Handbook 44 – 2017 Section 5

5-1

Section 5

Table of Contents

Page 

5.50. Fabric-Measuring Devices ............................................................................................ 5-3 

5.51. Wire- and Cordage-Measuring Devices ........................................................................ 5-9 

5.52. Linear Measures.......................................................................................................... 5-15 

5.53. Odometers................................................................................................................... 5-19 

5.54. Taximeters................................................................................................................... 5-25 

5.55. Timing Devices........................................................................................................... 5-35 

5.56.(a) Grain Moisture Meters ................................................................................................ 5-43 

5.56.(b) Grain Moisture Meters ................................................................................................ 5-55 

5.57. Near-Infrared Grain Analyzers .................................................................................. 5-65 

5.58. Multiple Dimension Measuring Devices .................................................................... 5-75 

5.59. Electronic Livestock, Meat, and Poultry Evaluation Systems and/or Devices .......... 5-89 
Section 5 Handbook 44 – 2017

5-2

THIS PAGE INTENTIONALLY LEFT BLANK
Handbook 44 – 2017 5.50. Fabric-Measuring Devices

5-3

Table of Contents

Section 5.50. Fabric-Measuring Devices.............................................................................. 5-5

A. Application .................................................................................................................................................... 5-5

A.1. General..................................................................................................................................................5-5

A.2. Devices Used to Measure Other Similar Material in Sheet, Roll, or Bolt Form. ..................................5-5

A.3. Additional Code Requirements.............................................................................................................5-5

S. Specifications................................................................................................................................................. 5-5

S.1. Units......................................................................................................................................................5-5

S.2. Design of Indicating Elements..............................................................................................................5-5

S.2.1. Graduations.............................................................................................................................5-5

S.2.2. Indicator..................................................................................................................................5-5

S.2.3. Money-Value Computations...................................................................................................5-6

S.2.4. Return to Zero.........................................................................................................................5-6

S.3. Marking Requirements..........................................................................................................................5-6

S.4. Design Accuracy...................................................................................................................................5-6

N. Notes............................................................................................................................................................... 5-6

N.1. Testing Medium....................................................................................................................................5-6

T. Tolerances...................................................................................................................................................... 5-7

T.1. Tolerance Values. .................................................................................................................................5-7

UR. User Requirements ....................................................................................................................................... 5-7

UR.1. Installation Requirements. ....................................................................................................................5-7

UR.1.1. Installation. .............................................................................................................................5-7

UR.2. Use Requirements. ................................................................................................................................5-7

UR.2.1. Limitation of Use. ...................................................................................................................5-7

UR.2.2. Return of Indicating Elements to Zero....................................................................................5-7
5.50. Fabric-Measuring Devices Handbook 44 – 2017

5-4

THIS PAGE INTENTIONALLY LEFT BLANK
Handbook 44 – 2017 5.50. Fabric-Measuring Devices

5-5

Section 5.50. Fabric-Measuring Devices

A. Application

A.1. General. – This code applies only to mechanisms and machines designed to indicate automatically (with or 

without value-computing capabilities) the length of fabric passed through the measuring elements.

A.2. Devices Used to Measure Other Similar Material in Sheet, Roll, or Bolt Form. – Insofar as they are clearly 

appropriate, the requirements and provisions of this code apply also to devices designed for the commercial 

measurement of other material similar to fabrics, in sheet, roll, or bolt form.

A.3. Additional Code Requirements. – In addition to the requirements of this code, Fabric-Measuring Devices 

shall meet the requirements of Section 1.10. General Code.

S. Specifications

S.1. Units. – A fabric-measuring device shall indicate lengths as follows:

(a) For devices indicating in metric units, lengths shall be indicated in terms of 5 cm; 10 cm; 25 cm; or 50 cm 

and meters.

In addition, lengths may be indicated in terms of any or all of the following subdivisions: millimeters and 

centimeters.

(b) For devices indicating in U.S. customary units, lengths shall be indicated in terms of 1/8 yd; 1⁄4 yd; or 1⁄2 yd; 

and yards.

In addition, lengths may be indicated in terms of any or all of the following subdivisions: 1/8 yd; 1

/16 yd;

inches and feet.

Digital indicators may indicate values in decimal fractions.

(Amended 1977)

S.2. Design of Indicating Elements.

S.2.1. Graduations.

S.2.1.1. Length. – Graduations shall be so varied in length that they may be conveniently read.

S.2.1.2. Width. – In any series of graduations, the width of a graduation shall in no case be greater than 

the width of the minimum clear interval between graduations, and the width of main graduations shall be not 

more than 50 % greater than the width of subordinate graduations. Graduations shall in no case be less than 

0.2 mm (0.008 in) in width.

S.2.1.3. Clear Interval between Graduations. – The clear interval between graduations shall be at least 

6 mm for cm graduations (1⁄4 in for 1

/8 yd graduations), and 3 mm for 20 cm graduations (1/8 in for 1 in 

graduations).

S.2.2. Indicator.

S.2.2.1. Symmetry. – The index of an indicator shall be symmetrical with respect to the graduations, at 

least throughout that portion of its length associated with the graduations.
5.50. Fabric-Measuring Devices Handbook 44 – 2017

5-6

S.2.2.2. Length. – The index of an indicator shall reach to the finest graduations with which it is used, 

unless the indicator and the graduations are in the same plane, in which case the distance between the end of 

the indicator and the ends of the graduations, measured along the line of the graduations, shall be not more 

than 1.0 mm (0.04 in).

S.2.2.3. Width. – The index of an indicator shall not be wider than the narrowest graduations with which 

it is used, and shall in no case exceed 0.4 mm (0.015 in).

S.2.2.4. Clearance. – The clearance between the index of an indicator and the graduations shall in no case 

be more than 1.5 mm (0.06 in).

S.2.2.5. Parallax. – Parallax effects shall be reduced to the practicable minimum.

S.2.3. Money-Value Computations.

S.2.3.1. Full-Computing Type. – In this type, the money value at each of a series of unit prices shall be 

computed automatically for every length within the range of measurement of the fabric-measuring device. 

Value graduations shall be provided and shall be accurately positioned. The value of each graduated interval 

shall be 1 cent at all prices per yard of 30 cents and less, and shall not exceed 2 cents at higher prices per 

yard. Five-cent intervals may be represented in the two-cent range by special graduations, but these shall not 

be positioned in the clear intervals between graduations of the regular series.

S.2.3.2. Limited-Computing Type. – In this type, the money value at each of a series of unit prices shall 

be computed automatically only for lengths corresponding to a definite series of length graduations. There 

shall be no value graduations. At no position that the chart can assume shall two value figures at the same 

price per yard be completely and clearly exposed to view at one time. Money values shown shall be 

mathematically accurate, except that a fraction of less than 1⁄2 cent shall be dropped and the next higher cent 

shall be shown in the case of a fraction of 1⁄2 cent or more. One of the following requirements shall be met:

(a) There shall be a money-value computation for each length graduation within the range of 

measurement of the device.

(b) No money-value computation shall be exposed to view except at such times as the device shows a 

length indication for which a corresponding series of value indications is computed.

(c) Each column or row of money-value computations shall be marked to show the length to which the 

computations correspond, the device shall be marked to show the character and limitations of the 

computations, and there shall be computations corresponding to at least 10 cm (1

/8 yd) throughout 

the range of measurement of the device.

S.2.4. Return to Zero. – Primary indicating elements shall be readily returnable to a definite zero indication. 

Means shall be provided to prevent the return of the indicating elements beyond their correct zero positions.

S.3. Marking Requirements. – If a device will not accurately measure all fabrics, it shall be marked to indicate 

clearly its limitations.

S.4. Design Accuracy. – Indications of length and money value shall be accurate whether the values of the 

indications are being increased or decreased.

N. Notes

N.1. Testing Medium. – A fabric-measuring device shall be tested with a suitable testing tape approximately 

7.62 cm (3 in) wide and with a graduated length of at least 11 m (12 yd), made from such material and having such 

surface finish as to provide dimensional stability and reduce slippage to the practicable minimum.
Handbook 44 – 2017 5.50. Fabric-Measuring Devices

5-7

T. Tolerances

T.1. Tolerance Values. – Maintenance and acceptance tolerances shall be as shown in Table 1. Maintenance and 

Acceptance Tolerances for Fabric-Measuring Devices.

Table 1.

Maintenance and Acceptance Tolerances for Fabric-Measuring Devices

Indication of

Device

(yards)

Maintenance Tolerance Acceptance Tolerance

On Underregistration

(inches)

On Overregistration

(inches)

On Underregistration

(inches)

On Overregistration

(inches)

2 or less 3/8 1⁄4 1⁄4 1/8

3 3/8 5/16 1⁄4 5/32

4 1⁄2 5/16 1⁄4 5/32

5 5/8 3/8 5/16 3/16

6 3⁄4 3/8 3/8 3/16

7 and 8 1 1⁄2 1⁄2 1⁄4

9 11⁄4 5/8 5/8 5/16

10 and 11 11⁄2 3⁄4 3⁄4 3/8

12 and 13 13⁄4 7/8 7/8 7/16

14 and 15 2 1 1 1⁄2

Over 15 Add 1

/8 inch

per indicated yard

Add 1/16 inch

per indicated yard

Add 1/16 inch

per indicated yard

Add 1/32 inch

per indicated yard

UR. User Requirements

UR.1. Installation Requirements.

UR.1.1. Installation. – A fabric-measuring device shall be securely supported and firmly fixed in position.

UR.2. Use Requirements.

UR.2.1. Limitation of Use. – A fabric-measuring device shall be used to measure only those fabrics that it was 

designed to measure, and in no case shall it be used to measure a fabric that a marking on the device indicates 

should not be measured.

UR.2.2. Return of Indicating Elements to Zero. – The primary indicating elements shall be returned to zero 

before each measurement.
5.50. Fabric-Measuring Devices Handbook 44 – 2017

5-8

THIS PAGE INTENTIONALLY LEFT BLANK