Unique Cool-ZoneTM on 

RapidVap N2

Insulates sample remaining in 

glassware stem to permit 

desired end point. 

Audible/visual alarms

Signal completion of run 

allowing samples to be left 

unattended while evaporating 

to desired end point. 

Dual lid clamps

Two clamps hold the lid 

securely in place.

Glass lid

Chemical-resistant and 

detaches for cleaning.

Microprocessor-controlled 

motor

Maintenance-free, brushless 

DC motor ensures safety and 

reproducibility for protocols.

Full one year warranty

Microprocessor-controlled 

nitrogen blow down manifold

Nitrogen helps facilitate phase 

change from liquid to gas.

PTFE*-coated chamber

Aluminum with PTFE for a 

broad range of chemical 

compatibility.

1000-watt dry block 

heating system

Speeds evaporation rate by 

supplying a controlled amount 

of heat from ambient up to a 

maximum of 100° C to the 

sample block. Heater and 

block move in tandem for more 

efficient heat transfer. Unlike 

water baths, the dry heating 

system adds no potential 

source of contamination and 

requires no maintenance.

Faster evaporation rates, 

greater throughput

Vortex action, heat and 

nitrogen blow down combine 

to accelerate evaporation.

Phenol-free gasket

Provides complete sealing 

under vacuum.

PTFE*-coated sample block

Provides chemical resistance 

(included with N2

 models).

Microprocessor-controlled 

vortex motion increases 

surface area for faster 

evaporation

Vortex action continually mixes 

sample, helps contain analytes 

in the solvent by constantly 

rinsing tube walls and 

maximizes sample recovery.

Program

LCD shows program from 1 to 9, R for run or S for stop.

Speed

LCD shows vortex speed from 0 to 100%.

Heat

LCD shows heat from OFF to 99 (° C) or HI (100° C).

Time

LCD shows time selected from 1 to 999 minutes.

Sample

LCD shows number of sample or clusters of samples 

selected to receive nitrogen blow down; 2, 4, 6 or 8.

*Polytetrafluoroethylene

RapidVap® N2 and N2/48 Evaporators

Features & Benefits

12 | labconco.com
All models require (not included):

• Sample Block. See page 21

• Glassware. See pages 18 and 21

• Nitrogen source with flow rate of 3.5 CFM/100 LPM minimum. See page 22 for Nitro 2LV Nitrogen Generator or contact your nitrogen 

gas supplier.

• Gas pressure regulator. Pressure of 5-10 psi is typical. Pressure must not exceed 20 psi. Contact a local supplier.

*International electrical configuration.

• Moderate sample 

throughput

• End-point determination

• Introduce nitrogen to 

tubes in sets of two

• Eight tube capacity

Catalog 

Number

Built-In Option Electrical Requirements Receptacle & Plug Type Overall Dimensions 

(w x d x h)

Shipping 

Weight

7910000 115V, 50/60 Hz, 9A North America, 115V, 20A 21.7" x 19.0" x 13.5" 107 lbs. (49 kg)

7910010 RS-232 Link 115V, 50/60 Hz, 9A North America, 115V, 20A 21.7" x 19.0" x 13.5" 107 lbs. (49 kg)

7910001* 230V, 50/60 Hz, 4.5A Schuko, 230V, 15A 21.7" x 19.0" x 13.5" 107 lbs. (49 kg)

7910011* RS-232 Link 230V, 50/60 Hz, 4.5A Schuko, 230V, 15A 21.7" x 19.0" x 13.5" 107 lbs. (49 kg)

RapidVap N2 Evaporators

• Small to moderate 

sample throughput

• Ideal for volatile 

samples

• Introduce nitrogen in 

clusters of six

• 48 tube capacity

RapidVap N2

 Evaporator RapidVap N2/48 Evaporator

Catalog 

Number

Built-In Option Electrical Requirements Receptacle & Plug Type Overall Dimensions 

(w x d x h)

Shipping 

Weight

7910012 115V, 50/60 Hz, 9A North America, 115V, 20A 21.7" x 19.0" x 13.5" 108 lbs. (49 kg)

7910014 RS-232 Link 115V, 50/60 Hz, 9A North America, 115V, 20A 21.7" x 19.0" x 13.5" 108 lbs. (49 kg)

7910013* 230V, 50/60 Hz, 4.5A Schuko, 230V, 15A 21.7" x 19.0" x 13.5" 108 lbs. (49 kg)

7910015* RS-232 Link 230V, 50/60 Hz, 4.5A Schuko, 230V, 15A 21.7" x 19.0" x 13.5" 108 lbs. (49 kg)

RapidVap N2/48 Evaporators

All models require (not included):

• Glassware. See pages 21-22

• Nitrogen source with flow rate of 0.6 CFM/17 LPM minimum. See page 22 for Nitro 2LV Nitrogen Generator or contact your nitrogen 

gas supplier.

• Gas pressure regulator. Pressure of 5-10 psi is typical. Pressure must not exceed 20 psi. Contact a local supplier.

RapidVap® N2 and N2/48 Evaporators

Ordering Information

labconco.com | 13