svitare

Avvitare
METTLER TOLEDO SevenEasy 7

Installazione

Collegamento connettore alimentazione da rete

Connettore per sonda temperatura ATC (compensazione temperatura in automatico)

Connettore per elettrodo combinato per misurazioni 

di pH Uscita dati RS232

4. Scollegare le pinze di corto circuito dal connettore pH.

5. Collegare l’elettrodo. Se si usa un elettrodo con sonda di temperatura integrata, collegare l’altro cavo 

al connettore ATC.

6. Se si usa una sonda di temperatura separata, collegarla al connettore ATC. 

7. Collegare l’alimentatore al connettore di alimentazione elettrica a corrente continua (CC). 
8 METTLER TOLEDO SevenEasy

5. Misurazione campioni 

5.1 Misurazione di pH 

Immergere l’elettrodo nel campione e premere per avviare la misurazione, il punto decimale 

lampeggia. 

L’indicatore mostra il valore di pH del campione, e il punto finale in automatico A è l’impostazione di 

default del misuratore. Appena i dati trasmessi dal sensore si stabilizzano, l’indicatore si blocca automaticamente e visualizza l’icona .

Il algoritmo del punto finale automatico è 0,1 mV per 5 secondi.

Tenendo premuto il tasto , è possibile commutare tra il modo a punto finale in automatico e manuale. 

Per introdurre manualmente il punto finale per una misurazione, premere il tasto . L’indicatore si 

blocca e visualizza l’icona .

5.2 Misurazione di mV 

Per effettuare misurazioni di mV, seguire la stessa procedura prevista per le misurazioni di pH. Per 

visualizzare il valore mV durante le misurazioni di pH, premere il tasto .

5.3 Regolazioni 

5.3.1 ATC

Per una maggiore precisione, si raccomanda l’uso di una sonda di temperatura integrata o separata. 

Se si utilizza una sonda di temperatura, sull’indicatore vengono visualizzati il simbolo ATC e la temperatura del campione.

5.3.2 MTC 

Se lo strumento non rileva una sonda di temperatura collegata, esso commuta automaticamente al modo 

di compensazione della temperatura manuale, e l’indicatore visualizza il simbolo MTC. 

Per impostare la temperatura MTC, premere il tasto . Utilizzare i tasti e , per aumentare o 

ridurre il valore di temperatura, premere il tasto per confermare la regolazione effettuata. L’impostazione 

di default è pari a 25 °C.

 

5.3.3 Emissione dati 

Se lo strumento è collegato ad un PC o ad una stampante, ciascuna misurazione del punto finale viene 

trasmessa al PC o alla stampata tramite il connettore RS232. 

Premendo e tenendo premuto il tasto , l’indicatore visualizza l’icona . Lo strumento trasmette 

una misurazione al secondo, fino a quando viene raggiunto il punto finale.

Misurazione 
METTLER TOLEDO SevenEasy 9

6. Calibrazione 

6.1 Regolazioni 

Il misuratore di pH SevenEasy consente di effettuare calibrazioni a 1, 2 e 3 punti. Quando si seleziona un 

gruppo di tamponi di calibrazione tra i quattro gruppi preimpostati nell strumento, durante la calibrazione 

lo strumento effettua il riconoscimento automatico del gruppo di tamponi.

I 4 gruppi di tamponi preimpostati sono:

B1: (25 °C) 7,00 4,00 10,01 1,68

B2: (25 °C) 7,00 4,01 9,21 2,00 11,00

B3: (20 °C) 7,00 4,00 9,00 2,00 12,00 

B4: (25 °C) 6,86 4,01 9,18 1,68 

Seguendo la procedura di regolazione tampone descritta sotto, l’utente può definire un proprio gruppo 

di tamponi personalizzato. Durante la calibrazione, il riconoscimento automatico del tampone non è 

attivo. 

6.2 Selezione di un gruppo di tamponi predefinito 

Premere il tasto , il tampone corrente inizia a lampeggiare. Se il gruppo di tamponi corrente è un 

gruppo definito dall’utente, il campo vuoto lampeggia.

Usare il tasto o per selezionare altri gruppi nella posizione superiore o inferiore. Quando il 

gruppo desiderato lampeggia, premere il tasto per confermare la selezione.

6.3 Impostazione di un gruppo di tamponi definito dall’utente

Se al punto 6.2, il campo vuoto lampeggia, premere il tasto , per avviare l’impostazione. Lo strumento visualizza il valore di temperatura corrente impostato, mentre il punto e il campo lampeggiano 

(la temperatura di default è 25 °C). Usare il tasto o per modificare il valore. Premere il tasto 

 per memorizzare il valore e continuare.

Dopo avere impostato il valore della temperatura, lo strumento visualizza l’impostazione corrente del 

primo tampone di calibrazione (valore di default 4.00). Usare il tasto o , per modificare il 

valore. Premere il tasto per memorizzare il valore e continuare.

Dopo aver impostato il primo tampone di calibrazione, premere il tasto per avviare la regolazione 

del punto successivo.

La procedura è identica a quella prevista per il primo punto. È possibile impostare fino a 3 tamponi di 

calibrazione definiti dall’utente. Dopo aver completato le impostazioni, premere il tasto per uscire 

dal menu.

Nota 

Se per la calibrazione si utilizza un gruppo di tamponi definito dall’utente, l’indicatore visualizzerà il 

valore del tampone che Lei ha impostato. Accertarsi che si stia utilizzando il tampone corretto. Inoltre, 

mantenere la temperatura del tampone al valore impostato. Se si installa una sonda di temperatura e 

la temperatura misurata differisce di oltre 1 grado rispetto al valore impostato, l’indicatore visualizza 

il messaggio Err 5.

Calibrazione 
10 METTLER TOLEDO SevenEasy

6.3 Calibrazione 

6.3.1 Calibrazione a 1 punto

Immergere l’elettrodo in un tampone di calibrazione e premere il tasto .

Durante la calibrazione, il misuratore di pH SevenEasy raggiunge automaticamente il punto finale. Per 

raggiungere manualmente il punto finale, premere il tasto . Lo strumento indica il valore del tampone 

corrispondente, e si ferma. L’indicatore mostra il punto zero dell’elettrodo. 

Per tornare alla misurazione del campione, premere il tasto .

6.3.2 Calibrazione a 2 punti 

Passo 1. Eseguire una “Calibrazione a 1 punto” come descritto sopra.

Passo 2. Lavare l’elettrodo con acqua distillata.

Passo 3. Immergere l’elettrodo nel tampone di calibrazione successivo e premere .

SevenEasy raggiunge automaticamente il punto finale. Per raggiungere manualmente il punto finale, 

premere il tasto . Lo strumento indica il valore del tampone corrispondente, e si ferma. Il punto zero 

dell’elettrodo viene aggiornato e l’indicatore visualizza la pendenza.

Per tornare alla misurazione del campione, premere il tasto .

6.3.3 Calibrazione a 3 punti 

Effettuare una “Calibrazione a 2 punti” secondo quanto descritto sopra. Per la calibrazione a 3 punti, 

ripetere i passi 2 e 3. 

Nota

Si raccomanda l’uso di una sonda di temperatura o di un elettrodo con sonda di temperatura integrata. 

Se si utilizza il modo MTC, conservare tutte le soluzioni tampone e campione alla temperatura impostata. Per garantire la massima precisione possibile della misurazione del pH, è necessario effettuare la 

calibrazione dello strumento a intervalli di tempo regolari.

Per indicare il numero e il tipo delle soluzioni tampone impiegate per la calibrazione, premere e mantenere 

premuto il tasto per 2 secondi. I valori delle soluzioni tampone sono indicati in alternanza.

Calibrazione 
METTLER TOLEDO SevenEasy 11

7. Autodiagnosi

Premere e tenere premuti contemporaneamente entrambi i tasti e fino a quando sull’indicatore 

compare l’icona di autodiagnosi del misuratore.

Sullo strumento compare prima lo schermo intero, poi le singole icone lampeggiano una dopo l’altra. 

Infine, è necessario che l’utente verifichi il corretto funzionamento dei tasti.

Quando lampeggia l’icona , entro 5 secondi premere il tasto corrispondente, come indicato nel 

seguente elenco:

a. Quando lampeggia l’icona , premere il tasto ;

b. Quando lampeggia l’icona , premere il tasto ;

c. Quando lampeggia l’icona , premere il tasto ;

d. Quando lampeggia l’icona , premere il tasto ;

e. Quando lampeggia l’icona , premere il tasto .

Una volta conclusa l’autodiagnosi, nell’indicatore compare un segno di spunta . Se l’autodiagnosi 

fallisce, consultare il paragrafo “9. Messaggi d’errore”, e adottare le misure necessarie.

8. Funzionamento opzionale con alimentazione a batterie

Il misuratore di pH SevenEasy consente il funzionamento opzionale con alimentazione a batterie. Installare 4 

batterie tipo AA (LR6) sul retro dello strumento. Se il cavo dell’alimentatore non è collegato, lo strumento 

funziona con alimentazione a batterie e sull’indicatore compare l’icona .

Quando le batterie sono scariche, nell’indicatore compare l’icona .

Durante il funzionamento con alimentazione a batterie, lo strumento dispone di una funzione di autospegnimento. Se per 10 minuti l’utente non preme alcun tasto, lo strumento si spegne automaticamente 

per risparmiare le batterie. Per disattivare la funzione di autospegnimento, tenere premuto il tasto 

per 2 secondi, fino a quando nell’indicatore compare l’icona .

Autodiagnosi/Funzionamento a batterie
12 METTLER TOLEDO SevenEasy

9. Messaggi d’errore

Errore 1 - Valore punto zero al di fuori dell’intervallo consentito

Accertarsi che si stia usando il tampone corretto e che lo stesso sia fresco. Pulire l’elettrodo oppure 

sostituirlo.

Errore 2 - Pendenza al di fuori dell’intervallo consentito

Accertarsi che si stia usando il tampone corretto e che lo stesso sia fresco. Pulire l’elettrodo oppure 

sostituirlo.

Errore 3 - Lo strumento non riconosce il tampone

Accertarsi che si stia usando il tampone corretto e che lo stesso sia fresco. Verificare se il tampone è 

stato utilizzato più volte per la calibrazione.

Errore 4 - Errore introduzione valore gruppo di tamponi definito dall’utente

Se l’utente introduce il valore del gruppo di tamponi definito dall’utente, lo strumento non accetta valori 

di pH inferiori, di più di un’unità di pH, rispetto ai valori definiti. Introdurre un nuovo valore.

Errore 5 - Temperatura ATC differisce dal valore definito dall’utente

Mantenere il tampone o il campione alla temperatura impostata. Se necessario, modificare l’impostazione della temperatura.

Errore 6 - Temperatura misurata del tampone al di fuori dell’intervallo consentito (5... 50°C)

Mantenere la temperatura del tampone entro l’intervallo.

Errore 7, (----) - Valore mV al di fuori dell’intervallo di misurazione

Accertarsi che l’elettrodo sia collegato. 

Se nessun elettrodo è collegato, inserire le pinze di corto circuito nel connettore. 

Errore 8 - Valore di pH al di fuori dell’intervallo di misurazione

Verificare se il coperchio dell’elettrodo è stato rimosso e se l’elettrodo è collegato e immerso correttamente in una soluzione campione. 

Errore 9 - Autodiagnosi fallita

Ripetere la procedura di autodiagnosi ed accertarsi di aver premuto i tasti corretti mentre l’icona 

lampeggiava. Se nell’indicatore compare nuovamente il messaggio Err 9, contattare il Servizio Assistenza METTLER TOLEDO.

Messaggi d’errore
METTLER TOLEDO SevenEasy 13

10. Manutenzione 

10.1 Manutenzione dello strumeno

Né lo strumento, né l’alimentatore contengono parti che possono essere sostituite dall’utente. Non 

aprire lo chassis. 

Il SevenEasy non richiede alcun intervento di manutenzione eccetto la pulizia occasionale, che dev’essere effettuata con un panno umido. Lo chassis è costituito da ABS/PC, che può essere corroso da 

alcuni solventi organici, come toluene, xilene, metiletilchetone. È buona pratica di laboratorio asciugare 

immediatamente eventuali spruzzi. 

10.2 Manutenzione dell’elettrodo

Accertarsi che l’elettrodo sia sempre riempito con la soluzione di riempimento. Per ottenere la massima precisione possibile, rimuovere con acqua distillata eventuali residui di soluzione di riempimento, che potrebbero 

essersi “infiltrati” ed essere rimasti incrostati sulla superficie esterna dell’elettrodo. 

Conservare sempre in modo adeguato l’elettrodo e non farlo asciugare. 

Se il valore di pendenza dell’elettrodo diminuisce rapidamente, il tempo di reazione aumenta o le 

misurazioni divengono imprecise, le seguenti procedure possono essere di aiuto. Esse devono essere 

eseguite nell’ordine indicato.

1. Sgrassare la membrana con lana di cotone impregnata con soluzione di acetone/sapone. 

2. Tenere la punta dell’elettrodo immersa in HCl 0,1 M HCl per una notte.

3. In caso di formazione di proteine, rimuovere i depositi immergendo l’elettrodo in una soluzione di 

HCl /pepsina (Cod. Ord.: 51340068).

4. In caso di contaminazioni di solfuro d’argento, immergere l’elettrodo in una soluzione di tiourea 

(Cod. Ord.: 51340070).

Nota 

Maneggiare le soluzioni per la pulizia ed il riempimento con la medesima cura prestata per le sostanze 

tossiche o corrosive.

10.3 Smaltimento

In conformità a quanto stabilito dalla Direttiva Europea 2002/96 CE in materia di apparecchi 

elettrici ed elettronici (RAEE), questo strumento non può essere smaltito come i normali rifiuti. 

Tale presupposto resta valido anche per i Paesi al di fuori dei confini della UE, conformemente 

alle norme nazionali in vigore.

Si prega quindi di smaltire questo prodotto separatamente e in modo specifico secondo le 

disposizioni locali relative alle apparecchiature elettriche ed elettroniche.

Per qualsiasi chiarimento, rivolgersi agli enti preposti o al rivenditore dell‘apparecchiatura stessa.

In caso di cessione dello strumento (per es. per ulteriore utilizzo privato o aziendale/industriale), si 

prega di comunicare anche questa disposizione.

Si ringrazia per il contributo alla tutela dell‘ambiente.

Manutenzione 
14 METTLER TOLEDO SevenEasy

Accessori

11. Accessori 

Cod. Ord.

Braccio elettrodo completo 51302820

Guide to pH measurement (English) 51300047

Guide to conductivity and dissolved oxygen (English) 51724716

Guide to ion selective measurement (English) 51300075

Sacchetti tampone pH 4,01 30 x 20mL 51302069

Flaconi tampone pH 4,01, 6 x 250mL 51340058

Sacchetti tampone pH 7,00, 30 x 20mL 51302047

Flaconi tampone pH 7,00, 6 x 250mL 51340060

Sacchetti tampone pH 9,21, 30 x 20mL 51302070

Flaconi tampone pH 9,21, 6 x 250mL 51300194

Sacchetti tampone pH 10,01, 30 x 20mL 51302079

Flaconi tampone pH 10,01, 6 x 250mL 51340231

InLab®410, Elettrodo pH, sonda temperatura, rabboccabile 52000118

InLa®b412, Elettrodo pH da laboratorio, bicchiere, rabboccabile 52000112

InLab®413, Elettrodo pH robusto, corpo in PEEK, ATC 52000100

InLab®420, Elettrodo pH con diaframma in PTFE 52000113

Soluzione HCl/pepsina (rimuove le contaminazioni dovute a proteine) 51340068

Soluzione di tiourea (rimuove le contaminzioni di solfuro d’argento) 51340070

Soluzione di rigenerazione per elettrodi pH 51340073
METTLER TOLEDO SevenEasy 15

Specifiche tecniche 

12. Specifiche tecniche 

pH mV Temperatura

Intervallo di misura 0,00 - 14,00 ± 1999 –5 ... 105 °C

Risoluzione 0,01 1 0,1 °C

Precisione relativa ± 0,01 ± 1 ± 0,5 °C

Calibrazione pH Punti 1, 2, o 3

Punto isopotenziale 7,00 pH

Tamponi di calibrazione B1: (25 °C) 7,00 4,00 10,01 1,68

B2: (25 °C) 7,00 4,01 9,21 2,00 11,00

B3: (20 °C) 7,00 4,00 9,00 2,00 12,00

B4: (25 °C) 6,86 4,01 9,18 1,68

B5: definito dall’utente

Compensazione temperatura da –5 ... 105 °C in automatico o manuale

Indicatore Indicatore a cristalli liquidi

Uscite RS232 seriale, 

Velocità di trasmissione: 1200

Bit di dati: 8

Bit di stop: 1

Parità: nessuna

Ingresso pH Impedenza > 1012 Ohm

Condizioni ambientali Temperatura ambiente: 5 °C ... 40 °C

Umidità relativa: 5 % ... 80 %, (senza condensa)

Categoria di sovratensione: II

Grado di inquinamento: 2

Dimensioni/Peso 180 x 180 x 65 (mm) / 0,61 kg

Materiali Chasis: ABS, rinforzato PC

Portaelettrodo: ABS, rinforzato PC 

Membrana tastiera: Poliestere

Alimentazione eettrica Il misuratore di pH SevenEasy è dotato di un alimentatore appropriato per la rete di alimentazione del luogo di installazione:

- Stati Uniti: 120 V / 60 Hz, 10 VA, 9 V DC

- Europa: 230 V / 50 Hz, 10 VA, 9 V DC

- Regno Unito: 240 V / 50 Hz, 10 VA, 9 V DC

- Giappone: 110 V / 50 Hz, 10 VA, 9 V DC

- Australia: 240 V / 50 Hz, 10 VA, 9 V DC

- Cina: 220 V / 50 Hz, 10 VA, 9 V DC

- Batterie (facolt.): 4 x AA (LR6)

Nota 

Il misuratore di pH SevenEasy pH deve essere utilizzato soltanto con l’alimentatore in dotazione o con 

batterie.
16 METTLER TOLEDO SevenEasy

13. Guida Rapida 1. Preparazione dell’elettrodo

Rimuovere il cappuccio

2. Calibrazione a 1 punto

pH tampone 7

Agitare, poi 

lasciare 

Punto finale in automatico o

3. Lavaggio dell’elettrodo 4. Calibrazione a 2 o 3 Punti

5. Misurazione campione

Campione

Agitare, poi 

lasciare

Punto finale in 

automatico o

6. Lavaggio dell’elettrodo 7. Conservazione dell’elettrodo

Elettrolita

Guida Rapida

Per una calibrazione a 2 o 3 punti, ripetere il punto 

2, quindi effettuare il lavaggio
METTLER TOLEDO SevenEasy 17

14. Gruppi di tamponi

Il misuratore di pH SevenEasy compensa automaticamente la temperatura impostata sulla base dei 

valori indicati nelle tabelle:

Gruppo di tamponi 1 (rif. 25 °C) MT USA

 5 7,09 4,00 10,25 1,67

 10 7,06 4,00 10,18 1,67

 15 7,04 4,00 10,12 1,67

 20 7,02 4,00 10,06 1,68

25 7,00 4,00 10,01 1,68

 30 6,99 4,01 9,97 1,68

 35 6,98 4,02 9,93 1,69

 40 6,97 4,03 9,89 1,69

 45 6,97 4,04 9,86 1,70

 50 6,97 4,06 9,83 1,71

Gruppo di tamponi 2 (rif. 25 °C) MT Europa

 5 7,09 4,01 9,45 2,02 11,72

 10 7,06 4,00 9,38 2,01 11,54

 15 7,04 4,00 9,32 2,00 11,36

 20 7,02 4,00 9,26 2,00 11,18

 25 7,00 4,01 9,21 2,00 11,00

 30 6,99 4,01 9,16 1,99 10,82

 35 6,98 4,02 9,11 1,99 10,64

 40 6,97 4,03 9,06 1,98 10,46

 45 6,97 4,04 9,03 1,98 10,28

 50 6,97 4,06 8,99 1,98 10,10

Gruppo di tamponi 3 (rif. 20 °C) Tamponi Merck

 5 7,07 4,04 9,16 2,01 12,41

 10 7,05 4,02 9,11 2,01 12,26

 15 7,02 4,01 9,05 2,00 12,10

20 7,00 4,00 9,00 2,00 12,00

 25 6,98 4,01 8,95 2,00 11,88

 30 6,98 4,01 8,91 2,00 11,72

 35 6,96 4,01 8,88 2,00 11,67

 40 6,95 4,01 8,85 2,00 11,54

 45 6,95 4,01 8,82 2,00 11,44

 50 6,95 4,00 8,79 2,00 11,33

Gruppo di tamponi 4 (ref. 25 °C) Giappone JIS

 5 6,95 4,00 9,40 1,67 

 10 6,92 4,00 9,33 1,67 

 15 6,90 4,00 9,28 1,67 

 20 6,88 4,00 9,22 1,68 

 25 6,86 4,01 9,18 1,68 

 30 6,85 4,02 9,14 1,68 

 35 6,84 4,02 9,10 1,69

 40 6,84 4,04 9,07 1,69

 45 6,83 4,05 9,04 1,70

 50 6,83 4,06 9,01 1,71 

Gruppi di tamponi
18 METTLER TOLEDO SevenEasy





© Mettler-Toledo AG 2003, 2004, 2006 ME-51710234C Printed in China 0607/2.45

Mettler-Toledo AG, Analytical, Sonnenbergstrasse 74, CH-8603 Schwerzenbach, 

Tel. +41 (0)44 806 77 11, Fax +41 (0)44 806 73 50, Internet: http://www.mt.com

Subject to technical changes and to the availability

of the accessories supplied with the instruments.

*P51710234*

Quality certificate. Development, production and testing according to ISO9001. 

Environmental management system according to ISO14001.

Worldwide service. Our extensive service network is among the best in the world and 

ensures maximum availability and service life of your product.

European conformity. The CE conformity mark provides you with the assurance that 

our products comply with the most recent EU directives.

On the Internet. You will quickly find lots of essential information about our products, 

our services, and our company at 

http://www.mt.com