Handbook 44 - 2007 3.38. Carbon Dioxide Liquid-Measuring Devices 

Section 3.38. Carbon Dioxide Liquid-Measuring Devices 

A. Application 

A.1. - This code applies to carbon dioxide liquid-measuring devices used for the measurement of liquid carbon dioxide. 

A.2. - This code does not apply to devices used solely for dispensing a product in connection with operations in which the 

amount dispensed does not affect customer charges. 

A.3. - See also, Sec. 1.10. General Code requirements. 

A.4. Type Evaluation. - The National Type Evaluation Program will accept for type evaluation only those devices that 

comply with all requirements of this code. 

(Added 1998) 

S. Specifications 

S.1. Design of Indicating and Recording Elements and of Recorded Representations. 

S.1.1. Primary Elements. -

S.1.1.1. General. - A device shall be equipped with a primary indicating element and may also be equipped 

with a primary recording element. 

S.1.1.2. Units. - A device shall indicate and record, if equipped to record, its deliveries in terms of pounds or 

kilograms or decimal subdivisions or multiples thereof. 

S.1.1.3. Value of Smallest Unit. - The value of the smallest unit of indicated delivery, and recorded delivery, if 

the device is equipped to record, shall not exceed the equivalent of: 

(a) for small delivery devices: 

(1) 1 kilogram, or 

(2) 1 pound 

(b) for large delivery devices: 

(1) 10 kilograms , or 

(2) 10 pounds 

S.1.1.4. Advancement of Indicating and Recording Elements. - Primary indicating and recording elements 

shall be susceptible to advancement only by the normal operation of the device. However, a device may be 

cleared by advancing its elements to zero, but only if: 

(a) the advancing movement, once started, cannot be stopped until zero is reached, or 

(b) in the case of indicating elements only, such elements are automatically obscured until the elements 

reach the correct zero position. 

S.1.1.5. Return to Zero. - Primary indicating and recording elements shall be readily returnable to a definite 

zero indication. Means shall be provided to prevent the return of primary indicating elements and of primary 

recording elements beyond their correct zero position. 

3-83 
3.38. Carbon Dioxide Liquid-Measuring Devices Handbook 44 - 2007 

S.1.2. Graduations. -

S.1.2.1. Length. - Graduations shall be so varied in length that they may be conveniently read. 

S.1.2.2. Width. - In any series of graduations, the width of a graduation shall in no case be greater than the 

width of the minimum clear interval between graduations. The width of main graduations shall be not more than 

50 % greater than the width of subordinate graduations. Graduations shall in no case be less than 0.2 mm 

(0.008 in) in width. 

S.1.2.3. Clear Interval Between Graduations. - The clear interval shall be not less than 1.0 mm (0.04 in). If 

the graduations are not parallel, the measurement shall be made: 

(a) along the line of relative movement between the graduations at the end of the indicator, or 

(b) if the indicator is continuous, at the point of widest separation of the graduations. 

(See also S.1.3.6.) 

S.1.3. Indicators. -

S.1.3.1. Symmetry. - The index of an indicator shall be of the same shape as the graduations at least throughout 

that portion of its length associated with the graduations. 

S.1.3.2. Length. - The index of an indicator shall reach to the finest graduations with which it is used, unless the 

indicator and the graduations are in the same plane, in which case the distance between the end of the indicator 

and the ends of the graduations, measured along the line of the graduations, shall be not more than 1.0 mm 

(0.04 in). 

S.1.3.3. Width. - The width of the index of the indicator in relation to the series of graduations with which it is 

used shall be not greater than: 

(a) the width of the narrowest graduation*, and 

[*Nonretroactive as of January 1, 2002]

(Amended 2001) 

(b) the width of the minimum clear interval between graduations. 

When the index of an indicator extends along the entire length of a graduation, that portion of the index of the 

indicator that may be brought into coincidence with the graduation shall be of the same width throughout the 

length of the index that coincides with the graduation. 

S.1.3.4. Clearance. - The clearance between the index of an indicator and the graduations shall in no case be 

more than 1.5 mm (0.06 in). 

S.1.3.5. Parallax. - Parallax effects shall be reduced to the practicable minimum. 

S.1.3.6. Travel of Indicator. - If the most sensitive element of the primary indicating element utilizes an 

indicator and graduations, the relative movement of these parts corresponding to the smallest indicated value 

shall be no less than 5 mm (0.20 in). 

S.1.4. Computing-Type Devices. - 

S.1.4.1. Printed Ticket. - Any printed ticket issued by a device of the computing type on which there is printed 

the total computed price shall have printed clearly thereon also the total quantity of the delivery and the price per 

unit. 

S.1.4.2. Money-Value Computations. - Money-value computations shall be of the full-computing type in 

which the money value at a single unit price, or at each of a series of unit prices, shall be computed for every 

3-84 
Handbook 44 - 2007 3.38. Carbon Dioxide Liquid-Measuring Devices 

delivery within either the range of measurement of the device or the range of the computing elements, whichever 

is less. 

The total price shall be computed on the basis of the quantity indicated when the value of the smallest division 

indicated is equal to or less than the value specified in S.1.1.3. 

S.1.4.3. Money-Values, Mathematical Agreement. - Any digital money-value indication and any recorded 

money value on a computing-type device shall be in mathematical agreement with its associated quantity 

indication or representation to within 1 cent of money value. 

S.2. Design of Measuring Elements.

S.2.1. Vapor Elimination. - A measuring system shall be equipped with an effective vapor eliminator or other 

effective means to prevent the measurement of vapor that will cause errors in excess of the applicable tolerances. 

S.2.2. Reverse Flow Measurement. - Effective means, automatic in operation, shall be installed to prevent reverse 

flow measurement. 

S.2.3. Maintenance of Liquid State. - A device shall be so designed that the product being measured will remain in 

a liquid state during passage through the device. 

S.2.4. Automatic Temperature or Density Compensation. - A volumetric device shall be equipped with automatic 

means for adjusting the indication and recorded representation of the measured quantity of the product to indicate or 

record the quantity of the product measured in terms of pounds. 

S.2.5. Provision for Sealing. - Adequate provision shall be made for an approved means of security (e.g., data 

change audit trail) or for physically applying a security seal in such a manner that requires the security seal to be 

broken before an adjustment or interchange may be made of: 

(a) any measuring or indicating element, 

(b) any adjustable element for controlling delivery rate when such rate tends to affect the accuracy of deliveries, 

(c) any automatic temperature or density compensating system, and 

(d) any metrological parameter that will affect the metrological integrity of the device or system. 

When applicable any adjusting mechanism shall be readily accessible for purposes of affixing a security seal. 

[Audit trails shall use the format set forth in Table S.2.5.]* 

[*Nonretroactive as of January 1, 1995] 

(Amended 2006) 

3-85 
3.38. Carbon Dioxide Liquid-Measuring Devices Handbook 44 - 2007 

Table S.2.5. Categories of Device and Methods of Sealing 

Category of Device Method of Sealing 

Category 1: No remote configuration capability. Seal by physical seal or two event counters: one for 

calibration parameters and one for configuration 

parameters. 

Category 2: Remote configuration capability, but access 

is controlled by physical hardware. 

The device shall clearly indicate that it is in the remote 

configuration mode and record such message if capable 

of printing in this mode or shall not operate while in this 

mode.

The hardware enabling access for remote communication 

must be on-site. The hardware must be sealed using a 

physical seal or an event counter for calibration 

parameters and an event counter for configuration 

parameters. The event counters may be located either at 

the individual measuring device or at the system 

controller; however, an adequate number of counters 

must be provided to monitor the calibration and 

configuration parameters of the individual devices at a 

location. If the counters are located in the system 

controller rather than at the individual device, means 

must be provided to generate a hard copy of the 

information through an on-site device.

Category 3: Remote configuration capability access may 

be unlimited or controlled through a software switch 

(e.g., password). 

The device shall clearly indicate that it is in the remote 

configuration mode and record such message if capable 

of printing in this mode or shall not operate while in this 

mode. 

An event logger is required in the device; it must include 

an event counter (000 to 999), the parameter ID, the date 

and time of the change, and the new value of the 

parameter. A printed copy of the information must be 

available through the device or through another on-site 

device. The event logger shall have a capacity to retain 

records equal to ten times the number of sealable 

parameters in the device, but not more than 1000 records 

are required. (Note: Does not require 1000 changes to 

be stored for each parameter.)

[Nonretroactive as of January 1, 1995] 

(Table Added 2006) 

S.3. Design of Discharge Lines and Discharge Line Valves.

S.3.1. Diversion of Measured Liquid. - No means shall be provided by which any measured liquid can be diverted 

from the measuring chamber of the device or the discharge line therefrom, except that a manually controlled outlet 

that may be opened for purging or draining the measuring system shall be permitted. Effective means shall be 

provided to prevent the passage of liquid through any such outlet during normal operation of the device and to 

indicate clearly and unmistakably when the valve controls are so set as to permit passage of liquid through such 

outlet. 

S.3.2. Discharge Hose. - The discharge hose of a measuring system shall be of a wet hose type with a shutoff valve 

at its outlet end. 

S.4. Marking Requirements.

S.4.1. Limitation of Use. - If a measuring system is intended to measure accurately only liquids having particular 

properties, or to measure accurately only under specific installation or operating conditions, or to measure accurately 

only when used in conjunction with specific accessory equipment, these limitations shall be clearly and permanently 

marked on the device. 

S.4.2. Discharge Rates. - A meter shall be marked to show its designed maximum and minimum discharge rates. 

The marked minimum discharge rate shall not exceed 20 % of the marked maximum discharge rate. 

3-86 
Handbook 44 - 2007 3.38. Carbon Dioxide Liquid-Measuring Devices 

Note: See example in Section 3.30. Liquid-Measuring Devices Code, Paragraph S.4.4.1. 

(Note Added 2003) 

N. Notes 

N.1. Test Liquid. - The test liquid shall be carbon dioxide in a compressed liquid state. 

N.2. Vaporization and Volume Change. - Care shall be exercised to reduce vaporization and volume changes to a 

minimum. When testing by weight, the weigh tank and transfer systems shall be pre-cooled to liquid temperature prior to 

the start of the test to avoid the venting of vapor from the vessel being weighed. 

N.3. Test Drafts.

N.3.1. Gravimetric Test. - Weight test drafts shall be equal to at least the amount delivered by the device in 

2 minutes at its maximum discharge rate. 

N.3.2. Transfer Standard Test. - When comparing a meter with a calibrated transfer standard, the test draft shall be 

equal to at least the amount delivered by the device in 2 minutes at its maximum discharge rate. 

 

N.3.3. Volumetric Prover Test Drafts. - Test drafts shall be equal to at least the amount delivered in 1 minute at 

normal discharge rate. 

N.4. Testing Procedures.

N.4.1. Normal Tests. - The "normal" test of a device shall be made at the maximum discharge flow rate developed 

under the conditions of installation. Any additional tests conducted at flow rates down to and including one-half of 

the sum of the maximum discharge flow rate and the rated minimum discharge flow rate shall be considered normal 

tests. 

N.4.1.1. Repeatability Tests. - Tests for repeatability should include a minimum of three consecutive test drafts 

of approximately the same size and be conducted under controlled conditions where variations in factors, such as 

temperature, pressure, and flow rate are reduced to the extent that they will not affect the results obtained. 

(Added 2002) 

N.4.2. Special Tests. - Any test except as set forth in N.4.1. shall be considered a special test. Tests shall be 

conducted, if possible, to evaluate any special elements or accessories attached to or associated with the device. A 

device shall be tested at a minimum discharge rate of: 

(a) not less than the marked minimum discharge rate or 20 % of the maximum rated discharge rate of the 

device, whichever is less, or 

(b) the lowest discharge rate practicable under the conditions of installation. 

"Special" tests may be conducted to develop any characteristics of the device anticipated under the conditions of 

installation. 

N.4.3. Density. - Temperature and pressure of the metered test liquid shall be measured during the test for the 

determination of density or volume correction when applicable. The appropriate correction values shall apply as 

specified in Table N.4.4. 

N.4.4. Automatic Temperature or Density Compensation. - If a device is equipped with an automatic temperature 

or density compensator, the compensator shall be tested by comparing the quantity indicated or recorded by the 

device (with the compensator connected and operating) with the actual delivered quantity. The appropriate 

correction values shall apply as specified in Table N.4.4. 

3-87 
3.38. Carbon Dioxide Liquid-Measuring Devices Handbook 44 - 2007 

Table N.4.4.

Pressure Liquid Density Vapor Density Temp °F PSIA PSIG lb/gal (lb-oz)/gal lb/cu ft lb/gal 

Vapor 

Displacement % 

-30.00 177.89 163.19 9.127 9 - 2.0 1.989 0.266 2.9 

-29.75 178.75 164.05 9.122 9 - 2.0 1.999 0.267 2.9 

-29.50 179.62 164.92 9.117 9 - 1.9 2.008 0.268 2.9 

-29.25 180.49 165.79 9.113 9 - 1.8 2.018 0.270 3.0 

-29.00 181.36 166.67 9.108 9 - 1.7 2.028 0.271 3.0 

-28.75 182.24 167.54 9.103 9 - 1.7 2.038 0.272 3.0 

-28.50 183.12 168.42 9.098 9 - 1.6 2.048 0.274 3.0 

-28.25 184.00 169.31 9.094 9 - 1.5 2.058 0.275 3.0 

-28.00 184.89 170.19 9.089 9 - 1.4 2.067 0.276 3.0 

-27.75 185.78 171.08 9.084 9 - 1.3 2.077 0.278 3.1 

-27.50 186.67 171.98 9.080 9 - 1.3 2.087 0.279 3.1 

-27.25 187.57 172.87 9.075 9 - 1.2 2.098 0.280 3.1 

-27.00 188.47 173.77 9.070 9 - 1.1 2.108 0.282 3.1 

-26.75 189.37 174.67 9.065 9 - 1.0 2.118 0.283 3.1 

-26.50 190.28 175.58 9.061 9 - 1.0 2.128 0.284 3.1 

-26.25 191.18 176.49 9.056 9 - 0.9 2.138 0.286 3.2 

-26.00 192.10 177.40 9.051 9 - 0.8 2.148 0.287 3.2 

-25.75 193.01 178.32 9.046 9 - 0.7 2.159 0.289 3.2 

-25.50 193.93 179.23 9.041 9 - 0.7 2.169 0.290 3.2 

-25.25 194.85 180.16 9.037 9 - 0.6 2.179 0.291 3.2 

-25.00 195.78 181.08 9.032 9 - 0.5 2.190 0.293 3.2 

-24.75 196.70 182.01 9.027 9 - 0.4 2.200 0.294 3.3 

-24.50 197.64 182.94 9.022 9 - 0.4 2.211 0.296 3.3 

-24.25 198.57 183.87 9.017 9 - 0.3 2.221 0.297 3.3 

-24.00 199.51 184.81 9.013 9 - 0.2 2.232 0.298 3.3 

-23.75 200.45 185.75 9.008 9 - 0.1 2.243 0.300 3.3 

-23.50 201.39 186.70 9.003 9 - 0.0 2.253 0.301 3.3 

-23.25 202.34 187.64 8.998 9 - 0.0 2.264 0.303 3.4 

-23.00 203.29 188.60 8.993 8 - 15.9 2.275 0.304 3.4 

-22.75 204.25 189.55 8.989 8 - 15.8 2.286 0.306 3.4 

-22.50 205.20 190.51 8.984 8 - 15.7 2.296 0.307 3.4 

-22.25 206.16 191.47 8.979 8 - 15.7 2.307 0.308 3.4 

-22.00 207.13 192.43 8.974 8 - 15.6 2.318 0.310 3.5 

-21.75 208.09 193.40 8.969 8 - 15.5 2.329 0.311 3.5 

-21.50 209.06 194.37 8.964 8 - 15.4 2.340 0.313 3.5 

-21.25 210.04 195.34 8.959 8 - 15.4 2.351 0.314 3.5 

-21.00 211.02 196.32 8.955 8 - 15.3 2.362 0.316 3.5 

-20.75 212.00 197.30 8.950 8 - 15.2 2.374 0.317 3.5 

-20.50 212.98 198.28 8.945 8 - 15.1 2.385 0.319 3.6 

-20.25 213.97 199.27 8.940 8 - 15.0 2.396 0.320 3.6 

-20.00 214.96 200.26 8.935 8 - 15.0 2.407 0.322 3.6 

-19.75 215.95 201.26 8.930 8 - 14.9 2.419 0.323 3.6 

-19.50 216.95 202.25 8.925 8 - 14.8 2.430 0.325 3.6 

-19.25 217.95 203.25 8.920 8 - 14.7 2.441 0.326 3.7 

-19.00 218.95 204.26 8.915 8 - 14.6 2.453 0.328 3.7 

-18.75 219.96 205.27 8.911 8 - 14.6 2.464 0.329 3.7 

-18.50 220.97 206.28 8.906 8 - 14.5 2.476 0.331 3.7 

-18.25 221.99 207.29 8.901 8 - 14.4 2.488 0.333 3.7 

-18.00 223.01 208.31 8.896 8 - 14.3 2.499 0.334 3.8 

-17.75 224.03 209.33 8.891 8 - 14.3 2.511 0.336 3.8 

-17.50 225.05 210.36 8.886 8 - 14.2 2.523 0.337 3.8 

3-88 
Handbook 44 - 2007 3.38. Carbon Dioxide Liquid-Measuring Devices 

Table N.4.4.

Pressure Liquid Density Vapor Density Temp °F PSIA PSIG lb/gal (lb-oz)/gal lb/cu ft lb/gal 

Vapor 

Displacement % 

-17.25 226.08 211.38 8.881 8 - 14. 2.534 0.339 3.8 

-17.00 227.11 212.42 8.876 8 - 14.0 2.546 0.340 3.8 

-16.75 228.15 213.45 8.871 8 - 13.9 2.558 0.342 3.9 

-16.50 229.18 214.49 8.866 8 - 13.9 2.570 0.344 3.9 

-16.25 230.23 215.53 8.861 8 - 13.8 2.582 0.345 3.9 

-16.00 231.27 216.58 8.856 8 - 13.7 2.594 0.347 3.9 

-15.75 232.32 217.62 8.851 8 - 13.6 2.606 0.348 3.9 

-15.50 233.37 218.68 8.846 8 - 13.5 2.618 0.350 4.0 

-15.25 234.43 219.73 8.841 8 - 13.5 2.630 0.352 4.0 

-15.00 235.49 220.79 8.836 8 - 13.4 2.643 0.353 4.0 

-14.75 236.55 221.86 8.831 8 - 13.3 2.655 0.355 4.0 

-14.50 237.62 222.92 8.826 8 - 13.2 2.667 0.357 4.0 

-14.25 238.69 223.99 8.821 8 - 13.1 2.680 0.358 4.1 

-14.00 239.76 225.07 8.816 8 - 13.1 2.692 0.360 4.1 

-13.75 240.84 226.14 8.811 8 - 13.0 2.704 0.362 4.1 

-13.50 241.92 227.22 8.806 8 - 12.9 2.717 0.363 4.1 

-13.25 243.00 228.31 8.801 8 - 12.8 2.729 0.365 4.1 

-13.00 244.09 229.39 8.796 8 - 12.7 2.742 0.367 4.2 

-12.75 245.18 230.49 8.791 8 - 12.7 2.755 0.368 4.2 

-12.50 246.28 231.58 8.786 8 - 12.6 2.767 0.370 4.2 

-12.25 247.37 232.68 8.781 8 - 12.5 2.780 0.372 4.2 

-12.00 248.48 233.78 8.776 8 - 12.4 2.793 0.373 4.3 

-11.75 249.58 234.89 8.771 8 - 12.3 2.806 0.375 4.3 

-11.50 250.69 236.00 8.765 8 - 12.2 2.819 0.377 4.3 

-11.25 251.80 237.11 8.760 8 - 12.2 2.832 0.379 4.3 

-11.00 252.92 238.22 8.755 8 - 12.1 2.845 0.380 4.3 

-10.75 254.04 239.34 8.750 8 - 12.0 2.858 0.382 4.4 

-10.50 255.16 240.47 8.745 8 - 11.9 2.871 0.384 4.4 

-10.25 256.29 241.60 8.740 8 - 11.8 2.884 0.386 4.4 

-10.00 257.42 242.73 8.735 8 - 11.8 2.897 0.387 4.4 

-9.75 258.56 243.86 8.730 8 - 11.7 2.911 0.389 4.5 

-9.50 259.70 245.00 8.725 8 - 11.6 2.924 0.391 4.5 

-9.25 260.84 246.14 8.719 8 - 11.5 2.937 0.393 4.5 

-9.00 261.98 247.29 8.714 8 - 11.4 2.951 0.394 4.5 

-8.75 263.13 248.44 8.709 8 - 11.3 2.964 0.396 4.5 

-8.50 264.29 249.59 8.704 8 - 11.3 2.978 0.398 4.6 

-8.25 265.44 250.75 8.699 8 - 11.2 2.991 0.400 4.6 

-8.00 266.60 251.91 8.694 8 - 11.1 3.005 0.402 4.6 

-7.75 267.77 253.07 8.688 8 - 11.0 3.019 0.404 4.6 

-7.50 268.93 254.24 8.683 8 - 10.9 3.032 0.405 4.7 

-7.25 270.11 255.41 8.678 8 - 10.8 3.046 0.407 4.7 

-7.00 271.28 256.59 8.673 8 - 10.8 3.060 0.409 4.7 

-6.75 272.46 257.76 8.668 8 - 10.7 3.074 0.411 4.7 

-6.50 273.64 258.95 8.662 8 - 10.6 3.088 0.413 4.8 

-6.25 274.83 260.13 8.657 8 - 10.5 3.102 0.415 4.8 

-6.00 276.02 261.32 8.652 8 - 10.4 3.116 0.417 4.8 

-5.75 277.21 262.52 8.647 8 - 10.3 3.130 0.418 4.8 

-5.50 278.41 263.72 8.641 8 - 10.3 3.144 0.420 4.9 

-5.25 279.61 264.92 8.636 8 - 10.2 3.159 0.422 4.9 

-5.00 280.82 266.12 8.631 8 - 10.1 3.173 0.424 4.9 

-4.75 282.03 267.33 8.626 8 - 10.0 3.187 0.426 4.9 

-4.50 283.24 268.55 8.620 8 - 9.9 3.202 0.428 5.0 

3-89 
3.38. Carbon Dioxide Liquid-Measuring Devices Handbook 44 - 2007 

Table N.4.4.

Pressure Liquid Density Vapor Density Temp °F PSIA PSIG lb/gal (lb-oz)/gal lb/cu ft lb/gal 

Vapor 

Displacement % 

-4.25 284.46 269.76 8.615 8 - 9.8 3.216 0.430 5.0 

-4.00 285.68 270.98 8.610 8 - 9.8 3.231 0.432 5.0 

-3.75 286.90 272.21 8.604 8 - 9.7 3.245 0.434 5.0 

-3.50 288.13 273.44 8.599 8 - 9.6 3.260 0.436 5.1 

-3.25 289.37 274.67 8.594 8 - 9.5 3.275 0.438 5.1 

-3.00 290.60 275.91 8.589 8 - 9.4 3.289 0.440 5.1 

-2.75 291.84 277.15 8.583 8 - 9.3 3.304 0.442 5.1 

-2.50 293.09 278.39 8.578 8 - 9.2 3.319 0.444 5.2 

-2.25 294.33 279.64 8.573 8 - 9.2 3.334 0.446 5.2 

-2.00 295.58 280.89 8.567 8 - 9.1 3.349 0.448 5.2 

-1.75 296.84 282.14 8.562 8 - 9.0 3.364 0.450 5.3 

-1.50 298.10 283.40 8.556 8 - 8.9 3.379 0.452 5.3 

-1.25 299.36 284.67 8.551 8 - 8.8 3.395 0.454 5.3 

-1.00 300.63 285.93 8.546 8 - 8.7 3.410 0.456 5.3 

-0.75 301.90 287.21 8.540 8 - 8.6 3.425 0.458 5.4 

-0.50 303.18 288.48 8.535 8 - 8.6 3.440 0.460 5.4 

-0.25 304.46 289.76 8.530 8 - 8.5 3.456 0.462 5.4 

0.00 305.74 291.74 8.524 8 - 8.4 3.471 0.464 5.4 

0.25 307.03 292.33 8.519 8 - 8.3 3.487 0.466 5.5 

0.50 308.32 293.62 8.513 8 - 8.2 3.503 0.468 5.5 

0.75 309.61 294.92 8.508 8 - 8.1 3.518 0.470 5.5 

1.00 310.91 296.21 8.502 8 - 8.0 3.534 0.472 5.6 

1.25 312.21 297.52 8.497 8 - 8.0 3.550 0.475 5.6 

1.50 313.52 298.82 8.491 8 - 7.9 3.566 0.477 5.6 

1.75 314.83 300.13 8.486 8 - 7.8 3.582 0.479 5.6 

2.00 316.15 301.45 8.480 8 - 7.7 3.598 0.481 5.7 

2.25 317.46 302.77 8.475 8 - 7.6 3.614 0.483 5.7 

2.50 318.79 304.09 8.469 8 - 7.5 3.630 0.485 5.7 

2.75 320.11 305.42 8.464 8 - 7.4 3.646 0.487 5.8 

3.00 321.45 306.75 8.458 8 - 7.3 3.662 0.490 5.8 

3.25 322.78 308.08 8.453 8 - 7.2 3.679 0.492 5.8 

3.50 324.12 309.42 8.447 8 - 7.2 3.695 0.494 5.8 

3.75 325.46 310.77 8.442 8 - 7.1 3.712 0.496 5.9 

4.00 326.81 312.11 8.436 8 - 7.0 3.728 0.498 5.9 

4.25 328.16 313.46 8.431 8 - 6.9 3.745 0.501 5.9 

4.50 329.52 314.82 8.425 8 - 6.8 3.761 0.503 6.0 

4.75 330.88 316.18 8.420 8 - 6.7 3.778 0.505 6.0 

5.00 332.24 317.54 8.414 8 - 6.6 3.795 0.507 6.0 

5.25 333.61 318.91 8.408 8 - 6.5 3.812 0.510 6.1 

5.50 334.98 320.28 8.403 8 - 6.4 3.829 0.512 6.1 

5.75 336.35 321.66 8.397 8 - 6.4 3.846 0.514 6.1 

6.00 337.73 323.04 8.392 8 - 6.3 3.863 0.516 6.2 

6.25 339.12 324.42 8.386 8 - 6.2 3.880 0.519 6.2 

6.50 340.51 325.81 8.380 8 - 6.1 3.897 0.521 6.2 

6.75 341.90 327.20 8.375 8 - 6.0 3.915 0.523 6.3 

7.00 343.30 328.60 8.369 8 - 5.9 3.932 0.526 6.3 

7.25 344.70 330.00 8.363 8 - 5.8 3.949 0.528 6.3 

7.50 346.10 331.41 8.358 8 - 5.7 3.967 0.530 6.3 

7.75 347.51 332.82 8.352 8 - 5.6 3.984 0.533 6.4 

8.00 348.92 334.23 8.346 8 - 5.5 4.002 0.535 6.4 

8.25 350.34 335.65 8.341 8 - 5.4 4.020 0.537 6.4 

8.50 351.76 337.07 8.335 8 - 5.4 4.038 0.540 6.5 

3-90 
Handbook 44 - 2007 3.38. Carbon Dioxide Liquid-Measuring Devices 

Table N.4.4.

Pressure Liquid Density Vapor Density Temp °F PSIA PSIG lb/gal (lb-oz)/gal lb/cu ft lb/gal 

Vapor 

Displacement % 

8.75 353.19 338.49 8.335 8 - 5.4 4.038 0.540 6.5 

9.00 354.62 339.92 8.323 8 - 5.2 4.073 0.545 6.5 

9.25 356.06 341.36 8.318 8 - 5.1 4.091 0.547 6.6 

9.50 357.49 342.80 8.312 8 - 5.0 4.110 0.549 6.6 

9.75 358.94 344.24 8.306 8 - 4.9 4.128 0.552 6.6 

10.00 360.38 345.69 8.300 8 - 4.8 4.146 0.554 6.7 

10.25 361.84 347.14 8.295 8 - 4.7 4.164 0.557 6.7 

10.50 363.29 348.60 8.289 8 - 4.6 4.183 0.559 6.7 

10.75 364.75 350.06 8.283 8 - 4.5 4.201 0.562 6.8 

11.00 366.22 351.52 8.277 8 - 4.4 4.220 0.564 6.8 

11.25 367.68 352.99 8.271 8 - 4.3 4.238 0.567 6.8 

11.50 369.16 354.46 8.266 8 - 4.2 4.257 0.569 6.9 

11.75 370.64 355.94 8.260 8 - 4.2 4.276 0.572 6.9 

12.00 372.12 357.42 8.254 8 - 4.1 4.295 0.574 7.0 

12.25 373.60 358.91 8.248 8 - 4.0 4.314 0.577 7.0 

12.50 375.09 360.40 8.242 8 - 3.9 4.333 0.579 7.0 

12.75 376.59 361.89 8.236 8 - 3.8 4.352 0.582 7.1 

13.00 378.09 363.39 8.230 8 - 3.7 4.371 0.584 7.1 

13.25 379.59 364.89 8.224 8 - 3.6 4.390 0.587 7.1 

13.50 381.10 366.40 8.219 8 - 3.5 4.410 0.589 7.2 

13.75 382.61 367.91 8.213 8 - 3.4 4.429 0.592 7.2 

14.00 384.13 369.43 8.207 8 - 3.3 4.449 0.595 7.2 

14.25 385.65 370.95 8.201 8 - 3.2 4.468 0.597 7.3 

14.50 387.17 372.48 8.195 8 - 3.1 4.488 0.600 7.3 

14.75 388.70 374.01 8.189 8 - 3.0 4.508 0.603 7.4 

15.00 390.24 375.54 8.183 8 - 2.9 4.527 0.605 7.4 

15.25 391.78 377.08 8.177 8 - 2.8 4.547 0.608 7.4 

15.50 393.32 378.62 8.171 8 - 2.7 4.567 0.611 7.5 

15.75 394.87 380.17 8.165 8 - 2.6 4.587 0.613 7.5 

16.00 396.42 381.72 8.159 8 - 2.5 4.608 0.616 7.5 

16.25 397.98 383.28 8.153 8 - 2.4 4.628 0.619 7.6 

16.50 399.54 384.84 8.147 8 - 2.3 4.648 0.621 7.6 

16.75 401.10 386.41 8.141 8 - 2.2 4.669 0.624 7.7 

17.00 402.67 387.98 8.134 8 - 2.2 4.689 0.627 7.7 

17.25 404.25 389.55 8.128 8 - 2.1 4.710 0.630 7.7 

17.50 405.82 391.13 8.122 8 - 2.0 4.731 0.632 7.8 

17.75 407.41 392.71 8.116 8 - 1.9 4.751 0.635 7.8 

18.00 409.00 394.30 8.110 8 - 1.8 4.772 0.638 7.9 

18.25 410.59 395.89 8.104 8 - 1.7 4.793 0.641 7.9 

18.50 412.19 397.49 8.098 8 - 1.6 4.814 0.644 7.9 

18.75 413.79 399.09 8.092 8 - 1.5 4.835 0.646 8.0 

19.00 415.39 400.70 8.085 8 - 1.4 4.857 0.649 8.0 

19.25 417.00 402.31 8.079 8 - 1.3 4.878 0.652 8.1 

19.50 418.62 403.92 8.073 8 - 1.2 4.900 0.655 8.1 

19.75 420.24 405.54 8.067 8 - 1.1 4.921 0.658 8.2 

20.00 421.86 407.17 8.061 8 - 1.0 4.943 0.661 8.2 

3-91 
3.38. Carbon Dioxide Liquid-Measuring Devices Handbook 44 - 2007 

T. Tolerances 

T.1. Application.

T.1.1. To Underregistration and to Overregistration. - The tolerances hereinafter prescribed shall be applied to 

errors of underregistration and errors of overregistration. 

T.2. Tolerance Values. - The maintenance and acceptance tolerances for normal and special tests shall be as shown in 

Table T.2. 

Table T.2. Accuracy Classes and Tolerances for Carbon Dioxide Liquid-Measuring Devices 

Accuracy 

Class Application Acceptance Tolerance 

Maintenance 

Tolerance 

Special Test 

Tolerance 

2.5 Liquid carbon dioxide 1.5 % 2.5 % 2.5 % 

(Table Added 2003) 

(Amended 2003) 

T.2.3. Repeatability. - When multiple tests are conducted at approximately the same flow rate and draft size, the 

range of the test results for the flow rate shall not exceed 40 % of the absolute value of the maintenance tolerance and 

the results of each test shall be within the applicable tolerance. See also N.4.1.1. 

(Added 2002) 

T.3. On Tests Using Transfer Standards. - To the basic tolerance values that would otherwise be applied, there shall be 

added an amount equal to two times the standard deviation of the applicable transfer standard when compared to a basic 

reference standard. 

UR. User Requirements 

UR.1. Installation Requirements.

UR.1.1. Discharge Rate. - A device shall be so installed that the actual maximum discharge rate will not exceed the 

rated maximum discharge rate. If necessary, means for flow regulation shall be incorporated in the installation. 

UR.1.2. Length of Discharge Hose. - The discharge hose shall be of such a length and design as to keep 

vaporization of the liquid to a minimum. 

UR.1.3. Maintenance of Liquid State. - A device shall be so installed and operated that the product being measured 

shall remain in the liquid state during passage through the meter. 

UR.2. Use Requirements.

UR.2.1. Return of Indicating and Recording Elements to Zero. - The primary indicating elements (visual) and 

the primary recording elements shall be returned to zero immediately before each delivery. 

UR.2.2. Condition of Discharge System. - The discharge hose, up to the valve at the end of the discharge hose, 

shall be completely filled and pre-cooled to liquid temperatures before a "zero" condition is established and prior to 

the start of a commercial delivery. Means shall be provided to fill the discharge hose with liquid prior to the start of a 

delivery. 

UR.2.3. Vapor Equalization Line. - A vapor equalization line shall not be used during a metered delivery unless 

the quantity of vapor displaced from the buyer's tank to the seller's tank is deducted from the metered quantity. The 

appropriate correction values shall apply as specified in Table N.4.4. 

3-92 
Handbook 44 - 2007 3.38. Carbon Dioxide Liquid-Measuring Devices 

UR.2.4. Temperature or Density Compensation.

UR.2.4.1. Use of Automatic Temperature or Density Compensators. - Devices equipped with an automatic 

temperature or density compensator shall have the compensator connected, operable, and in use at all times. 

Such automatic temperature or density compensator may not be removed. 

UR.2.4.2. Tickets or Invoices. - Any written invoice or printed ticket based on a reading of a device that is 

equipped with an automatic temperature or density compensator shall have shown thereon that the quantity 

delivered has been temperature or density compensated. 

UR.2.5. Ticket in Printing Device. - A ticket shall not be inserted into a device equipped with a ticket printer until 

immediately before a delivery is begun, and in no case shall a ticket be in the device when the vehicle is in motion 

while on a public street, highway, or thoroughfare. 

UR.2.6. Sale by Weight. - All quantity determinations shall be made by means of an approved and sealed weighing or 

measuring device. All sales shall be stated in kilograms or pounds. 

3-93 
3.38. Carbon Dioxide Liquid-Measuring Devices Handbook 44 - 2007 

THIS PAGE LEFT INTENTIONALLY BLANK. 

3-94