Handbook 44 – 2012 5.52. Linear Measures

5-15

Table of Contents

Section 5.52. Linear Measures............................................................................................... 5-17

A. Application .................................................................................................................................................. 5-17

A.1. General................................................................................................................................................5-17

A.2. Additional Code Requirements...........................................................................................................5-17

S. Specifications............................................................................................................................................... 5-17

S.1.M.Units....................................................................................................................................................5-17

S.1. Units....................................................................................................................................................5-17

S.2. Material...............................................................................................................................................5-17

S.2.1. Flexible Tape. .......................................................................................................................5-17

S.2.2. End Measure. ........................................................................................................................5-17

S.3. Finish. .................................................................................................................................................5-17

S.4. Design. ................................................................................................................................................5-17

S.4.1. Rigid Measure.......................................................................................................................5-17

S.4.2. Folding Measure. ..................................................................................................................5-17

S.5. Graduations.........................................................................................................................................5-18

S.5.1. General..................................................................................................................................5-18

S.5.2. Width. ...................................................................................................................................5-18

T. Tolerances.................................................................................................................................................... 5-18

T.1. For Measures Except Metal Tapes......................................................................................................5-18

T.2. For Metal Tapes..................................................................................................................................5-18
5.52. Linear Measures Handbook 44 – 2012

5-16

THIS PAGE INTENTIONALLY LEFT BLANK
Handbook 44 – 2012 5.52. Linear Measures

5-17

Section 5.52. Linear Measures

A. Application

A.1. General. – This code applies to any linear measure or measure of length, whether flexible or inflexible, 

permanently installed or portable.

A.2. Additional Code Requirements. – In addition to the requirements of this code, Linear Measures shall meet 

the requirements of Section 1.10. General Code.

S. Specifications

S.1.M. Units. – A linear measure may be in total length, and the total length may be subdivided in any or all of the 

following:

(a) centimeters and tenths of the centimeter;

(b) meters; and

(c) multiples of meters.

A 1-meter measure may be graduated, in addition, to show 0.1 m and multiples of 0.1 m subdivisions.

S.1. Units. – A linear measure may be in total length, and the total length may be subdivided in any or all of the 

following:

(a) inches and binary submultiples of the inch;

(b) feet;

(c) yards and multiples of yards.

A 1-yard measure may be graduated, in addition, to show 1⁄3 yd and 2⁄3 yd subdivisions. A flexible tape may be 

graduated in tenths or hundredths of a foot, or both tenths and hundredths of a foot. (Any other subdivisions are 

allowable only on measures of special purposes and when required for such purposes.)

S.2. Material.

S.2.1. Flexible Tape. – A flexible tape shall be made of metal.

S.2.2. End Measure. – If an end measure is made of material softer than brass, the ends of the measure shall 

be protected by brass (or other metal at least equally hard) securely attached.

S.3. Finish. – Measures shall be smoothly finished.

S.4. Design.

S.4.1. Rigid Measure. – A rigid measure shall be straight.

S.4.2. Folding Measure. – A folding measure shall open to a definite stop, and when so opened shall be 

straight.
5.52. Linear Measures Handbook 44 – 2012

5-18

S.5. Graduations.

S.5.1. General. – Graduations shall be perpendicular to the edge of the measure.

S.5.2. Width. – The width of the graduations on any measure shall not exceed one-half the width of the 

smallest graduated interval on the measure, and in no case shall be wider than 0.75 mm (0.03 in).

(Amended 1982)

T. Tolerances

T.1. For Measures Except Metal Tapes. – Maintenance tolerances in excess and in deficiency for measures 

except metal tapes shall be as shown in Table 1. Maintenance Tolerances, in Excess and in Deficiency, for Linear 

Measures Except Metal Tapes. Acceptance tolerances shall be one-half the maintenance tolerances.

Table 1.

Maintenance Tolerances, in Excess and in Deficiency, for Linear Measures Except Metal Tapes

Nominal Interval

from Zero Tolerance

feet inch

1⁄2 or less

1

2

3

4

5

6

1

/64

1

/32

1

/16

3

/32

1

/8

5

/32

3

/16

T.2. For Metal Tapes. – Maintenance and acceptance tolerances in excess and in deficiency for metal tapes shall 

be as shown in Table 2. Maintenance and Acceptance Tolerances, in Excess and in Deficiency, for Metal Tapes. 

Tapes of 10 m (25 ft) or over shall be tested at a tension resulting from a load of 5 kg (10 lb). Tapes less than 10 m 

(25 ft) shall be tested at a tension resulting from a load of 2.5 kg (5 lb). However, flexible metal tapes of 10 m 

(25 ft) or less that are not normally used under tension shall be tested with no tension applied. All tapes shall be 

supported throughout on a horizontal flat surface whenever tested.

(Amended 1972)

Table 2.

Maintenance and Acceptance Tolerances, in Excess and in Deficiency, for Metal Tapes

Nominal Interval

from Zero Tolerance

feet inch

6 or less

7 to 30, inclusive

31 to 55, inclusive

56 to 80, inclusive

81 to 100, inclusive

1

/32

1

/16

1

/8

3

/16

1⁄4