Handbook 44 – 2011 3.39. Hydrogen Gas-Measuring Devices – Tentative Code

3-127

Table of Contents

Section 3.39. Hydrogen Gas-Measuring Devices – Tentative Code ................................. 129

A. Application ....................................................................................................................................................129

A.1. General.................................................................................................................................................129

A.2. Exceptions............................................................................................................................................129

A.3. Additional Code Requirements............................................................................................................129

A.4. Type Evaluation...................................................................................................................................129

S. Specifications.................................................................................................................................................129

S.1. Indicating and Recording Elements. ....................................................................................................129

S.1.1. Indicating Elements ...............................................................................................................129

S.1.2. Vehicle Fuel Dispensers ........................................................................................................129

S.1.3. Units.......................................................................................................................................129

S.1.4. Value of Smallest Unit...........................................................................................................130

S.2. Operating Requirements. .....................................................................................................................130

S.2.1. Return to Zero........................................................................................................................130

S.2.2. Indicator Reset Mechanism....................................................................................................130

S.2.3. Provision for Power Loss.......................................................................................................130

S.2.4. Display of Unit Price and Product Identity. ...........................................................................130

S.2.5. Money-Value Computations..................................................................................................131

S.2.6. Recorded Representations, Point of Sale Systems. ................................................................131

S.2.7. Indication of Delivery............................................................................................................131

S.3. Design of Measuring Elements and Measuring Systems. ....................................................................131

S.3.1. Maximum and Minimum Flow-Rates....................................................................................131

S.3.2. Adjustment Means. ................................................................................................................131

S.3.3. Provision for Sealing..............................................................................................................131

S.3.4. Automatic Density Correction. ..............................................................................................132

S.3.5. Pressurizing the Discharge Hose............................................................................................132

S.3.6. Zero-Set-Back Interlock, Retail Vehicle Fuel Devices..........................................................132

S.4. Discharge Lines and Valves.................................................................................................................133

S.4.1. Diversion of Measured Product. ............................................................................................133

S.4.2. Directional Flow Valves. .......................................................................................................133

S.4.3. Other Valves. .........................................................................................................................133

S.5. Markings. .............................................................................................................................................133

S.5.1. Location of Marking Information; Hydrogen-Fuel Dispensers. ............................................133

S.6. Printer. .................................................................................................................................................134

S.6.1. Printed Receipt.......................................................................................................................134

S.7. Totalizers for Vehicle Fuel Dispensers................................................................................................134

S.8. Minimum Measured Quantity..............................................................................................................134

N. Notes...............................................................................................................................................................134

N.1. Minimum Measured Quantity..............................................................................................................134

N.2. Test Medium........................................................................................................................................134

N.3. Test Drafts. ..........................................................................................................................................134

N.4. Tests.....................................................................................................................................................134

N.4.1. Master Meter (Transfer) Standard Test..................................................................................134

N.4.2. Gravimetric Tests...................................................................................................................135

N.4.3. PVT Pressure Volume Temperature Test. .............................................................................135

N.5. Minimum Measured Quantity..............................................................................................................135

N.6. Testing Procedures...............................................................................................................................135
3.39. Hydrogen Gas-Measuring Devices – Tentative Code Handbook 44 – 2011

3-128

N.6.1. General...................................................................................................................................135

N.7. Density.................................................................................................................................................135

T. Tolerances......................................................................................................................................................135

T.1. Tolerances, General. ............................................................................................................................135

T.2. Tolerances............................................................................................................................................135

T.3. Repeatability. .......................................................................................................................................136

T.4. Tolerance Application on Test Using Transfer Standard Test Method................................................136

T.5. Tolerance Application in Type Evaluation Examinations for Devices. ...............................................136

UR. User Requirements .......................................................................................................................................136

UR.1. Selection Requirements. ......................................................................................................................136

UR.1.1. Computing-Type Device; Retail Dispenser. ..........................................................................136

UR.1.2. Discharge Hose-Length. ........................................................................................................136

UR.1.3. Minimum Measured Quantity................................................................................................136

UR.2. Installation Requirements. ...................................................................................................................137

UR.2.1. Manufacturer’s Instructions. ..................................................................................................137

UR.2.2. Discharge Rate.......................................................................................................................137

UR.2.3. Low-Flow Cut-Off Valve. .....................................................................................................137

UR.3. Use of Device.......................................................................................................................................137

UR.3.1. Unit Price and Product Identity for Retail Dispensers. ..........................................................137

UR.3.2. Vehicle-mounted Measuring Systems Ticket Printer.............................................................137

UR.3.3. Printed Ticket.........................................................................................................................137

UR.3.4. Steps After Dispensing. .........................................................................................................137

UR.3.5. Return of Indicating and Recording Elements to Zero. .........................................................138

UR.3.6. Return of Product to Storage, Retail Hydrogen Gas Dispensers............................................138

UR.3.7. Conversion Factors. ...............................................................................................................138

Appendix D. Definitions.........................................................................................................................................138

audit trail.........................................................................................................................................................138

automatic temperature or density compensation.............................................................................................138

calibration parameter ......................................................................................................................................138

discharge hose.................................................................................................................................................138

discharge line..................................................................................................................................................138

event counter...................................................................................................................................................138

event logger. ...................................................................................................................................................138

indicating element...........................................................................................................................................139

minimum measured quantity (MMQ).............................................................................................................139

nonresettable totalizer.....................................................................................................................................139

point-of-sale system........................................................................................................................................139

remote configuration capability. .....................................................................................................................139

retail device.....................................................................................................................................................139

wet hose. .........................................................................................................................................................139

wet-hose type..................................................................................................................................................139
Handbook 44 – 2011 3.39. Hydrogen Gas-Measuring Devices – Tentative Code

3-129

Section 3.39. Hydrogen Gas-Measuring Devices – Tentative Code

This tentative code has only a trial or experimental status and is not intended to be enforced. The requirements are 

designed for study prior to the development and adoption of a final code. Requirements that apply to wholesale 

applications are under study and development by the U.S. National Work Group for the Development of 

Commercial Hydrogen Measurement Standards. Officials wanting to conduct an official examination of a device or 

system are advised to see paragraph G-A.3. Special and Unclassified Equipment.

(Tentative Code Added 2010)

A. Application

A.1. General. – This code applies to devices that are used for the measurement of hydrogen gas in the vapor state 

used as a vehicle fuel.

A.2. Exceptions. – This code does not apply to:

(a) Devices used solely for dispensing a product in connection with operations in which the amount dispensed 

does not affect customer charges.

(b) The wholesale delivery of hydrogen gas.

(c) Devices used for dispensing a hydrogen gas with a hydrogen fuel index lower than 99.97 % and 

concentrations of specified impurities that exceed level limits.

(d) Systems that measure pressure, volume, and temperature with a calculating device to determine the mass of 

gas accumulated in or discharged from a tank of known volume.

A.3. Additional Code Requirements. – In addition to the requirements of this code, Hydrogen Gas-Measuring 

Devices shall meet the requirements of Section 1.10. General Code.

A.4. Type Evaluation. – The National Type Evaluation Program (NTEP) will accept for type evaluation only 

those devices that comply with all requirements of this code.

S. Specifications

S.1. Indicating and Recording Elements.

S.1.1. Indicating Elements. – A measuring assembly shall include an indicating element that continuously 

displays measurement results relative to quantity and total price. Indications shall be clear, definite, accurate, and 

easily read under normal conditions of operation of the device.

S.1.2. Vehicle Fuel Dispensers. – A hydrogen gas dispenser used to fuel vehicles shall be of the computing 

type and shall indicate the mass, the unit price, and the total price of each delivery.

S.1.3. Units.

S.1.3.1. Units of Measurement. – Deliveries shall be indicated and recorded in kilograms and decimal

subdivisions thereof.

S.1.3.2. Numerical Value of Quantity-Value Divisions. – The value of an interval (i.e., increment or

scale division) shall be equal to:

(a) 1, 2, or 5; or

(b) a decimal multiple of submultiple of 1, 2, or 5.
3.39. Hydrogen Gas-Measuring Devices – Tentative Code Handbook 44 – 2011

3-130

Examples: quantity-value divisions may be 10, 20, 50, 100; or 0.01, 0.02, 0.05; or 0.1, 0.2, or 0.5 

etc.

S.1.3.3. Maximum Value of Quantity-Value Divisions. – The maximum value of the quantity-value 

division shall be not greater than 0.5% of the minimum measured quantity.

S.1.3.4. Values Defined. – Indicated values shall be adequately defined by a sufficient number of 

figures, words, symbols, or combinations thereof. A display of “zero” shall be a zero digit for all displayed 

digits to the right of the decimal mark and at least one to the left.

S.1.4. Value of Smallest Unit. – The value of the smallest unit of indicated delivery, and recorded delivery if 

the device is equipped to record, shall not exceed the equivalent of: 

(a) 0.001 kg on devices with a marked maximum flow rated of 30 kg/min or less; or

(b) 0.01 kg on devices with a marked maximum flow rate of more than 30 kg/min.

S.2. Operating Requirements.

S.2.1. Return to Zero. 

(a) The primary indicating and the primary recording elements, if the device is equipped to record, shall be 

provided with a means for readily returning the indication to zero either automatically or manually.

(b) It shall not be possible to return primary indicating elements, or primary recording elements, beyond 

the correct zero position.

S.2.2. Indicator Reset Mechanism. – The reset mechanism for the indicating element shall not be operable 

during a delivery. Once the zeroing operation has begun, it shall not be possible to indicate a value other than 

the latest measurement, or “zeros” when the zeroing operation has been completed.

S.2.3. Provision for Power Loss. 

S.2.3.1. Transaction Information. – In the event of a power loss, the information needed to complete 

any transaction in progress at the time of the power loss (such as the quantity and unit price, or sales price) 

shall be determinable for at least 15 minutes at the dispenser or at the console if the console is accessible to 

the customer. 

S.2.3.2. User Information. – The device memory shall retain information on the quantity of fuel 

dispensed and the sales price totals during power loss.

S.2.4. Display of Unit Price and Product Identity. 

S.2.4.1. Unit Price. – A computing or money-operated device shall be able to display on each face the 

unit price at which the device is set to compute or to dispense.

S.2.4.2. Product Identity. – A device shall be able to conspicuously display on each side the identity of 

the product being dispensed.

S.2.4.3. Selection of Unit Price. – When a product is offered for sale at more than on unit price through 

a computing device, the selection of the unit price shall be made prior to delivery using controls on the 

device or other customer-activated controls. A system shall not permit a change to the unit price during 

delivery of a product.
Handbook 44 – 2011 3.39. Hydrogen Gas-Measuring Devices – Tentative Code

3-131

S.2.4.4. Agreement Between Indications. – All quantity, unit price, and total price indications within a 

measuring system shall agree for each transaction.

S.2.5. Money-Value Computations. – A computing device shall compute the total sales price at any singlepurchase unit price for which the product being measured is offered for sale at any delivery possible within 

either the measurement range of the device or the range of the computing elements, whichever is less.

S.2.5.1. Auxiliary Elements. – If a system is equipped with auxiliary indications, all indicated money 

value and quantity divisions of the auxiliary element shall be identical with those of the primary element.

S.2.5.2. Display of Quantity and Total Price. – When a delivery is completed, the total price and 

quantity for that transaction shall be displayed on the face of the dispenser for at least 5 minutes or until the 

next transaction is initiated by using controls on the device or other user-activated controls. 

S.2.6. Recorded Representations, Point of Sale Systems. – A printed receipt shall be available through a 

built-in or separate recording element for transactions conducted with point-or-sale systems or devices activated 

by debit cards, credit cards, and/or cash. The printed receipt shall contain the following information for products 

delivered by the dispenser:

(a) the total mass of the delivery;

(b) the unit price;

(c) the total computed price; and

(d) the product identity by name, symbol, abbreviation, or code number.

S.2.7. Indication of Delivery. – The device shall automatically show on its face the initial zero condition and 

the quantity delivered (up to the nominal capacity).

S.3. Design of Measuring Elements and Measuring Systems.

S.3.1. Maximum and Minimum Flow-Rates. – The ratio of the maximum to minimum flow-rates specified 

by the manufacturer for devices measuring gases shall be 10:1 or greater.

S.3.2. Adjustment Means. – An assembly shall be provided with means to change the ratio between the 

indicated quantity and the quantity of gas measured by the assembly. A bypass on the measuring assembly 

shall not be used for these means.

S.3.2.1. Discontinuous Adjusting Means. – When the adjusting means changes ratio between the 

indicated quantity and the quantity of measured gas in a discontinuous manner, the consecutive values of 

the ratio shall not differ by more than 0.1 %.

S.3.3. Provision for Sealing. – Adequate provision shall be made for an approved means of security (e.g., 

data change audit trail) or physically applying security seals in such a manner that no adjustment may be made 

of:

(a) each individual measurement element;

(b) any adjustable element for controlling delivery rate when such rate tends to affect the accuracy of 

deliveries;

(c) the zero adjustment mechanism; and

(d) any metrological parameter that detrimentally affects the metrological integrity of the device or 

system.
3.39. Hydrogen Gas-Measuring Devices – Tentative Code Handbook 44 – 2011

3-132

When applicable, the adjusting mechanism shall be readily accessible for purposes of affixing a security seal. Audit 

trails shall use the format set forth in Table S.3.3. Categories of Device and Methods of Sealing.

Table S.3.3.

Categories of Device and Methods of Sealing

Categories of Device Method of Sealing

Category 1: No remote configuration 

capability.

Seal by physical seal or two event counters: one for calibration 

parameters and one for configuration parameters.

Category 2: Remote configuration capability, 

but access is controlled by physical hardware. 

The device shall clearly indicate that it is in 

the remote configuration mode and record 

such message if capable of printing in this 

mode or shall not operate while in this mode.

The hardware enabling access for remote communication must be 

on-site. The hardware must be sealed using a physical seal or an 

event counter for calibration parameters and an event counter for 

configuration parameters. The event counters may be located 

either at the individual measuring device or at the system 

controller; however, an adequate number of counters must be 

provided to monitor the calibration and configuration parameters 

of the individual devices at a location. If the counters are located 

in the system controller rather than at the individual device, means 

must be provided to generate a hard copy of the information 

through an on-site device.

Category 3: Remote configuration capability 

access may be unlimited or controlled through 

a software switch (e.g., password).

The device shall clearly indicate that it is in 

the remote configuration mode and record 

such message if capable of printing in this 

mode or shall not operate while in this mode.

An event logger is required in the device; it must include an event 

counter (000 to 999), the parameter ID, the date and time of the 

change, and the new value of the parameter. A printed copy of the 

information must be available through the device or through 

another on-site device. The event logger shall have a capacity to 

retain records equal to 10 times the number of sealable parameters 

in the device, but not more than 1000 records are required. (Note: 

Does not require 1000 changes to be stored for each parameter.)

S.3.4. Automatic Density Correction.

(a) An automatic means to determine and correct for changes in product density shall be incorporated in 

any hydrogen gas-measuring system where measurements are affected by changes in the density of the 

product being measured.

(b) Volume-measuring devices with automatic temperature compensation used to measure hydrogen gas 

as a vehicle fuel shall be equipped with an automatic means to determine and correct for changes in 

product density due to changes in the temperature, pressure, and composition of the product. 

S.3.5. Pressurizing the Discharge Hose. – The discharge hose for hydrogen gas shall automatically 

pressurize to a pressure equal to or greater than the receiving vessel prior to the device beginning to register the 

delivery. The indications shall not advance as a result of the initial pressurization or the purging/bleeding of the 

discharge hose.

S.3.6. Zero-Set-Back Interlock, Retail Vehicle Fuel Devices. 

(a) A device shall be constructed so that:

(1) when the device is shut-off at the end of a delivery an automatic interlock prevents a subsequent 

delivery until the indicating element and recording elements, if the device is equipped and activated to record, have been returned to their zero positions; and

(2) it shall not be possible to return the discharge nozzle to its start position unless the zero set back 

interlock is engaged or becomes engaged.
Handbook 44 – 2011 3.39. Hydrogen Gas-Measuring Devices – Tentative Code

3-133

(b) For systems with more than one:

(1) dispenser supplied by a single measuring element, an effective automatic control valve in each 

dispenser prevents product from being delivered until the indicating elements on that dispenser are 

in a correct zero position; or 

(2) hose supplied by a single measuring element, effective automatic means must be provided to 

prevent product from being delivered until the indicating element(s) corresponding to each hose 

are in a correct zero position.

S.4. Discharge Lines and Valves.

S.4.1. Diversion of Measured Product. – No means shall be provided by which any measured product can 

be diverted from the measuring device. 

S.4.2. Directional Flow Valves. – If a reversal of flow could result in errors that exceed the tolerance for the 

minimum measured quantity, a valve or valves or other effective means, automatic in operation (and equipped 

with a pressure limiting device, if necessary) to prevent the reversal of flow shall be properly installed in the 

system. (See N.1. Minimum Measured Quantity)

S.4.3. Other Valves. – Check valves and closing mechanisms that are not used to define the measured 

quantity shall have relief valves (if necessary) to dissipate any abnormally high pressure that may arise in the 

measuring assembly.

S.5. Markings. – A measuring system shall be conspicuously, legibly, and indelibly marked with the following 

information:

(a) pattern approval mark (i.e., type approval number);

(b) name and address of the manufacturer or his trademark and, if required by the weights and measures 

authority, the manufacturer's identification mark in addition to the trademark;

(c) model designation or product name selected by the manufacturer;

(d) nonrepetitive serial number;

(e) the accuracy class of the device as specified by the manufacturer consistent with Table T.2. Accuracy 

Classes and Tolerances for Hydrogen-Gas Measuring Devices;

(f) maximum and minimum flow rates in kilograms per unit of time;

(g) maximum working pressure;

(h) applicable range of ambient temperature if other than –10 °C to + 50 °C;

(i) minimum measured quantity; and

(j) product limitations (such as fuel quality), if applicable.

S.5.1. Location of Marking Information; Hydrogen-Fuel Dispensers. – The marking information required 

in General Code, paragraph G S.1. Identification shall appear as follows:

(a) within 60 cm (24 in) to 150 cm (60 in) from the base of the dispenser;

(b) either internally and/or externally provided the information is permanent and easily read; and 

accessible for inspection; and
3.39. Hydrogen Gas-Measuring Devices – Tentative Code Handbook 44 – 2011

3-134

(c) on a portion of the device that cannot be readily removed or interchanged (i.e., not on a service access 

panel).

Note: The use of a dispenser key or tool to access internal marking information is permitted for retail hydrogen-measuring 

devices.

S.6. Printer. – When an assembly is equipped with means for printing the measured quantity, the printed 

information must agree with the indications on the dispenser for the transaction and the printed values shall be 

clearly defined.

S.6.1. Printed Receipt. – Any delivered, printed quantity shall include an identification number, the time and 

date, and the name of the seller. This information may be printed by the device or pre-printed on the ticket.

S.7. Totalizers for Vehicle Fuel Dispensers. – Vehicle fuel dispensers shall be equipped with a nonresettable 

totalizer for the quantity delivered through each separate measuring device.

S.8. Minimum Measured Quantity. – The minimum measured quantity shall satisfy the conditions of use of the 

measuring system as follows:

(a) Measuring systems having a maximum flow rate less than or equal to 4 kg/min shall have a minimum 

measured quantity not exceeding 0.5 kg.

(b) Measuring systems having a maximum flow rate greater than 4 kg/min but not greater than 12 kg/min shall 

have a minimum measured quantity not exceeding 1.0 kg.

N. Notes

N.1. Minimum Measured Quantity. – The minimum measured quantity shall be specified by the manufacturer.

N.2. Test Medium. – The device shall be tested with the product commercially measured except that, in a type 

evaluation examination, hydrogen gas as specified in NIST Handbook 130 shall be used.

Note: Corresponding requirements are under development and this paragraph will be revisited.

N.3. Test Drafts. – The minimum test shall be one test draft at the declared minimum measured quantity and one 

test draft at approximately ten times the minimum measured quantity or 1 kg, whichever is greater. More tests may 

be performed over the range of normal quantities dispensed. (See T.3. Repeatability)

The test draft shall be made at flows representative of that during normal delivery. The pressure drop between the 

dispenser and the proving system shall not be greater than that for normal deliveries. The control of the flow (e.g., 

pipework or valve(s) size, etc.) shall be such that the flow of the measuring system is maintained within the range 

specified by the manufacturer.

N.4. Tests.

N.4.1. Master Meter (Transfer) Standard Test. – When comparing a measuring system with a calibrated 

transfer standard, the minimum test shall be one test draft at the declared minimum measured quantity and one 

test draft at approximately ten times the minimum measured quantity or 1 kg, whichever is greater. More tests 

may be performed over the range of normal quantities dispensed.

N.4.1.1. Verification of Master Metering Systems. – A master metering system used to verify a 

hydrogen gas-measuring device shall be verified before and after the verification process. A master 

metering system used to calibrate a hydrogen gas-measuring device shall be verified before starting the 

calibration and after the calibration process.
Handbook 44 – 2011 3.39. Hydrogen Gas-Measuring Devices – Tentative Code

3-135

N.4.2. Gravimetric Tests. – The weight of the test drafts shall be equal to at least the amount delivered by 

the device at the declared minimum measured quantity and one test draft at approximately ten times the 

minimum measured quantity or 1 kg, whichever is greater. More tests may be performed over the range of 

normal quantities dispensed. 

N.4.3. PVT Pressure Volume Temperature Test. – The minimum test with a calibrated volumetric standard 

shall be one test draft at the declared minimum measured quantity and one test draft at approximately ten times 

the minimum measured quantity or 1 kg, whichever is greater. More tests may be performed over the range of 

normal quantities dispensed.

N.5. Minimum Measured Quantity. – The device shall be tested for a delivery equal to the declared minimum 

measured quantity when the device is likely to be used to make deliveries on the order of the declared minimum 

measured quantity.

N.6. Testing Procedures.

N.6.1. General. – The device or system shall be tested under normal operating conditions of the dispenser.

The test draft shall be made at flows representative of that during normal delivery. The pressure drop between 

the dispenser and the proving system shall not be greater than that for normal deliveries. The control of the 

flow (e.g., pipework or valve(s) size, etc.) shall be such that the flow of the measuring system is maintained 

within the range specified by the manufacturer.

N.6.1.1. Repeatability Tests. – Tests for repeatability should include a minimum of three consecutive 

test drafts of approximately the same size and be conducted under controlled conditions where variations in 

factors are reduced to minimize the effect on the results obtained.

N.7. Density. – Temperature and pressure of hydrogen gas shall be measured during the test for the determination 

of density or volume correction factors when applicable. For the thermophysical properties of hydrogen the 

following publications shall apply: for density calculations at temperatures above 255 K and pressures up to 120 

MPa, a simple relationship may be used that is given in the publication of Lemmon et al., J. Res. NIST, 2008. 

Calculations for a wider range of conditions and additional thermophysical properties of hydrogen are available free 

of charge online at the “NIST Chemistry WebBook” http://webbook.nist.gov/chemistry, or available for purchase 

from NIST as the computer program NIST Standard Reference Database 23 “NIST Reference Fluid 

Thermodynamic and Transport Properties Database (REFPROP): Version 8.0” http://www.nist.gov/srd/nist23.htm. 

These calculations are based on the reference Leachman, J.W., Jacobsen, R.T, Lemmon, E.W., and Penoncello, S.G. 

“Fundamental Equations of State for Parahydrogen, Normal Hydrogen, and Orthohydrogen" to be published in the 

Journal of Physical and Chemical Reference Data. More information may be obtained from NIST online at 

http://www.boulder.nist.gov/div838/Hydrogen/Index.htm.

T. Tolerances

T.1. Tolerances, General.

(a) The tolerances apply equally to errors of underregistration and errors of overregistration.

(b) The tolerances apply to all products at all temperatures measured at any flow rate within the rated 

measuring range of the device.

T.2. Tolerances. – The tolerances for hydrogen gas measuring devices are listed in Table T.2. Accuracy Classes 

and Tolerances for Hydrogen Gas-Measuring Devices. (Proposed tolerance values are based on previous work with 

compressed gas products and will be confirmed based on performance data evaluated by the U.S. National Work 

Group.)
3.39. Hydrogen Gas-Measuring Devices – Tentative Code Handbook 44 – 2011

3-136

Table T.2.

Accuracy Classes and Tolerances for Hydrogen Gas-Measuring Devices

Accuracy Class Application or Commodity Being 

Measured

Acceptance 

Tolerance

Maintenance 

Tolerance

2.0 Hydrogen gas as a vehicle fuel 1.5 % 2.0 %

T.3. Repeatability. – When multiple tests are conducted at approximately the same flow rate and draft size, the 

range of the test results for the flow rate shall not exceed 40 % of the absolute value of the maintenance tolerance

and the results of each test shall be within the applicable tolerance. See also N.6.1.1. Repeatability Tests.

T.4. Tolerance Application on Test Using Transfer Standard Test Method. – To the basic tolerance values 

that would otherwise be applied, there shall be added an amount equal to two times the standard deviation of the 

applicable transfer standard when compared to a basic reference standard.

T.5. Tolerance Application in Type Evaluation Examinations for Devices. – For type evaluation examinations,

the tolerance values shall apply under the following conditions:

(a) at any temperature and pressure within the operating range of the device, and

(b) for all quantities greater than the minimum measured quantity.

UR. User Requirements

UR.1. Selection Requirements.

UR.1.1. Computing-Type Device; Retail Dispenser. – A hydrogen gas dispenser used to refuel vehicles 

shall be of the computing type and shall indicate the mass, the unit price, and the total price of each delivery.

UR.1.2. Discharge Hose-Length. – The length of the discharge hose on a retail fuel dispenser:

(a) shall not exceed 4.6 m (15 ft) unless it can be demonstrated that a longer hose is essential to permit 

deliveries to be made to receiving vehicles or vessels; 

(b) shall be measured from its housing or outlet of the discharge line to the inlet of the discharge nozzle; 

and

(c) shall be measured with the hose fully extended if it is coiled or otherwise retained or connected inside 

a housing.

An unnecessarily remote location of a device shall not be accepted as justification for an abnormally long hose.

UR.1.3. Minimum Measured Quantity.

(a) The minimum measured quantity shall be specified by the manufacturer. 

(b) The minimum measured quantity appropriate for a transaction may be specified by the weights and 

measures authority. A device may have a declared minimum measured quantity smaller than that 

specified by the weights and measures authority; however, the device must perform within the 

performance requirements for the declared or specified minimum measured quantity up to deliveries at 

the maximum measurement range.
Handbook 44 – 2011 3.39. Hydrogen Gas-Measuring Devices – Tentative Code

3-137

(c) The minimum measured quantity shall satisfy the conditions of use of the measuring system as 

follows:

(1) Measuring systems having a maximum flow rate less than or equal to 4 kg/min shall have a 

minimum measured quantity not exceeding 0.5 kg

(2) Measuring systems having a maximum flow rate greater than 4 kg/min but not greater than 

12 kg/min shall have a minimum measured quantity not exceeding 1.0 kg

UR.2. Installation Requirements.

UR.2.1. Manufacturer’s Instructions. – A device shall be installed in accordance with the manufacturer’s 

instructions, and the installation shall be sufficiently secure and rigid to maintain this condition.

UR.2.2. Discharge Rate. – A device shall be installed so that after initial equalization the actual maximum 

discharge rate will not exceed the rated maximum discharge rate. Automatic means of flow regulation shall be 

incorporated in the installation if necessary.

UR.2.3. Low-Flow Cut-Off Valve. – If a measuring system is equipped with a programmable or adjustable 

"low-flow cut-off" feature:

(a) the low-flow cut-off value shall not be set at flow rates lower than the minimum operating flow rate 

specified by the manufacturer on the measuring device; and

(b) the system shall be equipped with flow control valves which prevent the flow of product and stop the 

indicator from registering product flow whenever the product flow rate is less than the low-flow cutoff value.

UR.3. Use of Device.

UR.3.1. Unit Price and Product Identity for Retail Dispensers. – The unit price at which the dispenser is 

set to compute shall be conspicuously displayed or posted on the face of a retail dispenser used in direct sale.

UR.3.2. Vehicle-mounted Measuring Systems Ticket Printer.

UR.3.2.1. Customer Ticket. – Vehicle-mounted measuring systems shall be equipped with a ticket printer 

which shall be used for all sales where product is delivered through the device. A copy of the ticket issued 

by the device shall be left with the customer at the time of delivery or as otherwise specified by the 

customer.

UR.3.2.2. Ticket in Printing Device. – A ticket shall not be inserted into a device equipped with a ticket 

printer until immediately before a delivery is begun, and in no case shall a ticket be in the device when the 

vehicle is in motion while on a public street, highway, or thoroughfare.

UR.3.3. Printed Ticket. – The total price, the total quantity of the delivery, and the price per unit shall be 

printed on any ticket issued by a device of the computing type and containing any one of these values.

UR.3.4. Steps After Dispensing. – After delivery to a customer from a retail dispenser:

(a) the device shall be shut-off at the end of a delivery, through an automatic interlock that prevents a 

subsequent delivery until the indicating elements and recording elements, if the device is equipped and 

activated to record, have been returned to their zero positions; and

(b) the discharge nozzle shall not be returned to its start position unless the zero set-back interlock is 

engaged or becomes engaged by the act of disconnecting the nozzle or the act of returning the 

discharge nozzle. 
3.39. Hydrogen Gas-Measuring Devices – Tentative Code Handbook 44 – 2011

3-138

UR.3.5. Return of Indicating and Recording Elements to Zero. – The primary indicating elements 

(visual), and the primary recording elements shall be returned to zero immediately before each delivery. 

UR.3.6. Return of Product to Storage, Retail Hydrogen Gas Dispensers. – Provisions at the site shall be 

made for returning product to storage or disposing of the product in a safe and timely manner during or 

following testing operations. Such provisions may include return lines, or cylinders adequate in size and 

number to permit this procedure.

UR.3.7. Conversion Factors. – Established correction values (see references in N.7. Density.) shall be used 

whenever measured hydrogen gas is billed. All sales shall be based on kilograms.

Appendix D. Definitions 

The specific code to which the definition applies is shown in [brackets] at the end of the definition. Definitions for 

the General Code [1.10] apply to all codes in Handbook 44.

A

audit trail. – An electronic count and/or information record of the changes to the values of the calibration or 

configuration parameters of a device.[1.10, 2.20, 2.21, 2.24, 3.30, 3.37, 3.39, 5.56(a)] 

automatic temperature or density compensation. – The use of integrated or ancillary equipment to obtain from 

the output of a volumetric meter an equivalent mass, or an equivalent liquid volume at the assigned reference 

temperature below and a pressure of 14.696 lb/in2 absolute. 

Cryogenic liquids 21 °C (70 °F) [3.34]

Hydrocarbon gas vapor 15 °C (60 °F) [3.33]

Hydrogen gas 21 °C (70 °F) [3.39]

Liquid carbon dioxide 21 °C (70 °F) [3.38]

Liquefied petroleum gas (LPG) and Anhydrous ammonia 15 °C (60 °F) [3.32]

Petroleum liquid fuels and lubricants 15 °C (60 °F) [3.30]

C

calibration parameter. – Any adjustable parameter that can affect measurement or performance accuracy and, due 

to its nature, needs to be updated on an ongoing basis to maintain device accuracy (e.g., span adjustments, 

linearization factors, and coarse zero adjustments).[2.20, 2.21, 2.24, 3.30, 3.37, 3.39, 5.56(a)] 

D

discharge hose. – A flexible hose connected to the discharge outlet of a measuring device or its discharge line.[3.30, 

3.31, 3.32, 3.34, 3.37, 3.38, 3.39]

discharge line. – A rigid pipe connected to the outlet of a measuring device.[3.30, 3.31, 3.32, 3.34, 3.37, 3.39]

E

event counter. – A nonresettable counter that increments once each time the mode that permits changes to sealable 

parameters is entered and one or more changes are made to sealable calibration or configuration parameters of a 

device.[2.20, 2.21, 3.30, 3.37, 3.39, 5.54, 5.56(a), 5.56(b), 5.57] 

event logger. – A form of audit trail containing a series of records where each record contains the number from the 

event counter corresponding to the change to a sealable parameter, the identification of the parameter that was 
Handbook 44 – 2011 3.39. Hydrogen Gas-Measuring Devices – Tentative Code

3-139

changed, the time and date when the parameter was changed, and the new value of the parameter.[2.20, 2.21, 3.30, 

3.37, 3.39, 5.54, 5.56(a), 5.56(b), 5.57]

I

indicating element. – An element incorporated in a weighing or measuring device by means of which its 

performance relative to quantity or money value is "read" from the device itself as, for example, an 

index-and-graduated-scale combination, a weighbeam-and-poise combination, a digital indicator, and the like. 

(Also see "primary indicating or recording element.")[1.10]

M

minimum measured quantity (MMQ). – The smallest quantity delivered for which the measurement is to within 

the applicable tolerances for that system.[3.37, 3.39]

N

nonresettable totalizer. – An element interfaced with the measuring or weighing element that indicates the 

cumulative registration of the measured quantity with no means to return to zero.[3.30, 3.37, 3.39]

P

point-of-sale system. – An assembly of elements including a weighing or measuring element, an indicating element, 

and a recording element (and may also be equipped with a “scanner”) used to complete a direct sales 

transaction.[2.20, 3.30, 3.32, 3.37, 3.39]

R

remote configuration capability. – The ability to adjust a weighing or measuring device or change its sealable 

parameters from or through some other device that is not itself necessary to the operation of the weighing or 

measuring device or is not a permanent part of that device.[2.20, 2.21, 2.24, 3.30, 3.37, 3.39, 5.56(a)]

retail device. – A measuring device primarily used to measure product for the purpose of sale to the end user.[3.30, 

3.32, 3.37, 3.39]

W

wet hose. – A discharge hose intended to be full of product at all times. (See "wet-hose type.")[3.30, 3.31, 3.38, 

3.39]

wet-hose type. – A type of device designed to be operated with the discharge hose full of product at all times. (See 

"wet hose.")[3.30, 3.32, 3.34, 3.37, 3.38, 3.39]
3.39. Hydrogen Gas-Measuring Devices – Tentative Code Handbook 44 – 2011

3-140

THIS PAGE INTENTIONALLY LEFT BLANK