U.S. flagAn official website of the United States government
                    Here's how you know **Official websites use .gov**
                            
                            A **.gov** website belongs to an official government organization in the United States.
 **Secure .gov websites use HTTPS**
                            
                            A **lock** (                            
                              A locked padlock                                                      
                            
                            ) or **https://** means you’ve safely connected to the .gov website. Share sensitive information only on official, secure websites.


Menu Agricultural Research Service
						
                    U.S. DEPARTMENT OF AGRICULTURE
 Hydrology and Remote Sensing Laboratory: Beltsville, MD
 close



    Research

                    Research Projects
                                    Publications
                                    Collaborations
                                    News
                                    Software
                People

                    Staff Listing
                                    CareersSearch small
                    Search
            
            ARS Home
                                About ARS
                                Contact Us




ARS Home Northeast Area Beltsville, Maryland (BARC) Beltsville Agricultural Research Center Hydrology and Remote Sensing Laboratory Research Publications at this Location





- ARS National Programs
- Search for a scientific manuscript
- Search for a research project
- ARS Software
- Programs and Projects
- Collaborations
- Subjects of Investigation
- Publications
- People
- Research Sites
- Pesticide Volatilization
- Watershed Modeling
- Climate Change
- Soil Moisture
- Drought and ET
- HRSL Brochure

Title:   Author  item
                                            Cosh, Michaelitem
                                            OCHSNER, TYSON - Oklahoma State Universityitem
                                            McKee, Lynnitem
                                            DONG, GEANO - Oklahoma State Universityitem
                                            BASARA, J. - University Of Oklahomaitem
                                            Evett, Steven

                                                         - *Steve*item
                                            HATCH, CHRISTINE - University Of Nevadaitem
                                            SMALL, ERIC - University Of Coloradoitem
                                            STEELE-DUNNE, SUSAN - Delft Universityitem
                                            ZREDA, MAREK - University Of Arizonaitem
                                            SAYDE, CHADI - Oregon State University
  Submitted to: Vadose Zone Journal
                                    Publication Type: Peer Reviewed Journal
                                    Publication Acceptance Date: 2/14/2016
                                    Publication  Date: 4/18/2016
                                    Citation: Cosh, M.H., Ochsner, T., Mckee, L.G., Dong, G., Basara, J., Evett, S.R., Hatch, C., Small, E., Steele-Dunne, S., Zreda, M., Sayde, C. 2016. The Soil Moisture Active Passive Marena Oklahoma In Situ Sensor Testbed (SMAP-MOISST): Design and initial results. Vadose Zone Journal. 15(4). doi: 10.2136/vzj2015.09.0122.Interpretive Summary: In situ soil moisture monitoring systems rely on different technological principles, depending on the manufacturer. This can lead to variations in the soil moisture record as different networks are integrated together into a single data record.  The Marena Oklahoma In Situ Sensor Testbed (MOISST), a comprehensive study, was developed to inter-compare these different technologies at a single location over a long time period and determine how this would impact studies based on the time series. A location near Stillwater, Oklahoma was instrumented in 2010 with eleven different sensor technologies.  Most of the sensors were calibrated to the specific soil type at the study site and physical collection of volumetric soil moisture was collected as ground truth for the larger field scale.  Most sensors could  reach accuracy targets and were able to be scaled to the field scale with limited effort in scaling activity.  The results of this study will be useful to the international soil moisture community as they work toward standardized methodologies and unified data records.  The remote sensing community will also find this work helpful in their calibration and validation programs.
                                Technical Abstract: In situ soil moisture monitoring networks are critical to the development of soil moisture remote sensing missions as well as agricultural and environmental management, weather forecasting and many other endeavors.  These in situ networks are composed of a variety of sensors and installation practices, which confounds the development of a unified reference database for satellite calibration and validation programs.  As part of the Soil Moisture Active Passive Mission, the Marena, Oklahoma, In Situ Sensor Testbed (SMAP-MOISST) was initiated to perform inter-comparisons and study sensor limitations.  Soil moisture sensors that are deployed in major monitoring networks were included in the study, along with new and emerging technologies, such as the Cosmic Ray Soil Moisture Observing System (COSMOS), Passive/Active Distributed Temperature Sensing System (DTS), and Global Positioning System Reflectometers (GPSR).  Four profile stations were installed in May of 2010 and soil moisture was monitored to a depth of 1 m on an hourly basis.  The four stations were distributed within a circular domain of approximately 600 meters diameter, adequate to encompass the sensing range of COSMOS. The sensors included in the base station configuration included the Stevens Water Hydra Probe, Campbell Scientific 616 and 229, Decagon EC-TM, Delta-T Theta Probe, Acclima, and Sentek EnviroSMART capacitance system.  In addition, the Pico TRIME system and additional time domain reflectometry (TDR) systems were deployed when available.  Using gravimetric calibrations, the majority of the sensors achieved accuracies better than 0.04 m3 m-3.  It is also shown that in situ measurement errors can be greatly reduced when compared to local gravimetric sampling by using a simple linear regression equation.


Return to top


Connect with ARS Sign up


ARS Home
                            USDA.gov
                            Plain Writing
                            Policies & Links
                            Civil Rights Statements
                            FOIA
                            Accessibility Statement
                            Privacy Policy
                            Non-Discrimination Statement
                            Quality of Information
                            USA.gov
                            WhiteHouse.gov