PureLab HE Glovebox

User Manual
ii of 82 Innovative Technology, Inc.

Copyright

© Innovative Technology, Inc. 1994 - 2006

No part of this manual may be reproduced in any form, electrical or mechanical, including photocopy, recording, or 

any information storage and retrieval system, without permission in writing from Innovative Technology, Inc.

Company Information

Innovative Technology, Inc.

One Industrial Way

Amesbury, MA 01913

USA

Phone: 978-462-4415

Fax: 978-462-3338

www.Gloveboxes.com

www.solventpurification.com

www.isolatorgloves.com

Document Information

Version Revision Date

1.00 06/29/2006

2.00 <mm/dd/yyyy>
Innovative Technology, Inc. iii of 82

Contents

Figures .............................................................................................................................................. vi

Tables............................................................................................................................................... vii

1 About This Manual....................................................................................................................... 9

1.1 Firmware Revisions............................................................................................................. 9

1.2 Optional Equipment............................................................................................................. 9

1.3 Terms .................................................................................................................................. 9

1.4 Symbols Used ..................................................................................................................... 9

2 System Overview ....................................................................................................................... 10

2.1 The PLC Enclosure ........................................................................................................... 12

2.2 Exterior Controls................................................................................................................ 12

2.3 Circuit Breakers................................................................................................................. 12

2.4 Foot Pedal ......................................................................................................................... 12

3 Installation.................................................................................................................................. 14

3.1 Pre-Installation Requirements........................................................................................... 14

3.1.1 Working Gas.............................................................................................................. 14

3.1.2 Regeneration Gas ..................................................................................................... 14

3.1.3 Electrical .................................................................................................................... 14

3.1.4 System Exhaust......................................................................................................... 14

3.2 Unpacking the System ...................................................................................................... 15

3.3 Assembly........................................................................................................................... 15

3.4 Electrical Connections....................................................................................................... 17

3.5 Gas Connections............................................................................................................... 17

3.6 Venting Connections ......................................................................................................... 17

3.7 Commissioning the System............................................................................................... 17

3.7.1 Switching Power On .................................................................................................. 18

3.7.2 Pressure Testing the System .................................................................................... 18

3.7.3 Adjusting the Box Pressures ..................................................................................... 19

3.7.4 Purging ...................................................................................................................... 21

3.7.5 Circulation.................................................................................................................. 22

4 Control Panel Functions ........................................................................................................... 24

4.1 The Main Control Screen .................................................................................................. 24

4.2 Initial Power On ................................................................................................................. 25

4.3 Control Sub-Screens......................................................................................................... 27

4.3.1 Regen Control ........................................................................................................... 27

4.3.2 Regen Timers ............................................................................................................ 28

4.3.3 Solvent Regen........................................................................................................... 29

4.3.4 Solvent Regen Timers............................................................................................... 30

4.3.5 Blower Control ........................................................................................................... 30

4.3.6 Purge Control ............................................................................................................ 31

4.3.7 Automatic Antechamber Control ............................................................................... 32

4.3.8 Antechamber Config.................................................................................................. 33

4.3.9 System Config ........................................................................................................... 34

4.3.10 Alarms ....................................................................................................................... 35

4.3.11 Freezer Control.......................................................................................................... 36

4.3.12 Box Cooling Control................................................................................................... 37

4.3.13 Heater Control ........................................................................................................... 37

5 System Operation...................................................................................................................... 39

5.1 Adjusting the System Pressure ......................................................................................... 39
Contents

iv of 82 Pure_Lab_2012-09-18.docx

5.2 Using the Antechambers................................................................................................... 40

5.2.1 Manual Antechamber Control ................................................................................... 40

5.3 Removing an Item from the Glovebox Using the Large Antechamber............................. 41

5.4 Mini Antechamber............................................................................................................. 41

5.4.1 Introducing an Item into the Glovebox Using the Mini Antechamber........................ 41

5.4.2 Removing an Item from the Glovebox Using the Mini Antechamber........................ 42

5.5 Automatic Antechamber Control....................................................................................... 42

5.6 Introducing an Item into the Glovebox Using the Large Antechamber............................. 43

5.7 Removing an Item from the Glovebox Using the Large Antechamber............................. 45

5.8 Purge Valves..................................................................................................................... 45

5.8.1 Automatic Purge Valve.............................................................................................. 46

5.9 Regeneration .................................................................................................................... 47

5.9.1 Single-Column Purification System Regeneration Procedure .................................. 48

5.10 Dual-Column Purification System Regeneration Procedure............................................. 49

5.11 Solvent Regeneration ....................................................................................................... 49

5.11.1 Solvent-Column System Regeneration Procedure ................................................... 49

5.12 Analyzers .......................................................................................................................... 51

5.12.1 O2 Analyzer .............................................................................................................. 51

5.13 Moisture Analyzer ............................................................................................................. 52

5.13.1 Box Cooling Control .................................................................................................. 53

5.14 Heater Control................................................................................................................... 54

5.14.1 Parallel – Piped Systems .......................................................................................... 55

5.15 Solvent Removal Systems................................................................................................ 57

5.15.1 Large Capacity Solvent Removal.............................................................................. 57

5.15.2 Opening the Solvent Removal Column..................................................................... 57

5.15.3 Isolating the Solvent Removal Column and Removing Solvent Vapors ................... 58

5.15.4 Replacing the Carbon Charge in a Solvent Removal Column.................................. 59

5.16 Large Capacity Regenerable Solvent Removal................................................................ 60

5.16.1 Small Capacity Solvent Removal – External............................................................. 61

6 Routine Maintenance ................................................................................................................ 63

6.1 Daily Tasks........................................................................................................................ 63

6.2 Weekly Tasks.................................................................................................................... 63

6.3 Annual or Semi Annual Tasks........................................................................................... 63

6.4 Safety Inspections............................................................................................................. 63

7 Troubleshooting ........................................................................................................................ 64

8 System Control Electronics ..................................................................................................... 66

8.1 General Description .......................................................................................................... 66

8.1.1 PLC Controller........................................................................................................... 66

8.1.2 Display panel............................................................................................................. 66

8.2 Lower Control Box ............................................................................................................ 66

8.2.1 Power Supply ............................................................................................................ 66

8.2.2 Box Pressure Sensor ................................................................................................ 66

8.2.3 24 Volt Low Power Relays ........................................................................................ 66

8.2.4 Circuit Breakers......................................................................................................... 68

8.3 Specifications.................................................................................................................... 68

9 System Components................................................................................................................. 69

9.1 Box Flow ........................................................................................................................... 69

9.1.1 Piping and Valves ..................................................................................................... 69

9.1.2 Column...................................................................................................................... 69

9.1.3 The Blower................................................................................................................ 70

9.1.4 Vacuum Pump........................................................................................................... 70

9.1.5 Chambers.................................................................................................................. 70
Contents

Pure_Lab_2012-09-18.docx v of 82

10 Removal and Replacement Procedures.................................................................................. 71

10.1 Blower................................................................................................................................ 71

10.2 Column .............................................................................................................................. 71

10.3 Filters................................................................................................................................. 72

10.4 Gloves ............................................................................................................................... 72

10.4.1 Replacement with an Internal Glove Port Cover....................................................... 72

10.4.2 Replacement without an Internal Glove Port Cover.................................................. 73

10.4.3 Control Box Panel...................................................................................................... 73

10.4.4 Circuit Breakers ......................................................................................................... 73

10.4.5 Power Supply ............................................................................................................ 73

10.4.6 Solid State Relays ..................................................................................................... 74

10.4.7 Valves........................................................................................................................ 74

10.4.8 Vacuum Pump........................................................................................................... 74

10.5 Window.............................................................................................................................. 75

10.6 System Diagrams .............................................................................................................. 76

10.6.1 Front View ................................................................................................................. 76

10.6.2 Back View.................................................................................................................. 77

10.6.3 Pure Lab HE 2GB and Pure Lab HE 4GB-2500 Overall Dimensions ....................... 78

10.6.4 Pure Lab HE 3GB Overall Dimensions ..................................................................... 80

10.6.5 Pure Lab HE 4GB-1800 Overall Dimensions ............................................................ 81

10.6.6 Pure Lab HE 4GB-1950 Overall Dimensions ............................................................ 82
Contents

vi of 82 Innovative Technology, Inc.

Figures

Figure 1 PureLab HE ............................................................................................................................. 11

Figure 2 Assembled Glovebox............................................................................................................... 15

Figure 3 Glovebox Control screen ......................................................................................................... 18

Figure 4 Login/Logout button ................................................................................................................. 19

Figure 5 Main Glovebox Control screen ................................................................................................ 19

Figure 6 Glovebox Config Screen.......................................................................................................... 20

Figure 7 Numerical input screen ............................................................................................................ 20

Figure 8 Oxygen analyzer valves - closed............................................................................................. 22

Figure 9 Oxygen analyzer valves - Flow In............................................................................................ 22

Figure 10 Glove Box Control System..................................................................................................... 23

Figure 11 Bower Control screen ............................................................................................................ 23

Figure 12 Glove Box Control System...................................................................................................... 24

Figure 13 Glovebox powered on............................................................................................................ 25

Figure 14 Login/Logout screen .............................................................................................................. 25

Figure 15 Glove Box Control System with fully functional buttons ........................................................ 26

Figure 16 Regeneration Control screen................................................................................................. 28

Figure 17 Regen Timers screen ............................................................................................................ 29

Figure 18 Solvent Regeneration Control screen.................................................................................... 29

Figure 19 Solvent Regen screen ........................................................................................................... 30

Figure 20 Blower Control screen ........................................................................................................... 31

Figure 21 Purge Control screen............................................................................................................. 32

Figure 22 Antechamber Control screen................................................................................................. 33

Figure 23 Antechamber Config screen .................................................................................................. 33

Figure 24 Glove Box Config screen ....................................................................................................... 34

Figure 25 Alarms screen........................................................................................................................ 35

Figure 26 Trend screen.......................................................................................................................... 36

Figure 27 Freezer Control screen .......................................................................................................... 36

Figure 28 Box Cooling Control screen................................................................................................... 37

Figure 29 Heater Control screen............................................................................................................ 38

Figure 30 Glove Box Config screen ....................................................................................................... 39

Figure 31 Numerical Input screen.......................................................................................................... 39

Figure 32 Large and Mini Antechambers............................................................................................... 40

Figure 33 Inside Antechambers ............................................................................................................. 41

Figure 34 Antechamber Control screen................................................................................................. 42

Figure 35 Antechamber Config screen .................................................................................................. 43

Figure 36 Inserting an object into the Antechamber.............................................................................. 44

Figure 37 Removing object from Antechamber ..................................................................................... 44

Figure 38 Hand-operated 2-Way Ball Valve .......................................................................................... 45

Figure 39 Automatic purge valve ........................................................................................................... 46

Figure 40 Purge Control screen............................................................................................................. 46

Figure 41 Regeneration Control screen................................................................................................. 47

Figure 42 Regen Timers screen ............................................................................................................ 48

Figure 43 Glove Box Config screen ....................................................................................................... 49

Figure 44 Gas purification module ......................................................................................................... 50

Figure 45 Solvent Regeneration Control screen.................................................................................... 50

Figure 46 Oxygen analyzer valves – “Closed” position ......................................................................... 51

Figure 47 Oxygen analyzer valves – “Flow Out” position ...................................................................... 51

Figure 48 Moisture Analyzer.................................................................................................................. 52

Figure 49 Freezer Control screen .......................................................................................................... 53

Figure 50 Box Cooling Control............................................................................................................... 53

Figure 51 Box Cooking Control screen.................................................................................................. 54
Contents

Innovative Technology, Inc. vii of 82

Figure 52 Heater Control screen............................................................................................................ 55

Figure 53 Blower Control screen............................................................................................................ 56

Figure 54 Purge Control screen ............................................................................................................. 57

Figure 55 Solvent Removal Column - Open .......................................................................................... 58

Figure 56 Solvent Removal Column – Closed ....................................................................................... 59

Figure 57 Fill and Empty Ports ............................................................................................................... 60

Figure 58 Evac/Refill Valve at Evacuate ................................................................................................ 60

Figure 59 Small Capacity Solvent Removal – External.......................................................................... 61

Figure 60 Small Capacity Solvent Removal – Internal........................................................................... 62

Figure 61 General Box Flow................................................................................................................... 69

Figure 62 Gloves .................................................................................................................................... 72

Figure 63 Front view............................................................................................................................... 76

Figure 64 Back view ............................................................................................................................... 77

Figure 65 Pure Lab HE 2GB .................................................................................................................. 78

Figure 66 Pure Lab HE 4GB-2500 ......................................................................................................... 79

Figure 67 Pure Lab HE 3GB .................................................................................................................. 80

Figure 68 Pure Lab HE 4GB-1800 ......................................................................................................... 81

Figure 69 Pure Lab HE 4GB-1950 ......................................................................................................... 82

Tables

Table 1 Circuit Breakers......................................................................................................................... 12

Table 2 Relay Functions and Connections............................................................................................. 67

Table 3 Solid State Relays ..................................................................................................................... 68

Table 4 Circuit Breakers......................................................................................................................... 68

Table 5 Specifications ............................................................................................................................ 68

Pure_Lab_2012-09-18.docx 9 of 82

1 About This Manual

This document explains how to install and operate your Glovebox system and accessories. 

References are made in the Installation section to features in the Operational section. It is 

imperative that you refer to them as indicated. If you have any further questions please call our 

offices at 978-462-4415

1.1 Firmware Revisions

This User Manual is based on a minimum PLC firmware Version 7 Revision 1. Enhancements 

made to this version may result in operator display screens or other operational functions that 

vary slightly from this manual’s figures and descriptions.

1.2 Optional Equipment

This User Manual includes information about optional equipment that may not be installed on 

your system. It is presented for clarity and completeness. We have indicated which equipment is 

optional and which equipment is standard. In addition, a system that has been ordered with 

custom options may vary from what is documented in this User Manual.

There may be slight differences due to manufacturing changes.

This User Manual includes information on Dual Column systems, Automatic Antechambers, 

and Automatic Purge systems, all of which are optional equipment.

1.3 Terms

The terms “box”, “Glovebox”, and “system” all refer to the entire standard PureLab HE 

configuration. The terms “regeneration,” “regeneration cycle,” and “regen” all refer to the 

components or process of regenerating the system’s purifier or column.

1.4 Symbols Used

The following are symbols used in this manual and on the Glovebox itself:

Circuit breaker

Electrical Ground

Electrical AC current
System Overview

10 of 82 Pure_Lab_2012-09-18.docx

2 System Overview

PureLab HE is designed as a complete package including box, Antechamber(s), gas purification 

system and vacuum pump. The system is capable of removing O2 and H2O to levels less than 1 

ppm. O2 and H2O are removed from the inert atmosphere by a combination of molecular sieve 

and copper catalyst. Once saturated, these materials are easily regenerated to their original state.

The PureLab HE has been designed with the user in mind.

• The Operator Interface comprises a color HMI (Human Machine Interface) touch screen that 

is mounted on a swing arm for ease of use.

• The foot pedal allows the operator hands free control of the box pressure.

• Interior and exterior doors are easy to operate.

• The controls and indicators are simple yet fully functional.

The PureLab HE (Figure 1) is designed to be a completely modular system. The gas purification 

system is contained within a separate module that is located beneath the large Antechamber. 

This module has casters allowing easy movement for servicing.

All service connections are located on the rear of the gas purification system and are clearly 

labeled.
System Overview

Pure_Lab_2012-09-18.docx 11 of 82

Figure 1 PureLab HE

The vacuum pump is located on the Glovebox stand, allowing for easy access and oil change.

The Gas Purification system module has removable side panels and a top panel, allowing easy 

access to the Valve Assembly, Purifier Column, Valves, and Blower.

In addition you will find:

• Visual indicators on the Automatic Column valves.

• Manual override should the Automatic Column Pilot Solenoids fail.

• Circuit breakers.

• Solid State Relays.
System Overview

12 of 82 Pure_Lab_2012-09-18.docx

• Easy access for replacement of valves.

• The Antechamber has a door lifting mechanism so the door lifts gently and cannot close 

unless pulled down by the user.

• Gloveboxes are modular in design; there are no bolts through the end panel or front window 

that could cause potential leaks.

• All connections are stainless steel tubing, which is either welded or connected via KF-40 

clamps or compression connectors. There are no rubber hoses or hose clamps which, over 

time, can develop leaks or cracks.

• The PureLab HE control system is PLC-based and has a color HMI touch screen interface. 

This system monitors and controls all box functions. Extensive functionality and user 

features are available in the System Controller.

2.1 The PLC Enclosure

The PLC enclosure is located on the front of the gas purification module.

2.2 Exterior Controls

On the PLC enclosure, mounted on the side wall, you will find several circuit breakers and an 

illuminated main circuit breaker that doubles as the main power on switch.

2.3 Circuit Breakers

There are seven (7) circuit breakers on the side wall. The function of each breaker is indicated 

on a label at the side of the breaker. When they are in the tripped position, a white colored tab 

extends from the middle of the breaker. To reset them, press the tab back in until it locks in 

place. A label on the inside of the enclosure door contains specific information regarding the 

circuit breakers.

Table 1 Circuit Breakers

Circuit Breaker # Amperage

110/220 VAC

Function

1 15A/10A Illuminated Main Power Switch

2 10A/5A Vacuum Pump

3 6A/3A 24V Power Supply

4 5A/3A Blower

5 4A/2A Heater for Column 1

6 4A/2A Heater for Column 2

7 5A/3A Lighthood

2.4 Foot Pedal

A foot pedal is provided so that the box pressure can be increased or decreased without the 

operator having to remove their arms from the gloves. It is a dual pedal switch. The right pedal 
System Overview

Pure_Lab_2012-09-18.docx 13 of 82

increases the box pressure and the left pedal decreases it. Note that the working minimum and 

maximum pressure settings are still being maintained by the PLC control. The foot pedal allows 

the user to add or remove pressure within the working minimum and maximum set points.
Installation

14 of 82 Pure_Lab_2012-09-18.docx

3 Installation

This section explains how to install a PureLab HE Glovebox

3.1 Pre-Installation Requirements

3.1.1 Working Gas

The working gas requirement is Nitrogen, Argon or Helium with a suitable regulator to adjust 

pressure to 55-60 psi. The working gas must be connected to the gas inlet on the Glovebox

either by flexible hose or hard copper plumbing to a 3/8" compression connector. Four (4) 

cylinders are required: three (3) for purging and one (1) for the final operation of the Glovebox.

Note 1: Refer to system specific installation requirements for further details. This document was 

supplied prior to shipping the system. Please contact your local Innovative Technology 

representative if you need another copy.

Note 2: The working gas tubing must be 3/8” outside diameter for the entire distance from the 

regulator to the working gas connection. Any reductions in tubing diameter will result in 

insufficient gas flow for optimum system performance. Outside North America the connections 

and tubing should be 10mm

3.1.2 Regeneration Gas

Note: The gas purification system has been regenerated and performance tested to < 1ppm of 

O2 ands H2O at our factory prior to shipment. It does NOT need to be regenerated prior to first 

use upon installation.

One cylinder of Nitrogen or Argon containing approximately 3 to 7% Hydrogen.

3.1.3 Electrical

All Innovative Technology, Inc. Glovebox systems are configured to operate using the electrical 

power standard for the region in which the Glovebox has been ordered.

Note: The standard 2-glove system requires 2 electrical outlets, one for the main system power 

and one for the power supplied to the power strip inside the Glovebox. Larger systems and those 

configured with other options require additional outlets.

3.1.4 System Exhaust

Note: The Regeneration Gas, Vacuum Pump, and Purge exhausts should be handled in 

accordance with your facility’s required specifications.

The regeneration gas exhaust is 3/8" I.D., 19/32" O.D. Tygon tubing. The Vacuum Pump 

exhausts are 1" OD tubing. The purge valve should also be vented, 1" Female NPT (manual 

purge) or KF-40mm flange (automatic purge).

WARNING: Proper ventilation of the exhausted gases is imperative as high levels of inert gas 

can cause asphyxiation.
Installation

Pure_Lab_2012-09-18.docx 15 of 82

3.2 Unpacking the System

All systems shipped outside North America are packaged in specially designed shipping crates. 

These systems require a forklift or similar lifting device to safely remove the Glovebox and Gas 

Purification module from the crate. For more information, please ask your Innovative 

Technology, Inc. representative.

WARNING: Glovebox systems are typically top-heavy.

All systems shipped within North America are shipped on a special-designed pallet that includes 

detachable ramps to enable the Glovebox and Gas Purification module to be rolled off the pallet. 

Please ensure that you have sufficient personnel available to assist in the unloading process to 

ensure personnel safety.

To unpack the Glovebox system

1 Place the Glovebox and gas purification module on a level floor in the desired location.

2 Remove the shrink-wrap and any other strapping that may be supporting the vacuum pump 

or any other accessories.

3 Depending on the exact system configuration purchased, there will be one or more boxes 

containing items such as gloves, power strips, etc. inside the large Antechamber and on the 

pallet. Carefully inspect all packaging materials before discarding to ensure that any small 

components are not misplaced.

3.3 Assembly

Figure 2 Assembled Glovebox
Installation

16 of 82 Pure_Lab_2012-09-18.docx

Note: For double box systems, the two boxes need to be joined prior to performing the 

following procedure. See the next procedure, “To assemble a double box system”.

To install the Glovebox system

1 See Glovebox layout drawing (Figure 1).

The Glovebox system has lockable casters on each leg to facilitate its placement. Once in 

place, the casters can be adjusted to lock the system in place.

2 Level the system by adjusting the leveling feet underneath each leg.

3 Place the Gas Purification module underneath the Large Antechamber.

4 Connect the Gas Purification module to the Glovebox. Match up the labels on the piping and 

components to the labels on the Glovebox and the gas purification module. e.g., “1” connects 

to “1”, “2” connects to “2”, “A” connects to “A”, etc.

5 Lock the Purifier Cart into place by adjusting the casters.

6 Locate and install the shelves inside the box at your preferred height. The shelves are 

adjustable and can be relocated at any time. Fit the shelves into the box by sliding them 

through the Antechamber. It is not necessary to remove the window.

7 Place the power strip in the box as desired. Connect the male plug to the female outlet, which 

leads to the feed-through on the side of the box.

8 The oxygen analyzer requires that the small sampling pump be connected to the black tubing 

connected to the bulkhead fitting on the inside of the Antechamber wall. The batching pump 

should also be plugged into the electrical power strip inside the Glovebox.

9 Attach the purge valve to the corresponding KF-40 flange located on the top of the 

Glovebox. The purge valve and flange are labeled like the removed plumbing. For the autopurge version, locate and connect the electrical and pneumatic lines.

10 Install the gloves. Gloves are placed inside the box through each glove port. The cuff end of 

the glove is stretched over the glove port until the lip fits into the last groove. Two O-rings 

are then stretched over each glove and placed in the outer two grooves.

To assemble a double box system

1 Remove the window from one box. See “Removal and Replacement Procedures” on page 

71.

2 Press both boxes together and make them level on the inside by adjusting the leveling feet. 

After the boxes are level, lock the leveling feet into position.

3 Press both boxes apart far enough so that you can fit between them. Apply a bead of DAP 

latex sealant, approximately 1/8"-3/16" wide continuously around the flange of one box in 

the area between the inside of the box and the bolt holes.

4 Press the boxes together and put in the four corner bolts. After the four are in and hand tight, 

insert the rest of the hardware and tighten all bolts. Excess sealant will squeeze out from 

between the flanges.

5 Clean the excess sealant; it is easier to do this when the sealant is slightly dry. Clean the 

entire interior of the Glovebox.
Installation

Pure_Lab_2012-09-18.docx 17 of 82

6 Place the window on the box. See “Removal and Replacement Procedures” on page 71.

7 Install the gloves and glove O-rings. See “Removal and Replacement Procedures” on page 

71.

3.4 Electrical Connections

The primary power cord has been retracted into the gas purifier module to prevent damage 

during shipment. Simply loosen the strain relief, pull the cord through the strain relief, and plug 

it into the proper wall receptacle.

The plug that is connected to the power strip is also inserted into the proper wall receptacle.

Note: Glove boxes are shipped with many different options. Please review each section in this 

User Manual that applies to the configuration of the system being installed.

3.5 Gas Connections

IMPORTANT: Male sure you have installed the proper regulator on the working gas cylinder 

and on the regeneration gas cylinder, connected tubing of sufficient length, and ensured

that there are no leaks.

The fittings for both gases are located on the rear wall of the gas purifier module. Connect the 

working gas to the appropriately-labeled compression fitting and tighten it 1.25 turns from 

finger tight. Over tightening compression fittings can cause problems.

Note: If you are not familiar with compression fittings please call your Innovative Technology 

representative for further assistance.

Set the Working Gas pressure to 55 psi. Set the Regen Gas pressure to 7 psi. These connections 

are 3/8” in North America and 10mm outside North America.

3.6 Venting Connections

All exhaust connections should be made in accordance with your internal HVAC guidelines.

The fitting on the vacuum pump exhaust is 1” stainless steel tube stub.

The exhaust from the purifier is 3/8" ID hose.

The exhaust from the purge valve is 1" Female NPT.

WARNING: Proper ventilation of the exhausted gases is essential as high levels of inert gas can 

cause asphyxiation.

3.7 Commissioning the System

You are now ready to begin the commissioning the system. This section assumes that all prior 

sections have been completed as described.
Installation

18 of 82 Pure_Lab_2012-09-18.docx

3.7.1 Switching Power On

The main power ON/OFF switch is accessible from the left side of the gas purifier module on 

the side wall of the PLC enclosure.

Switch the power on. The main power switch light ups. All box electrical equipment is enabled 

with this switch. There is no other action necessary to turn power on with the possible exception 

of optional equipment.

The fluorescent light in the hood is controlled by its own manual switch located on the top of 

the light hood.

3.7.2 Pressure Testing the System

The Glovebox system is a leak-tight, hermetically sealed system. The entire system has been 

pressure and performance-tested prior to shipment. After assembly, the system must be pressure 

tested to ensure that all of the fittings and pipework have been connected properly.

After initial power on, the Glovebox Control screen shows the following display (Figure 3). In 

order to change any setting, the user must log in to the system.

Figure 3 Glovebox Control screen

1 Press the Login/Logout button (Figure 4).
Installation

Pure_Lab_2012-09-18.docx 19 of 82

Figure 4 Login/Logout button

2 To Log in as a User, press the Login User button and then enter the code 7990. The Login 

User button turns Green and flashes Accepted.

3 Press the Main Screen button to reach the Main Glovebox Control screen (Figure 5).

Figure 5 Main Glovebox Control screen

3.7.3 Adjusting the Box Pressures

The Glovebox Control system is programmed to maintain the Glovebox pressure between the 

minimum and maximum working pressure settings. To adjust these pressure settings:

1 Press the System Config button. The Glovebox Config Screen appears (Figure 6).
Installation

20 of 82 Pure_Lab_2012-09-18.docx

Figure 6 Glovebox Config Screen

2 Press the Numeric blue display button next to Working Maximum SP. A numerical input 

screen appears (Figure 7). Type in 12.0 and press the Enter key.

3 Press the Numeric blue display button next to Working Minimum SP. A numerical input 

screen appears. Type in 10.0 and Press the Enter key.

Figure 7 Numerical input screen

4 You will hear working gas enter the box automatically, and the pressure indicators will read 

11mbar.

a Ensure that the inside and outside doors to the Antechambers are closed.

b Turn on the vacuum pump by pressing the Start button on the main screen.

c Open the evacuate valve on the Large Antechamber and allow the Antechamber to 

evacuate to -30.

d Close the evacuate valve.

e Repeat this process for the mini Antechamber.

5 Wait 10 minutes to allow the Glovebox to stabilize.
Installation

Pure_Lab_2012-09-18.docx 21 of 82

6 Use the appropriate foot pedal to add/remove gas to/from the Glovebox to reach a pressure 

of 11.0 mbar.

7 Using a timer, wait 25 minutes and observe the current box pressure.

• If the pressure has not decreased more than 1 mbar in this time period, the Glovebox is 

leak tight.

• If the pressure drops more than 1 mbar, check all of the connections that were made 

during the assembly process. Contact your Innovative Technology, Inc. representative if 

you require assistance in troubleshooting.

3.7.4 Purging

The purpose of purging the system is to displace the air in the box with the inert gas of your 

choice. It is extremely important to displace as much of the air as possible prior to circulating 

the inlet gas through the purifier column. It is advisable to use at least three cylinders of gas per 

Glovebox module during this purging procedure and to reduce the O2 and H2O levels to less 

than 50 ppm.

Note: If you have a freezer, the door should be open during purging.

The purging method is as follows:

1 Set the Working Maximum pressure to +10.0 mbar and the Working Minimum Pressure to 

+5.0 mbar See “Adjusting the Box Pressures” on page 19.

2 Open the manual purge valve located on the top left of the box. If your Glovebox system is 

fitted with an Auto Purge valve, refer to “Purge Control” on page 31.

3 Adjust the flow using the purge valve so that there is a steady flow of incoming gas and the 

box pressure is maintained at >1mbar.

4 Create turbulence in the box to increase the efficiency of the purge. This can be done by 

pressing in the gloves and waving inside the box or installing a small electric fan inside the 

box.

5 During the purge process you will need to change the working gas cylinder as it empties. 

You must close the purge valve before changing the cylinder.

6 Re-open the Purge valve after replacing the cylinder and opening the valve on the cylinder.

7 Purge the system with at least 3 cylinders of inert gas for a standard 2-glove system.

Note: If you do not have an Oxygen Analyzer, it can be assumed that after purging with 

three (3) cylinders (for a single 2-glove system), the Oxygen content in the box is 25 ppm or 

less. Close the Manual Purge valve.

8 Open the outlet valve of the O2 analyzer, then open the inlet valve. Make sure that the 

batching pump has power. It should be plugged into the power strip. The power strip should 

be plugged into the wall outlet. Within 15 minutes, you can read the Oxygen content on the 

display.
Installation

22 of 82 Pure_Lab_2012-09-18.docx

9 After 15 minutes if the O2 level is below 50 ppm close the Purge valve and start the 

circulation. See “Circulation” below on page 22.

10 The O2 level may increase after the blower has been turned on. If the O2 level exceeds 

100 ppm, stop the blower and continue to purge. Repeat steps 9 and 10 as necessary.

WARNING: Do not open O2 analyzer valves until Glovebox has been properly purged. 

Exposure to high O2 levels will diminish the lifetime of the O2 sensor.

When shipped, the oxygen analyzer valves are closed as shown in Figure 8:

Figure 8 Oxygen analyzer valves - closed

In order to monitor the oxygen content inside the Glovebox the left Valve must be turned to 

the “Flow Out” Position and the Right Valve must be turned to the “Flow In” position as 

shown in Figure 9:

Figure 9 Oxygen analyzer valves - Flow In

Note: When purging is complete you may adjust the Minimum and Maximum Working 

pressures to attain the operating pressure that best suits your needs and comfort.

3.7.5 Circulation

The blower is designed to circulate the Glovebox atmosphere continuously through the purifier 

column. For optimum performance the blower must run constantly. Turning off the blower 

results in the atmosphere within the Glovebox degrading as it is not being circulated through the 

O2 and H2O absorbing materials contained within the purifier column. O2 and H2O levels 

inside the Glovebox will rise if the blower is switched off.

The blower can be switched on and off from the main control screen. The blower control screen 

can be accessed by Pressing the “Blower Control” button on the main control screen (Figure 

10).
Installation

Pure_Lab_2012-09-18.docx 23 of 82

Figure 10 Glove Box Control System

Figure 11 Bower Control screen

Starting the blower automatically opens the valves on the column and circulates the box 

atmosphere through the purifier column. There will be a momentary rise in oxygen because of 

the small amount of air trapped in the blower.

The blower provided is continuously variable; it can be set at a very low speed if you are 

handling sensitive crystals or during weighing. It can be set at a maximum to recover quickly 

from an operator error.

We suggest it be set at 50% during normal operation. Blower speed can be changed by pressing 

the numeric value next to the Change Blower Speed. Pressing this numeric value opens up a 

numeric entry screen where the desired blower speed can be entered. The value entered is a 

percentage. 0 means the blower is not circulating. 100% means that the blower is circulating at 

full speed.
Control Panel Functions

24 of 82 Pure_Lab_2012-09-18.docx

4 Control Panel Functions

All Glovebox functions are controlled from the touch screen (HMI – Human Machine 

Interface). This section describes the touchscreen and all the functions associated with it. The 

HMI is typically located on the side panel that has the large Antechamber attached. Custom 

systems may have the HMI mounted in a different location.

The HMI communicates with the Programmable Logic Controller (PLC) that is contained 

within the PLC enclosure on the gas purification module.

4.1 The Main Control Screen

The main control screen (Figure 12) is the primary source of information for the system.

This screen provides real time display of all key Glovebox operating parameters.

Figure 12 Glove Box Control System
Control Panel Functions

Pure_Lab_2012-09-18.docx 25 of 82

4.2 Initial Power On

When the system is powered on the following screen (Figure 13) is appears:

Figure 13 Glovebox powered on

This screen indicates the basic status parameters but does not allow any interaction until the user 

logs in to the HMI. Customers may only log in as a User.

To log in, press the Login Logout button. The following screen (Figure 14) appears:

Figure 14 Login/Logout screen

To log in as User, press the gray Login User button. A numeric data entry screen appears. The 

User Login password is 7990. After pressing the Enter button the Main Control Screen appears 

with fully functional buttons (Figure 15).
Control Panel Functions

26 of 82 Pure_Lab_2012-09-18.docx

Figure 15 Glove Box Control System with fully functional buttons

Note: Depending on the particular configuration of your system not all buttons may be 

available.
Control Panel Functions

Pure_Lab_2012-09-18.docx 27 of 82

Box Pressure (mbar) The Main Control Screen continuously displays the current Glovebox

pressure. This pressure is displayed in mbar relative to atmospheric pressure. 

The current pressure is displayed in the blue bar.

For information on adjusting the box pressure, see “Commissioning the 

System” on page 17.

O2 Level (ppm) If your system is equipped with an O2 Analyzer, the current real time O2 

level in ppm will be displayed here.

H20 Level (ppm) If your system is equipped with a H2O Analyzer, the current real time H2O 

level in ppm will be displayed here.

Pump Status The Vacuum Pump is turned on and off by pressing the START/STOP 

button. The current Vacuum Pump Status is shown in the Blue Box. If the 

Blue Box indicates OFF then the Green button next to it will display 

START. Pressing the START button will turn on the vacuum pump. The 

pump status will now show as ON and the Red STOP button will be visible. 

Pressing this Red STOP button will turn off the vacuum pump. The pump 

also has its own on/off switch mounted on the motor. This should be in the 

on position at all times.

The vacuum pump has three purposes in the Glovebox operation.

• Removal of excess pressure from the Glovebox

• Evacuation of the Antechambers

• Evacuation of the purifier column during the Regeneration cycle

Regen Status Indicates whether the purifier column is currently being regenerated. It 

displays the current stage of regeneration and the time in minutes that has 

elapsed in that stage. Refer to the regeneration section for further details 

regarding catalyst regeneration procedures.

Blower The blower is a vital component in the Glovebox gas purification system 

because it is responsible for continuously circulating the Glovebox

atmosphere through the purifier column to remove oxygen and moisture. 

The blower is comprised of a fan hermetically sealed inside an enclosure 

that is connected to the circulation pipework to create a leak-tight closedloop. The blower is located inside the gas purification module, typically 

located beneath the large Antechamber.

The Blower is turned on and off by Pressing the START/STOP button. The 

current Blower Status is shown in the Blue Box. If the Blue Box indicates 

OFF then the Green button next to it will display START. Pressing the 

START button will turn on the Blower. The Blower status will now show as 

ON and the Red STOP button will be visible. Pressing this Red STOP button 

turns off the Blower. When the BLOWER switch is activated, the PLC 

opens the column valves to enable circulation. The Blower Control button 

opens a control screen that allows further interaction with the blower 

settings. See “Blower Control” on page 30.

Note: For the blower to operate correctly the system must be supplied with 

at least 55 psi of inert working gas pressure.

4.3 Control Sub-Screens

4.3.1 Regen Control

Pressing the “Regen Control” button brings up the following screen (Figure 16):
Control Panel Functions

28 of 82 Pure_Lab_2012-09-18.docx

This screen allows the user to initiate a regeneration of the purification column.

Figure 16 Regeneration Control screen

Start Button Initiates a regeneration.

Are the Column 

Valves Closed?

Prompts the user to acknowledge that the column valves are closed.

Is Flow OK? Prompts the user to acknowledge that the regeneration gas is flowing.

Last Regen Indicates when the last regeneration occurred.

Regen Status Indicates the current step in the regeneration cycle.

Step Time Indicates the elapsed time for the current step.

Regen Column Indicates which column will be regenerated.

Abort Button Allows the user to abort a regeneration within the first 5 minutes of starting.

Regen Timers Button Accesses sub-screen below.

4.3.2 Regen Timers

The Regen timers are set in minutes. Below are the factory default settings (Figure 17). Do not 

change these timers without contact your Innovative Technology service provider.
Control Panel Functions

Pure_Lab_2012-09-18.docx 29 of 82

Figure 17 Regen Timers screen

Step 1: Heat Indicates duration of Heating Cycle in minutes.

Step 2: Heat & Purge Indicates duration of Heat & Purge Cycle in minutes.

Step 3: Purge Indicates duration of Purge Cycle in minutes.

Step 4: Evacuate Indicates duration of Evacuate Cycle in minutes.

Step 5: Cooling Indicates duration of Cooling Cycle in minutes.

Regen Screen Button Returns to the Regeneration Control screen.

Solvent Regen Button Accesses sub-screen below (Figure 18).

Note 1: All times shown are factory defaults.

Note 2: This feature is only available on systems fitted with a Large Capacity Regenerable 

Solvent Removal System.

4.3.3 Solvent Regen

Figure 18 Solvent Regeneration Control screen
Control Panel Functions

30 of 82 Pure_Lab_2012-09-18.docx

Start Button Initiates a solvent regeneration.

Are the Column Valves 

Closed?

Prompts the user to CLOSE the manual isolation valves on Large Capacity 

Regenerable Solvent Removal Column.

Is Flow OK? Prompts the user to acknowledge that the inert gas is flowing.

Regen Status Indicates the current step in the regeneration cycle.

Step Time Indicates the elapsed time for the current step.

Abort Button Allows the user to abort a regeneration within the first 5 minutes of 

starting.

Regen Timers Button Accesses sub-screen below (Figure 19).

4.3.4 Solvent Regen Timers

Figure 19 Solvent Regen screen

Step 1: Heat Indicates duration of Heating Cycle in minutes.

Step 2: Evacuate Indicates duration of Evacuate Cycle in minutes.

Step 3: Purge Indicates duration of Purge Cycle in minutes.

Step 4: Evacuate 2 Indicates duration of Evacuate 2 Cycle in minutes.

Step 5: Cooling Indicates duration of Cooling Cycle in minutes.

Sol Regen Control 

Button

Returns to the Solvent Regeneration Control screen.

Note: All times shown are factory defaults.

4.3.5 Blower Control

Pressing the Blower Control button brings up the following screen (Figure 20):
Control Panel Functions

Pure_Lab_2012-09-18.docx 31 of 82

Figure 20 Blower Control screen

The blower can be turned on or off from this screen as well as from the Main Screen.

O2 Blower Speed Enabling this feature automatically adjusts the blower speed based on the 

oxygen level within the Glovebox. If the oxygen level is above 10 ppm the 

blower speed increases in an attempt to scrub out the excess oxygen more 

quickly.

Blower Timer Enabling this feature automatically returns the Blower speed to 50% after 

30 minutes.

Change Blower 

Speed

Pressing this button brings up a numeric keypad that allows the blower 

speed to be changed between 0 and 100%.

Blower Mode Displays either Automatic (AUTO) or Manual (MANUAL).

This is not directly adjustable and is used in diagnostics.

Blower Speed (%) Displays the current speed of the blower.

Regen Blower Restart Disabled – Blower will not automatically restart after regeneration cycle is 

complete.

Enabled – Blower automatically restarts after regeneration cycle is 

complete.

Press for Box 1 Refer to section concerning parallel-piped Glovebox systems. See “Parallel 

– Piped Systems” on page 55.

Press for Box 2 Refer to section concerning parallel-piped Glovebox systems. See “Parallel 

– Piped Systems” on page 55.

Press for Both Refer to section concerning parallel-piped Glovebox systems. See “Parallel 

– Piped Systems” on page 55.

4.3.6 Purge Control

This feature only displays if your system has been factory-fitted with an Automatic Purge valve.

Pressing the Purge Control button brings up the following screen (Figure 21):
Control Panel Functions

32 of 82 Pure_Lab_2012-09-18.docx

Figure 21 Purge Control screen

O2 Level Purge SP 

(ppm)

If the system is fitted with an oxygen analyzer this setting will open the 

automatic purge valve if the oxygen level exceeds the set value. This value 

can be changed by pressing the value and entering the desired value on the 

numeric keypad. In order to operate this feature the “O2 Level Purge” must 

be ENABLED.

Timed Purge Duration 

(min)

This value sets the length of time that the automatic purge valve will be 

open if the “Timed Purge” is started.

Timed Purge Pressing START initiates the timed purge.

Timed Purge Counter The counter indicates the elapsed time since the Timed Purge was started. 

After the time limit is reached the automatic purge valve closes.

Manual Purge Pressing START will open the automatic purge valve. The purge valve will 

remain open until STOP is Pressed.

Box 1 Active/Inactive Indicates whether Box 1 can be purged using the auto purge valve.

Box 2 Active/Inactive Indicates whether Box 2 can be purged using the auto purge valve.

4.3.7 Automatic Antechamber Control

This feature is only displayed if your system has been factory-fitted with automatic 

Antechamber control.

Pressing the “Automatic Antechamber Control” button brings up the following screen (Figure 

22):
Control Panel Functions

Pure_Lab_2012-09-18.docx 33 of 82

Figure 22 Antechamber Control screen

Use Antechamber 

Number

Allows the user to select which Antechamber to control. This is only 

utilized if your system is configured with multiple automatic 

Antechambers.

Current Cycle Indicates how many evacuate/refill cycles have already been completed.

Pressure (mbar) Indicates the absolute pressure within the Antechamber as measured with 

the digital vacuum gauge.

EVAC Pressing this evacuates the Antechamber. Press again to stop evacuating.

REFILL Pressing this refills the Antechamber with inert gas from the Glovebox. 

Press again to stop refilling.

AUTO Pressing this causes the button to turn Green and start the evacuate/refill 

sequence. The individual steps associated with this are accessed and 

changed in the Antechamber Config screen (Figure 23).

4.3.8 Antechamber Config

Figure 23 Antechamber Config screen
Control Panel Functions

34 of 82 Pure_Lab_2012-09-18.docx

Auto Evacuate SP 

(mbar)

The set point that the Antechamber will be evacuated to.

Antechamber Cycles Determines the number of evacuate/refill cycles that occur under Automatic 

operation.

Auto Refill Duration 

(sec)

Determines the duration of all refills except the final refill.

Auto Final Refill Determines the duration of the final refill.

4.3.9 System Config

Pressing the System Config button brings up the following screen (Figure 24):

Figure 24 Glove Box Config screen

Working Gas Type Allows selection of Nitrogen or Argon as working gas.

Column in Use Indicates which purifier column is currently in use.

Blower Restart Status When enabled, the blower will automatically restart after a power failure.

Vacuum Pump Timer When enabled, the vacuum pump will automatically power down after 10 

minutes. The vacuum pump will be automatically powered on as required 

by the system.

Working Minimum SP The lower working pressure set point.

Working Maximum 

SP

The higher working pressure set point.

Note: The PLC control system will maintain the Glovebox pressure 

between these Minimum and Maximum set points.

IT Info Page Contains contact information for customer support.

Screen Setup Allows the contrast of the HMI screen to be adjusted.
Control Panel Functions

Pure_Lab_2012-09-18.docx 35 of 82

4.3.10 Alarms

Pressing the Alarms button brings up the following screen (Figure 25):

Figure 25 Alarms screen

O2 Alarm Enabling this feature turns on the O2 level alarm.

Set point allows the alarm to be adjusted.

Actual is the real-time O2 level.

State indicates whether the alarm has been triggered.

H2O Alarm Enabling this feature turns on the H2O level alarm.

Set point allows the alarm to be adjusted.

Actual is the real-time H2O level.

State indicates whether the alarm has been triggered.

Auto Antechamber 

Complete

Enabling this feature triggers a pop-up banner indicating the cycle is 

complete.

Alarm History A log of the various alarms that have been triggered.

Note: When an alarm is triggered a pop-up banner displays on the current 

screen to alert the user.

Acknowledge Alarm This button must be pressed to clear any active alarms.

Trend Screen

This button access graph (Figure 26) logs 02 and H20 levels over time.
Control Panel Functions

36 of 82 Pure_Lab_2012-09-18.docx

Figure 26 Trend screen

4.3.11 Freezer Control

This feature only displays if your system has been factory-fitted with a freezer.

Pressing the Freezer Control button brings up the following screen (Figure 27):

Figure 27 Freezer Control screen

Current Temp Indicates the actual temperature (°C) inside the freezer.

Set point Allows the temperature inside the freezer to be adjusted.

Range The deviation +/- °C from the set point.

Status Turns the freezer On and Off.

Note: If your Glovebox is equipped with a freezer do not plug in or turn on the freezer until the 

box has been purged with the freezer door open.
Control Panel Functions

Pure_Lab_2012-09-18.docx 37 of 82

Turning the freezer on while the Glovebox contains room atmosphere will result in the 

condensation of moisture preventing the Glovebox from achieving optimum atmosphere 

conditions, i.e. preventing the system from being able to reach 1 ppm moisture content.

4.3.12 Box Cooling Control

This feature is only displayed if your system has been factory-fitted with Box Cooling.

Pressing the Box Cooling Control button brings up the following screen (Figure 28):

Figure 28 Box Cooling Control screen

Current 

Temp

Indicates the actual temperature (°C) inside the Glovebox.

Set point Allows the temperature inside the Glovebox to be adjusted.

Range The deviation +/- °C from the set point.

Status Turns the Box Cooling On and Off.

4.3.13 Heater Control

This feature only displays if your system has been factory-fitted with a heated Antechamber.

Pressing the Heater Control button brings up the following screen (Figure 29):
Control Panel Functions

38 of 82 Pure_Lab_2012-09-18.docx

Figure 29 Heater Control screen

Current Temp Indicates the actual temperature (°C) inside the heated Antechamber.

Set point Allows the temperature inside the Antechamber to be adjusted.

Range The deviation +/- °C from the set point.

Status Turns the Heater On and Off.
System Operation

Pure_Lab_2012-09-18.docx 39 of 82

5 System Operation

5.1 Adjusting the System Pressure

The Glovebox control system is programmed to maintain the Glovebox pressure between the 

minimum and maximum working pressure settings.

To adjust these pressure settings

1 Press the System Config button.

Figure 30 Glove Box Config screen

2 Press the Numeric blue display button next to Working Maximum SP. A Numerical Input 

screen appears. Type in the desired maximum pressure and Press the Enter key.

3 Press the Numeric blue display button next to Working Minimum SP. A Numerical Input 

screen will appear. Type in the desired minimum pressure and Press the Enter key.

Figure 31 Numerical Input screen
System Operation

40 of 82 Pure_Lab_2012-09-18.docx

The Glovebox will automatically adjust the Glovebox pressure to be between these two 

limits.

5.2 Using the Antechambers

5.2.1 Manual Antechamber Control

This section describes the manual operation of the large Antechamber. Specifically, it lists the 

steps necessary to transfer an object into and out of the Glovebox. These steps assume that the 

atmosphere inside the chamber is pure and not open to the outside air.

Introducing an item into the Glovebox using the Large Antechamber

Figure 32 Large and Mini Antechambers

1 Ensure that the inner door is fully closed.

2 Ensure that the refill valve is closed.

3 Open the outside door and insert the object into the Antechamber.

4 Close the outside door.

5 Open the vacuum valve to evacuate the Antechamber (-29 in. HG ).

6 Close the vacuum valve and open the refill valve until the pressure reads –15 in HG.

7 Repeat steps 5) and 6 at least three times, finally refilling to atmospheric pressure.

8 Close the refill valve.

9 Open the inner door and remove the object from the Antechamber.

Large Antechamber

Mini Antechamber
System Operation

Pure_Lab_2012-09-18.docx 41 of 82

Figure 33 Inside Antechambers

5.3 Removing an Item from the Glovebox Using the Large Antechamber

It is extremely important that the Antechamber contains purified gas since opening the inner 

door will expose the box to the Antechamber. If you are not certain of the status of the 

Antechamber, then evacuate and refill 3 times before proceeding.

We suggest that you keep the Antechamber under vacuum. This will prompt you to refill before 

proceeding.

1 Ensure that the outside door is fully closed.

2 Ensure that the refill valve is closed.

3 Open the inner door.

4 Insert the object into the Antechamber and close the inner door.

5 Open the outside door and remove the object.

5.4 Mini Antechamber

The same precautions above are true for the mini Antechamber.

This section describes the manual operation of the mini Antechamber. Specifically, it lists the 

steps necessary to transfer an object into and out of the Glovebox. These steps assume that the 

atmosphere inside the chamber is pure and not open to the outside air.

5.4.1 Introducing an Item into the Glovebox Using the Mini Antechamber

1 Ensure that the inner door is fully closed.

2 Ensure that the 3-way valve for mini is closed.

3 Open outside door and insert the object.

Inside Antechamber with tray

Inside Mini Antechamber

0.3 micron box filter
System Operation

42 of 82 Pure_Lab_2012-09-18.docx

4 Close outside door.

5 Turn 3-way valve to evacuate.

6 Evacuate to best level of vacuum (-29 hg).

7 Turn 3-way valve to refill.

8 Repeat steps 5 through 7 at least three times.

9 Close 3-way valve.

10 Open inner door and remove object from chamber.

5.4.2 Removing an Item from the Glovebox Using the Mini Antechamber

1 Ensure that the outside door is fully closed.

2 Ensure that the 3-way valve for mini is closed.

3 Open inner door and insert the object.

4 Close inner door.

5 Open the outer door and remove the object.

Since the chambers share common vacuum and gas piping, we do not recommend using both at 

the same time.

5.5 Automatic Antechamber Control

This feature is only displayed if your system has been factory-fitted with automatic 

Antechamber control.

Pressing the “Automatic Antechamber Control” button brings up the following screen:

Figure 34 Antechamber Control screen
System Operation

Pure_Lab_2012-09-18.docx 43 of 82

Use Antechamber 

Number

Allows the user to select which Antechamber to automatically control.

Typical systems will have one large Antechamber equipped with automatic 

control.

Current Cycle Indicates how many evacuate/refill cycles have already been completed.

Pressure (mbar) Indicates the absolute pressure within the Antechamber as measured with 

by the digital vacuum gauge.

EVAC Pressing this evacuates the Antechamber. Press again to stop evacuating.

REFILL Pressing this refills the Antechamber with inert gas from the Glovebox. 

Press again to stop refilling.

AUTO Pressing the AUTO button causes the button to turn Green and start the 

evacuate/refill sequence. The individual steps associated with this are 

accessed and changed in the ‘Antechamber Config” screen.

Antechamber Config

Figure 35 Antechamber Config screen

Auto Evacuate SP 

(mbar)

The set point that the Antechamber will be evacuated to.

Antechamber Cycles Determines the number of evacuate/refill cycles that occurs under 

Automatic operation.

Auto Refill Duration 

(sec)

Determines the duration of all refills except the final refill.

Auto Final Refill Determines the duration of the final refill.

5.6 Introducing an Item into the Glovebox Using the Large Antechamber

1 Ensure that the inner door is fully closed.

2 Ensure that the refill valve is closed.

3 Open the outside door and insert the object into the Antechamber.
System Operation

44 of 82 Pure_Lab_2012-09-18.docx

Figure 36 Inserting an object into the Antechamber

4 Close the outside door.

5 Press the Auto button.

The system evacuates and refills as many cycles as programed.

The system final refills.

6 Open the inner door and remove the object from the Antechamber

Figure 37 Removing object from Antechamber

Large Antechamber

Mini Antechamber
System Operation

Pure_Lab_2012-09-18.docx 45 of 82

5.7 Removing an Item from the Glovebox Using the Large Antechamber

It is extremely important that the Antechamber contains purified gas because opening the inner 

door will expose the box to the Antechamber. If you are not certain of the status of the 

Antechamber, then evacuate and refill 3 times before proceeding.

We suggest that you keep the Antechamber under vacuum. This will prompt you to refill before 

proceeding.

1 Ensure that the outside door is fully closed.

2 Ensure that the refill valve is closed.

3 Open the inner door.

4 Insert the object into the Antechamber and close the inner door.

5 Open the outside door and remove the object.

5.8 Purge Valves

Manual Purge Valve – As standard the Glovebox is fitted with a hand-operated 2-Way Ball 

Valve that is used to manually displace atmosphere from the Glovebox. The valve must be 

closed fully after purging process to prevent loss of box pressure.

Figure 38 Hand-operated 2-Way Ball Valve
System Operation

46 of 82 Pure_Lab_2012-09-18.docx

5.8.1 Automatic Purge Valve

Figure 39 Automatic purge valve

This feature is only available if your system has been factory-fitted with an automatic purge 

valve.

Pressing the “Purge Control” button brings up the following screen (Figure 40):

Figure 40 Purge Control screen
System Operation

Pure_Lab_2012-09-18.docx 47 of 82

O2 Level Purge SP 

(ppm)

If the system is fitted with an oxygen analyzer this setting will open the 

automatic purge valve if the oxygen level exceeds the set value. This value 

can be changed by pressing the value and entering the desired value on the 

numeric keypad. In order to operate this feature the “O2 Level Purge” must 

be ENABLED.

Timed Purge Duration 

(min)

This value sets the length of time that the automatic purge valve will be 

open if the “Timed Purge” is started.

Timed Purge Pressing START initiates the timed purge.

Timed Purge Counter The counter indicates the elapsed time since the Timed Purge was started. 

After the time limit is reached, the automatic purge valve closes.

Manual Purge Pressing START opens the automatic purge valve. The purge valve remains

open until STOP is pressed.

Box 1 Active/Inactive Indicates whether Box 1 can be purged using the auto purge valve.

Box 2 Active/Inactive Indicates whether Box 2 can be purged using the auto purge valve.

Note: Initiating any purge sequence while the blower is running will switch off the blower until 

the purge sequence is complete. The blower will automatically restart after the purge sequence 

is complete.

5.9 Regeneration

Pressing the “Regen Control” button brings up the following screen (Figure 41):

This screen allows the user to initiate a regeneration of the purification column.

Figure 41 Regeneration Control screen

The purification column contains a mixture of copper-based catalyst to absorb oxygen and 

molecular sieve to absorb moisture. Over time these materials will become saturated and will no 

longer be able to absorb the oxygen and water. The regeneration process allows these materials 

to be returned to their original state thus allowing continued scrubbing of oxygen and moisture. 
System Operation

48 of 82 Pure_Lab_2012-09-18.docx

To maintain optimal performance it is recommended to change the purifier catalyst every three

years.

WARNING: The final step (step 5 Cooling) of the regeneration process exposes the freshlyregenerated purifier material to the Glovebox atmosphere. The Glovebox must contain less than 

100 ppm oxygen prior to starting the regeneration.

A regeneration cycle comprises five steps as shown below (Figure 42). The default times are 

shown and are measured in minutes.

Figure 42 Regen Timers screen

5.9.1 Single-Column Purification System Regeneration Procedure

1 Connect a cylinder of regeneration gas (forming gas) to the regen gas connection on the rear 

of the gas purification module. The regeneration gas must contain 3 to 7% hydrogen with the 

balance being nitrogen or argon. Ensure that the connections are tightened correctly. Open 

the regulator and set it to 7 psi.

2 Turn off the Blower on the Main screen. On a single column system this automatically closes 

the purification column valves.

3 Press the START button.

4 Press the Yes button next to “Are the Column Valves Closed?” A “click” will be heard as the 

solenoid valves open to allow regeneration gas to flow.

Note: All column isolation valves have a yellow indicator tab. This yellow tab is not visible 

when the valve is closed.

5 Adjust needle valve on the regeneration flow meter located on the front of the gas 

purification module to set the flow to 25-30 scfh on the graduated scale on the flow meter.

6 Press Yes next to “Is Flow OK?”. At this point the regeneration gas will stop flowing. “Step 

1. Heating” begins. The Regen Status displays as Heating and the timer begins to count up. 

The Regeneration Status is also shown on the Main Screen.

7 The system automatically progresses through the regeneration steps 1 to 5. At the conclusion 

of the regeneration sequence the blower automatically turns back on if this feature has been 
System Operation

Pure_Lab_2012-09-18.docx 49 of 82

enabled in the “System Config” Screen. The blower can also be turned on using the Blower 

Start button.

5.10 Dual-Column Purification System Regeneration Procedure

Dual column purification systems allow the regeneration of one purifier column while still 

circulating the Glovebox atmosphere through the second column. This permits constant uptime.

Initiating a regeneration on a dual column system automatically regenerates the column that is 

NOT in use. The column in use can be selected from the “System Config” Screen (Figure 43).

Figure 43 Glove Box Config screen

If the “Column In Use” is shown as “1” then pressing the blue bar brings up a numeric entry 

keypad. To select column 2, press number 2 and then enter.

To regenerate perform steps 1, 3,4,5,6 as for the single column purification system regeneration. 

The blower should remain switched on during the regeneration of a dual column system,

5.11 Solvent Regeneration

5.11.1 Solvent-Column System Regeneration Procedure

Note: Blower does not need to be turned off.

1 Connect a cylinder of high purity Inert Gas (Nitrogen or Argon) to the regen gas connection 

on the rear of the gas purification module. The Inert gas should be 99.998 nitrogen or argon. 

Ensure that the connections are tightened correctly. Open the regulator and set it to 7 psi.

2 Open the by-pass valve (labeled A in Figure 44 below). This is located in the Gas Purifier. 

The top must be removed to access valve
System Operation

50 of 82 Pure_Lab_2012-09-18.docx

Figure 44 Gas purification module

3 Close the isolation valves (labeled B & C) connected to the solvent column. This is located 

in the Gas Purifier. The top must be removed to access valve.

4 From the Main Control Screen select Regen Control. Then select Solvent Regen.

Figure 45 Solvent Regeneration Control screen

5 Press the START button (Figure 45).

6 Press the Yes button next to “Are the Column Valves Closed?” A “click” will be heard as the 

solenoid valves open to allow regeneration gas to flow.

7 Adjust needle valve on the regeneration flow meter located on the front of the gas 

purification module to set the flow to 25-30 scfh on the graduated scale on the flow meter.
System Operation

Pure_Lab_2012-09-18.docx 51 of 82

8 Press Yes next to “Is Flow OK?”. At this point the regeneration gas stops flowing. Step 1. 

Heating begins. The Regen Status shows as Heating and the timer begin to count up. The 

Regeneration Status is also shown on the Main Screen.

9 The system automatically progresses through the regeneration steps 1 to 5.

10 Upon completion of the solvent regeneration, open the column isolation valves (B & C), then 

close the Solvent by-pass valve (A).

5.12 Analyzers

5.12.1 O2 Analyzer

When shipped, the oxygen analyzer valves are closed as shown in Figure 46 below:

Figure 46 Oxygen analyzer valves – “Closed” position

WARNING: The analyzer isolation valves should remain closed until the Glovebox has been 

properly purged.

In order to monitor the oxygen content inside the Glovebox, the left Valve must be turned to the 

“Flow Out” Position and the Right Valve must be turned to the “Flow In” position as shown in 

Figure 47 below:

Figure 47 Oxygen analyzer valves – “Flow Out” position
System Operation

52 of 82 Pure_Lab_2012-09-18.docx

The Oxygen analyzer is designed specifically for detecting trace amounts of oxygen in inert 

gases. The operational range of this analyzer is 0.1 to 1,000 ppm.

Note 1: The analyzer requires that the small batching pump inside the Glovebox be connected 

and powered on at all times.

Note 1: The analyzer is calibrated at the factory prior to shipment.

5.13 Moisture Analyzer

The moisture analyzer sensor is aluminum oxide ultra-high capacitance designed. It incorporates 

automatic temperature compensation. It is mounted in the gas stream piping.

It is shipped with a calibration certificate traceable to international standards.

The moisture content is shown on Main Screen of the Color HMI touch screen in ppm.

Freezer

Figure 48 Moisture Analyzer

The freezer is controlled from the HMI.

Pressing the “Freezer Control” button brings up the following screen (Figure 49):
System Operation

Pure_Lab_2012-09-18.docx 53 of 82

Figure 49 Freezer Control screen

Current Temp Indicates the actual temperature (°C) inside the freezer.

Set point Allows the temperature inside the freezer to be adjusted.

Max set point is -35 c.

Range The deviation +/- °C from the set point.

Status Turns the freezer On and Off. By pressing the green start button or red stop 

button.

Shelves

Cylinderical freezer has a sliding tray with 2 or 3 shelves.

Rectangle freezer has 5 removable shelves.

5.13.1 Box Cooling Control

Figure 50 Box Cooling Control

This feature is only displayed if your system has been factory-fitted with Box Cooling.
System Operation

54 of 82 Pure_Lab_2012-09-18.docx

Pressing the “Box Cooling Control” button brings up the following screen (Figure 51):

Figure 51 Box Cooking Control screen

Current Temp Indicates the actual temperature (°C) inside the Glovebox.

Set point Allows the temperature inside the Glovebox to be adjusted.

Range The deviation +/- °C from the set point.

Status Turns the Box Cooling On and Off. By pressing the green start button or red 

stop button.

Note: Boxing will remove 10 degrees for ambient system temp.

5.14 Heater Control

This feature is only displayed if your system has been factory-fitted with a heated Antechamber.

Pressing the “Heater Control” button brings up the following screen (Figure 52):
System Operation

Pure_Lab_2012-09-18.docx 55 of 82

Figure 52 Heater Control screen

Current Temp Indicates the actual temperature (°C) inside the heated Antechamber.

Set point Allows the temperature inside the Antechamber to be adjusted up to 250 C.

Range The deviation +/- °C from the set point.

Status Turns the Heater On and Off. By pressing green start button or red stop 

button.

WARNING: Users can be burned if Ante chamber is not operated properly.

5.14.1 Parallel – Piped Systems

Parallel-piping offers the convenience of using a single purifier to control two separate 

Glovebox modules. It is intended that the primary mode of operation will be both Glovebox

modules running simultaneously. In this case the atmosphere inside each Glovebox module is 

being circulated through the purifier column. The O2 and H2O values displayed are 

representative of the atmosphere in both Gloveboxes as they share a common atmosphere. Each 

Glovebox module (1 & 2) is fitted with automatic valves on the gas inlet and outlet to allow 

isolation of the 2 Glovebox modules.

One Glovebox module is defined as Box 1 and the other module is Box 2.

Under the sub screen Blower Control. The system can operate 3 ways.

Press for Box 1 – Selecting this will change the button to green to indicate that the purifier will 

only interact with Box 1. Pressure will only be controlled in Box1 and the displayed O2 and 

H2O values are for Box 1 only.

WARNING: In this state Box 2 is completely isolated and should not be used. This mode would 

typically be used to enable Box 2 to be opened to air for cleaning while continuing to circulate 

Box 1 atmosphere through the purifier.
System Operation

56 of 82 Pure_Lab_2012-09-18.docx

Press for Box 2 – Selecting this will change the button to green to indicate that the purifier will 

only interact with Box 2. Pressure will only be controlled in Box 2 and the displayed O2 and 

H2O values are for Box 2 only.

WARNING: In this state Box 1 is completely isolated and should not be used. This mode would 

typically be used to enable Box 1 to be opened to air for cleaning while continuing to circulate 

Box 2 atmosphere through the purifier.

Press for Both – Selecting this changes the button to green to indicate that the purifier will 

circulate through both Box 1 AND Box 2 when the blower is started.

Note: For the three modes of operation described above the Blower Start button must be used to 

initiate circulation.

Figure 53 Blower Control screen

Purging Parallel Piped System

The user must select which Box (1 or 2) to purge.

Manual Purge – Only open the purge valve on the selected Glovebox module (1 or 2).

See “Purge Control” on page 31 for further details.

Auto Purge

Press for Box 1 – Selects the module labeled Box 1 to be purged (See “Purge Control” on page 

31) (Purge control – refer to Purge Control Screen in Figure 54 below) Box 1 Active button will 

be Green indicating that it will be purged. Box 2 will be isolated.

Press for Box 2 – Selects the module labeled Box 2 to be purged (See “Purge Control” on page 

31) (Purge control – refer to Purge Control Screen in Figure 54 below) Box 2 Active button will 

be Green indicating that it will be purged. Box 1 will be isolated

Note: Only one module should be selected to purge at a given time.
System Operation

Pure_Lab_2012-09-18.docx 57 of 82

Figure 54 Purge Control screen

5.15 Solvent Removal Systems

Various sizes of solvent removal systems are available. These are optional extras that can absorb 

solvent vapors that would otherwise contaminate the purification media.

5.15.1 Large Capacity Solvent Removal

There are three isolation valves labeled A, B, and C. The isolation valves dictate the flow of box 

atmosphere, routing it directly to the purification column or through the solvent removal column 

prior to the purification column. There are arrows on these valves, indicating the direction of 

flow through them. To open them, turn the black knob counterclockwise. To close them, turn 

the black knob clockwise.

The evac/refill valve is located on the front of the unit. To the left of the valve is a pressure 

gauge that indicates the pressure inside the solvent removal column. Evacuate will open the 

solvent removal column directly to the vacuum pump. Refill will expose it to the box 

atmosphere as well as the incoming working gas supply.

5.15.2 Opening the Solvent Removal Column

1 Refill the column by turning the evac/refill valve to refill until the pressure gauge has 

returned to 0, and then close the valve.

2 Open isolation valves B and C.

3 Close isolation valve A. It is important that B and C be opened prior to A being closed so 

that flow to the blower will not be interrupted.

4 The flow of box atmosphere is now passing through the solvent removal column prior to 

passing through the purification column.
System Operation

58 of 82 Pure_Lab_2012-09-18.docx

Figure 55 Solvent Removal Column - Open

5.15.3 Isolating the Solvent Removal Column and Removing Solvent Vapors

1 Open isolation valve A. This allows the system to continue to circulate through the 

purification column.

2 Close isolation valves B and C. This isolates the solvent removal column so that you can 

evacuate it without affecting the rest of the system. It is important to open valve A prior to 

closing B and C so that flow to the blower is not interrupted.

3 Turn evac/refill valve to evacuate.

4 Allow vacuum to pull on the column for at least 24 hours.

5 To begin circulation through the solvent removal column again, see “Opening the Solvent 

Removal Column” on page 57.
System Operation

Pure_Lab_2012-09-18.docx 59 of 82

Figure 56 Solvent Removal Column – Closed

5.15.4 Replacing the Carbon Charge in a Solvent Removal Column

1 Follow the instructions for isolating the solvent column and removing solvent vapors on 

page 58.

2 Turn the evac/refill valve to refill until the pressure in the column has returned to 0.

3 Close the evac/refill valve.

4 Place receptacle below the Empty Port to catch the old carbon charge.

5 Remove the Fill Port clamp and blank.

6 Remove the Empty Port blank and begin to collect the old carbon charge.

7 Replace the Empty Port clamp and blank. Make sure the centering ring is clean.

8 Pour new carbon charge into Fill Port. Use a funnel if necessary.

9 Turn evac/refill valve to evacuate and allow vacuum to pull on the column for at least 48 

hours.

10 To begin circulation through the solvent removal column again, see “Opening the Solvent 

Removal Column” on page 57.
System Operation

60 of 82 Pure_Lab_2012-09-18.docx

Figure 57 Fill and Empty Ports

Figure 58 Evac/Refill Valve at Evacuate

5.16 Large Capacity Regenerable Solvent Removal

Operates identical as the large capacity solvent removal while having the added ability to 

regenerate the material with in the column.

See “Solvent Regeneration” on page 49 for further details.
System Operation

Pure_Lab_2012-09-18.docx 61 of 82

5.16.1 Small Capacity Solvent Removal – External

Figure 59 Small Capacity Solvent Removal – External

This contains an activated carbon-impregnated filter element. It is located in the Glovebox

circulation piping.

WARNING: It is fitted with an isolation valve that MUST remain OPEN at all times when the 

Glovebox is running.

Replacing the filter element

1 Turn off the blower. Power down the Glovebox using the main power switch on the gas 

purification module.

2 Close the isolation valve on the housing of the solvent removal system. See Figure 59 above.

3 Twist the clear base counter-clockwise to open the trap.

4 Replace the filter element.

5 Screw the clear base back into the lid.

6 Open the isolation valve.

7 Power the system on.

8 Purge the Glovebox until the O2 level is less than 50 ppm. See “Purging” on page 21.

9 Turn on the blower.
System Operation

62 of 82 Pure_Lab_2012-09-18.docx

Small Capacity Solvent Removal – Internal

Figure 60 Small Capacity Solvent Removal – Internal

This replaces the Glovebox HEPA filter on the suction side of the gas purification system. It is 

filled with activated carbon. It should be unscrewed before purging the Glovebox to prevent 

activated carbon particles being dispersed inside the Glovebox. Cartridge must be removed from 

Glovebox to be emptied and refilled.
Routine Maintenance

Pure_Lab_2012-09-18.docx 63 of 82

6 Routine Maintenance

Regular preventive maintenance will help to reduce box problems that cause down time and 

increase the overall performance of the system. Different environments require different 

maintenance intervals, but the following are the recommended minimum levels of service 

required.

6.1 Daily Tasks

• Check the gloves for wear or holes. Check that they are seated properly and that the exterior 

“O” rings are in place. Replace worn or defective gloves or “O” rings as soon as possible. 

See “Removal and Replacement Procedures” on page 71.

• Inspect the exterior of the system. Ensure that it is kept free from excessive dirt. Check that 

the piping is intact and that all gas supply and vent lines are well connected. If necessary, 

clean the exterior. Tighten any loose connections.

• Check that the gas supply is sufficient and the flow is adequate. If necessary, replace the gas 

supply and adjust the flow rate.

6.2 Weekly Tasks

• Check and replace, if necessary, the vacuum pump oil. This step is crucial to the life of the 

pump and its ability to perform to specifications. Because each pump is different, refer to the 

pump manual for service procedures.

• Check the ante chambers “O” rings. Replace them if they are worn.

• Check the box filters. Replace if necessary.

6.3 Annual or Semi Annual Tasks

• If the purification capability of the column is not sufficient (it can no longer maintain low 

oxygen and moisture levels), regenerate the filter column material.

• If time between regenerations has become minimal, replace filter column material.

• Replace box filters.

6.4 Safety Inspections

• Inspect all box wiring for signs of wear or damage. Replace any suspect wiring found during 

the inspection.

• Inspect the window for signs of stress or cracks. Replace as needed.

• Never place or stack materials, tools, or documentation on any part of the Glovebox exterior 

surfaces except for those that are designed for such purposes. Vibration or contact with 

people or other objects may cause the items to fall.

Follow all of the routine maintenance tasks listed in this section.
Troubleshooting

64 of 82 Pure_Lab_2012-09-18.docx

7 Troubleshooting

These are some of the most frequently-asked questions and our responses.

Problem: There is no working gas going into the Glovebox.

Possible causes/solutions:

• If cylinder gas, is the tank empty? If empty, replace the tank.

• Is the regulator open? If closed, open the regulator.

• Is the needle valve on the regulator open? If not, open the valve.

• What is the gas pressure? If the pressure is too high the valve won't open. Pressure on a 

Purelab HE should be 55 PSI.

• Is the electrical signal getting to the gas (GA) valve? The Valve will illuminate if it is 

receiving the proper voltage.

• Is the Working Minimum Pressure set higher than the current pressure box pressure? The 

system will not call for gas unless the box pressure drops below the Working Minimum set 

point.

Problem: There is no regen gas flowing.

Note: Regen gas will only flow after you have confirmed that the column valves are closed. 

After the regen starts the gas will no longer flow until the Purge stage of the regen begins.

Possible causes/solutions:

• If cylinder gas, is the tank empty? If empty, replace the tank.

• Is the regulator open? If closed, open the regulator.

• Is the needle valve on the regulator open? If not, open the valve.

• Are the RG and EX valves illuminated?

• Is the flow meter on the front of the purifier cart open?

Problem: The Antechamber won't hold a vacuum.

Possible causes/solutions:

• Clean the Antechamber door and O-rings of debris. Replace the O-rings if necessary.

• Verify the refill valve is closed.

• Verify both doors are closed.

• Verify all fittings on the chamber are tight.
Troubleshooting

Pure_Lab_2012-09-18.docx 65 of 82

Problem: Box is leaking.

Possible causes/solutions:

• Normally a box will not begin to leak on its own. If none of the fittings have been modified 

the leak is most likely in the gloves. Inspect the gloves closely for leaks. Replace gloves as 

required.

• Verify that all Antechambers are left under a static vacuum and are not leaking.

• Refer to pressure test section of this manual. Record the rate of leakage.

Problem: Vacuum pump is making noise or has reduced performance.

Possible causes/solutions:

• Change the pump oil if it looks bad.

• Verify that all fittings connected to the vacuum pump are tight.

• If there is no improvement after changing the pump oil and checking the fittings upstream 

from the pump, put a vacuum gauge directly on the inlet of the pump. Refer to the pump 

manual for obtainable vacuum. If the pump cannot obtain specified levels repair or replace 

the pump.

Problem: The window is "cloudy" or scratched.

Possible causes/solutions:

• Safety glass can be washed with alcohol. Scratches in glass are not easily repaired, especially 

if they are on the inside. A glass company professional might repair scratches on the outside 

of the glass.

• Lexan, can be washed gently with mild soap or detergent using a soft cloth. Organic solvents 

such as aliphatic hydrocarbons, kerosene, or naphtha may also be used. To remove light 

scratches try automotive wax.

When a Service Call is Necessary

Should a service call be necessary, the more information that can be provided about the 

problem, the better we will be able to respond with a quality answer in a timely fashion. If a 

service call is necessary or you cannot resolve one of the above issues, call your local 

Innovative Technology representative for help.
System Control Electronics

66 of 82 Pure_Lab_2012-09-18.docx

8 System Control Electronics

8.1 General Description

The Glovebox is controlled through the use of the Color HMI Touch Screen. It directs box 

operation through the lower PLC control box, which generally turns on or off relays to control 

box pressure, circulation, purifier regeneration and other functions. Each of these subsystems 

are described in this section.

8.1.1 PLC Controller

All inputs and Output are controlled by the PLC.

8.1.2 Display panel

The display panel is Color HMI Touch Screen.

8.2 Lower Control Box

The Lower Control Box houses the following components:

• Power Supply

• PLC

• Connections

• Solid State Relays

8.2.1 Power Supply

The power supply generates 24 volts DC from an input of 115 volts AC. It can be strapped for 

operation at other input voltages. The 24 VDC is for the operation of the upper control 

electronics, the lower PC board, and for other miscellaneous components. It can easily be 

removed for repair or replacement.

8.2.2 Box Pressure Sensor

This section connects the tubing from the box directly to the on board pressure sensor. The 

sensor input is calibrated and provided to the microprocessor.

8.2.3 24 Volt Low Power Relays

This section of the board is where the PLC control lines activate or deactivate the low power 

relays. These relays then switch on and off other relays in the system. Note that there is an LED 

for each relay on the PLC. When this LED is illuminated, it indicates that 24 VDC is present on 

the switched or output side of the relay. When trouble shooting a control problem, this is the 

best indication that the box electronics are functioning properly, and that perhaps the higher 

power relay is not working.
System Control Electronics

Pure_Lab_2012-09-18.docx 67 of 82

Table 2 Relay Functions and Connections

Connector Function Connection

GAS Gas valve, opens the box to the working gas source, 

increases box pressure.

Valve block, GA

VAC Vacuum valve, opens the box to the vacuum pump, reduces 

box pressure.

Valve block, VA

AE1 Antechamber 1 evacuate, opens the chamber to the vacuum 

pump.

Antechamber 1

AE2 Antechamber 2 evacuate, opens the chamber to the vacuum 

pump.

Antechamber 2

AR1 Antechamber 1 refill, allows the Antechamber to refill from 

the box.

Antechamber 1

AR2 Antechamber 2 refill, allows the Antechamber to refill from 

the box.

Antechamber 2

BL1 Blower, activates the column 1 valves to enable the blower 

speed control to turn on the blower.

Column 1 valves

BL2 Blower, activates the column 2 valves to enable the blower 

speed control to turn on the blower.

Column 2 valves

CO1 Cooling, vents column 1. Valve block, CO

CX1 Column 1 flow control, used in dual column systems. Dual column in-line pipe

CX2 Column 2 flow control, used in dual column systems. Dual column in-line pipe

EX1 Regeneration gas exit, column 1. Main cell block, EX

FRZ Freezer control. Freezer Power Crydom

HTR Heater control. Furnace power Crydom

HT1 Heater control, column 1. Heater Crydom

HT2 Heater control, column 2. Heater Crydom

PG1 Automatic purge valve control. Purge valve

PMP Vacuum pump control. Vacuum pump Crydom

RG1 Regeneration gas, opens column 1 to regeneration gas. Valve block,RG

RG2 Regeneration gas, opens column 2 to regeneration gas. Valve block,RG 

RV Regeneration vacuum, opens column 1 to the vacuum pump. Valve Block RV

PG2 Automatic purge valve control. Purge Valve

8.2.3.1 Solid State Relays

This section of the board is where the PLCcontrol lines activate or deactivate the solid state 

relays (SSR’s). These SSR’s then switch on and off motors and heaters in the system.
System Control Electronics

68 of 82 Pure_Lab_2012-09-18.docx

Table 3 Solid State Relays

Solid State Relay Function

FRZ Turn on AC power for the freezer motor. Only used in some systems.

BXC Turn on AC power for the box cooling motor. Only used in some systems.

HTR Turn on AC power for the heater element. Only used in some systems.

HT2 Turn on AC power for the heater for the second column in a dual column 

system.

HT1 Turn on AC power for a single column system’s heater.

VP Turn on AC power for the vacuum pump.

8.2.4 Circuit Breakers

Table 4 Circuit Breakers

Circuit Breaker # Amperage 110/220 VAC Function

1 15A/10A Illuminated Main Power Switch

2 10A/5A Vacuum Pump

3 6A/3A 24V Power Supply

4 5A/3A Blower

5 4A/2A Heater for Column 1

6 4A/2A Heater for Column 2

7 5A/3A Lighthood

8.3 Specifications

Table 5 Specifications

Input Operating Voltage and Frequency 110-120, 200-240 V∼, 50/60Hz

Operating Temperature 10-40° Celsius

Relative Humidity Non condensing 5% to 95% RH
System Components

Pure_Lab_2012-09-18.docx 69 of 82

9 System Components

9.1 Box Flow

The general box flow is described below. When circulation is on, gas is drawn by the blower 

from the box, through the column, and then back into the box. When the box requires a more 

positive pressure, valve ‘GA’ opens enabling working gas to enter the circulation loop prior to 

the column. This allows the gas to be purified before entering the box. When the box requires 

less pressure, valve ‘VA’ opens enabling the gas from the box to be drawn out by the vacuum 

pump. The remaining valves are for system regeneration purposes.

Figure 61 General Box Flow

9.1.1 Piping and Valves

All piping and tubing is stainless steel unless otherwise specified. All electromagnetic valves are 

stainless steel with brass bases unless otherwise specified. All electro pneumatic valves are 

stainless steel unless otherwise specified.

9.1.2 Column

The column contains the material that purifies the box gas, a heater for regeneration purposes, 

and a set of either manual or automatic valves that switch on or off the circulation of the box 

through the column. Construction is of stainless steel.
System Components

70 of 82 Pure_Lab_2012-09-18.docx

9.1.3 The Blower

The blower circulates the gas in the Glovebox through the column for purification. It is mounted 

inside an enclosure that is attached to the floor of the purifier module. The blower enclosure is 

connected through two pipes using two KF-40 connections.

9.1.3.1 Variable Speed

The variable speed blower is fully variable from off through full speed. There is one electrical 

connection containing five wires. Two of the wires are for 115 VAC input power and the other 

two are for a 0 to 10 VDC reference voltage that controls the blower's speed. This reference 

voltage is provided by the PLC controller.

9.1.4 Vacuum Pump

The vacuum pump provides the means to reduce overall pressure levels in the main box and the 

Antechamber(s). It is electrically turned on via a Press button on the main control screen. In 

response to this user action, the PLC controller turns on a solid state relay. The relay supplies 

line voltage to turn on the pump. In response to a requirement to lower the box pressure, or 

pump down an Antechamber, the PLC will open valve, “VA”, or “AE”, respectively. The pump 

will turn on automatically if it is needed by the system.

9.1.5 Chambers

There are many different types of Antechambers available from Innovative Technology, Inc. 

However, the standard chamber is a 15" I.D. by 24" long cylinder mounted on the left or right 

side of the box. In addition to this, a 6" x 15" mini Antechamber is also available on either side.
Removal and Replacement Procedures

Pure_Lab_2012-09-18.docx 71 of 82

10 Removal and Replacement Procedures

These procedures are intended to be a guideline for removing and replacing various components 

of the system, for either routine maintenance or basic repair of the system. They are not 

intended to be exact step-by-step instructions. It is assumed that the person using these 

procedures is capable of performing basic mechanical and electrical tasks.

10.1 Blower

To replace the blower, proceed as follows.

1 Turn off the blower.

2 Turn off the system main power.

3 Remove the left side panel from the purifier module.

4 Remove the electrical connector from the blower.

5 Loosen and remove the two clamps that attach the piping. Cover the opening with KF-40 

blanks or plastic caps to prevent excess outside air from entering the system.

6 Remove the blower box assembly and replace it with a new one.

7 If possible, purge the box piping with working gas.

8 Connect the piping with the clamps.

9 Connect the electrical connector.

10 Power the system on and check for leaks.

10.2 Column

To replace the filter material in the purifier column you will need a vacuum cleaner, funnel, and 

a waste bins to collect old material.

1 Turn off the blower.

2 Turn off the system main power.

3 Place the waste bin under the bottom port of the column.

4 Remove the two KF-40 clamps and blanks from the side of the column.

Note: As soon as the bottom blank is removed the filter material will begin to come out.

5 Drain the old material and use the vacuum to remove any remaining material.

6 Empty the entire contents of the purifier according to your safety regulations.

7 Replace the blank onto the bottom port of the column.

8 Use the funnel in the top port of the column and add half of the molecular sieve, all of the 

copper catalyst, and then the rest of the sieve.

9 Replace the blank to the top port of the column. Be certain that KF-40 connections are tight.

10 Restore the main power to the system.
Removal and Replacement Procedures

72 of 82 Pure_Lab_2012-09-18.docx

11 Perform two regenerations of the column for optimum performance.

12 Resume normal system operation.

10.3 Filters

To replace the box filters, simply exchange them with new filters. They are installed hand tight 

and should be replaced the same way.

10.4 Gloves

Replacing a glove increases the risk of exposing the system to the outside environment. Prior to 

proceeding, ensure that every precaution is taken to protect the contents of the box.

Figure 62 Gloves

10.4.1 Replacement with an Internal Glove Port Cover

An internal glove port cover is used to seal off the glove port from the inside of the Glovebox.

1 If the glove port cover is not inside the Glovebox, bring it in via the Antechamber.

2 Install and tighten the glove port cover on the port that has the glove that is to be replaced.

3 Remove the “O” rings that hold the glove onto the glove port.

4 Remove the old glove and discard it properly.

5 Compress the new glove as much as possible to remove excess air.

6 Install the new glove over the glove port.

7 If possible, purge the new glove by venting working gas into it via a separate gas line (this 

removes as much air as possible, and hence reduces the amount of air being introduced into 

the system).

8 Install the “O” rings onto the new glove port, over the new glove.

9 Remove the internal glove port.

Glove Port

Glove O-Rings

Internal Glove Port Cover

Butyl Gauntlets
Removal and Replacement Procedures

Pure_Lab_2012-09-18.docx 73 of 82

10.4.2 Replacement without an Internal Glove Port Cover

Without an internal glove port cover, the old glove can be used to cover the port while the new 

glove is installed.

1 Press the glove that is to be replaced into the box.

2 Remove the inner glove port “O” ring (that which is closest to the window).

3 Fold back the glove onto the outer “O” ring, taking care not to let the glove come off of the 

port.

4 Compress the new glove as much as possible to remove excess air.

5 Install the new glove over the glove port, over the old glove.

6 Install the inner “O” ring onto the glove port, over the new glove.

7 If possible, purge the new glove by venting working gas into it via a separate gas line. (This 

removes as much air as possible, and hence reduces the amount of air being introduced into 

the system.)

8 From the inside of the Glovebox, using the other glove, remove the old glove by taking it 

into the Glovebox.

9 Install the outer “O” ring onto the glove port, over the new glove.

10 Remove the old glove from the box via the Antechamber, and discard it properly.

10.4.3 Control Box Panel

The control box consists of the circuit breakers, electronics module, power supply, and the 

solid-state relays. Some systems will require the white backing plate to be removed prior to 

removing the power supply and the solid-state relays.

10.4.4 Circuit Breakers

When replacing a circuit breaker, always use a replacement part of the same rating.

1 Remove system AC power from the source.

2 Disconnect the two fast-on type connectors using pliers. Do not remove them by pulling on 

the wire to which they are attached.

3 Depress the two tabs on the breaker body and Press out through the front door.

4 Install new circuit breaker in reverse order.

10.4.5 Power Supply

To replace the 24 VDC power supply, proceed as follows:

1 Remove system AC power at the source.

2 Disconnect the DC power cable (right side, Blue & White with Blue wires).

3 Disconnect the AC power cable.

4 Release the clip on the bottom of the power supply. This will release it from the DIN rail.
Removal and Replacement Procedures

74 of 82 Pure_Lab_2012-09-18.docx

5 Reinstall in reverse order.

10.4.6 Solid State Relays

Mark all wiring before removal. To replace the SSRs, proceed as follows:

1 Remove system AC power at the source.

2 Disconnect the DC control wiring, right side of SSR, screws 3 & 4.

3 Disconnect the AC control wiring, left side of SSR, screws 1 & 2.

4 Remove the screws holding the SSR to the white backing panel.

5 Reinstall in reverse order.

10.4.7 Valves

If an electrical valve should fail, it can be replaced as follows.

1 Remove Glovebox AC power. Do not assume that the valve is off and will stay off, because

the PLC can switch a valve on due to an external event.

2 If the GA valve is being replaced, ensure that the gas supply is off.

3 If the VA valve is being replaced, ensure that the vacuum source is off.

4 Remove the electrical connection by loosening the center screw.

5 Remove the solenoid by removing the four top screws.

6 Replace the new solenoid in reverse order.

7 Turn on the gas and vacuum supply, and power on the system.

8 Test the new valve for operation and for leaks.

10.4.8 Vacuum Pump

To replace the vacuum pump, proceed as follows.

1 Remove box AC power. If the vacuum pump is powered separately from the system, ensure 

that the power is removed.

2 Disconnect the clamp from the pipe.

3 Disconnect the AC line cord.

4 Remove the bolts that secure the pump to the box frame.

5 Remove the old pump.

6 Install the new pump in reverse order.

7 Ensure that the new pump has the proper oil and that the oil level is correct.

8 Power on the system and test for proper operation.
Removal and Replacement Procedures

Pure_Lab_2012-09-18.docx 75 of 82

10.5 Window

Replacing a window requires the box be exposed to open air, and later purged and possibly 

regenerated.

1 Stop box circulation by turning off the blower switch.

2 Remove all of the bolts from the window frame, and remove the frame.

3 Remove the window and discard properly.

4 Ensure that the window gasket is completely attached to the box frame and that it is free 

from all dirt or other material that would prevent a good seal.

5 Mount the new window. Ensure that the window is sitting above the window shims (clear 

plastic material along the bottom edge).

6 Position the window frame over the window and insert all corner bolts, but do not tighten 

them.

7 Insert the remaining bolts while ensuring the window is properly positioned.

8 Tighten all bolts.
Removal and Replacement Procedures

76 of 82 Pure_Lab_2012-09-18.docx

10.6 System Diagrams

10.6.1 Front View

Figure 63 Front view
Removal and Replacement Procedures

Pure_Lab_2012-09-18.docx 77 of 82

10.6.2 Back View

Figure 64 Back view
Removal and Replacement Procedures

78 of 82 Pure_Lab_2012-09-18.docx

10.6.3 Pure Lab HE 2GB and Pure Lab HE 4GB-2500 Overall Dimensions

 

Figure 65 Pure Lab HE 2GB
Removal and Replacement Procedures

Pure_Lab_2012-09-18.docx 79 of 82

Figure 66 Pure Lab HE 4GB-2500
Removal and Replacement Procedures

80 of 82 Pure_Lab_2012-09-18.docx

10.6.4 Pure Lab HE 3GB Overall Dimensions

Figure 67 Pure Lab HE 3GB
Removal and Replacement Procedures

Pure_Lab_2012-09-18.docx 81 of 82

10.6.5 Pure Lab HE 4GB-1800 Overall Dimensions

Figure 68 Pure Lab HE 4GB-1800
Removal and Replacement Procedures

82 of 82 Pure_Lab_2012-09-18.docx

10.6.6 Pure Lab HE 4GB-1950 Overall Dimensions

Figure 69 Pure Lab HE 4GB-1950