ORAU TEAM

Dose Reconstruction

Project for NIOSH

Oak Ridge Associated Universities I Dade Moeller I MJW Technical Services

Page 1 of 135

DOE Review Release 11/19/2012

Document Title:

Nevada Test Site – Occupational External 

Dose

Document Number: ORAUT-TKBS-0008-6

Revision: 03

Effective Date: 11/09/2012

Type of Document: TBD

Supersedes: Revision 02

Subject Expert(s): Cheryl Y. Smith

Approval: Signature on File Approval Date: 10/25/2012

Eugene M. Rollins, Document Owner

Concurrence: Signature on File Concurrence Date: 10/25/2012

John M. Byrne, Objective 1 Manager

Concurrence: Signature on File Concurrence Date: 10/25/2012

Edward F. Maher, Objective 3 Manager

Concurrence: Theresa Yisrael Signature on File for Concurrence Date: 11/01/2012

Kate Kimpan, Project Director

Approval: Signature on File Approval Date: 11/09/2012

James W. Neton, Associate Director for Science

New Total Rewrite Revision Page Change

FOR DOCUMENTS MARKED AS A TOTAL REWRITE, REVISION, OR PAGE CHANGE, REPLACE THE PRIOR 

REVISION AND DISCARD / DESTROY ALL COPIES OF THE PRIOR REVISION.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 2 of 135

PUBLICATION RECORD

EFFECTIVE 

DATE

REVISION 

NUMBER DESCRIPTION

09/21/2004 00 Information contained in Revision 00 is applicable only to 

employment periods after 1962 (post-atmospheric testing phase) and 

to workers not identified as involved with drillback activities prior to 

1965. In addition, Revision 00 is not applicable to dose 

reconstruction for (1) workers involved with weapons testing at 

locations other than the NTS (South Pacific, Alaska, etc.); (2) 

workers affected by any of the following 10 underground tests that 

resulted in unexpected release of radioactive material:

1. BLANCA (October 30, 1958)

2. DES MOINES (June 13, 1962)

3. BANEBERRY (December 18, 1970)

4. CAMPHOR (June 29, 1971)

5. DIAGONAL LINE (November 24, 1971)

6. RIOLA (September 25, 1980)

7. AGRINI (March 31, 1984)

8. MIDAS MYTH (February 15, 1984)

9. MISTY RAIN (April 6, 1985)

10. MIGHTY OAK (April 10, 1986) 

It is also assumed that, because they were such a rarity, it would be 

likely that if a claimant was involved in one of these events, he [or his

survivor] would mention it in his CATI. All of these conditions are 

likely verifiable by careful examination of the DOL, DOE (incident 

reports and dosimetry records), and OCAS documents. Incorporates 

internal and NIOSH comments. Initiated by Eugene M. Rollins.

07/21/2006 00 PC-1 Approved page change revision incorporates definitions and 

directions for dose reconstruction for non-presumptive cancers that 

are excluded from the 1951 through 1962 Special Exposure Cohort. 

Text was added or modified on pages 8-9 in Section 6.1. 

Incorporates NIOSH formal review comments. No sections were 

deleted. This revision results in no change to the assigned dose and 

no PER is required. Training required: As determined by the Task 

Manager. Initiated by Eugene M. Rollins. Approval:

Signature on File 07/12/2006

Eugene M. Rollins, TBD Team Leader

Signature on File 07/12/2006

John M. Byrne, Task 3 Manager

Signature on File 07/14/2006

Edward F. Maher, Task 5 Manager

Signature on File 07/20/2006

Kate Kimpan, Project Director

Signature on File 07/21/2006

James W. Neton, Associate Director for Science
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 3 of 135

EFFECTIVE 

DATE

REVISION 

NUMBER DESCRIPTION

01/11/2007 00 PC-2 Approved page change revision incorporates coworker data for 

completion of external dose prior to universal badging in April 1957. 

Text was added or modified on pages 40-42 in Sections 6.3.4.1. 

Added DeMarre reference on page 50. Deleted the term, claimant 

favorable, from the Glossary on page 54. As a result of changes, 

corrected table references on page 46 and 48. No further changes 

occurred as a result of internal formal review. Incorporates NIOSH 

formal review comments. No sections were deleted. This revision 

results in an increase in assigned dose and a PER is required. 

Training required: As determine by the Task Manager. Initiated by 

Eugene M. Rollins. Approval:

Signature on File 12/15/2006

Eugene M. Rollins, TBD Team Leader

Signature on File 12/12/2006

John M. Byrne, Task 3 Manager

Signature on File 12/12/2006

Edward F. Maher, Task 5 Manager

Signature on File 01/11/2007

Kate Kimpan, Project Director

Brant A. Ulsh Signature on File for 01/11/2007

James W. Neton, Associate Director for Science

07/30/2007 01 Approved Revision 01 initiated to add Section 6.4 Dose 

Reconstruction Recommendations, Section 6.5 Claim Analysis 

Methods. Added Attachments A through D. Constitutes a total 

rewrite of the document. Incorporates formal internal and NIOSH 

review comments. Adds Attributions and Annotations section. 

Training required: As determine by the Task Manager. Initiated by 

Eugene M. Rollins. 

05/28/2008 01 PC-1 Approved page change revision to incorporate expanded co-worker 

data on pages 45 - 47, 49, and 50 in Section 6.4. Text was added to 

Section 6.5.1 (page 57) and Section 6.5.2 (page 58) to address hot 

particle issues and the use of the document Hazards to Personnel 

Re-entering the Nevada Test Site Following Nuclear Reactor Tests. 

Changes to attributions and annotations numbering were made on 

pages 59-61, 67, 68, 92, and 95. No sections were deleted. Made 

additional changes to correct the wording in Section 6.5.1 on page 57 

to be consistent with the information in Section 6.3.5.1 on page 36 

regarding the lead covering on the dosimetry and identification of the 

25% as a dose correction factor between the years 1960 and 1965. 

Corrected the dates in Section 6.3.5.1 on page 36 to be consistent 

with the dates in Table 6-1. Added information for Operations BREN 

and HENRE and the Super Kukla reactor. As a result of NIOSH 

formal review, the following pages were changed: 9, 12, 13, 17-19, 

21, 24 - 34, 36, 37, 39 - 42, 44 - 47, 49 - 66, 68, 71 - 78, 93 - 95, 101, 

102, 104, 105, and 110. These changes occurred in the Acronyms 

and Abbreviations Section, Sections 6.1, 6.3, 6.4, 6.5, 6.6, Reference 

Section, Glossary Section and Attachment A. Training required: As 
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 4 of 135

EFFECTIVE 

DATE

REVISION 

NUMBER DESCRIPTION

determined by the Task Manager. Initiated by Eugene M. Rollins. 

Approval:

Signature on File 04/23/2008

Eugene M. Rollins, Document Owner

Signature on File 05/14/2008

John M. Byrne, Task 3 Manager

Signature on File 04/22/2008

Edward F. Maher, Task 5 Manager

Signature on File 05/14/2008

Kate Kimpan, Project Director

Brant A. Ulsh Signature on File for 05/28/2008

James W. Neton, Associate Director for Science

04/15/2010 02 Revision initiated to incorporate definitions and directions for dose 

reconstruction for nonpresumptive cancers that are excluded from 

the 1963 through 1992 Special Exposure Cohort. Changes were 

made to Sections 6.3.5.1 and 6.5.1 to clarify the application of a 25% 

dosimeter correction factor for all energy ranges prior to 1966 to 

account for lead shielding in some dosimeters. Table 6-1, footnote d 

was revised to reflect this change. Added Section 6.1.2. 

Incorporates formal internal and NIOSH review comments. Training 

required: As determined by the Task Manager. Initiated by Eugene 

M. Rollins. 

11/09/2012 03 Revision initiated to fix Table 6-1 for neutron dosimeters (1961–1979, 

1979–1986, and 1987–2000) that included a factor of 2 (for claimant

favorability) based on the discussion in Sections 6.3.4.2.1, 6.3.4.2.2, 

and 6.3.4.2.3. This multiplier has been removed and the language in 

these sections modified. Section 6.3.1.4 – Language was added to 

clarify REECo reporting practices for workers visiting the site and to 

document the correct source of information when evaluating REECo 

employees assigned to TTR. Clarification of the beta-to-gamma ratio 

to be applied before 1966 during most dose reconstructions. 

Language was added to Section 6.4.2.1 to direct the reader to 

Attachment C if a more accurate estimate of the beta to gamma ratio 

was required. Section 6.4.2.2 was moved to Attachment C, Section 

C.2, and Section 6.4.2.3 was moved to Attachment C, Section C. 3. 

In Section 6.5.2 the beta-to-gamma ratio to be used before 1966 was 

clarified. Uncertainty paragraph was deleted because for beta dose 

OTIB-ORAUT-0017 is the methodology used. Section 6.5.1 –

Language added about unmonitored workers and when no external 

monitoring indicated a worker was not on NTS. Incorporates formal 

internal and NIOSH review comments. Constitutes a total rewrite of 

the document. Training required: As determined by the Objective 

Manager.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 5 of 135

TABLE OF CONTENTS

SECTION TITLE PAGE

Acronyms and Abbreviations ............................................................................................................... 9

6.1 Introduction ............................................................................................................................ 12

6.1.1 Purpose...................................................................................................................... 13

6.1.2 Special Exposure Cohort Petition Information for the Nevada Test Site...................... 13

6.1.3 Scope......................................................................................................................... 14

6.2 Dosimetric Basis of Comparison ............................................................................................ 14

6.2.1 Photon Measurement Quantities ................................................................................ 15

6.2.2 Beta Measurement Quantities .................................................................................... 16

6.2.3 Neutron Measurement Quantities ............................................................................... 16

6.3 Dose Reconstruction Parameters........................................................................................... 16

6.3.1 Administrative Practices ............................................................................................. 23

6.3.1.1 Exposure Limits ........................................................................................... 24

6.3.1.2 Dosimeter Exchange.................................................................................... 25

6.3.1.3 Dosimetry Codes for External Monitoring ..................................................... 25

6.3.1.4 Duplicate Dosimetry Records....................................................................... 26

6.3.1.5 Special Issues.............................................................................................. 28

6.3.1.5.1 External Exposure to Hot Particles.............................................. 28

6.3.1.5.2 Intentional Nonuse of Dosimeters ............................................... 29

6.3.2 Individual Monitoring Methods .................................................................................... 29

6.3.2.1 Beta/Gamma Dosimeters ............................................................................. 29

6.3.2.1.1 Beta Dosimetry with Film Badges ............................................... 32

6.3.2.2 Neutron Dosimeters ..................................................................................... 33

6.3.2.2.1 Film Dosimeters.......................................................................... 33

6.3.2.2.2 Thermoluminescent Albedo Dosimeters...................................... 34

6.3.2.2.3 Track-Etch Dosimeters ............................................................... 34

6.3.2.2.4 Panasonic 809............................................................................ 34

6.3.2.3 Extremity Monitoring .................................................................................... 34

6.3.3 Calibration .................................................................................................................. 35

6.3.3.1 Photons........................................................................................................ 35

6.3.3.2 Neutrons ...................................................................................................... 36

6.3.4 Limit of Detection........................................................................................................ 36

6.3.4.1 Gamma Dosimeters ..................................................................................... 36

6.3.4.1.1 Film............................................................................................. 36

6.3.4.1.2 TLD............................................................................................. 37

6.3.4.2 Neutron Dosimeters ..................................................................................... 38

6.3.4.2.1 NTA Film..................................................................................... 38

6.3.4.2.2 TLD Albedo................................................................................. 39

6.3.4.2.3 Track-Etch Dosimeters ............................................................... 39

6.3.5 Workplace Radiation Fields ........................................................................................ 39

6.3.5.1 Photon Fields ............................................................................................... 40

6.3.5.1.1 Nuclear and Ramjet Engine Tests and Other Reactor Tests....... 40

6.3.5.1.2 Noble Gas Exposure................................................................... 41

6.3.5.2 Beta Particle Fields ...................................................................................... 42

6.3.5.3 Neutron Fields.............................................................................................. 43

6.3.5.3.1 Weapons-Related Neutron Fields ............................................... 43

6.3.5.3.2 Isotopic Neutron Sources............................................................ 44
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 6 of 135

6.3.5.3.3 Reactor Operations..................................................................... 44

6.3.5.3.4 Dosimetric Characteristics of Neutron Fields .............................. 44

6.4 Dose Reconstruction Recommendations ............................................................................... 46

6.4.1 Photon Radiation........................................................................................................ 46

6.4.1.1 Unmonitored External Radiation Photon Dose ............................................. 46

6.4.1.2 Adjustments to Recorded External Photon Radiation Dose.......................... 46

6.4.1.3 Missed Recorded External Radiation Dose .................................................. 47

6.4.1.4 IREP Input Photon Radiation Energy Fractions............................................ 50

6.4.1.5 Determination of Radiation Quantities and Exposure Geometry................... 50

6.4.1.6 Correction Factors for External Environmental Dose.................................... 50

6.4.2 Beta Radiation............................................................................................................ 51

6.4.2.1 Evaluation of Beta Exposure Geometries..................................................... 51

6.4.2.2 Beta-Photon Ratios for Other NTS Operational Areas.................................. 52

6.4.2.3 Estimation of Beta-Photon Exposure............................................................ 52

6.4.2.4 Skin Contamination...................................................................................... 53

6.4.3 Neutron Radiation....................................................................................................... 54

6.4.3.1 Energy Dependence of Dosimetric Quantities .............................................. 54

6.4.3.2 Neutron-Photon Dose Ratios ....................................................................... 54

6.4.3.3 Special Considerations ................................................................................ 56

6.5 Claim Analysis Methods......................................................................................................... 56

6.5.1 Photon Dose............................................................................................................... 57

6.5.2 Beta Dose................................................................................................................... 58

6.5.3 Neutron Dose ............................................................................................................. 60

6.6 Attributions and Annotations .................................................................................................. 61

References ........................................................................................................................................ 68

Glossary ............................................................................................................................................ 80

ATTACHMENT A, DOSIMETRY TECHNOLOGY.............................................................................. 90

ATTACHMENT B, ENERGY GROUP ALLOCATION OF PHOTON EXPOSURE AND 

PERSONAL DOSE EQUIVALENT BY NTS WORK AREA AND 

OPERATION...................................................................................................... 111

ATTACHMENT C, BETA-PHOTON RATIO ESTIMATES FOR NTS WORK AREAS AND 

OPERATIONS ................................................................................................... 113

ATTACHMENT D, NEUTRON DOSES FROM ATMOSPHERIC NUCLEAR TESTS ....................... 130
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 7 of 135

LIST OF TABLES

TABLE TITLE PAGE

6-1 NTS external dosimetry, 1951 to present ............................................................................... 17

6-2 NTS film badge issue locations .............................................................................................. 24

6-3 Type of dose codes for external dosimetry............................................................................. 26

6-4 Film damage and irregularity codes ....................................................................................... 26

6-5 Characteristics of ISO reference radiation radionuclide sources used for neutron 

dosimeter calibration..............................................................................................................36

6-6 Background dose equivalent rates ......................................................................................... 37

6-7 NTS Panasonic UD-802 detection limits for DOELAP categories........................................... 38

6-8 Primary iodines and fission product noble gases with half-lives greater than 1 hour............... 41

6-9 Areas and operations where neutron exposure was possible................................................. 43

6-10 RBE, quality factors, or weighting factors for neutrons ........................................................... 45

6-11 NTS calculated dose to apply as coworker dose .................................................................... 47

6-12 NTS associated measured doses........................................................................................... 48

6-13 NTS work area and operation-dependent photon fractions..................................................... 50

6-14 Fraction of Hp(10,0°) by energy band for selected spectra ..................................................... 54

6-15 Neutron field characteristics associated with NTS operational areas...................................... 55

6-16 Neutron-photon dose ratios for isotopic sources .................................................................... 56

6-17 Beta dose assignment............................................................................................................ 59

A-1 Panasonic UD-802 dosimeter configuration ........................................................................... 96

A-2 Photon exposure to dose correction factors for NIST reference radiations ............................. 98

A-3 Bias values and uncertainties for NTS DOELAP accreditation categories ............................ 105

A-4 Fraction of Hp(10,0) for neutrons with energies less than En................................................. 110

B-1 Work area and operation-dependent photon fractions.......................................................... 112

C-1 Calculated beta-photon ratios for mixed fission and activation products from Sedan 

and tests from Operations RANGER, BUSTER-JANGLE, TUMBLER-SNAPPER, 

AND UPSHOT-KNOTHOLE-contaminated surfaces ............................................................ 118

C-2 Beta-photon ratios for exposure from surface contamination................................................ 120

C-3 Calculated beta-photon ratios for mixed fission and activation products from 

Operations RANGER, BUSTER-JANGLE, TUMBLER-SNAPPER, and UPSHOTKNOTHOLE tests – immersion in a contaminated semi-infinite cloud................................... 121

C-4 Photon dose constants, beta dose rates, and beta-photon ratios for NTS 

radionuclides........................................................................................................................ 124

C-5 Beta-photon ratios for exposure from immersion in an infinite cloud of mixed fission 

and activation products ........................................................................................................ 126

C-6 Geometry-dependent beta-photon ratios for NTS radionuclides........................................... 128
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 8 of 135

LIST OF FIGURES

FIGURE TITLE PAGE

6-1 Security badge for NRDS TLD ............................................................................................... 30

6-2 REECo badge holder .............................................................................................................31

6-3 REECo security badge holder with Panasonic UD-802 dosimeter.......................................... 31

6-4 X-ray energy dependence of PICs ......................................................................................... 32

6-5 Panasonic UD-807 extremity dosimeter ................................................................................. 35

6-6 Ratio of fluence to dose conversion coefficients for personal dose equivalent Hp(10, 

0°) to conversion coefficients for dose equivalent H ............................................................... 45

A-1 Typical photon energy response curves for Li2B4O7 and CaSO4 TLD elements...................... 96

A-2 Panasonic UD-802 energy responses.................................................................................... 97

A-3 NTA film neutron energy response......................................................................................... 99

A-4 Neutron energy response of NTS albedo dosimeters ........................................................... 101

A-5 Neutron energy response of NTS TED................................................................................. 102

A-6 Electrochemically etched CR-39 dose response .................................................................. 103

A-7 Lethargy neutron spectra for 252Cf calibration source with and without room scatter............. 107

A-8 Lethargy neutron spectra for 252Cf with various thickness moderators.................................. 108

A-9 Lethargy neutron spectra for an 241AmBe calibration source with and without room 

scatter.................................................................................................................................. 108

A-10 Fraction of Hp(10,0) due to neutrons with energies less than En ........................................... 109

C-1 Beta-gamma dose ratios for contaminated surfaces used for Pacific tests ........................... 115

C-2 Calculated beta-photon ratios for skin for 11 atmospheric tests from Operation 

UPSHOT-KNOTHOLE ......................................................................................................... 116

C-3 Calculated beta-photon ratios for skin for nuclear rockets and ramjet test effluents ............. 117

C-4 Calculated beta-photon ratios for immersion in a contaminated semi-infinite cloud –

maximum of values from 11 atmospheric test radionuclide inventories ................................ 123

D-1 Neutron spectra for fission weapon and thermonuclear weapon per kiloton total 

energy yield.......................................................................................................................... 131

D-2 Initial neutron dose per kiloton as a function of slant range from fission and 

thermonuclear air bursts, based on 0.9 normal sea-level air density .................................... 132

D-3 Comparison of the initial neutron dose per kiloton as a function of slant range from 

fission and thermonuclear air bursts..................................................................................... 133

D-4 Neutron dose equivalent as a function of slant range for atmospheric detonations 

with a range of weapon yields .............................................................................................. 133

D-5 1-mrem neutron isodose curves for continental detonations................................................. 135
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 9 of 135

ACRONYMS AND ABBREVIATIONS

ABS acrylonitrile butadiene styrene

AEC U.S. Atomic Energy Commission

ALARA as low as is reasonably achievable

BREN Bare Reactor Experiment Nevada

C coulomb

cm centimeter

CR-39 Columbia Resin-39 

d day

DOD U.S. Department of Defense

DOE U.S. Department of Energy

DOELAP DOE Laboratory Accreditation Program

DOL U.S. Department of Labor

ECF element correction factor

EEOICPA Energy Employees Occupational Illness Compensation Program Act

EG&G Edgerton, Germeshausen and Grier

eV electron-volt

ft foot

g gram

GM geometric mean

GSD geometric standard deviation

Gy gray

H&N Holmes & Narver

HENRE High Energy Neutron Reactions Experiment 

Hp(d) personal dose equivalent (d = depth in millimeters)

HPRR Health Physics Research Reactor

hr hour

HRA High Radiation Area

Hz hertz

IARC International Agency for Research on Cancer

ICRP International Commission for Radiological Protection

ICRU International Commission on Radiological Units and Measurements

IDO Idaho Operations Office

in. inch

ISO International Organization for Standardization

IREP Interactive RadioEpidemiological Program

keV kiloelectron-volt, 1,000 electron-volts

kg kilogram

km kilometer

kT kiloton

kV kilovolt

LANL Los Alamos National Laboratory 
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 10 of 135

LASL Los Alamos Scientific Laboratory

LLD lower limit of detection

LLNL Lawrence Livermore National Laboratory

LRL Lawrence Radiation Laboratory

m meter

MBq megabecquerel

MDL minimum detection level

MeV megaelectron-volt, 1 million electron-volts

mg milligram

mGy milligray

mi mile

min minute

mm millimeter

MMD maximum missed dose

mo month

mR milliroentgen

mrem millirem

mSv millisievert

MT megaton

NBS National Bureau of Standards

NCRP National Council on Radiation Protection and Measurements

NIOSH National Institute for Occupational Safety and Health

NIST National Institute of Standards and Technology

nm nanometer

NPG Nevada Proving Ground

NRDS Nuclear Rocket Development Station

NTA Neutron Track Emulsion, type A

NTS Nevada Test Site

ORAU Oak Ridge Associated Universities

OW open window

Pan Am Pan American Airways

PIC pocket ionization chamber

POC probability of causation

QA quality assurance

QC quality control

R roentgen

RADEX controlled radiation exclusion (area)

RBE relative biological effectiveness

RCF run calibration factor

REECo Reynolds Electrical and Engineering Company

RSN Raytheon Services Nevada

SEC Special Exposure Cohort

SNL Sandia National Laboratories

Sv sievert

TBD technical basis document
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 11 of 135

TED track-etch dosimeter

TLD thermoluminescent dosimeter

TRU transuranic

TTR Tonopah Test Range

U.S.C. United States Code

wk week

yd yard

YMP Yucca Mountain Project

yr year

Z atomic number

μCi microcurie

μm micrometer

°C degree Celsius

°F degree Fahrenheit

§ section or sections
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 12 of 135

6.1 INTRODUCTION

Technical basis documents and site profile documents are not official determinations made by the 

National Institute for Occupational Safety and Health (NIOSH) but are rather general working 

documents that provide historical background information and guidance to assist in the preparation of 

dose reconstructions at particular sites or categories of sites. They will be revised in the event 

additional relevant information is obtained about the affected site(s). These documents may be used 

to assist NIOSH staff in the completion of the individual work required for each dose reconstruction.

In this document the word “facility” is used as a general term for an area, building, or group of 

buildings that served a specific purpose at a site. It does not necessarily connote an “atomic weapons 

employer facility” or a “Department of Energy [DOE] facility” as defined in the Energy Employees 

Occupational Illness Compensation Program Act [EEOICPA; 42 U.S.C. § 7384l(5) and (12)]. 

EEOICPA defines a DOE facility as “any building, structure, or premise, including the grounds upon 

which such building, structure, or premise is located ... in which operations are, or have been, 

conducted by, or on behalf of, the Department of Energy (except for buildings, structures, premises, 

grounds, or operations ... pertaining to the Naval Nuclear Propulsion Program)” [42 U.S.C. § 

7384l(12)]. Accordingly, except for the exclusion for the Naval Nuclear Propulsion Program noted 

above, any facility that performs or performed DOE operations of any nature whatsoever is a DOE 

facility encompassed by EEOICPA.

For employees of DOE or its contractors with cancer, the DOE facility definition only determines 

eligibility for a dose reconstruction, which is a prerequisite to a compensation decision (except for 

members of the Special Exposure Cohort). The compensation decision for cancer claimants is based 

on a section of the statute entitled “Exposure in the Performance of Duty.” That provision [42 U.S.C. § 

7384n(b)] says that an individual with cancer “shall be determined to have sustained that cancer in the 

performance of duty for purposes of the compensation program if, and only if, the cancer ... was at 

least as likely as not related to employment at the facility [where the employee worked], as 

determined in accordance with the POC [probability of causation1

] guidelines established under 

subsection (c) ...” [42 U.S.C. § 7384n(b)]. Neither the statute nor the probability of causation 

guidelines (nor the dose reconstruction regulation, 42 C.F.R. Pt. 82) restrict the “performance of duty”

referred to in 42 U.S.C. § 7384n(b) to nuclear weapons work (NIOSH 2010a).

The statute also includes a definition of a DOE facility that excludes “buildings, structures, premises, 

grounds, or operations covered by Executive Order No. 12344, dated February 1, 1982 (42 U.S.C. 

7158 note), pertaining to the Naval Nuclear Propulsion Program” [42 U.S.C. § 7384l(12)]. While this 

definition excludes Naval Nuclear Propulsion Facilities from being covered under the Act, the section 

of EEOICPA that deals with the compensation decision for covered employees with cancer [i.e., 42 

U.S.C. § 7384n(b), entitled “Exposure in the Performance of Duty”] does not contain such an 

exclusion. Therefore, the statute requires NIOSH to include all occupationally-derived radiation 

exposures at covered facilities in its dose reconstructions for employees at DOE facilities, including 

radiation exposures related to the Naval Nuclear Propulsion Program. As a result, all internal and 

external occupational radiation exposures are considered valid for inclusion in a dose reconstruction. 

No efforts are made to determine the eligibility of any fraction of total measured exposure for inclusion 

in dose reconstruction. NIOSH, however, does not consider the following exposures to be 

occupationally derived (NIOSH 2010a):

• Background radiation, including radiation from naturally occurring radon present in 

conventional structures

• Radiation from X-rays received in the diagnosis of injuries or illnesses or for therapeutic 

reasons

 1 The U.S. Department of Labor (DOL) is ultimately responsible under the EEOICPA for determining the POC.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 13 of 135

6.1.1 Purpose

The purpose of this document is to describe Nevada Test Site (NTS) external dosimetry systems and 

practices. This information will be used as needed to evaluate external occupational doses for 

EEOICPA claimants. 

Attributions and annotations, indicated by bracketed callouts and used to identify the source, 

justification, or clarification of the associated information, are presented in Section 6.6.

6.1.2 Special Exposure Cohort Petition Information for the Nevada Test Site

The status of Special Exposure Cohort (SEC) petitions for NTS is:

Classes Added to the SEC

• Department of Energy (DOE) employees or DOE contractor or subcontractor employees who 

worked at the Nevada Test Site from January 27, 1951 through December 31, 1962 for a

number of work days aggregating at least 250 work days, either solely under this employment 

or in combination with work days within the parameters (excluding aggregate work day 

requirements) established for other classes of employees included in the SEC, and who were 

monitored or should have been monitored (Leavitt 2006). 

• All employees of the Department of Energy, its predecessor agencies, and its contractors and 

subcontractors who worked at the Nevada Test Site from January 1, 1963 through December 

31, 1992 for a number of work days aggregating at least 250 work days, occurring either solely 

under this employment or in combination with work days within the parameters established for 

one or more other classes of employees included in the SEC (Sebelius 2010). 

NIOSH has determined, and the Secretary of Health and Human Services has concurred, that in the 

absence of bioassay results for the worker, internal doses cannot be reconstructed between 1951 and 

1962 inclusive for an Energy Employee (EE). Based on the SEC petition evaluation, internal dose is 

not to be reconstructed for work before 1963 unless a worker has specific bioassay results that can be 

directly related to an event or incident. Any bioassay results in the DOE files for NTS workers before 

1963 should be assumed to be valid [HASL-300 (Harley 1976) procedures were used in the early 

bioassay program)]; therefore, these results can be used to evaluate internal dose. Much of the 

internal monitoring for individuals during the SEC period was event related. However, certain job 

classifications required routine monitoring. These included radiation safety personnel, industrial 

hygienist, and security personnel. 

NIOSH has determined, and the Secretary of Health and Human Services has concurred, that NIOSH 

lacks sufficient information that would allow it to adequately estimate internal exposures during the 

period 1963 through 1992 (Sebelius 2010). NIOSH believes that the cessation of nuclear testing, 

coupled with the implementation of the 1993 NTS internal technical basis document that 

demonstrates NTS compliance with 10 CFR Part 835, supports NIOSH’s ability to bound internal dose 

for the evaluated class starting in 1993.

Dose reconstruction guidance in this technical basis document (TBD) for periods before January 1, 

1993 is presented to provide a technical basis for partial dose reconstructions for nonpresumptive 

cancers not covered in the SEC classes through December 31, 1992. Although NIOSH found that it is 

not possible to bound total dose for the proposed classes, it intends to use internal and external 

monitoring data that might become available for an individual claim (and that can be interpreted using 

its existing dose reconstruction processes or procedures). Therefore, dose reconstructions for 

individuals employed at NTS during the period from 1951 through December 31, 1992, but who do not 

qualify for inclusion in the SEC, can be performed using these data as appropriate.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 14 of 135

6.1.3 Scope

NTS operations played an important role in the U.S. nuclear weapons program. During the period 

when the United States tested nuclear weapons, radiation exposure monitoring of Energy Employee 

civilian and U.S. military personnel associated with that testing was performed using portable radiation 

instrumentation and personnel film dosimeters (Lalos 1989). Operations at NTS involved atmospheric 

and underground weapons tests, experimental reactor tests intended for aircraft and rocket 

propulsion, and low-level transuranic (TRU) waste disposal (REECo 1995a; Allen and Schoengold 

1995). NTS workers received exposures from a range of fission and activation products from test 

programs beginning in the early 1950s. Exposure venues of particular significance include sample 

and measurement device recovery to assess weapon yields, terrestrial and airborne fallout plume 

tracking, aircraft operations (when used) for weapon assembly deployment and surveillance, and 

postevent decontamination processes [1].

Radiation monitoring and control programs instituted with the mission of NTS included personal 

dosimetry, area monitoring, source term characterization, and measurements of fallout 

(contamination) dispersion [1]. As NTS test programs progressed, efforts to measure exposures and 

limit dose improved (Allen and Schoengold 1995; Author unknown undated). The atmospheric 

nuclear test series consisted of a number of operations between 1951 and 1958 and from 1961 to 

1963 (DOE 2000). Each operation consisted of a number of individual tests. Underground nuclear 

testing occurred at NTS as early as 1958 and continued to 1992 (DOE 2000).

This TBD contains supporting documentation to assist in the evaluation of occupational external 

doses from these processes using the methodology in OCAS-IG-001, External Dose Reconstruction 

Implementation Guideline (NIOSH 2007b). NIOSH considers the available data and methods for 

performing external dose reconstruction to be adequate for estimating with sufficient accuracy the 

external radiation doses at NTS from 1951 to the present.

6.2 DOSIMETRIC BASIS OF COMPARISON

Since the start of the Manhattan Engineer District program in the early 1940s, various dosimetric 

concepts and quantities have been used to measure and record occupational exposure from external 

radiation sources. The selection of the measurement quantities to be used for radiation protection 

was initially based on the radiation interaction properties of the primary radiations of interest –

photons (X- and gamma rays), electrons (beta particles), and neutrons – and the measurement 

methods employed. Sections 6.2.1 to 6.2.3 discuss these quantities briefly.

The problem with the use of different quantities for different radiations is that it prevented direct 

comparison of measurements of the three primary radiation qualities. In the 1950s, the industry 

recognized a need for a special quantity to facilitate comparison of measurements of the various 

radiation qualities. The concepts of quality factor and dose equivalent were introduced by the 

International Commission for Radiological Protection (ICRP) and the International Commission on 

Radiation Units and Measurements (ICRU) in 1962 (ICRU 1962). The special unit of dose equivalent

was the rem. As early as 1961, NTS radiation exposure criteria and guidance were given in rem 

(AEC 1961).

In 1985, the ICRU defined a new set of operational quantities defined as radiation quantities for 

operational radiation protection measurement purposes (ICRU 1993). These quantities have a 

common definition for the three primary radiation qualities, so they have the advantage that they 

provide a means of direct comparison of measurements for these radiations.

The operational quantity recommended for individual or personal monitoring is the personal dose 

equivalent Hp(d), where d is the depth (in millimeters) and represents the point of reference for dose in 
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 15 of 135

tissue. For weakly penetrating radiation of significance to skin dose, d is 0.07 mm and the operational 

quantity is noted as Hp(0.07). For strongly penetrating radiation of significance to whole-body dose, d

is 10 mm and the operational quantity is noted as Hp(10). Both Hp(0.07) and Hp(10) are 

recommended as the operational quantities to be recorded for radiological protection proposed by the 

ICRU (1993).

These personal dose equivalents, Hp(0.07) and Hp(10), have been used in the DOE Laboratory 

Accreditation Program (DOELAP) for accreditation of the Department’s personnel dosimetry systems 

since the 1980s (DOE 1986a). The International Agency for Research on Cancer (IARC) Three 

Country Combined Study (Fix et al. 1997; Fix, Wilson, and Baumgartner 1997) and the IARC 

Collaborative Study (Thierry-Chef et al. 2002) selected Hp(10) as the quantity to assess error in 

recorded whole-body dose for workers in IARC nuclear worker epidemiologic studies.

6.2.1 Photon Measurement Quantities

From the beginning of operations at NTS, exposure was used as the basis of photon measurement. 

When in italics, the term exposure designates a radiation measurement quantity based on the 

electrical charges created in air due to interaction of photons. It is specific to air and photon

measurement. The unit of exposure is coulomb per kilogram, and the special unit of exposure is the 

roentgen (1 R = 2.58 × 10-4 C/kg). Although the term exposure dose was used in the mid-1950s, 

exposure is not dose because it is not a measure of energy deposition in mass of material. The ICRU 

recommended the use of exposure in 1962 (ICRU 1962). However, dose in soft tissue can be 

determined from exposure by the following relationship,

D = fX (6-1)

where f is a constant that relates the exposure X in roentgen for a given photon energy to dose in soft 

tissue, and was typically taken to be 0.877 rad/R. Because 1 R is numerically slightly greater than 1 

rad, it is favorable to claimants to assume a numerical equivalence of the two quantities. Therefore, if 

the energy, and hence the constant f, of the exposing photon radiation are unknown, it is appropriate 

to assume that 1 R equals 1 rad, which equals 10 mGy. 

Instrument and dosimeter calibrations, dose measurements, and dose records were made in terms of 

exposure in units of milliroentgen. However, as noted above, from 1961 quarterly and annual limits 

were specified in millirem (AEC 1961), where the general term dose was used without specific 

reference to dose equivalent. As a result, individual monitoring results at NTS were recorded in 

millirem, although the measurements had been made in terms of exposure. However, for photons, 

the values of exposure and dose equivalent were considered to be essentially the same (Griffith 

2004a; Brady and Iverson 1968). In effect, a de facto conversion factor of 1 rem/R was used for dose 

recording purposes.

Until July 1970, individual monitoring at the Nuclear Rocket Development Station (NRDS) was 

provided by the Reynolds Electrical and Engineering Company (REECo) using the standard NTS film 

dosimeter (Boone, Bennett, and Adams 1970). From July 1970 until NRDS operations ended in 

January 1973, individual monitoring was conducted by Pan American World Airways (Pan Am) using 

thermoluminescent dosimeters (TLDs) (Boone, Bennett, and Adams, 1970). However, throughout 

NRDS operations, dosimeters were calibrated in terms of exposure and doses were recorded in 

millirem.

Beginning in 1987, NTS occupational exposures were recorded in terms of personal dose equivalent, 

Hp(d). For exposures during the period from 1962 to 1986, dose reconstructors should use the 

recorded photon dose values in terms of exposure, together with the Exposure to Organ Dose 

coefficients in Appendix B of OCAS-IG-001, External Dose Reconstruction Implementation Guideline
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 16 of 135

(NIOSH 2007b) to determine organ dose. Since 1987, the recorded values are in terms of Hp(10), 

and the Deep-Dose-Equivalent-to-Organ-Dose conversion factors (from NIOSH 2007b, Appendix B) 

should be used.

6.2.2 Beta Measurement Quantities

As was common practice, NTS beta particle measurements were made in terms of absorbed dose, D 

[2]. Until introduction of the ICRU-defined operational quantities in 1985, beta doses were recorded 

as millirad. Because the quality factor for electrons (beta particles) was set at 1, the absorbed dose 

values are considered to be numerically equivalent to dose equivalent (ICRU 1993).

6.2.3 Neutron Measurement Quantities

The basis for comparison for neutron radiation is complex because, historically, the calibration of 

dosimeters to measure neutron dose was based on different dosimetric quantities (such as first 

collision dose and multiple collision dose). However, the neutron dose equivalent specified by the 

National Council on Radiation Protection and Measurements (NCRP) has been used since 1971. 

Evaluation of the numerical difference in comparison with the Hp(10) dose used in DOELAP 

performance testing is used to establish relative values of the dose conversion factors for the dose 

quantities in conjunction with characteristics of the neutron dosimeter response characteristics and 

workplace radiation fields.

Dose reconstructors should convert recorded neutron dose to Hp(10) using the bias values in 

Table 6-1, and use the Deep-Dose-Equivalent-to-Organ-Dose conversion factors from Appendix B of 

OCAS-IG-001 (NIOSH 2007b) to calculate the appropriate organ doses.

6.3 DOSE RECONSTRUCTION PARAMETERS

Overall accuracy and precision of the original recorded individual worker doses and their 

comparability to be considered in using OCAS-IG-001 (NIOSH 2007b) guidelines depend on the 

following factors (Fix et al. 1997; Fix, Wilson, and Baumgartner 1997):

• Administrative practices adopted by facilities to calculate and record personnel dose based on 

technical, administrative, and statutory compliance considerations.

• Dosimetry technology, which includes physical capabilities of the dosimetry system such as 

the response to different types and energies of radiation, in particular to mixed radiation fields.

• Calibration of monitoring systems and similarity of the methods of calibration to sources of 

exposure in the workplace.

• Workplace radiation fields that could include mixed types of radiation, variations in exposure 

geometries, and environmental conditions.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 17 of 135

Table 6-1. NTS external dosimetry, 1951 to present (Griffith 2007; Allen and Schoengold 1995; BN 2001).

Dates Operation Dosimeter Description Issue and exchange

Measurement 

quantity Bias GSD MDL MMD

1/27/51–

2/6/51

RANGER, 

1/27/51–

2/6/51

DuPont 552 

Packet

DuPont 552 packet, including:

Type 502 low-range element (0.05 

to 10 R)

Type 510 high-range element (5 to 

50 R)

Brass/cadmium filters, 0.020-in. 

thick (symmetrical coverage on 

both sides with open area)

(Shipman et al. 1951)

Issued to personnel who 

entered radiation areas and 

air crews. Exchanged daily. 

Supplemented with selfreading pocket dosimeters 

for exposure control. 

(Sometimes underresponded in relation to 

film.)

Photon: 

Exposure

1.1 1.23a 40 mR 5.0 R

10/22/51

–

11/29/51

BUSTER–

JANGLE, 

10/22/51–

11/29/51

DuPont 553 

Packet

DuPont 553 packet, including:

Type 502 low-range element (0.02 

to 10 R)

Type 510 high-range element (5 to 

50 R)

Type 606 high-range element (10 

to 300 R)

Brass/cadmium filters, 0.020-in. 

thick (symmetrical coverage on 

both sides with open area). Lead 

filters, 0.020-in. thick, possibly used 

with badges issued to Camp Desert 

Rock personnel (Kean 1951; 

Shipman et al. 1951; Storm 1951)

Issued to personnel who 

entered radiation areas, air 

crews, and people with 

potential for exposure from 

experiments.

Exchanged daily.

Self-reading pocket 

dosimeters sometimes 

used.

Badges for Camp Desert 

Rock personnel were 

exchanged following 

deployment.

Photon: 

Exposure

1.1 1.23a 40 mR 5.0 R

4/1/52–

6/5/52

TUMBLER–

SNAPPER, 

4/1/52–

6/5/52

DuPont 558 

Packet

NPG personnel

DuPont 558 packet, including:

Lead filters, 0.028-in. thick, 

symmetrical on both sides (except 

first test – ABLE)

Type 508 low-range element (0.01 

to 6 R)

Type 1290 high-range element (20 

to 3,000 R)

(Brady and Nelson 1985)

Issued to NPG test 

participants. Exchanged 

daily.

Photon: 

Exposure

1.6 1.23a 40 mR 5.0 R
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 18 of 135

Dates Operation Dosimeter Description Issue and exchange

Measurement 

quantity Bias GSD MDL MMD

DuPont

Type 502

Debris cloud sampling pilots

LASL badge

Type 502 low-range element (0.02 

to 10 R)

Brass/cadmium filters, 0.020-in. 

thick (symmetrical coverage on 

both sides with open area)

Exchanged daily. Photon: 

Exposure

1.4 1.19a 40 mR 5.0 R

1953–

June 

1960

All of NTS

UPSHOT–

KNOTHOLE, 

3/17/53–

6/4/53

DuPont 559 

Packet

DuPont 559 packet, including:

Type 502 low-range element (0.02 

to 10 R)

Type 606 high-range element (10 

to 300 R)

Lead filters, 0.028-in. thick 

(symmetrical coverage on both 

sides with open area)

(Brady and Nelson 1985; Collison 

1953)

Exchanged daily. 

(Exposures might have 

included prompt radiation, 

including neutrons.)

Photon: 

Exposure

0.9 1.28a 40 mR 5.0 R

All of NTS

TEAPOT, 

2/18/55–

5/15/55

DuPont 559 packet, including:

Type 502 low-range element (0.02 

to 10 R)

Type 606 high-range element (10 

to 300 R)

Lead filters, 0.028-in. thick 

(symmetrical coverage on both 

sides with open area)

(Collison 1955)

Issued to essentially all test 

participants.

Exchanged daily.

(Exposures might have 

included prompt radiation, 

including neutrons.)

Photon: 

Exposure

0.9 1.28a 40 mR 5.0 R

All of NTS

PLUMBOB, 

4/24/57–

10/07/57

Nevada Test Organization

DuPont 559 packet, including:

Type 502 low-range element (0.02 

to 10 R)

Type 606 high-range element (10 

to 300 R)

Lead filters, 0.028-in. thick 

(symmetrical coverage on both 

sides with open area)

Exchanged monthly and 

following radiation area 

work.

Photon: 

Exposure

1.0 1.19a 40 mR 0.240 R
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 19 of 135

Dates Operation Dosimeter Description Issue and exchange

Measurement 

quantity Bias GSD MDL MMD

All of NTS,

PLUMBOB, 

4/24/57–

10/07/57

Camp Desert Rock personnel

DuPont 559 Packet, including:

Type 502 low-range element (0.02 

to 10 R)

Type 606 high-range element (10 

to 300 R)

Aluminum, copper, and tin/lead 

laminate filters with open area.

Exchanged at various 

intervals.

Photon: 

Exposure

1.0 1.19a 40 mR 5.0 R

All of NTS

HARDTACK 

2, 9/19/58–

10/30/58

DuPont 559 packet, including:

Type 502 low-range element (0.02 

to 10 R)

Type 834 high-range element (5 to 

800 R)

Lead filters, 0.028-in. thick 

(symmetrical coverage on both 

sides with open area). Packet and 

filter enclosed in 0.004-in.-thick 

plastic bag.

Exchanged monthly and on 

exit from radiation areas if 

≥100 mR was suspected.

Photon: 

Exposure

1.0 1.19a 40 mR 5.0 R

July 

1960 –

1965

All of NTS

DOMINIC II 

(SUNBEAM), 

7/7/62–

7/17/62

DuPont 301-

4 Packet 

(also known 

as DuPont 

Type 556)

DuPont 301-4 packet, including:

Type 508 low-range element (0.03 

to 5 R)

Type 834 high-range element (5 to 

800 R)

Lead filters, 0.028-in. thick 

(symmetrical coverage on both 

sides with open area)

Packet covered with 0.004-in.-thick 

plastic bag.

Exchanged monthly for 

general exposures and on 

exit from radiation areas for 

exposures likely to exceed 

100 mR.

Photon: 

Exposureb

1.0 1.23a Photon: 

40 mR

Photon:

240 mR

1966–

Feb. 

1971

All of NTS DuPont Type 

556 Packet

DuPont Type 556 film pack 

Type 508 (519 also referenced) 

low-range element (0.03 to 5 R)

Type 834 high-range element (10 

to 1,000 R)

Four-area filter described: 

tantalum-cadmium, tantalum, 

Teflon, open

With fast neutron pack, dosimeter 

was sensitive to mixed fields with 

thermal and fast neutrons, X-rays, 

beta, and gamma

Exchanged monthly for 

general exposures and on 

exit from radiation areas for 

exposures likely to exceed 

100 mR.

Photon: 

Exposure

Beta:

Absorbed 

dosec

1.0 1.23a,

d

Photon: 

40 mR

Beta: 

40 

mreme

Photon:

240 mR

Beta:

240 

mrem
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 20 of 135

Dates Operation Dosimeter Description Issue and exchange

Measurement 

quantity Bias GSD MDL MMD

March 

1971–

1986

All of NTS Kodak Type 

III

Low-range element (0.03 to 10 R)

High-range element (10 to 800 R)

Exchanged monthly for 

general exposures and on 

exit from radiation areas for 

exposures likely to exceed 

100 mR.

Photon: 

Exposure

Beta:

Absorbed 

dosec

1.0 1.23d Photon:

30 mRf

Beta: 

30 

mremf

Photon:

180 mR

Beta:

180 

mrem

1970–

1972

NRDS Pan Am TLD Two-element LiF, Type 700 Quarterly Photon: 

Exposure

Beta:

Absorbed 

dosec

0.8 1.23d Photon:

15 mRg

Beta: 

15 

mremg

Photon:

30 mR

Beta:

30 mrem

1987–

Present

All of NTS Panasonic 

802

Four-element TLD – two 

Li2B4O7:Cu chips and two 

CaSO4:Tm chips. Filtration 

provided to determine gamma, 

deep dose, shallow dose, beta, and 

eye dose. NTS badge holder used 

from 1987. Used for beta gamma 

only.

Quarterly Personal dose 

equivalent, 

Hp(d)h

0.9 1.23d Photon:

30 

mrem

Beta:

25 

mrem

Photon:

60 mrem

Beta:

50 mrem

2001–

present

All of NTS Panasonic 

809 

combination 

dosimeter

Multielement TLD containing four 

elements.

E1 - gamma-sensitive 7

Li2

11B4O7(Cu), enriched to 99.99% 

in 7

Li.

E2, E3 and E4 – neutron-sensitive 

6

Li2

10B4O7(Cu) chips. Li-6 enriched 

to 95.33% and B-10 enriched to 

94.64%. Elements are shielded 

with tin and cadmium on front and 

back, in various combinations. 

Issued for mixed beta, gamma, 

neutron.

Quarterly Personal dose 

equivalent, 

Hp(d)h

0.9 1.23d Photon:

30 

mremg

Beta:

25 

mremg

Photon:

60 mrem

Beta:

50 mrem
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 21 of 135

Dates Operation Dosimeter Description Issue and exchange

Measurement 

quantity Bias GSD MDL MMD

Neutron dosimeters

1961–

1979

Areas and 

operations 

where the 

potential for 

neutron 

exposure 

existed

Kodak NTA Responds to neutrons with 

energies above 0.8 MeV; range 

under near-ideal conditions 0.1 to 

few rem of neutrons; high gamma 

doses might mask neutron tracks.

Exchanged monthly for 

general exposures and on 

exit from radiation areas for 

exposures likely to exceed 

100 mR.

Dose 

equivalenti

0.5.j 1.52d 50 

mremj,k

300 

mrem

1979–

1986

Albedo 

dosimeter

Hankins-type albedo dosimeter. 

Consists of four pairs of TLD-600 

and TLD-700 (6

LiF and 7LiF) in 

cadmium pillbox for thermal 

neutron suppression. High 

sensitivity to low-energy neutrons, 

with decreasing response as 

energy increases.

Monthly

Issued only to individuals 

with potential for exposure 

to neutrons

Dose 

equivalenti

1.0.j 1.23d 10 

mremi,j

60 mrem

1987–

2000

TED Three pieces of CR-39 plastic used 

to detect neutrons with energies 

above 100 keV. 

Quarterly, except for limited 

number of workers 

(radiographers, well 

loggers, and personnel 

routinely entering HRAs).

Issued only to individuals 

with potential for exposure 

to neutrons

Dose 

equivalenti

0.9j 1.23d 30 

mremj,k

60 mrem
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 22 of 135

Dates Operation Dosimeter Description Issue and exchange

Measurement 

quantity Bias GSD MDL MMD

2001–

present

Panasonic Multielement TLD containing four 

elements.

E1 - gamma-sensitive 7

Li2

11B4O7(Cu), enriched to 99.99% 

in 7

Li.

E2, E3 and E4 – neutron-sensitive 

6

Li2

10B4O7(Cu) chips. Li-6 enriched 

to 95.33% and B-10 enriched to 

94.64%. Elements are shielded 

with tin and cadmium on front and 

back, in various combinations. 

Includes CR-39.

Quarterly, except for limited 

number of workers 

(radiographers, well 

loggers, and personnel 

routinely entering HRAs).

Issued only to individuals 

with potential for exposure 

to neutrons.

Dose 

equivalenth

1.0j 1.23d 40 

mremj,k

80 mrem

809 

combination 

dosimeter.

a. Based on uncertainty values provided by Lalos (1989) and ORAUT-TKBS-0008-4 (ORAUT 2010a, Equation 4-1) for the 95th-percentile estimate.

b. A contribution amounting to 25% of the total dose should be included for all energy ranges to account for low-energy photons attenuated by the lead filter that 

covered a portion of the film (Kathren 2004).

c. Numerically equivalent to dose equivalent (Q = 1).

d. Based on an assumption favorable to claimants of a sigma (standard deviation) equal to ±20% and 95th-percentile values enveloped by 2 sigma.

e. Assumed to be the same as that for DuPont Type 502 and 508 films (Lalos 1989).

f. From Author unknown (undated).

g. Assumed to be the same as Panasonic 802 dosimeter.

h. Deep dose equivalent = Hp(10).

i. Source: NCRP (1971).

j. See discussion in Section 6.4.3.1.

k. Based on reported values, corrected for potential energy-dependent under-response (NTA film and Panasonic TED) and ratio of the conversion coefficients for 

personal dose equivalent, Hp(10), to those for dose equivalent, H (NCRP 1971), in the 100- to 2,000-keV energy range.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 23 of 135

Examination of the beta and photon (X- and gamma rays) radiation type, energy and geometry of 

exposure in the workplace, and characteristics of the dosimeter response is crucial to assessment of 

bias and uncertainty of the original recorded dose. Table 6-1 lists the chronology of NTS external 

dosimetry methods and practices beginning in 1951. The parameters of significance are the bias, 

uncertainty, minimum detection level (MDL), and potential maximum missed dose (MMD). For this 

document, bias is defined as the ratio of the reported dose to the true dose (i.e., if the reported dose 

was an overestimate of the true dose, Bias >1). The values of uncertainty are presented as the 

geometric standard deviation (GSD). The values of Bias and GSD for 1951 to 1966 are based on the 

values of bias and uncertainty reported in Film Badge Dosimetry in Atmospheric Nuclear Tests (Lalos 

1989) for the dosimetry systems and practices in use during the period of atmospheric testing.

The bias factors listed in Table 6-1, are the factors by which recorded values in the dosimetry records 

should be divided to provide the best estimate of the measurement quantity [3]. The value of the

measurement quantity should then be multiplied by the dose conversion factors in Appendix B of 

OCAS-IG-001 (NIOSH 2007b) to obtain the organ dose [4]. The bias values for neutrons include an 

estimate of potential under-response due to the inherent energy response of the detector and spectral 

differences between the calibration sources and operational spectra. Potential detector overresponse (e.g., for TLD albedo dosimeters) maintains favorability for claimants.

The uncertainty in the best estimate of the measurement quantity is accounted for in the GSD. The 

GSD is discussed in more detail in ORAUT-TKBS-0008-4, Nevada Test Site – Occupational 

Environmental Dose (ORAUT 2010a), and defined in Equation 4-1 of that document as:

( ) ( ) 1.645

1

50th percentile

95th percentile GSD = (6-2)

Bias and uncertainty for current DOE dosimetry systems are well documented for Hp(0.07) and Hp(10)

under DOELAP (DOE 1986a,b, 1995). The performance of current dosimeters can be compared with 

performance characteristics of historical dosimetry systems in the same, or highly similar, workplaces. 

In addition, current performance testing techniques can be applied to earlier dosimetry systems to 

achieve a consistent evaluation of those systems. Dosimeter response characteristics for radiation 

types and energies in the workplace are crucial to the overall analysis of error in recorded dose.

The MDL is typically established at the point where the laboratory uncertainty of the readings at the 

95% confidence level is ±100% in normal distribution terms. The MMD is equal to half the MDL 

multiplied by the number of exchange or monitoring periods (NIOSH 2007b). With an MDL = 0.04 R, 

and 250 monitoring periods (5 days × 50 weeks), the MMD = 0.04/2 × 250 = 5 R.

6.3.1 Administrative Practices

When the testing program at NTS began in January 1951, the Los Alamos Scientific Laboratory 

[LASL; now the Los Alamos National Laboratory (LANL)] was responsible for administering the 

external dosimetry program (Allen and Schoengold 1995). While contractor organizations and the 

military were involved in issuing and collecting badges for some of the early operations at NTS, LASL 

performed calibration, processing, and interpretation work (Allen and Schoengold 1995). In July 

1955, REECo assumed responsibility for most onsite radiological safety functions (Allen and 

Schoengold 1995).

Until 1966, there was no determination of shallow or skin dose from the film badges (Author unknown 

undated; Allen and Schoengold 1995). With the introduction of a new multi-element film dosimeter in 

1966 until the conversion to TLDs in January 1987, the open window (OW) was used to make a 

separate determination of shallow dose and deep dose (Brady and Iverson 1968; Author unknown 
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 24 of 135

undated; Allen and Schoengold 1995). Shallow dose was determined by comparing the OW and 

closed-window readings and, therefore, did not include the penetrating photon component. This same 

type of determination was used for TLDs used by Pan Am at the NRDS from 1970 to 1972 (Pan Am 

1967; Boone, Bennett and Adams 1970). On the NV-185 form used for dose records at NTS, the 

shallow dose was incorrectly called the "Skin of the Whole-Body" dose when it was, in fact, only the 

beta and low-energy photon components. Beginning in 1987, procedures used currently for 

calculating shallow or skin dose were adopted (Allen and Schoengold 1995).

Beginning in 1987, with the introduction of Panasonic TLDs, the deep (or whole-body) dose and the 

shallow (or skin) dose were reported, but the shallow dose included the penetrating photon 

component and can therefore be considered the total skin dose (Author unknown undated; Allen and 

Schoengold 1995).

Before 1987, unexposed control films were processed together with the personnel dosimeters [5]. 

Two sets of control films were used. Personnel films were processed together with two unexposed 

films from the same emulsion series that had been stored at the Dosimetry Laboratory in Mercury. 

Additional unexposed “area” films that had been stored at NTS issue locations (Table 6-2) were 

processed as well. Additional badges could be obtained at Building 1000.

Table 6-2. NTS film badge issue locations (REECo 1961).

Area Building

Mercury 111, 155

Yucca and Frenchman’s Flats CP-2

12 Rad-safe trailers and stations

400 Rad-safe stations

401 Rad-safe stations

Readings from the Mercury control films were subtracted from the dosimeter readings to obtain a net 

reading for determining exposure (REECo 1961, 1962). The net reading represented the 

occupational exposure plus the differential between the Mercury and work area environmental levels. 

If there was an indication that there had been a problem with use or storage of the films at the issue 

locations, the “area” controls were used for background subtraction. However, such incidents were 

rare, and were noted in the dosimetry records when they occurred.

Before 1986, the film badge was used for the dose of record. If the film badge was lost or damaged, 

the health physicists preferred to use cohort dosimetry results and ambient radiation levels. Pocket 

dosimeter results could be a consideration, but they were used only for short work periods and might 

not have been dependable because they were potentially subject to discharge (Griffith 2006a).

Beginning in 1987, with the introduction of TLDs, the procedure was continued with TLD background 

dosimeters (REECo 1990; Allen and Schoengold 1995).

6.3.1.1 Exposure Limits

From 1951 to 1958, the allowable external exposure limits for occupational workers at NTS were 

generally consistent with NCRP recommendations (Allen and Schoengold 1995). During the 1951 to 

1952 test series, participants could receive up to 3 R of gamma exposure for a 13-week period (Allen 

and Schoengold 1995). Pilots and crew could receive up to 3.9 R of exposure (Allen and Schoengold 

1995). For the 1953 and 1955 series, workers could receive up to 3.9 R (Allen and Schoengold 

1995). Beginning in 1957, the maximum permissible exposure for test participants was limited to 

3 rem per 13-week period and 5 rem per calendar year (Allen and Schoengold 1995). In 1961, the 

U.S. Atomic Energy Commission (AEC; a DOE predecessor agency) Standard Operating Procedure, 

Nevada Test Site Organization (NTSO), Chapter 0524, Radiological Safety (AEC 1961), stated that 

the radiation exposure criteria for NTS personnel were 3 rem per quarter and 5 rem/yr. However, with 
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 25 of 135

the approval of the Test Manager, an NTS worker could receive as much as 12 rem/yr. DOE (1982), 

Radiological Safety, also set 3 rem per quarter and 5 rem/yr as limits for exposure for occupational 

workers.

6.3.1.2 Dosimeter Exchange

The dosimeter exchange procedures evolved over the period from 1951 to 1987. From 1951 through 

1954, 1-day film dosimeters were issued to personnel entering a controlled radiation exclusion 

(RADEX) area. The badges were collected when personnel left that area. Entry to a RADEX area 

was not allowed without a film dosimeter (LASL 1953). In 1955, there was a transition to a 

combination of 1-day, 1-week dosimeters for working in RADEX areas. In 1956, 1-week dosimeters 

were issued for exclusion areas and by December, monthly dosimeters were in use. In 1957, the

regular practice was to issue monthly dosimeters. Before 1957, the majority of workers were not 

badged because their job responsibilities did not require that they enter RADEX areas. 

On April 1, 1957, the issue card system came on line (Author unknown undated). All persons entering 

NTS had to have a film badge, with the correct monthly color coding on the exterior of the badge. An 

individual working in a RADEX area could have many more than 12 badges for the year. If the pocket 

dosimeter indicated a possible exposure on the work shift, the film dosimeter was pulled and 

replaced. From April 1957 through the end of testing in 1992, all NTS personnel had dosimeters 

(Author unknown undated). The front gate security officer would check to make sure each badge had 

the proper color. If it did not, it was necessary to go to the Badge Office in Building 1000 at the gate 

to get a current badge.

Until 1987, film badges were exchanged monthly for all individuals or on exit from radiation areas if an 

exposure of 100 mR or more was measured (or suspected) (Allen and Schoengold 1995). In addition 

to film badges, self-reading pocket dosimeters were issued to persons entering RADEX areas, which 

were controlled locations at which an exposure was usually expected. The purpose of issuing pocket 

ionization chambers (PICs) to persons entering a RADEX area was to provide an action alert.

Following the introduction of TLDs in January 1987, dosimeters were issued on a quarterly basis 

unless a particular job assignment indicated the need for more frequent issue and readout (REECo 

1990; Allen and Schoengold 1995). Measured exposures were added to the yearly and quarterly 

accumulated exposures (REECo 1990; Allen and Schoengold 1995).

6.3.1.3 Dosimetry Codes for External Monitoring

Employer codes and job titles for NTS contractors [REECo, Edgerton, Germeshausen and Grier 

(EG&G), Holmes & Narver (H&N), and Raytheon Services Nevada (RSN)] are available to dose 

reconstructors on the O: Drive maintained by the NIOSH Division of Compensation Analysis and 

Support.

Current NTS contractors include National Security Technologies, Bechtel Nevada; Lockheed Martin 

Nevada Technologies, Inc.; Johnson Controls Nevada, Inc.; and Wackenhut Services, Inc [6]. Former 

NTS contractors included H&N (1956–1990), Fenix & Scisson of Nevada (1963–1990), EG&G (1951–

1995), REECo (1953–1995), and RSN (1990–1995) [6]. EG&G had offices and shops in Las Vegas, 

but some EG&G workers worked at NTS for extended periods. Some REECo workers were assigned 

to the Tonopah Test Range (TTR); Sandia National Laboratories (SNL) is the custodian of TTR 

dosimetry records.

The computerized external dosimetry records contain the codes listed in Tables 6-3 and 6-4 (DeMarre 

2003). Body part codes can be found in Table A-2 of ORAUT-TKBS-0008-5, Nevada Test Site –

Occupational Internal Dose (ORAUT 2010b). However, the important body part codes for external 

dosimetry are 01 WB Whole body; 02 SK Skin; 03 LH L-HAND; and 04 RH R-HAND.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 26 of 135

Table 6-3. Type of dose codes for external 

dosimetry.

Code Definition

00000001 Dose type unknown

00000002 Gamma 

00000003 Beta

00000004 Thermal neutron 

00000005 Other neutron (other than 

thermal)

00000006 Body part-1

00000007 Body part-2

00000008 Initial gammaa

00000009 Initial neutrona

a. Refers to initial radiation at the time of 

detonation, and not normally used for workers 

because they were not present at the time.

Table 6-4. Film damage and irregularity codes.

Damage code Description

A Lost dosimeter

B Light damage

C Heat damage

D Pressure damage

E Factory damage

F Processing damage

G Medical exposure

H Non-personnel exposure X-ray

I Destroyed

J Water damage

K Age damage

L Undetermined damage

N Occupation damage

O Non-returned

X Dose by investigation

Y Late return

Additional information on recordkeeping practices from 1945 to 1962 is in Griffith (2006b).

6.3.1.4 Duplicate Dosimetry Records

Dosimetry for workers at NTS presents a particular problem for dose reconstruction because of the 

large number of DOE facilities where workers were involved with NTS activities. The standard 

operating practice at NTS called for wearing only the NTS personnel dosimeter (film or TLD) while on 

site (REECo 1995a). This posed no problem for NTS contractor employees assigned to the site who 

were routinely issued personnel dosimeters that served to provide the dose of record. Visitors, 

contractor employees from other sites, and temporary workers were also issued NTS dosimeters if 

called on to work on site (REECo 1995a). However, major laboratories such as LANL, Lawrence 

Livermore National Laboratory (LLNL), SNL, and their contractors (e.g., R E McKee) were heavily 

involved in NTS activities. Such temporary assignments might be of several weeks duration, perhaps 

punctuated by a return home over a weekend or for a few days [7].

Normal practice called for temporary or casual workers from other AEC/DOE contractor sites to obtain 

their NTS personnel dosimeter/security badge at the badge house in exchange for their regular, 

employer-issued dosimeter and security credential. Thus, workers on temporary assignment from, for 

example, LLNL would come to the main gate where they would exchange their Livermore-issued 
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 27 of 135

personnel dosimeter/security credentials for NTS-issued equivalents. When the workers left the site 

at the end of their temporary assignments, they would turn in their NTS-issued dosimeters in 

exchange for the LLNL dosimeters they had left at the badge house (REECo 1995a) [8].

This system worked well, but was not foolproof and, thus, there exists a small but real possibility that 

in some cases the NTS dosimeter and a dosimeter from the worker’s parent facility were both worn. 

Such might be the case if workers were flown into or out of the alternative air strip at Yucca Flats or in 

Area 17, and did not make the badge exchange at the Main Gate. In some cases, workers might 

have been issued a special dosimeter by the parent dosimetry laboratory as a means of tracking a 

special exposure away from the home facility. This was done from time to time when the potential for 

a high or unusual exposure was thought to be present, and the parent facilities desired a separate 

measurement that, if necessary, could be processed immediately on return from NTS. This would 

provide an immediate indication of the worker’s exposure status. There were also occasions when 

Lawrence Radiation Laboratory (LRL)/LLNL personnel, for example, were provided with test 

dosimeters for various reasons (REECo 1995a).

As a result, some individuals would have dosimetry records maintained at their parent laboratory’s 

dosimetry service and at those operated by NTS. In fact, there are cases in which duplicate records 

exist for the same exposure. This was particularly true in the period before REECo assumed 

responsibility for NTS activities (July 1955). For example, LANL-based employees reported to NTS 

for the RANGER Operation or the BUSTER-JANGLE Operation in 1951. LANL performed the 

dosimetry for NTS for those two operations and there are copies of the same data in both the LANL 

and NTS record systems [9]. Once REECo was responsible for Radiation Safety at the site, reports 

were sent to the other sites if they had employees who were based at NTS or visited NTS. These 

reports were often incorporated into the record for the laboratory employee and were retained by NTS 

in its records for that employee. The records should be evaluated to ensure that the dose is not 

credited to the employee twice. It is also possible that when the laboratory included the NTS reported 

dose in its records, the dose was rounded according to the scheme developed by the laboratory for 

including such dose. 

There were a few instances in which personnel wore duplicate dosimeters [9]. Although care was 

taken to avoid duplicate dose assignment, the obvious problem associated with double-badging is that 

the individual could be assigned the same dose twice. While this is favorable to claimants, it could 

result in a situation in which the unearned dose assigned to an individual might be sufficient to result 

in an unduly and inappropriately high POC leading to unwarranted compensation.

For situations in which an employee of one of the laboratories obtained an exposure that appears in 

the NTS records system, the dose reconstructor should review laboratory records for the same period 

to determine if the same exposure for the same period appears in those records as well. If so, a note 

should be made that the exposure was received at NTS and the exposure appearing in laboratory 

records should not be included.

Double-badging and associated doubled dose assignments are likely to be difficult to establish with 

any degree of certainty. Unless there is unequivocal documentation that two badges were issued and 

that two doses were assigned for the same period, the prudent action would be for the dose evaluator 

to assign all recorded dose to the claimant. However, if two personnel dosimeters were assigned to 

the same person and both were worn over the same or approximately the same period with 

approximately the same recorded dose, this is prima facie evidence that the person was doublebadged and highly suggestive that a doubled dose assignment was made. In such instances, a 

determination should be made if this was likely to be the case. If so, only the higher of the doses from 

the two dosimeters should be assigned to the claimant.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 28 of 135

Similarly, if documentation exists that a person was double-badged, with the timeframe for one badge 

overlapping the time frame for the other badge at both ends, the dose from the former dosimeter 

should be assigned unless it is the smaller of the two. This is illustrated by the following hypothetical 

example: Worker ABC was issued a personnel dosimeter on January 15, 19xx, by his parent facility 

and another by NTS on January 20, 19xx. The NTS dosimeter was turned in on January 30, 19xx, 

and from it a dose of 400 mrem was assigned. The dosimeter from the parent facility was turned in 

on January 31, 19xx, and indicated a dose of 600 mrem. Documentation indicates that both badges 

were worn or carried on the person while at NTS. In this case, the claimant should be assigned a 

dose of 600 mrem for the period from January 15 through January 31. If it cannot be established that 

both dosimeters were worn or carried at NTS, the readings from both dosimeters should be assigned 

to the claimant.

An additional concern arises because there were some REECo employees who were hired and 

terminated at NTS, but who were assigned to offsite locations. For example, REECo employees were 

assigned to support SNL at TTR, a situation that was possible until the mid-1990s [10]. Dose 

reconstructors should be aware that the only records NTS would have are the dosimetry results for 

the in- and out-processing period (usually 1 day each). If mention is made of working at TTR, the 

dose reconstructor might need to request additional information from SNL (e.g., either the dosimetry 

data or a clear statement that the employee was not monitored). If the TTR employment was not 

verified by DOL and no dosimetry records for TTR are included in the file, the TTR employment 

should not be included in the dose reconstruction and the entire period of DOL-verified employment 

should be evaluated as NTS employment. 

6.3.1.5 Special Issues

6.3.1.5.1 External Exposure to Hot Particles

Highly radioactive particles were produced by some of the operations at NTS (e.g., atmospheric 

testing or reactor operations such as the nuclear rocket tests) (NRDL 1968; NCRP 1990). The size of 

hot particles in nuclear fallout ranges from 10 nm to 20 μm for worldwide fallout (NCRP 1990). Local 

fallout particles are significantly bigger (100 μm to several millimeters) (NCRP 1990). When 

deposited on skin or clothing, they can produce high levels of localized exposure, primarily from beta 

or alpha particles (NRDL 1968; NCRP 1990).

Hot particle exposure is not easily identified because, in general, the dosimeter response cannot be 

used to distinguish between a hot particle exposure from a distributed radiation field exposure [11]. 

Because workers were normally monitored with hand-held survey instruments when leaving a 

radiation area, it was most likely that hot particles would be detected during this monitoring process 

(Rollins 2007). If such particles were detected, the worker would be asked to go through a 

decontamination procedure, including removal of outer clothing and showering as necessary (Allen 

and Schoengold 1995, Rollins 2007). However, it is not likely that such incidents would be entered in 

the dosimetry record unless the conditions were unusual (e.g., very high count rates).

Unless hot particles were detected shortly after deposition using survey instruments, they were

removed by normal washing or change of clothes, and their exposure was not recorded. Without a 

specific entry in the dosimetry record, the only evidence, if the hot particle exposure was high enough, 

could be subsequent formation of a lesion at the deposition site [12].

The dose reconstructor should be aware of the possibility of external exposure from hot particle 

deposition and document positive indications in the claimant’s dose or medical record that can be 

reasonably associated with hot particle deposition. Hot particle deposition issues should be 

addressed consistent with Project guidance on assigning shallow dose.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 29 of 135

6.3.1.5.2 Intentional Nonuse of Dosimeters

There is anecdotal evidence suggesting that workers might have intentionally failed to wear their 

dosimeters during specific operations, which could have caused them to exceed administrative limits 

[13]. This was said to have been done to avoid administrative actions that, inter alia, could have had 

adverse financial impact such as loss of overtime pay. There are no documented instances of this 

occurring (Rollins 2007); however, the practice could have taken place. In such situations, cohort 

dosimetry would be of questionable value because other members of the cohort are likely to have 

taken the same action. Dose reconstructors are encouraged to be aware that intentional nonuse of 

dosimeters could have taken place, and any indication of such occurrences should be documented in 

the claimant’s file. Because the personnel security credential holder (identification badge) was 

required for entry into controlled areas, workers needed to have their dosimeters with them, limiting 

their ability to avoid wearing the dosimeter.

6.3.2 Individual Monitoring Methods

The dosimetry methods employed initially at NTS were adopted from techniques implemented at 

LASL from the beginning of the atmospheric weapons testing program (Lalos 1989; Boone, Bennett, 

and Adams 1970). As the atmospheric test series progressed, dosimeter configurations improved in 

relation to the radiation fields encountered, and the responsibility for dosimetry programs was 

delegated among the participating agencies and military organizations [14]. In 1955, the site 

contractor, REECo, assumed responsibility for most onsite radiological safety functions; this included 

a site-wide service based on the use of film dosimetry for photons, betas, and neutrons. These 

methods evolved and eventually gave way to other methods including TLD and nuclear track 

detection or track-etch dosimeters (TEDs) (Allen and Schoengold 1995; Author unknown undated). 

From the beginning, PICs were used if necessary to augment the passive dosimeters issued to NTS 

workers and visitors. More detail on the Dosimetry Technology associated with the monitoring 

methods used at NTS can be found in Attachment A.

6.3.2.1 Beta/Gamma Dosimeters

The film dosimeter issued for the first test series at NTS in 1951 – RANGER – was a DuPont 552 

packet, which included two films: a Type 502 low-range component (0.05 to 10 R) and a Type 510 

high-range component (5 to 50 R). Brass/cadmium strips, 0.020-in. thick, provided symmetrical 

filtration on both sides of the film, with the remainder of the film open and unfiltered (Shipman et al. 

1951).

A modified dosimeter was used by both the Nevada Proving Ground (NPG) and Desert Rock RadSafe groups for the second NTS series in 1951 – BUSTER-JANGLE (Lalos 1989). The DuPont 553 

dosimeter included a third, higher range film component – Type 606 (10 to 300 R) (Kean 1951; 

Shipman et al. 1951; Storm 1951). NPG film packets were used in the LASL brass-cadmium badge 

with 0.020-inch-thick brass and cadmium filters plus an open window. Desert-Rock film packets were 

in sealed, clear plastic envelopes, and probably had 0.020-inch-thick lead filters, as were used a few 

months later at NPG during Operation TUMBLER-SNAPPER (Kean 1951; Shipman et al. 1951; Storm 

1951).

The film dosimeter used by NPG personnel for TUMBLER-SNAPPER was designated DuPont 558 

packet. With the exception of the first test, ABLE, it included lead filters, 0.028-in. thick, symmetrical 

on both sides; a Type 508 low-range component (0.01 to 6 R); and a Type 1290 high-range element 

(20 to 3,000 R) (Brady and Nelson 1985).

The use of DuPont 559 dosimeters began in 1953 and continued to July 1960. It consisted of a Type 

502 low-range component (0.02 to 10 R) and a Type 606 high-range element (10 to 300 R). Lead 
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 30 of 135

filters, 0.028-in. thick, were used to provide symmetrical coverage on both sides, leaving an open area 

(Brady and Nelson 1985; Collison 1953). It was used essentially unchanged until HARDTACK II in 

1958, when Type 834 high-range film (5 to 800 R) replaced Type 606 film to provide and overlap with 

Type 502 film.

The film badge dosimeter used from July 1960 through 1965 was a modification of the DuPont 559, 

designated DuPont Type 301-4. It consisted of a Type 508 low-range component (0.03 to 10 R) and 

a Type 834 high-range component (10 to 1,000 R) wrapped with a 28-mil (0.028-in.)-thick lead strip 

covering an area 0.5 in. by 1 in. on each side. The remainder of the 1 in. by 1.5 in. was uncovered. 

The packet was in a 4-mil (0.04-in.)-thick plastic bag sealed with colored tape to indicate the month of 

validity. The bag was clipped to the security badge, and all personnel entering NTS wore this 

dosimeter (Author unknown undated).

In 1966, NTS began using a combination personnel dosimeter and security credential holder (Brady 

and Iverson 1968) to provide the increased personnel dosimetry capability necessary to meet 

radiation exposure problems associated with nuclear rocket testing and underground nuclear 

detonations. The holder was designed to accommodate a DuPont Type 556 film packet, a fast 

neutron packet [containing Kodak nuclear track emulsion, type A (NTA) film], an identification plate, 

criticality accident components, the security credential, and a snap-type clip. The complete package 

could measure beta, gamma, X-ray, thermal neutron, fast neutron, high-range gamma, and highrange neutron exposures.

In March 1971, when the use of DuPont film ended, NTS dosimetry operations converted to Kodak 

Type Ill film packets (Author unknown undated). This two-component packet contained low-range (30 

mR to 10 R) and high-range (10 to 800 R) films. The other components of the badge remained 

essentially the same.

The first routine use of TLDs at NTS began in 1970 (Boone, 

Bennett, and Adams 1970). Starting in February 1966, Pan Am 

used TLDs at the NRDS as part of the site effluent monitoring 

program (Figure 6-1) (Pan Am 1967). These dosimeters 

contained a calcium-fluoride phosphor bound to a helically wound 

wire in an evacuated glass tube and were ideal for the intended 

purpose but unsuitable for personnel dosimetry (Pan Am 1967).

With the advent of DOE requirements to restrict personnel 

exposures to as low as is reasonably achievable (ALARA) and 

with emphasis on accurate dosimetry at low doses, REECo 

Environmental Sciences Department personnel began evaluating 

TLD systems and neutron dosimeters in the early 1980s to 

replace the film badge and neutron TLD (Author unknown 

undated).

After evaluating several dosimetry systems, the Environmental 

Sciences Department determined that the Panasonic 802 TLD 

and the neutron TED were the best combination for NTS 

exposure conditions. These were put into use on January 1, 

1987 (Author unknown undated). The security credential holder 

was redesigned to accommodate both dosimeters (Figures 6-1, 

6-2, and 6-3).

The four-element Panasonic UD-802 TLD was the primary dosimeter for routine use issued to all 

monitored personnel until 2001, when it was replaced by the Panasonic 809 dosimeter, which also 

Figure 6-1. Security badge for 

NRDS TLD.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 31 of 135

Figure 6-2. REECo badge holder (REECo 1990).

Figure 6-3. REECo security badge holder with Panasonic 

UD-802 dosimeter.

contained four elements (Table 6-1). The Panasonic UD-807 TLD is used to monitor personnel 

working in situations where the likelihood of exposure to an extremity is significantly greater than 

exposure to the whole body (BN 2001).

In addition to film and TLD badges, self-reading pocket dosimeters were issued to persons entering 

RADEX areas, which were controlled locations at which an exposure was usually expected (Author 

unknown undated). The maximum reading on the pocket dosimeters most commonly used was 200 

mrem. The pocket dosimeter could discharge when dropped, knocked, or exposed to water. Unless 

there was good evidence that the pocket dosimeter was damaged, a high reading of a PIC (≥ 100 

mrem) triggered the action of collecting and processing the personnel dosimeter being used at the 

time (Griffith 2006a). PIC results were not used unless the personnel dosimeter had been 

compromised. The PIC result would have been included in a special investigation of the incident.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 32 of 135

Although PIC results were only used as dose-of-record when film badge results were not available 

because of their typical over-response characteristics (Figure 6-4), PIC results should be evaluated 

carefully before including them in the dose reconstruction.

Figure 6-4. X-ray energy dependence of PICs (Sanders, Auxier, 

and Cheka 1959).

6.3.2.1.1 Beta Dosimetry with Film Badges

Film response to beta radiation is accomplished with an unshielded portion of a film packet. 

Multielement film badges have an OW or unshielded portion that allows beta and photon radiation to 

reach the film. The beta dose is interpreted by removing the density attributable to a concomitant 

photon exposure as determined by the densities on those portions of the film under the various filters. 

The process is subject to large uncertainties, perhaps as much as a factor of 2 [15].

Before 1966, film dosimetry at NTS was performed with a bare film packet partially covered by a lead 

strip. The exposed film was interpreted by attributing the density under the lead filter to photons. The 

density in the unshielded portion of the film was assumed to be due to both photons and betas, and 

was typically not measured. In those instances in which the unshielded portion was read, the beta 

dose was determined by subtracting the density under the lead strip from that in the unshielded 

portion of the film to obtain a net density. This procedure did not accurately determine beta dose 

because it attributed the entire net density of the unshielded portion to beta dose, ignoring the lowenergy photon dose contribution (e.g., energies less than about 80 keV, of concern primarily to 

posttest drillback crews before 1965 and tunnel reentry crews in situations when the tunnel was not 

vented before reentry), which did not penetrate the lead strip. As much as 25% of the total photon 

dose could have been missed as a result of this attenuation (Kathren 2004; Coryell and Sugarman 

1951; Nelms and Cooper 1959). Therefore, the beta dose could have been overestimated. Given the 

spectra and mix of radionuclides at NTS, dose estimates made in this fashion are likely to be within a 

factor of 2 of actual beta dose (Becker 1966, p. 102; Kathren and Larson 1969) and could slightly 

underestimate the overall photon dose. However, such dose estimates are likely to be unreliable and 
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 33 of 135

should be confirmed by other factors such as the beta-photon dose rate ratio obtained with monitoring 

instruments in the field, if such data are available. The value of a factor of 2 is an estimate of the 

range of uncertainty based on knowledge of the reported response characteristics of the dosimeters, 

and is presented for general information only. For purposes of dose reconstruction, prior to 1966, 

beta doses should be estimated by using the beta-to-gamma ratios discussed in Section 6.4.2.1 

below.

Multielement badges were introduced at NTS in 1966 (Brady and Iverson 1968; Author unknown 

undated). A single high-Z metallic filter provided a more or less flat energy response for photons with 

energies above the uncertainty edge of the filter (about 50 keV). The density under the filter was used 

to assess the dose to photons above this cutoff energy and to evaluate the doses to photons with 

energies below the cutoff using densities under the other filters. Two filters – an OW and a low-Z

material (typically plastic such as Teflon) – were used to determine the beta dose. The response of 

the film under these filters was approximately the same for photons, but the low-Z plastic filter 

essentially removed all the betas. The reading from the low-Z filter, which was considered to be 

attributable to photons only, was subtracted from the density under the unshielded portion, which had 

a comparable response to photons and also was responsive to beta radiation. The analysis can be 

performed manually with a set of calibration curves made at different photon energies and a beta 

calibration curve or with an algorithm developed from such a set of calibration curves. If it is 

necessary to make a beta dose estimate using the film badge results from shallow dose estimates 

from 1966 through 1986, dose reconstructors should double the reported value to ensure favorability 

to claimants and to account for uncertainties [15].

For external dose reconstruction, a positive indication of beta exposure in a dose record is due to 

betas with energies above 15 keV [16].

6.3.2.2 Neutron Dosimeters

A small fraction of the workers at NTS had potential for exposure to neutrons [17]. For workers with a 

possibility of neutron exposure, personnel neutron dosimeters were used to monitor exposure. 

Potential sources of neutron exposure at NTS were:

1. Direct production from a nuclear detonation

2. Spontaneous fission and subcritical multiplication in fissile materials (e.g., 235U, 239Pu)

3. Isotopic sources such as initiators and calibration sources

4. Reactor testing

6.3.2.2.1 Film Dosimeters

From 1961 to 1979, nuclear track emulsions in the form of Kodak NTA film packets were used for 

personnel fast neutron dosimetry at NTS. The NTA film packet was incorporated in the newly 

designed combination personnel dosimeter and security credential along with a DuPont Type 556 film 

packet for beta photon monitoring, an identification plate, criticality accident components, the security 

credential, and a snap-type clip (Brady and Iverson 1968). The NTA film packet consisted of a single 

thick-coated film of dental size wrapped in several layers of light-tight paper. More information on 

NTA film can be found in Attachment A, Section A.2.1.

From 1966 to 1986, every NTS badge had a cadmium strip that provided a thermal neutron-sensitive 

component that was only evaluated to provide verification that there had been no incidental neutron 

exposure (Brady and Iverson 1968). The density under a tantalum filter was compared to the density 

under the cadmium-tantalum filter to ensure that they were similar (equivalent densitometer reading of 

less than 80 mR). Cadmium readily absorbs thermal neutrons and after the absorption emits a 

photon of approximately 80 keV, which is of low enough energy to result in an over-response of the 
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 34 of 135

film badge in comparison with higher and lower photon energies. This energy is in the peak of the 

response curve, so low-level exposures to thermal neutrons were readily detected.

6.3.2.2.2 Thermoluminescent Albedo Dosimeters

The NTS albedo dosimeters adopted in 1979 were based on the design of Hankins (1977). They 

were used until 1987 and consisted of four TLD chips each of 6

LiF and 7LiF, placed in a cadmium box 

to suppress response to thermal neutrons. Albedo neutron dosimetry depends on the detection of 

low-energy neutrons reflected from the body (albedo neutrons) with a thermal neutron detector. 

Normally, a TLD with 6

LiF (TLD-600) is used to detect neutrons, while a companion 7LiF (TLD-700) 

detector that is insensitive to neutrons is used to subtract the photon/beta contribution.

6.3.2.2.3 Track-Etch Dosimeters

Gamma-insensitive Columbia Resin (CR)-39 TEDs were introduced for personnel neutron dosimetry 

at NTS in 1987. These were based on the dosimeter developed at LLNL (Hankins, Homann, and 

Westermark 1987; Hadlock et al. 1988). The CR-39 TED offered better fast neutron (>0.1 MeV) 

energy response characteristics for occupational monitoring than the TLD albedo dosimeter. The 

response curve is relatively flat, between 0.1 and 4 MeV (Attachment A, Figure A-5). Once a foil has 

been properly etched, acquiring information is a nondestructive process.

6.3.2.2.4 Panasonic 809

The Panasonic 809 replaced the Panasonic 802 dosimeter in 2001 (BN 2001). The 809 dosimeter 

contained three neutron-sensitive 6

Li2

10B4O7(Cu) TLD chips. This gives the dosimeter a neutron 

detection capability, so the use of the TED was discontinued. Photon compensation is provided by 

the neutron-insensitive 7

Li2

11B4O7(Cu) chip. The algorithm used to process the response of the four 

TLDs provides the neutron dose measurement. It is, in effect, an albedo detector.

6.3.2.3 Extremity Monitoring

Extremity dosimetry has been used at NTS to assess exposure to the finger, hand, forearms, and 

even the head (on rare occasions) that might have occurred during operations near or involving the 

manual manipulation of radioactive material and radiation-emitting objects. Extremity monitoring 

might be required, for example, if radiation technicians were involved in handling post-test core 

samples.

The dosimeter (film or TLD) was worn in a position that was intended to represent the highest 

exposure to the extremity, usually on the inside of the wrist in the case of film, or on the finger near 

the tip. The extremity being monitored is normally identified in the dose record using the codes listed 

in ORAUT-TKBS-0008-5, Nevada Test Site – Occupational Internal Dose (ORAUT 2010b, Table A-2).

In 1957, extremity limits were set at 1,500 mrem/wk (Griffith 2007). The film pack used at that time 

was the Film badge-DuPont 559 film packet (Table 6-1). This continued until July 1960 when the use 

of DuPont film packet Type 301-4 (also called DuPont Type 556) was adopted (Table 6-1). In 1964, 

the limit for extremities was set at 75 rem per quarter (Griffith 2007). TLD finger rings for extremity 

monitoring were used beginning in July 1967 (Griffith 2007).

Although the regular use of TLD finger rings was documented in 1967 (Griffith 2007), extremity 

monitoring with film and, later, TLD occurred on rare occasions before that time (DeMarre 2006c). 

ConRad 7

Li F and Teflon discs, 1.3 cm in diameter by 0.4 mm thick were used for measuring finger 

and hand exposures when personnel handled radioactive material in certain operations (REECo 

1968). The 7

LiF and Teflon disks were protected from the light by inserting them in black plastic 

pouches. The pouches were affixed to the adhesive portion of “band-aids”, which were attached to 
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 35 of 135

the fingers of the personnel being monitored. Extremity dose determined by the TLDs was not 

included in the routine dose reports. Rather, they were included with the bioassay data, and a card 

file of extremity doses was maintained.

The current extremity monitoring dosimeter is the single-element Panasonic UD-807 TLD shown in 

Figure 6-5. The Panasonic UD-807 TLD is similar to a single CaSO4:Tm element in the Panasonic 

UD-802 TLD. A single UD-807 TLD is sealed in a small, transparent, circular envelope embossed 

with the TLD serial number. Except when removed for placement in an envelope-type holder, the 

TLD is kept in a processing holder. The holder has a serial number identical to the number embossed 

on the TLD element encapsulation. All personnel monitoring, exposure checks, and calibrations are 

performed with the TLD in the envelope-type holder, while all processing is performed with the TLD in 

the Panasonic holder.

Figure 6-5. Panasonic UD-807 extremity dosimeter (Allen and Schoengold 1995).

6.3.3 Calibration

6.3.3.1 Photons

In the early 1950s, film dosimeters were calibrated to free-air exposures using 226Ra and 60Co sources 

with source strengths traceable to the National Bureau of Standards (NBS), which is now the National 

Institute of Standards and Technology (NIST) (Lalos 1989; Allen and Schoengold 1995). Exposures 

were quantified from inverse-square relationships over a fixed exposure interval at various distances 

and for exposures at fixed distances over various time intervals. Different approaches were used to 

compare dosimetry results, including processing badges exposed to calibration fields to badges 

exposed during test operations. In addition, collocated badges of different types and ionization 

chambers were exposed simultaneously during some test events to provide in-field calibration for 

realistic radiation spectra [18].
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 36 of 135

6.3.3.2 Neutrons

Neutron calibrations have been performed with isotopic sources, reactor beams, or accelerators. 

However, virtually all routine or practical radiation protection calibrations employ radioactive sources 

[19]. The preferred neutron production mechanisms are either alpha-neutron reactions with beryllium 

or boron, or spontaneous fission (252Cf). On rare occasions, gamma-neutron reactions have been 

used, but the associated high-intensity gamma fields generally make them unacceptable for 

calibration of dosimeters and instruments that have any degree of gamma sensitivity.

Table 6-5 summarizes the properties of commonly used isotopic sources from the International 

Organization for Standardization (ISO 2001). Probably the first isotopic source to be used was 

226RaBe (Roberts 1937). However, its neutron production was accompanied by a high photon 

emission. As 239Pu, 238Pu, and 241Am became available, they became the alpha-emitting radionuclides 

of choice. Because these radionuclides have similar alpha energies, the resultant neutron energies 

are similar and can be considered nearly identical for radiation protection purposes. In particular, 

210PoBe has been in use at NTS for several years (Griffith 2007). Later 238Pu(n,α)Be and 252Cf 

sources were used (Allen and Schoengold 1995).

Table 6-5. Characteristics of ISO reference radiation radionuclide 

sources used for neutron dosimeter calibration.

Source

Half-life

(yr)

Flux average energy 

(MeV)

AmBe-241 (α,n) 432.7 4.4

Cf-252 spontaneous fission 2.65 2.4

Cf-252 in 30-cm-diameter D2O sphere 2.65 0.54

Occupational neutron fields at NTS are commonly due to fission-origin neutrons. Therefore, 252Cf is 

particularly attractive as a calibration source because it offers better inherent spectral simulation than 

the alpha-neutron sources. However, the unmoderated californium neutron spectrum is significantly 

harder than most reactor spectra. Beginning in the 1980s, spheres filled with deuterated water (D2O) 

came into common use as moderators to soften the spectrum of neutrons from the 252Cf source in the 

center (Schwartz and Eisenhauer 1980).

For some neutron dosimeters such as TLD albedos, an alternative to source calibrations is the use of 

operational area calibrations. This involves exposure of dosimeters on appropriate phantoms in 

operational areas for extended periods (hours, days, or even weeks, depending on the ambient field 

intensities). The reading on the dosimeter when processed is compared with a measurement made 

with a reference instrument. This relationship is used to establish a calibration for that particular area. 

However, the use of operational area calibrations implies work areas that have sufficient field 

intensities, so this method was not commonly used at NTS [20].

6.3.4 Limit of Detection

6.3.4.1 Gamma Dosimeters

6.3.4.1.1 Film

The National Research Council (NRC) evaluation of Film Badge Dosimetry in Atmospheric Tests

(Lalos 1989) included the minimum detection levels for film dosimetry used during the period from 

1951 to 1962. Application of this concept to film dosimetry during atmospheric tests generally results 

in an MDL of about 40 mR, indicating that 95% of a series of exposures at 40 mR would yield 

readings between 0 and 80 mR. Although lower values have been cited elsewhere (Griffith 2007; 

Allen and Schoengold 1995), the NRC values are conservative, and they are listed in Table 6-1.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 37 of 135

6.3.4.1.2 TLD

Panasonic UD-802 TLDs have been tested to determine their lower limit of detection (LLD). The LLD 

is the minimum evaluated dose equivalent for which the readout value of a dosimeter is significantly 

different (at the 90% confidence level) from the mean readout of unirradiated dosimeters. LLD is a 

detection limit based on the standard deviation of background measurements and a 5% chance of 

reporting a false positive value.

DOE (1986a, Chapter 3) summarizes the procedure for determining the LLD for DOELAP personnel 

dosimetry systems. It provides two alternative equations to calculate LLD. Although DOE (1986a) 

requires determination of the LLD for accredited dosimetry systems, no performance criteria are 

applied to these results.

The method used to determine the LLD for the NTS personnel TLD system is consistent with the 

method described in DOE (1986a). The following equations are used for determining the LLD:

( ) ( ) 2 2 L tS H B S B D p 10 0 = + ÷+ ÷ −÷+ 2 1.75 1 1 1  ′        (6-3)

and

2 S0 L t D = p

(6-4)

where:

LD1 = LD where the probability of reporting a false positive or false negative result is 5%

LD2 = LD corresponding to a 5% chance of reporting a false positive

Tp = distribution factor for n – 1 degrees of freedom and a probability value of 0.95 = 1.68488

S0 = 0

1/2 2

1 0 ( ) ( 1) n

i i XH n =

  − ÷−   ∑ = standard deviation of measurement

H0 = H0′ = 0

Xi ÷ n = mean dose value

0

Xi = unirradiated dosimeter value background dose

N = number of dosimeters, number of measurements

B = bias for DOELAP categories

S = standard deviation for DOELAP categories

The LLD is primarily a function of the standard deviation in background dose rates. Table 6-6 lists 

selected NTS background dose equivalent rates. The values can vary over time. However, the 

temporal variation has been small since the end of atmospheric testing.

Table 6-6. Background dose equivalent rates (Allen and Schoengold 

1995).

Location

Dose equivalent rate (mrem/d)

Shallow Eye Deep

NTS Area 12, Camp 0.375 0.311 0.283

NTS Area 6, Building 2 0.327 0.246 0.211

NTS Area 6, Building 50 0.255 0.182 0.150

NTS Area 23, Building 650 0.271 0.199 0.168

Test Range Complex, DOD facility 0.249 0.193 0.169

Las Vegas, REECo Highland Building 0.181 0.139 0.121

Mean dose rate 0.276 0.212 0.184

Standard deviation 0.067 0.059 0.057
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 38 of 135

Table 6-7 lists detection limits for DOELAP categories determined during the 1990 accreditation 

process (Allen and Schoengold 1995). The LLD used at NTS as most appropriate for its operations is 

LD2, which is the most applicable value for determination of a dose likely to be higher than background 

levels and which increases the probability that a dose will be evaluated as positive. However, for 

dose reconstruction, the major concern is reporting a false negative – reporting an exposure as zero 

when it really was greater than zero. Therefore, the LD1 values are appropriate for determination of 

MDL. Values of 0.3 mrem/d for deep dose and eye dose equivalents and 0.27 mrem/d for shallow 

dose should be used for NTS personnel TLD measurements. For a quarterly exchange period 

(90 days), the equivalent LLD values are 27 mrem for deep dose and eye dose equivalents, and 

24.3 mrem for beta dose equivalent. Considering the conservatism of a 90-day exposure quarter, 

nominal values of 30 mrem for photon exposures and 25 mrem for beta exposures are recommended 

for dose reconstruction (Table 6-1).

Table 6-7. NTS Panasonic UD-802 detection limits for DOELAP categories (mrem/day).

Category

Shallow dose Eye dose Deep dose

LD1 LD2 LD1 LD2 LD1 LD2

I. Low-energy photons–X-ray high dose N/Aa N/A N/A N/A 0.21 0.21

II. High-energy photons–high dose N/A N/A N/A N/A 0.21 0.21

IIIa. Low-energy photons–X-ray general 0.31 0.13 0.23 0.10 0.23 0.10

IIIb. Low-energy photons X-ray–plutonium 

environments

0.33 0.13 0.25 0.10 0.24 0.10

IV. High-energy photons 0.28 0.13 0.21 0.10 0.21 0.10

V. Beta particles–general 0.27 0.13 N/A N/A N/A N/A

a. N/A = not applicable.

6.3.4.2 Neutron Dosimeters

Because of the energy dependence of the various neutron dosimeters used at NTS (Attachment A, 

Section A.2), as well as other facilities throughout the DOE complex, detection limits for neutron 

dosimeters are highly dependent on the operational neutron spectra encountered. The issue of 

operational neutron spectra is discussed in more detail in Section 6.3.5.3 and Attachment A.

6.3.4.2.1 NTA Film

NTA film has an effective energy threshold of about 0.5 MeV (Figure A-3). As a result, the limit of 

detection is directly proportional to the fraction of the neutron dose that is due to neutrons below 

0.5 MeV (i.e., if 50% of the occupational neutron dose is due to neutrons below 0.5, the limit of 

detection is twice as high as it would be for a PuBe calibration source).

NTA film dosimetry is relatively insensitive when compared with other methods. Under the best of 

conditions, with a fast neutron spectrum such as that from an α, n source or unmoderated 252Cf fission 

neutrons, a reasonable limit of detection is about 50 mrem (ICRU 2001). However, when the 

spectrum is degraded, the detection limit is increased significantly – by as much as a factor of 2 due 

to shielding and scatter from isotopic sources, and a factor of 5 for heavily shielded reactor operations 

(Section 6.3.5.3). The MDL recommended in Table 6-1 is 50 mrem. The 50 mrem MDA is based on 

the instructions found on page 2 of Form NV-185 (08/98), United States Department of Energy, 

Radiation Exposure History for an employee/visitor. The Instructions list the minimum reportable dose 

as 50 mrem for nuclear track emulsion results. For the purposes of dose reconstruction, the 

reportable dose has been assumed to be equivalent to the MDA. For workers identified with activities 

requiring NTA film dosimeters and having reported positive dose, it might be appropriate to apply one 

of the factors discussed above to account for a degraded spectrum.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 39 of 135

6.3.4.2.2 TLD Albedo

TLD albedo dosimetry is inherently much more sensitive than NTA film dosimetry. However, it suffers 

from an adverse energy dependence, with the response decreasing seriously with increasing energy 

(Figure A-4). A bias of about a factor of 2 (Section 6.3.5.3) and an MDL of 10 mrem is recommended 

in Table 6-1.

6.3.4.2.3 Track-Etch Dosimeters

Control foils processed with each TED are used as the basis for background subtraction for the TED 

system. The mean background dose equivalent to the control foils and the variance among the 

control foil dose equivalents are calculated for each processing. The background dose equivalent is 

determined for each processing because it can differ among foil sheets and within foils from the same 

sheet. Therefore, the TED background correction is not a single value, but a variable determined with 

each processing (Allen and Schoengold 1995; ICRU 2001).

The method for determining the LLD for the TED system, with a variable background correction, was 

determined from Chapter 7 of NCRP Report 58, A Handbook of Radioactivity Measurements 

Procedures (NCRP 1978). The appropriate equations are:

L V D1 = + 2.71 4.65 B (6-5)

and

L V D2 = 2.32 B (6-6)

where:

VB = background variance in units of dose equivalent squared (millirem squared).

The LD2 equation above was selected as the most appropriate LLD for the TED system. The selection 

reasons are the same as those discussed for the TLD system.

The control foil variance (VB) increases with TED foil age and typically ranges between 5 and 

25 mrem. Therefore, the TED system LLD is normally between 5 and 12 mrem. However, based on 

practical experience with these detectors, a value of 25 to 30 mrem is more realistic. That range of 

limits is reasonable for any TED issue period because it is primarily dependent on foil age and is only 

slightly dependent on the issue period. Considering the possibility of long-term variations and other 

contributing factors, potential energy-dependent under-response to moderated spectra, and the ratio 

of Hp(10) to dose equivalent H, dose reconstructors should use a value of 30 mrem for the MDL 

(Table 6-1) [21].

6.3.5 Workplace Radiation Fields

The radiation production characteristics at NTS have been outlined in ORAUT-TKBS-0008-2, Nevada 

Test Site – Site Description (ORAUT 2008). The potential for external radiation exposures arises 

primarily from the fission and activation products handled at NTS (ORAUT 2008, Table 2-2). In 

addition, there is limited potential for neutron exposure from handling TRU radionuclides, isotopic 

sources, and reactor operations.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 40 of 135

6.3.5.1 Photon Fields

The residual radiation field following detonation of a nuclear weapon consists of radiation from fission 

products, activation products, and unfissioned uranium or plutonium. During atmospheric testing of 

fission and fusion devices, differences in photon fields of residual radioactivity from detonations were 

observed. These differences are caused by the relative abundance of a few radionuclides that were 

produced in each atmospheric test (Hicks 1981a to 1981i). For example, a low-altitude detonation of 

a fusion weapon induces large quantities of activation products that emit high-energy gamma rays 

that dominate the residual radiation spectrum for the first few days after the detonation. In contrast, a 

low-altitude detonation of a fission weapon produces large quantities of fission products that emit a 

wide range of photon energies. With either type of weapon, depending on the design, there can be a 

large amount of activity from the 239Np produced, which can dominate the spectrum for several days.

Although the residual radiation intensity depends on a number of factors that can vary from test to 

test, relatively few radionuclides, common to all tests, contribute to the major part of the photon 

spectrum. The relative abundance of each of these radionuclides determines the spectrum. In all 

cases, the photon field is from photons with energies between about 100 keV and 2 MeV. There is 

very little contribution from photons with energies less than 100 keV with the exception of scattering 

from large area sources. In those cases, the scattered radiation was determined to have an energy of 

approximately 75 keV and to have contributed as much as 10% of the overall photon spectrum 

(Kathren 2004). The special case of exposure to noble gases during post-test drilling can involve a 

significant contribution to photons with energies below 100 keV (Section 6.3.5.1.2).

For external dose reconstruction, if the conditions of exposure (work area, operation, etc.) are 

unknown, dose reconstructors should use the assumption favorable to claimants that photon energies 

are between 30 and 250 keV. If the exposure was due to fresh fallout (early reentry teams), it would 

be reasonable and still favorable to claimants to assume that 75% of the photon dose was from 

photons with energies above 250 keV (Kathren 2004; Coryell and Sugarman 1951; Nelms and 

Cooper 1959). If there is adequate documentation linking exposures to a particular NTS work area or 

operation, Table 6-9 presents guidance on a reasonable allocation of the recorded exposure or 

personal dose equivalent to the energy groups 30 to 250 keV and >250 keV. The information in this 

table is based on the radionuclide inventories in ORAUT-TKBS-0008-2, Nevada Test Site – Site 

Description (ORAUT 2008, Table 2-2). For more detail, see Attachment B. Before 1966, an 

additional 25% of the total dose should be included as an adjustment to account for the lead shielding 

described for some of the dosimeters (see Table 6-1).

6.3.5.1.1 Nuclear and Ramjet Engine Tests and Other Reactor Tests

Nuclear rocket and ramjet engine tests were conducted on NTS in Areas 25 and 26, about 80 mi 

northwest of Las Vegas, Nevada, from July 1959 through September 1969 (Friesen 1995). 

Development of the nuclear rocket engine began in 1955 under joint sponsorship of the U.S. Air 

Force and AEC (Friesen 1995). LASL was responsible for developing reactor technology for the 

nuclear rocket engine (Project ROVER) while the Air Force was responsible for the nonnuclear portion 

of the project. In 1960, the AEC and the National Aeronautics and Space Administration formed the 

Space Nuclear Propulsion Office (changed later to the Space Nuclear Systems Office) to administer 

development of an operational nuclear rocket (NERVA, Nuclear Engine for Rocket Vehicle 

Application). Such a nuclear-powered rocket was visualized for use in space travel, provided the crew 

could be adequately protected from radiation produced by the operating engine.

In a somewhat parallel program (Project PLUTO), LRL in 1957 began development of a nuclear 

ramjet engine. Because this was designed as an air-breathing engine, it was visualized as being 

restricted to relatively low altitudes. No engine tests were conducted during 1970 and 1971. The final 

related reactor test was of a “nuclear furnace” with a replaceable core in a reusable test bed designed 
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 41 of 135

to provide an inexpensive approach to testing advanced fuels in full-scale reactor environments. The 

nuclear furnace was successfully tested in June and July 1972. All nuclear rocket engines and 

systems tests were terminated in January 1973 (Friesen 1995). During operation, the nuclear and 

ramjet assemblies were very well shielded with essentially no potential for external exposure of 

personnel (Rollins 2007). However, the maintenance that was required between runs often required 

hands-on manipulation that could result in significant exposure to the photon fields from fresh fission 

products (NRDL 1968).

Other reactor and linear accelerator tests were conducted by various agencies at NTS. Operation 

Bare Reactor Experiment Nevada (BREN) and Operation High Energy Neutron Reactions Experiment 

(HENRE) and the Super Kukla reactor operations are examples. There were many unique tests 

conducted at NTS and this TBD does not include a detailed account of all operations and activities. 

Additional information and references will be researched as appropriate as claims requiring such 

information are submitted.

6.3.5.1.2 Noble Gas Exposure

Exposure to significant concentrations of noble gases – isotopes of krypton and xenon – was possible 

after an underground test (LRL 1964; Eberhart 1994). While such exposure was not of concern from 

atmospheric tests because of rapid dispersion, certain post-test operations after an underground test 

carried the risk of relatively sudden exposure to large concentrations of such gases (Eberhart 1994). 

These exposures could generally occur in two ways.

Post-test drilling operations were often conducted as soon after detonation as possible. As soon as 

the cavity had collapsed and the crater was created, drill rigs were brought in to begin the process of 

core sampling. A rig basically consisted of the drill platform, with a structure to shelter the drillers 

some distance from the drill rod entry point, and a separate platform called the monkey board several 

feet above the platform. As drilling progressed, new sections of drill rod were added by workers on 

the monkey board. Drill mud was circulated down the drill hole for lubrication. When the drill rod 

reached the cavity, circulation was lost as the mud entered the cavity rather than returning up the drill 

hole. This allowed the gases trapped in the cavity to escape up the drill hole, with the potential for 

significant exposure to workers on the platform and monkey board. The magnitude of the exposure 

would decrease with time after the detonation in proportion to the radioactive decay of the volatile 

radionuclides released (noble gases, halogens, etc.).

While the iodine that escaped in this way was primarily considered an internal exposure hazard, the 

noble gases were of concern from an external exposure standpoint, enveloping workers in a cloud, 

but not having significant potential for intake (Eberhart 1994). The primary fission product noble gas 

radionuclides are summarized in Table 6-8. Of these, 133,133mXe are of primary concern. Their fission 

yields were large enough that they were produced in significant quantities, while their half-lives were 

long enough that they would be present in significant quantities several days after the test, but not so 

long as to significantly reduce their specific activity, compared with the much longer lived 85Kr.

Table 6-8. Primary iodines and fission product noble gases with half-lives greater than 1 hour (for 

yields ≥ 0.01 photon or beta per disintegration). 

Nuclide Half-life

Photon fractions Average maximum 

beta energy (keV)

Average beta 

<30 keV 30–250 keV >250 keV energy (keV)

Kr-85 10.7 yr 0.0 0.0 1.000 685 251

Kr-85m 4.5 hr 0.0 0.84 0.16 839 290

Kr-87 76.3 min 0.0 0.0 1.00 3,119 1,333

Kr-88 2.84 hr 0.015 0.235 0.75 968 362

I-131 8.02 d 0.041 0.036 0.923 578 182

I-132 2.295 hr 0.002 0.186 0.812 1,320 485

I-133 20.8 hr 0.009 0.001 0.990 1,142 367
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 42 of 135

I-134 52.5 min 0.20 0.30 0.950 1,630 610

I-135 6.57 hr 0.003 0.017 0.980 944 363

Xe-133 5.24 d 0.0 1.00 0.0 346 100

Xe-133m 2.19 d 0.0 1.00 0.0 - -

Xe-135 9.14 hr 0.0 0.97 0.03 890 301

The potential for noble gas exposure during post-test drilling was primarily limited to radiation 

technicians, drillers, and roughnecks. Such exposures were drastically reduced with the introduction 

of blow-out preventers in 1964 (LRL 1964; Johnson et al. 1966).

Reentry after tunnel tests had the potential for noble gas exposure to miners and other crew 

members. Ventilation was usually introduced into tunnels after the tests to reduce the potential for 

such exposure before reentry. However, if the ventilation was not working, or had not been provided 

at all, the potential existed. The potential for noble gas exposure during tunnel reentries was primarily 

limited to radiation technicians and miners [22].

6.3.5.2 Beta Particle Fields

This section applies only to whole-body external doses from beta radiation that could be encountered 

from a fallout field or residual fission product activity at a work location. It does not apply to situations 

in which beta radiation was deposited directly on the skin or clothing, but rather concerns in-air doses 

to which a worker could have been exposed. Although there is a large and highly useful body of 

literature pertaining to skin doses from deposition of beta-emitting contamination on the skin, less 

information is available on external (i.e., in-air) beta doses from fallout fields or residual fission product 

activity.

There are about 250 individual fission-produced radionuclides, most of which are beta particle 

emitters.2

 Many of these beta-emitting fission product radionuclides have very short half-lives – on 

the order of a few seconds or less – and quickly decay into other nuclides. Others decay over longer 

periods. Therefore, the composition and beta energy spectrum of fallout changes over time. In 

addition to beta-emitting fission products, activation products are produced by a nuclear detonation, 

although their contribution to overall beta dose is relatively small [23]. Unlike photons, which are 

emitted with discrete quantum energies, beta emission is characterized by a distribution of energies 

ranging from zero to a maximum value that is commonly used to characterize the spectrum. The 

average energy of a beta spectrum is typically about one-third of the maximum energy, and the total 

energy produced by beta particles from fission products is essentially the same as that from fission 

product gamma rays, namely about 7 MeV per fission (Glasstone and Dolan 1977).

In contrast to gamma rays and neutrons, whose attenuation in matter is exponential, beta rays have a 

finite range in matter determined by the energy of the beta particle. Because beta particles emitted by 

radioactive species are not monoenergetic, the range is usually specified in terms of the maximum 

energy of the beta particle spectrum. For the beta produced by the decay of fission and activation 

products, the maximum energy typically does not exceed 3 MeV,3 and the range of a 3-MeV particle in 

air is approximately 36 ft. Therefore, an individual at a distance greater than 36 ft from a fallout field 

would not receive an external dose from beta radiation associated with the decay of radionuclides 

produced by fission and fission-produced activation products. Similarly, an individual exposed to beta 

particles with energies below 70 keV would receive no beta dose to the skin because beta particles 

with energies below 70 keV have insufficient energy to penetrate the cornified outer layer of the skin. 

 2 Although the term beta particle can refer to both positron and negatron emission from the nucleus of an excited atom, as 

applied to fission-produced radionuclides it refers only to negatron (i.e., electron) emission because no known fission 

products emit positrons. 3 The few exceptions are fission products with very short half-lives, which can be ignored for all practical purposes.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 43 of 135

More details related to beta particle ranges and beta-photon ratios can be found in Attachment A, 

Section A.4, and Attachment C.

For external dose reconstruction, a positive indication of beta exposure in a dose record is due to 

betas with energies greater than 15 keV.

6.3.5.3 Neutron Fields

Table 6-9 lists areas and operations where neutron exposure could have occurred at NTS. From 

1961 to 1986, fast neutron monitoring was conducted using NTA film. Personnel who did not work in 

the areas listed in Table 6-9 or were not directly involved in operations during the periods indicated 

were not issued personal neutron dosimeters. However, there was a thermal-neutron-sensitive 

component in every NTS film badge to record any indication of neutron exposure.

Table 6-9. Areas and operations where neutron exposure was possible (Allen and Schoengold 1995).

Area Operation Neutron sources Beginning Final

4a Operation BREN Fission neutrons (HPRR) 1962 1962

5b Low-level waste site TRU waste 1974 Present

6 Nuclear device assembly Fission neutrons 1951 1962

25 NRDSc Fission neutrons and neutron 

sources Cf-252, PuBe, AmBe

1959 1969

Operation HENREd Linear accelerator 1966 1968

26c PLUTO Reactor (nuclearpowered ramjet engine) Fission neutrons and neutron 

sources Cf-252, PuBe, AmBe 

1959 1969

27 Nuclear explosive assembly 

using Special Nuclear 

Material 

Fission neutrons and neutron 

sources Cf-252, PuBe, AmBe

1951 1992

Super Kuklae Super Kukla reactor 1964 1979

Variousf Down-hole well logging PuBe-238 or Cf-252 isotopic sources 1951 Present

Variousg Neutron detection instrument 

calibration facilities

PuBe-238 or Cf-252 isotopic sources 1955 Present

a. Auxier, Haywood, and Gilley (1963).

b. Norton (2006).

c. Friesen (1995).

d. Butler and Haywood (1971).

e. DOE (2007a).

f. Griffith (2007).

g. DeMarre (2007a). 

6.3.5.3.1 Weapons-Related Neutron Fields

Exposure of NTS workers to neutrons from nuclear detonations, while a theoretical possibility, was for 

all practical purposes nonexistent [24]. As shown in Attachment D, if an individual was more than 

6 km from a detonation site, the neutron dose would have been less than 1 mrem. Strict measures 

were taken to ensure that personnel were not exposed to the prompt radiation from the detonation 

and, at the locations where personnel could have been exposed to prompt neutrons from the blast, air 

attenuation and similar attenuation mechanisms would have reduced the energy and the fluence of 

the neutrons to negligible levels. The closest workers (test personnel) were at CP-2, and later CP-1 in 

Area 6 (Griffith 2006c). Therefore, dose reconstructors should ignore neutron exposure from this 

source unless there is evidence the claimant was within 6 km of one or more of the atmospheric test 

detonation points at the time of detonation. CP-1 and CP-2 were more than 6 km from atmospheric 

test locations.

Neutron exposures were possible in the vicinity of test shapes or other significant quantities of fissile 

materials. Plutonium pits that are not associated with high explosives are referred to as “bare pits,” 
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 44 of 135

although all pits are sealed or encapsulated (Shipler 2004). Assembly and disassembly operations 

comprise the only times workers have been exposed to neutrons from bare pits.

Maximum radiation dose rates occur when workers handled bare pits. Only a few individuals were 

associated with final assembly, arming, and firing of test weapons (Rollins 2007). Assembly 

operations were the only times that workers, in proximity with the weapons components, were 

exposed to neutrons from bare pits. The operations often involved direct hands-on manipulation in 

which the distance from the surface of the pit to the dosimeter was approximately 30 cm. Lead 

aprons or other shielding was used to reduce photon dose rates. In assembly or disassembly 

operations, where high explosives (HE) or other materials surround the pit, photon and neutron dose 

rates decreased significantly, although photon dose rates decreased more rapidly with increased 

shielding. Most workers involved in final assembly, arming, and firing of test weapons were national 

laboratory employees who traveled to NTS for these operations [25] (Rollins 2007).

For exposures beginning in 1961, if neutron dose information is not specifically available for those 

involved with final assembly and arming operations, photon exposure records, together with neutronphoton dose ratios, can be used. The neutron-photon ratios can be derived from experience at the 

DOE Pantex Plant, where weapons assembly operations were conducted. Analysis of dose records 

for each Pantex worker in which neutron and photon doses were equal to or greater than 50 mrem for 

the period from 1994 to 2004 yields a geometric mean of 0.8 and GSD of 1.6. An upper 95thpercentile value of 1.7 should be used for the neutron-photon dose ratio (ORAUT 2007) for neutron 

exposures beginning in 1961.

Assuming that 100% of the neutron doses were delivered by neutrons in the 0.1- to 2-MeV range is 

favorable to claimants. Although there are more penetrating neutrons with higher energies at NTS, 

the POC for deeper organs, such as the liver, is much larger, in the 0.1- to 2 MeV-energy range than 

any other energy group, thus offsetting the higher dose at depth for the more energetic energy 

ranges.

6.3.5.3.2 Isotopic Neutron Sources

A more significant potential source of neutron exposure was from isotopic neutron sources such as 

238PuBe or 252Cf. These sources were used in specific activities such as instrument calibration and 

well logging. Only a few highly trained and specialized individuals, however, had access to such 

sources [26].

6.3.5.3.3 Reactor Operations

The final source of potential neutron exposure was reactor test operations. These occurred in specific 

areas (Areas 4, 25, 26, and 27) designated for that purpose (Allen and Schoengold 1995). The 

number of individuals with the potential for neutron exposure was relatively small. The potential for 

significant neutron exposure was further mitigated by the fact that simultaneous gamma exposures 

were much greater and therefore likely to be the governing factor for exposure control (Allen and 

Schoengold 1995). Neutrons were produced only during reactor operation, and personnel were 

always at locations remote from the reactors. Therefore, neutron exposures were low or negligible 

during reactor operations or test periods (Rollins 2007). Additional information on reactor operations 

is available in ORAUT-TKBS-0008-2, Nevada Test Site – Site Description (ORAUT 2008).

6.3.5.3.4 Dosimetric Characteristics of Neutron Fields

The dose from neutrons is a function of neutron fluence and neutron energy. Fewer fast neutrons 

(i.e., neutrons with kinetic energies exceeding several hundred electron-volts) are required to produce 

a given level of dose in comparison with slow or moderated neutrons. This is reflected in the use of 

quality factors or neutron weighting factors applied to the absorbed dose to arrive at dose equivalent. 
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 45 of 135

These energy-dependent factors have changed in the last 50 years as a result of new information on 

the relative biological effectiveness (RBE) of neutrons. Table 6-10 summarizes historical changes in 

the factors used in the United States to adjust measured absorbed doses for the higher radiological 

impact associated with neutron exposures. They are listed in the neutron energy groups used in dose 

reconstruction.

Table 6-10. RBE, quality factors, or weighting factors for neutrons.

Neutron 

energy

(MeV)

Dosimetry 

guidelinea

RBE

Quality 

factorsb Average quality factors used at NTS

Neutron weighting 

factorc wr

Factor to be applied 

to NTS neutron dose

2.5E10-8 3 2

2.35 5 2.13

1E10-7

10

2

1E10-6 2

1E10-5 2

1E10-4 2

1E10-3 2

1E10-2 2.5 5.38 10 1.86

1E10-1 7.5

5E10-1 11 10.49 20 1.91

1 11

2 10

7.56 10 1.32

2.5 9

5 8

7 7

10 6.5

14 7.5

20 8

40 7 Not applicable 5 Not applicable

60 5.5

a. Trilateral meeting in 1949 radiation protection guidelines (Fix, Gilbert, and Baumgartner 1994).

b. Recommendations of NCRP Report 38 (NCRP 1971).

c. ICRP Publication 60 (ICRP 1991). Factors implemented at the NTS on January 1, 2010.

The RBE was used until 1971, after which quality factors were used. The ICRU introduced the 

radiation weighting factor wr in 1990 as part of its definition of Hp(10). Although wr has not been 

adopted in U.S. regulations, it is necessary to convert from neutron doses obtained using NCRP 

Report 38 quality factors to Hp(10) (NCRP 1971). Figure 6-6 shows the ratio of conversion 

coefficients for personal dose equivalent, Hp(10, 0°), to conversion coefficients for dose equivalent, H

(NCRP 1971). Project guidance provides conversions from radiation weighting factors for respective 

Interactive RadioEpidemiological Program (IREP) input neutron energy ranges.

0

0.2

0.4

0.6

0.8

1

1.2

1.4

1.6

1.8

2

1.E-07 1.E-06 1.E-05 1.E-04 1.E-03 1.E-02 1.E-01 1.E+00 1.E+01 1.E+02

Neutron energy - MeV

Hp(10)/H

 Figure 6-6. Ratio of fluence to dose conversion coefficients for personal dose 

 equivalent Hp(10, 0°) to conversion coefficients for dose equivalent H
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 46 of 135

6.4 DOSE RECONSTRUCTION RECOMMENDATIONS

6.4.1 Photon Radiation

Photon radiation dose reconstruction for NTS workers requires consideration of several factors:

• Determination of additional dose to unmonitored workers before the routine assignment of 

personnel dosimeters to all workers beginning in April 1957

• Adjustments to the reported DOE photon dose associated with the dosimeter response and 

radiation field characteristics

• Determination of the missed dose for monitored workers for low dose results (less than MDL of 

the personnel dosimeter)

• Determination of the IREP input photon radiation energy fraction

• Determination of the appropriate radiation quantities and exposure geometries to be used to 

determine the target organ dose

6.4.1.1 Unmonitored External Radiation Photon Dose

Assignment of dosimeters to all persons at NTS began in April 1957 and ended in December 1992

[27]. After 1992, only persons identified as having the potential for exposure were issued dosimeters; 

however, any person who requested a dosimeter would be assigned one. As noted above, all 

significantly exposed persons are likely to have been monitored (Allen and Schoengold 1995). For 

potentially exposed unmonitored workers before 1958, an estimated dose favorable to the claimant 

can be assigned based on the 50% dose of the monitored workers noted in the second column of 

Table 6-11. This provides an estimate of the potential unmonitored dose favorable to claimants, 

particularly if assigned for several years of employment. This dose can be adjusted during dose 

reconstruction to reflect the actual annual employment of the exposed person as verified by DOL. 

Most workers were not continuously employed at this site.

A 50th- and 95th-percentile dose has been calculated for the badge exchanges of persons at NTS 

and is listed in Table 6-11. For an unmonitored worker with verified employment before 1957, the 

50th-percentile dose and missed dose for each month employed during the years evaluated should be 

assigned. With the exception of the fourth plutonium dispersal event (January 18, 1956), there was 

no device testing during 1954 and 1956. Table 6-12, used in developing Table 6-11, is based on a 

query of the NTS historical dosimetry (1945 to 1983) database (DeMarre 2006) and summaries of 

AEC-190 reports (1984 to 1992) sent to AEC/DOE by NTS (DeMarre 2007b). The values include 

civilian workers and visitors and the military. An extensive review of test series measured doses can 

be viewed at http://www.dtra.mil/rd/programs/nuclear_personnel/atr.cfm.

6.4.1.2 Adjustments to Recorded External Photon Radiation Dose

Table 6-1 provides bias and uncertainty factors determined for the test series. The factors are 

dependent on the radiation field specifics of the test series and the dosimeters in use. These values

are likely to provide the most reasonable option to adjust the NTS claimant reported dose. The 

annual reported penetrating dose for each of the years of employment is divided by the bias (B) 

factors listed in Table 6-1 to arrive at the annual estimated exposure. The bias should be applied to 

the reported and the missed dose when applying the bias increases the dose being assigned.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 47 of 135

Table 6-11. NTS calculated dose to apply as coworker dose.

a

Year

Coworker dose (rem)

50% 95%

1945 0.001 4.251

1946 0.147 0.856

1947 0.036 0.499

1948-1950 No data available No data available

1951 0.058 0.941

1952 0.106 0.665

1953 0.048 3.186

1954 0.001 0.172

1955 0.014 1.173

1956 0.026 0.323

1957 0.001 0.481

a. Summaries of external dose measurements were provided as a dose 

histogram. Doses values for the 50th and 95th percentiles were calculated by 

finding the numerical position of each percentile value in the histogram to find 

the correct bin, then extrapolating to the point in the bin to calculate the dose. 

The method chosen to use to determine the 95th and 50th percentiles should 

be inherently favorable to claimants. It is likely the doses represented in each 

bin of the histogram would have been skewed toward the lower end of the bin 

(i.e., lognormally distributed). In the analysis, it was assumed that the dose 

points were equally spaced between the lower and upper limits of the bin,

which should have resulted in a higher calculated dose. 

Example:

In 1951, there were 21,086 recorded dose measurements. If the 

measurements were sorted from smallest to largest, the position that would 

represent the 95th percentile would be 20,032 (21,086 × 0.95). Knowing the 

position, you also know the histogram bin, which in this case is 0.500 – 0.999 

rem. We now extrapolate to position 20,032 in that bin. The position number 

representing the bottom of the bin is 19,987, which means the 95th percentile 

value is in position 844.7 (20,032 – 19,987) for a bin that contains 965 

measurements. Extrapolation results in a dose of 0.941 rem (844.7/965) ×

(0.999 – 0.5) + 0.5. 

6.4.1.3 Missed Recorded External Radiation Dose

Missed dose occurs when the dose of record is less than the MDL (i.e., considered to be zero) 

because the dosimeter response was less than the MDL or there is no dose of record for an assigned 

badge for a monitoring period. This kind of missed dose is most important for earlier years when 

MDLs were higher and dosimeter exchange was more frequent. 

Dose reconstructors should follow the guidance in OCAS-IG-001 (NIOSH 2007b) and use the MDL 

data in Table 6-1 to calculate the missed photon dose as follows. Missed photon dose can be 

assigned based on the MDL/2 method and project guidance regarding reported dose under the 

MDL/2, and the number of exchange periods given in Table 6-1 for the dosimetry systems or the dose 

of record provided by DOE. For many workers, the number of dosimeter exchanges will be greater 

than the routine monthly exchange. The dose reconstructor should evaluate the exchange cycles and 

assign the appropriate number of zero cycles or default values in Table 6-1 as applicable.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 48 of 135

Table 6-12. NTS associated measured doses.

a

Dose (rem)

Lower0 0.001 0.050 0.1 0.15 0.2 0.25 0.3 0.4 0.5 1.0 1.5 2.0 2.5 5.0 7.5

Higher 0 <0.05 <0.1 <0.15 <0.2 <0.25 <0.3 <0.4 <0.5 <1.0 <1.5 <2.0 <2.5 <5.0 <7.5 <10 >10

Year Number of persons

b

1945

c 776

8 23 31 14 24 14 23 17 94 87 22 25 1572 107

1946

c 16 12 23 24 21

579 15 142000000

1951 6,849 2,730 5,851 1,735 537 421 382 457 225 956 311 137 49 444200

1952 978 4,972 1,569 3,970 319 225 134 2,781 109 348 228 125 98 104722

1953

b 18,746 6,351 3,448 2,134 951 1,219 592 1,295 708 3,723 2,856 202 4,305 3,266 100

9 18

1954

b 622 205 102 32 31 13

55154001000

1955

b 16,122 4,652 1,922 3,094 595 352 325 906 1,326 3,187 1,490 407 102 292 60

2 10

1956 254 471 114 56 25 168 137 187301004

1957 42,999 9,617 9,046 5,559 2,615 2,094 1,249 1,868 1,205 2,727 676 183 98 77934

1958 19,986 4034 3696 2119 603 527 254462 231 380 54 27 112330

1959 5,537 692 313 126 70 49 37 56 46 118 44 25 247000

1960 6,820 497 205 123 44 31 18 29 14 39 21 134 11101

1961 10,842 348 302 162 86 57 49 86 44 164 96 72 146 180000

1962 15,791 838 1456 677 442 366 280 402 260 799 401 313 215 279 1500

1963 16,408 358 407 222 128 99 84 107 94 188 81 28 19 25000

1964 18,452 452 812 409 236 164 103 170 113 271 125 42 27 45000

1965 18,527 381 758 341 219 171 128 147 97 346 200 89 40 81420

1966 18,838 547 687 370 256 157 138 203 113 284 151 94 55 115000

1967 18,128 630 421 226 161 109 99 127 86 206 103 57 52 39000

1968 19,792 1158 576 260 155 88 85 134 89 203 88 50 107000

1969 16,828 339 226 106 71 43 33 56 31 110 26 1353000

1970 16,595 253 257 12574 61 41 47 41 110 34832000

1971 15,701 307 178 84 45 31 15 37 21 79 30400000

1972 91,158 414 496 290 103 42 44 46 1480000000

1973 83,227 238 248 99 32 15 10 19 1540000000

1974 82,441 262 264 77 18 10 116560000000

1975 83,775 249 262 73 34 12 126620000000

1976 79,430 151 122 329513420000000

1977 74,782 173 164 48 27 13 118930000000

1978 76,949 467 224 51 19622110000000

1979 81,266 163 67 142222010000000

1980 93,552 565 205 33 11531200000000

1981 11,0066 286 108 27 16745340000000

1982 11,9911 222 129 32 12423100000000

1983 11,2934 190 104 339312000000000

1984 25,144

d

1985 28,863

d

1

1986 28,506

d 14

1987 26,752 542 66 15811
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 49 of 135

Dose (rem)

Lower 0 0.001 0.050 0.1 0.15 0.2 0.25 0.3 0.4 0.5 1.0 1.5 2.0 2.5 5.0 7.5

Higher 0 <0.05 <0.1 <0.15 <0.2 <0.25 <0.3 <0.4 <0.5 <1.0 <1.5 <2.0 <2.5 <5.0 <7.5 <10 >10

Year Number of personsb

1988 28,060 337 45 23 2

1989 28,198 155 29 7 2

1990 25,213 141 32 2 2

1991 26,239 79 18 1 2

1992 29,559 64 12 2 1

1993e 25,429 28 5 2 2

a. Results of query for NTS personnel from NTS historical dosimetry (1945–1983) database (DeMarre 2006).

b. The number of individuals is inflated for 1953 to 1955 because persons were counted by account number and could have more than one account number due to multiple identifications 

(NTS number, employee number, social security number). Dose reconstruction records are also counted (provided by the Defense Threat Reduction Agency for military participants) and 

there are visitors and vendors who were included as well. This would make the totals higher than just film badges alone.

c. No testing occurred at NTS in 1945 and 1946; however, a continental atmospheric test occurred in New Mexico as Operation TRINITY. Because some of the same individuals were 

potentially present during later tests, coworker data for them have been included for completeness.

d. The DOE 190 statistics consisted of the following (DeMarre 2007b): 

• 1984 – 25,144 workers and visitors between 0 and 0.999 rem with collective dose of 22.620 rem

• 1985 – 28,864 workers and visitors between 0 and 0.999 rem with collective dose of 30.130 rem

• 1986 – 28,521 workers and visitors between 0 and 0.999 rem with collective dose of 57.810 rem. 

Note: The DOE 190 data (DeMarre 2007b) differ slightly from those in the NTS historical dosimetry database (DeMarre 2006). 

e. With the ending of testing, universal badging ended in 1992. Dosimetry was not required for any nonradiation worker; however, a nonradiation worker requesting a dosimeter could have 

one assigned.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 50 of 135

6.4.1.4 IREP Input Photon Radiation Energy Fractions

Attachment A of ORAUT-TKBS-0008-5, Nevada Test Site – Occupational Internal Dose (ORAUT 

2010b) presents an inventory of the radionuclides at NTS by area and operation. This information is a 

basis for estimating the IREP input fraction of the dose in each of the three energy groups – less than 

30 keV, 30 to 250 keV, and greater than 250 keV. Default values are 100% 30 to 250 keV for 

efficiently processing a claim or 25% 30 to 250 keV and 75% greater than 250 keV. The results are 

listed in Table 6-13.

Table 6-13. NTS work area and operation-dependent photon fractions.

a

Operation Area

Assigned fraction

<30 keV 30-250 keV >250 keV

Drillback operations 1–10 and 18-20 0.03 0.50 0.47

Reentry and mineback operations 1, 12, 15, and 16 0.04 0.38 0.58

Routine tunnel operations 1, 12, 15, and 0.00 0.22 0.78

Decontamination facility 6 0.14 0.43 0.44

Treatability test facility 25 0.12 0.46 0.42

Atmospheric safety test areas 5 and 11 None indicated

Atmospheric weapons test areas 1–5, 7–11, and 18 0.10 0.45 0.45

Low-level waste site 3 0.31 0.40 0.29

Low-level waste site 5 0.20 0.45 0.34

Radiation instrument calibration 6 and 23 0.36 0.40 0.24

Radiograph operations 23 0.00 0.23 0.76

Well logging operations 1–10 and 18–20 0.32 0.38 0.30

Nuclear explosive/device assembly 6 and 27 0.57 0.43 0.00

Nuclear rocket development 25 and 26 0.02 0.43 0.55

Radioactive source storage 6 and 23 0.26 0.41 0.33

Radiochemistry and counting laboratories 6 and 23 0.09 0.40 0.51

a. See Attachment B for derivation of partition fraction. 

6.4.1.5 Determination of Radiation Quantities and Exposure Geometry

As noted in Attachments A, B, and C, the conversion coefficients for exposure and personal dose 

equivalent vary widely in each of the three energy groups – a factor of more than 10 for <30 eV (with 

a 10-keV cutoff), as much as a factor of 4 for the 30-to-250-keV group, and up to a factor of 10 above 

250 keV (4,000-keV upper cutoff). The recommended dose quantities to be used in selecting dose 

conversion factors from OCAS-IG-001, External Dosimetry Implementation Guideline (NIOSH 2007b) 

are:

• 1951–1986: Exposure (R) to Organ Dose(HT) 

• 1987–present: Deep Dose Equivalent [Hp(10)] to Organ Dose (HT)

The anterior-posterior geometry should generally be selected because higher doses are typically 

recorded while workers are directly working with and facing sources of radiation.

6.4.1.6 Correction Factors for External Environmental Dose

External environmental exposures, as recorded by the dosimeter, have geometries that are different 

from those used for dosimeter calibrations. The sources are highly extended, and isotropic in contrast 

with point sources placed relatively near the dosimeter for calibration. The dose assessment for 

external environmental exposures requires attention to (1) angular or directional dependence of the 

dosimeter, (2) angular or directional dependence of the dosimetric quantity used for monitoring 

purposes, and (3) the dose of specific organs in relation to the dosimeter. Environmental exposures, 

particularly at NTS, are characterized by complex photon spectra due to the presence of a large 
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 51 of 135

number of fission and activation product radionuclides. In addition, these spectra are significantly 

modified by the attenuation of lower energy photons by several centimeters of intervening soil or 

several meters of intervening air.

The evaluation of geometric considerations and correction factors for assessment of external 

environmental exposures at NTS has been addressed by Griffith (2007). The assessment included 

three distinct exposure geometries:

1. Exposure to ground surface contamination (infinite plane surface), characteristic of fresh 

fallout.

2. Exposure to soil contaminated to an infinite depth, characteristic of fallout that had been in 

place for several days to weeks, and had been “weathered in.”

3. Submersion in a semi-infinite cloud, characteristic of airborne radioactivity as might be 

encountered in a release after an underground test.

Data on the angular dependence of the dose conversion factors is from ICRU (1988, 1992, 1998) and 

ICRP (1987). Angular dependence of the dosimeters is based on dosimeter design. Although 

specific information for NTS dosimeters was not available, data for dosimeters of similar design were

taken from results of the International Collaborative Study of Cancer Risk among Workers in the 

Nuclear Industry (Cardis et al. 2005; Thierry-Chef et al. 2002).

Calculations of the correction factors for organ geometry in relation to the dosimeter have been 

performed for exposure to (1) fallout from atmospheric tests (Hicks 1981c,i), (2) radionuclides 

released by underground test leaks (Hicks 1981a), and (3) nuclear rocket and ramjet tests (Hicks 

1981a), for exposures from time of test to 50 years after the test. The radionuclide- and geometrydependent dose conversion factors were taken from Eckerman et al. (1999).

Results of these calculations (Griffith 2007) show that the correction factors for external exposure 

environmental radiation fields at NTS are not significantly different from unity for most organs, and, in 

most cases, these values are less than 1. Given the low environmental external exposure rates at 

NTS, it appears that the new DCFs would not have a significant impact on the assigned 

environmental doses in comparison with the NIOSH dose estimates that are favorable to claimants.

6.4.2 Beta Radiation

6.4.2.1 Evaluation of Beta Exposure Geometries

If personal dose records do not include estimates of beta dose, an estimate can be made from the 

associated photon (gamma + X-ray) exposure or dose equivalent using beta-photon ratios. The betaphoton ratios are highly variable and generally decrease with distance from the source. Three 

common geometries associated with beta exposures are (1) standing on a contaminated surface, 

(2) immersion in contaminated clouds, or (3) exposure to discrete sources (Cross et al. 1982).

To determine the best value of beta-photon to be applied to the relevant photon dose, it is necessary 

to identify clearly the claimant’s work requirements as they relate to the most likely exposure

conditions. The beta-photon values in this section are in terms of sievert beta per sievert photons. 

However, with a quality factor equal to 1 for beta particles, 1 Sv = 1 Gy = 100 rad. For photons, 

1 Sv = 1 Gy. Because 1 Gy = 100 rad = 114 R, within the uncertainties associated with establishing 

beta-photon ratios, the numerical value of the beta-photon ratio can be considered valid for the period 

covering NTS operations [28].
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 52 of 135

If claimants were working in connection with the nuclear testing or the nuclear rocket and ramjet 

development program or if they were primarily working in areas where fission and activation products 

from previous operations were a significant, if not the most significant, source of exposure, data for 

one of the geometries primarily associated with such exposure – surface contamination or plume/

cloud immersion – could be used. If the claimant had work assignments at other areas of NTS, 

factors related to source inventories in those areas (see Attachment C) could be used.

Detailed inventories of fission and activation products after an atmospheric nuclear detonation, 

release of contamination after detonation, or nuclear rocket and ramjet operation are also available 

(Hicks 1981a,b,c,d,e,f,g,h,i, 1982, 1984). The development of beta-photon ratios for estimating 

external dose to the skin and eye from standing on contaminated surfaces or immersion in 

contaminated clouds is summarized in Attachment C. These values apply only to skin and eye. 

Separate beta-photon values for the gonads were not calculated (Kocher and Eckerman 1981; 

Eckerman et al. 1999) because the beta particles were not considered to contribute to the gonad dose 

beyond bremsstrahlung production in the soil and air (Eckerman 2006a).

No routine beta monitoring data exist for NTS before 1966. For the period from 1966 to 1987, 368 

data pairs were identified from 84 claim files with positive beta and gamma results (i.e., results higher 

than the applicable MDA). Based on these data, a lognormal distribution was calculated with a 50thpercentile beta-to-gamma ratio of 1.04, a 95th-percentile ratio of 4.59, a GSD of 2.41, and a mode 

value between 0.5 and 1. Thus, the use of the 50th-percentile value on an annual basis is assumed 

to be reasonable, yet favorable to claimants. The regulations at 42 CFR Part 82 allow claims to be 

completed using efficiency methods if precise estimates cannot realistically be developed due to all 

the variables that modify the potential for exposure to beta radiation. 

If a more precise estimate is required, the values in Attachment C can be adjusted and used. The 

beta-photon ratios would be reduced by attenuation of clothing [anticontamination clothing (e.g., 

coveralls), shoes, gloves, etc.], increased separation between the contaminated surface and the 

individual (e.g., working on an elevated structure), or any intervening material (surface coatings, 

floorings, etc.), time after the event, location of the cancer in relation to the source of the 

contamination, and stay times. 

6.4.2.2 Beta-Photon Ratios for Other NTS Operational Areas

If the presumed exposure was likely to be due to discrete sources or radioactive material with limited 

spatial distribution, application of beta-photon ratios associated with such exposure conditions can be 

used. The beta-photon ratio can be estimated from the radionuclide inventories known to be 

associated with a particular operation or work area. In some work situations, neither a contaminated 

ground surface nor immersion in a contaminated plume might adequately represent the exposure 

geometry. Attachment C contains detailed development of beta-photon ratios for a point-source 

geometry where the source is 1 m from the body surface. If sufficient information about the nature of 

a presumed exposure is available, more detailed calculations can be made on a case-by-case basis 

for individual dosimeter cycles. 

6.4.2.3 Estimation of Beta-Photon Exposure

The simplest calculation is for a short-term exposure to beta and gamma radiation. The total beta 

plus gamma dose to the skin or lens of the eye is estimated as (Barss 2000):

skin/lens γ/ub/fall β/γ( , ) ( , ) ( ) [ 1] Dose D t R M = ×+ xt xt (6-7)
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 53 of 135

where:

D(t)γ/ub/fall = upper-bound gamma dose due to external exposure to fallout or other beta radiation 

field

Rβ/γ(x,t) = β/γ ratio = bare skin or lens beta-gamma dose ratio at distance x and time t

M(x,t) = combined modifying factor that accounts for differences from the simple case of 

standing on contaminated ground with bare skin (e.g., attenuation by clothing, 

position of the body, and location above or below deck on a ship or in an aircraft) 

(see Barss 2000, Attachment C, Table 12)

D(t)γ/ub/total = upper-bound gamma dose from all sources

Dose to skin is reduced by clothing, and Barss (2000) provides a method for determining the 

reduction as a function of clothing thickness. However, the corrections at a point 1 m above the 

contaminated surface reduce the beta exposure on the order of 20% to 30%. In view of the 

associated uncertainties, attenuation is recommended consistent with Project guidance on shallow 

dose in ORAUT-OTIB-0017, Interpretation of Dosimetry Data for Assignment of Shallow Dose

(ORAUT 2005).

Because of the uncertainties associated with the application of beta-photon ratios to estimate potential 

beta exposures to NTS workers, dose reconstructors should only use these ratios if other relevant 

information, such as the optical density under the open area of a film badge, is not available.

If the worker dosimetry record contains no other information, including portable survey instrument 

results, beta-photon ratio estimates can be used with caution, recognizing that the results can be 

considered semiquantitative at best, with likely errors and uncertainties in the beta dose estimate 

easily exceeding a factor of 2. Uncertainty factors of 10, considering the lack of information, are not 

unreasonable [29].

6.4.2.4 Skin Contamination

For skin contamination (e.g., from fallout or resuspended radioactive soil), the film badge gamma 

dose is a highly inaccurate indicator of skin dose, so beta-gamma ratios are not appropriate for such 

applications. However, the beta energy spectrum due to radioactive material on the surface of the 

skin can be determined as a function of time after detonation for each radionuclide. This allows beta 

doses to be directly calculated by using dose coefficients from Kocher and Eckerman (1987).

These dose coefficients are based on radionuclides deposited on or near the skin surface, and are 

nearly constant for average beta energies greater than 0.1 MeV. Therefore, if a measurement of the 

skin contamination levels was recorded, it is possible to estimate an average dose coefficient for skin. 

That dose coefficient is about 9 rem/hr (beta plus gamma) per μCi/cm2 of skin (Barss 2000). This 

includes a 30% to 35% overestimate due to the potential presence of an external backscatter surface 

(e.g., a contaminated surface or tool) and a gamma contribution of about 5%. For contaminated 

gloves, a dose reduction factor of 0.5 is assumed. However, without recorded skin contamination 

levels, the skin dose is virtually impossible to determine. Skin contamination estimates should be 

based on measurements if they are available. The VARSKIN computer program (Durham 1992) can 

be used for additional calculations of skin dose, presumably for source geometries if an assumption of 

uniform large-area contamination is inappropriate.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 54 of 135

6.4.3 Neutron Radiation

6.4.3.1 Energy Dependence of Dosimetric Quantities

The neutron spectra at NTS have been generally unmoderated and rich in fast neutrons, so much so 

that doses from thermal neutrons were most certainly trivial. However, in some operational situations, 

considerable scattering could have occurred, resulting in some softening of the spectra. Attachment 

A, Section A.5 presents more detail on neutron spectral characteristics. Figure A-10 and Table A-4 

indicate that the Hp(10, 0°) contribution from neutrons with energies less than 10 keV can be ignored 

for NTS dose reconstruction. The primary Hp(10) contributions fall in the energy ranges 100 keV to 2 

MeV and 2 to 20 MeV. This information is summarized by energy band for a variety of spectra in 

Table 6-14 and NTS operational areas in Table 6-15.

Table 6-14. Fraction of Hp(10,0°) by energy band for selected spectra.

Spectral description

Neutron energy bands

<100 keV 100 keV–2 MeV 2 MeV–20 MeV

Cf-252 0.003 0.659 0.338

Cf-252 with room scatter 0.017 0.691 0.292

Cf-252, 15 cm D2O 0.089 0.579 0.332

Cf-252, D2O with room scatter 0.106 0.669 0.225

AmBe-241 0.001 0.261 0.738

AmBe with room scatter 0.010 0.366 0.624

PuBe at 1m 0.005 0.368 0.627

TRU Plant, Pu Repro. Plant, heavily shielded 0.080 0.734 0.186

Pu-238 O2 at 100 cm 0.021 0.821 0.158

Operation BREN HPRRa 0.15 0.62 0.23

Operation HENRE linear acceleratorb N/A 0.05 0.95

Super Kukla reactorc 0.10 0.86 0.04

Godiva spectrum 0.008 0.773 0.219

a. These data are representative for the HPRR at an angle of incidence of zero. For other configurations, refer to 

Sims and Ragan (1987).

b. These data are representative for the linear accelerator at an angle of incidence of zero. For other 

configurations, refer to Burson (1971).

c. Source: Wimett (1965).

The last column of Table 6-15 lists factors to make the conversion from dose equivalent (H as given in 

NCRP Report 38; NCRP 1971) to personal dose equivalent Hp(10,0°) for each of the four neutron 

energy groups for dose reconstruction at NTS (none of the neutron-producing facilities generated a 

significant number of neutrons above 20 MeV). If the neutron spectra are not known, dose 

reconstructors should treat the exposure as having come from neutrons in the 0.1- to 2-MeV range, 

and should use a factor of 2 to convert from NCRP (1971)-based values to Hp(10). If the spectral 

characteristics of the operational neutron field are known, dose reconstructors can use Table 6-17 in 

Section 6.5.2 to assign neutron dose fractions to the primary energy bands – 10 to 100 keV, 100 keV 

to 2 MeV and 2 to 20 MeV. In the absence of such information, however, it is recommended that 

dose reconstructors make the assumption, which is favorable to claimants, that neutron dose is 

entirely due to neutrons in the 100-keV to 2-MeV energy band.

6.4.3.2 Neutron-Photon Dose Ratios

The use of NTA film at NTS began in 1961 (REECo 1961). Before that time, an estimation of neutron 

doses to workers known to have worked with neutron-emitting sources or in operations where 

neutrons were present (Table 6-15) could be made from the recorded photon dose, together with an 

appropriate neutron-photon dose ratio. The nominal neutron-photon dose ratios for isotopic neutron 

sources are listed in Table 6-16. These are based on literature values for sources free in air. 

However, these values are not appropriate for estimation of the neutron dose from recorded photon 
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 55 of 135

dose because the photon usually includes a large fractional contribution from other photon sources 

that are not accompanied by neutrons. Therefore, if the neutron exposure is associated with the use 

of isotopic sources, use of a neutron-photon dose ratio of 5 is recommended. These operations 

include neutron instrument calibration and down-hole well logging.

Table 6-15. Neutron field characteristics associated with NTS operational areas.

a

Area Operation Neutron sources

Neutron energy fractional 

distribution

Neutron energy (MeV) Fraction

4 Operation BREN, 1962 Fission neutrons (HPRRb) <10 keV

10–100 keV

0.1–2.0

2.0–20

0.10

0.05

0.62

0.23

5 Low-level waste site (TRU 

waste) (1970–present)

TRU waste 0.1–2.0 1.00

6 Nuclear device assembly (1951–

1992)

Fission neutrons 0.1–2.0 1.00

25 NRDS (1959–1973) Fission neutrons and 

neutron sources Cf-252, 

PuBe, AmBe

0.1–2.0 1.00

Operation HENRE (1966–1968) Fission neutrons (linear 

acceleratorc)

0.1–2.0

2.0–20

0.05

0.95

26 PLUTO Reactor (nuclearpowered ramjet engine) (1960–

1964)

Fission neutrons and 

neutron sources Cf-252, 

PuBe, AmBe 

0.1–2.0 1.00

27 Nuclear explosive assembly 

using Special Nuclear Material 

(1958–1975)

Fission neutrons & 

neutron sources Cf-252, 

PuBe, AmBe

0.1–2.0 1.00

Super Kukla (1964–1979) Fission neutrons (Super 

Kukla reactord)

10–100 keV

0.1–2.0

2–20

0.10

0.86

0.04

Various Down-hole well logging (1951–

present)

PuBe-238 isotopic 

sources

0.1–2.0

2.0–20

0.50

0.50

Various Down-hole well logging (1951–

present)

Cf-252 isotopic sources 0.1–2.0

2.0–20.

0.75

0.25

Various Neutron detection instrument 

calibration facilities (1955–

present)

PuBe-238 isotopic 

sources

0.1–2.0

2.0–20.

0.50

0.50

Various Neutron detection instrument 

calibration facilities (1955–

present)

Cf-252 isotopic sources 0.1–2.0

2.0–20.

0.75

0.25

a. Neutron field characteristics are applicable to certain timeframes as indicated in Table 6-9.

b. These data are representative for the HPRR at an angle of incidence of zero. For other configurations, refer to Sims 

and Ragan (1987).

c. These data are representative for the linear accelerator at an angle of incidence of zero. For other configurations, refer 

to Burson (1971).

d. Source: Wimett (1965).

Of the other neutron-related operations listed in Table 6-15, only the nuclear device assembly in 

Area 6 and nuclear explosive assembly using Special Nuclear Material were carried out before the 

introduction of personal dosimetry in 1961 (Allen and Schoengold 1995). Specific information on the 

neutron-photon dose equivalent ratios is not readily available for these operations at NTS. However, 

similar operations were carried out at the Pantex Plant. Based on analysis of dosimetry records at 

Pantex, as well as at the Hanford and Savannah River Sites, where similar materials were handled, 

this resulted in a recommendation in ORAUT-TKBS-0013-6, Pantex Plant – Occupational External 
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 56 of 135

Dose (ORAUT 2007) that a neutron-photon dose equivalent ratio of 1.7 be adopted for Pantex

workers. 

Table 6-16. Neutron-photon dose ratios for isotopic 

sources.

Neutron sources

Neutron-photon dose 

equivalent rate ratio

Bare Cf-252a 20

Cf-252 moderated by 15 cm D2Oa 5.6

PuBe-238 29b

AmBe-241 >5c

a. Used for neutron dosimeter calibration starting in the early 1980s.

b. Mayer, Otto, and Golnik (2004).

c. ISO (2001).

Reactor operations in Areas 4, 25, 26, and 27 were conducted remotely, so workers were protected 

by a significant separation distance and by shielding in the case of the ramjet operations in Area 26 

(Allen and Schoengold 1995; Friesen 1995). Moreover, the workers wore neutron dosimeters during 

reactor operation. When the reactor was shut down, the neutron production was drastically reduced 

to the point that the photon dose from fresh fission and activation products far exceeded any 

remaining neutron production. Because of this and the standard use of personal neutron dosimetry, 

the use of neutron-photon ratios is not recommended for these reactor operations. 

6.4.3.3 Special Considerations

No individual had access to areas in which there was potential for neutron exposure. Work in these 

areas always involved pairs of workers; two knowledgeable persons had to be involved with potential 

operational activities. This group probably involved only a few hundred workers and was limited to 

persons who worked on specific tasks in specific areas [26]. Neutron doses, for the most part, were 

low. In addition, neutron exposure was not possible without a concomitant gamma exposure. 

Although information on nominal neutron-gamma ratios appears later in this document, in practice 

these ratios can vary widely. Caution should be exercised before applying these ratios to gamma 

exposures for those who worked in these operations (Table 6-9). Application to low-level photon 

exposures that were not associated with neutron source operations could lead to unrealistic estimates 

of neutron exposure [31].

If workers were unmonitored for fast neutrons, based on NTS personal dosimeter issue practices, and 

if there was no indication of exposure based on the thermal-neutron-sensitive component, it is highly 

unlikely that neutron exposure occurred [32]. If a worker’s duties did not involve access to fissile 

materials or isotopic neutron sources, neutron exposures should not be considered in dose 

reconstruction. Therefore, dose reconstructors should not consider missed neutron dose for NTS 

personnel who were not monitored specifically for neutrons using NTA film or TLDs depending on the 

timeframe (see Table 6-1). 

6.5 CLAIM ANALYSIS METHODS

Guidance in external radiation dose reconstruction is presented in ORAUT-PROC-0006, External 

Dose Reconstruction (ORAUT 2006). NTS-specific recommendations are presented in the following 

sections. Basics of the dosimetry system through different types and operational periods are listed in 

Table 6-1. There were no ongoing processes at NTS. The events that occurred in the various 

operations were discrete occurrences. The NTS site description TBD (ORAUT 2008) includes a list of 

radionuclides of concern for NTS activities and summarizes job descriptions for categories of workers 

that can help the dose reconstructor determine the most suitable method to use when evaluating the 

claim (i.e., minimizing, maximizing, or best estimate). 
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 57 of 135

6.5.1 Photon Dose

Beginning in 1987, NTS occupational exposures were recorded in terms of personal dose equivalent, 

Hp(d). For exposures from 1951 to 1986, dose reconstructors should use the recorded photon dose 

values in terms of exposure, together with the Exposure to Organ Dose coefficients in Appendix B of 

OCAS-IG-001 (NIOSH 2007b) to determine organ dose. Since 1987, the recorded values are in 

terms of Hp(10), and the Deep-Dose-Equivalent-to-Organ-Dose conversion factors (NIOSH 2007b, 

Appendix B) should be used.

An exception to this guidance is at the NRDS facility from 1970 to 1972, where a TLD was used and 

the Deep-Dose-Equivalent-to-Organ-Dose conversion factors (NIOSH 2007b, Appendix B) should be 

used.

Recorded doses less than the LOD/2 for the era of the dosimetry should be included as missed dose 

rather than recorded dose (NIOSH 2007b).

Energy Range. For external dose reconstruction, if the conditions of exposure (work area, operation, 

etc.) are unknown, dose reconstructors should use the assumption favorable to claimants that photon 

energies are between 30 and 250 keV (NIOSH 2007b). If the exposure was due to fresh fallout (e.g., 

early reentry teams), it would be reasonable and favorable to claimants to assume that 75% of the 

photon dose was from photons with energies above 250 keV (Kathren 2004; Coryell and Sugarman

1951; Nelms and Cooper 1959). If there is adequate documentation linking exposures to a particular 

NTS work area or operation, Table 6-9 lists guidance on a reasonable allocation of the recorded 

exposure or personal dose equivalent to the energy groups 30 to 250 keV and greater than 250 keV.

Before 1966, an additional 25% of the total dose should be included as an adjustment to account for 

the lead shielding described for some of the dosimeters (see Table 6-1). This is not an uncertainty 

and should be assigned to measured, missed, and unmonitored dose (Kathren 2004). The 

multielement dosimeter was introduced in 1966 (Author unknown undated). 

Default values are 100% 30 to 250 keV for efficiently processing a claim or 25% 30 to 250 keV and 

75% greater than 250 keV for a reasonable application. Table 6-13 lists more specific values that can 

be used if the location of the worker is clearly identified.

Bias. The bias should be applied to both the reported and the missed dose, when applying the bias 

increases the dose being assigned (NIOSH 2007b).

Uncertainty. To process a claim efficiently as noncompensable, an uncertainty of 1.3 can be applied 

as maximizing to the measured dose (NIOSH 2007b). For a reasonable estimate, an uncertainty 

based on the GSD in Table 6-1 could be applied to the measured dose.

Missed Dose. Missed photon dose can be assigned based on the MDL/2 method and Project 

guidance on reported dose under MDL/2, and the number of exchange periods (NIOSH 2007b) listed 

in Table 6-1 for the dosimetry systems or the dose of record provided by DOE. For many workers, the 

number of dosimeter exchanges will be greater than the routine monthly exchange. The dose 

reconstructor should evaluate the exchange cycles and assign the appropriate number of zero cycles 

or default values in Table 6-1 as applicable.

Special Situations

Noble Gas. The potential for noble gas exposure during post-test drilling was primarily limited to 

radiation technicians, drillers, and roughnecks. Such exposures were drastically reduced with the 

introduction of blowout preventers in 1964 (LRL 1964; Johnson et al. 1966). Reentry after tunnel 

events had the potential for noble gas exposure to reentering miners and other crew members. 
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 58 of 135

Ventilation was usually introduced in tunnels after the events to reduce the potential for such 

exposure before reentry. However, if the ventilation was not working, or had not been provided, the 

potential for exposure existed. The number of individuals involved in such operations was small [33].

Unmonitored Workers. For potentially exposed unmonitored workers before 1958, an estimated dose 

favorable to the claimant can be assigned based on the 50% dose listed in the last column of 

Table 6-11. This average annual dose can be adjusted during dose reconstruction to reflect the 

actual annual employment of the exposed person, as verified by DOL. Many of the unmonitored 

workers were hired for a specific project and terminated at the end of the project. In addition, a 

missed dose of 20 mrem/mo (LOD/2) can be assigned as favorable to the claimant. From April 1957 

through 1992, all workers entering the site were provided with a dosimeter. At NTS after 1992, only 

persons identified as having the potential for exposure were issued dosimeters; however, any person 

who requested a dosimeter would be assigned one.

Nuclear Rocket Development Station. The dose models in NRDL (1968) are applicable only to the 

relatively small population of workers who were involved in reentry operation after a nuclear 

propulsion rocket test at NRDS. The data are experiment-specific and require knowledge of specific 

physical parameters. It would be difficult to use these models for other NRDS experiments or 

reentries because the empirical data necessary to develop the curves and tables for other events 

cannot be extracted from existing NTS records. 

NRDL (1968) provides methods of computing doses from individual particles and infinite field 

contamination levels from the nuclear rocket tests. The model provides methods for calculating doses 

to the lungs, gastrointestinal tract, skin, and gonads for particular exposure scenarios. If a claimant 

who participated in NRDS nuclear rocket reentries has a covered cancer in a location where a hot 

particle exposure has been documented, and the claimant can be associated with a specific test, and 

the requisite physical parameters for the test are available from NTS records, dose reconstructors can 

consider using the models and methods in the NRDL report. If this information is not available for 

NRDS workers, external exposures can be addressed through the procedures in ORAUT-OTIB-0017, 

Interpretation of Dosimetry Data for Assignment of Shallow Dose (ORAUT 2005); OCAS-IG-001, 

External Dose Reconstruction Implementation Guide (NIOSH 2007); and the VARSKIN model. If the 

requisite information is not available, NIOSH will not speculate on potential exposure scenarios from 

large hot particles [34].

Hot Particles. NTS sampling data, bioassay or environmental, does not indicate that hot particles 

were an issue at NTS other than during the NRDS nuclear rocket test program. Historically,

measurement of hot particles was not conducted at NTS. Although insufficient or nonexistent hotparticle data from NTS make dose calculations impractical, any documented hot-particle external 

exposures can be addressed through the procedures in ORAUT-OTIB-0017, Interpretation of 

Dosimetry Data for Assignment of Shallow Dose (ORAUT 2005); OCAS-IG-001, External Dose 

Reconstruction Implementation Guide (NIOSH 2007); and the VARSKIN model. 

6.5.2 Beta Dose

If an electron component needs to be assigned based on the cancer location or type, Project 

guidance can be found in the current revision of ORAUT-OTIB-0017 (ORAUT 2005) and site-specific 

guidance can be found in Attachments A and C. These provide the dose reconstructor with options 

dependent on the circumstances of the Energy Employee (e.g., what ratio to apply depends on duties, 

location, and period of employment). The following general guidelines can be applied:

• During atmospheric testing, if little information is available about worker activities, electron 

dose can be assigned using the electron-photon dose ratio in Table 6-17.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 59 of 135

Table 6-17. Beta dose assignment.

Period Technique

Before 1966 Work area unidentified – 1.04 GM with GSD of 2.41

Evidence of exposure during a drillback or tunnel reentry – values appropriate to the 

period after the event in Attachment C. These values to be applied to the dosimeter 

exchange for the drillback or tunnel reentry.

1966 and later Measured beta dose adjusted consistent with Project guidance (ORAUT 2005).

• If the worker has skin dose recorded in 1966 and later – and has no change in job title – an 

electron-photon ratio can be developed from the worker’s data and applied to the reported 

photon dose for the period before 1966. 

• In a few cases before 1966, experimental dosimetry can be found in the files provided by 

DOE. Due to a potential for large errors in the reported dose, beta dose should be assigned 

using the beta-to-gamma ratios discussed in Section 6.4.2.1.

• After 1966, for a worker with reported beta or skin dose, the dose reconstructor should assign 

this as electron dose.

• For a reasonable approach if the location of the employee at a specific point in time can be 

determined, the ratios in Attachment C for contaminated surfaces and immersion in a cloud 

can be used. Depending on the type of contamination environment assigned to the worker, 

credit can be assigned for shielding, attenuation, length of exposure (stay time), and geometry 

when determining the organ dose correction factor.

Because of the uncertainties associated with the application of beta-photon ratios to estimate potential 

beta exposures to NTS workers, dose reconstructors should use these ratios only when other relevant 

information, such as the optical density under the open area of a film badge, is not available.

For the beta produced by the decay of fission and activation products, the maximum energy typically 

does not exceed 3 MeV, and the range of a 3-MeV particle in air is approximately 36 ft. Fission 

products with very short half-lives can exceed the maximum energy of 3 MeV, but these can be 

ignored for all practical purposes. Therefore, an individual at a distance greater than 36 ft from a 

fallout field would not receive an external dose from beta radiation associated with the decay of 

radionuclides produced by fission and fission-produced activation products. Similarly, an individual 

exposed to beta particles with energies below 70 keV would receive no beta dose to the skin because 

beta particles with energies below 70 keV have insufficient energy to penetrate the cornified outer 

layer of the skin [30].

Bias. The bias should be applied to both the reported and the missed dose when applying the bias 

increases the dose being assigned. Bias from Table 6-1 should only be applied to electrons during 

those eras for which an LOD for electrons is identified.

Special Situations

Unmonitored Workers. Before universal badging, only workers directly involved with tests were 

provided with dosimetry (Author unknown undated). From Table 6-11, photon dose is assigned to 

unmonitored workers based on the 50% dose to those who did receive monitoring. When the 

worker’s photon dose is assigned in this manner, it is favorable to the claimant to assign an electron 

dose equal to the unmonitored photon dose. This is likely to contribute a greater electron component 

than the worker received given that the worker status was unmonitored. A minimizing approach is to 

assign no unmonitored electron dose. Given the range of electrons in air and the likelihood of an 

unmonitored worker being within that range, this is a reasonable approach [31]. 
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 60 of 135

Hot Particles. NTS sampling data, bioassay or environmental, does not indicate that hot particles 

were an issue at NTS other than during the NRDS nuclear rocket test program. Historically, the 

measurement of hot particles was not conducted at NTS. Any documented hot-particle NTS external 

exposures can be addressed through the procedures in ORAUT-OTIB-0017, Interpretation of 

Dosimetry Data for Assignment of Shallow Dose (ORAUT 2005); OCAS-IG-001, External Dose 

Reconstruction Implementation Guide (NIOSH 2007); and the VARSKIN model. 

6.5.3 Neutron Dose

Dose reconstructors should convert recorded neutron dose to Hp(10) using the bias values in 

Table 6-1, and use the Deep-Dose-Equivalent-to-Organ-Dose conversion factors from Appendix B of 

OCAS-IG-001 (NIOSH 2007b) to calculate the appropriate organ doses. In addition to the bias, the 

factors in Table 6-10 for conversion from NCRP Report 38 (NCRP 1971) and ICRP Publication 60 

(ICRP 1991) neutron factors are to be applied. Table 6-15 provides neutron field characteristics 

associated with NTS operational areas and Table 6-9 provides the timeframe for application.

Energy. Assuming that 100% of the neutron doses were delivered by neutrons in the 0.1-to-2-MeV 

range is favorable to claimants. Table 6-15 lists neutron energy ranges that can be applied if more 

information is available on location and activities of the worker.

Missed Dose. Assign any missed dose based on the number of “OT” (other than thermal) neutrons in 

the records provided by DOE. If workers were unmonitored for fast neutrons, based on NTS personal

dosimeter issue practices, and if there was no indication of exposure based on the thermal-neutronsensitive component, it is highly unlikely that neutron exposure occurred [32]. If a worker’s duties did 

not involve access to fissile materials or isotopic neutron sources, neutron exposures should not be 

considered in dose reconstruction. 

There was a thermal neutron component in the NTS film badge packet from 1966 through 1986 to 

record neutron dose [34]. Every film badge packet issued on site had this component regardless of 

potential for exposure; therefore, monitoring data are not necessarily an indicator of exposure to 

thermal neutrons during routine activities. Most of the neutron exposure at NTS in this timeframe 

occurred in Area 25 (test reactor area). Low-level exposure to neutrons in Area 12 would have 

occurred only intermittently and involved only workers who directly handled the fissile materials used 

for testing. During this activity, fast neutron monitoring was conducted using NTA film, and thermal 

neutrons were monitored using the standard NTS film badge. Therefore, dose reconstructors should 

not consider missed neutron dose for NTS personnel who were not monitored specifically for neutrons 

using NTA film or TLDs, depending on the timeframe (see Table 6-1). Starting in 1961, NTA film was 

assigned to workers with the potential for exposure to neutrons (Griffith 2007). Before that, based on 

Table 6-9, it might be appropriate to assign neutrons based on a neutron-photon ratio applied to the

reported photon dose.

Dose reconstructors should ignore neutron exposure unless there is evidence the claimant was within 

6 km of one or more of the atmospheric test detonation points at the time of detonation.

Bias. The bias should be applied to both the reported and the missed dose if applying the bias 

increases the dose being assigned. When using the neutron-photon ratio, the bias should be from the 

appropriate neutron dosimeter era (Table 6-1) and the correction factor (conversion from NCRP to 

ICRP) applied based on the information in Table 6-10.

Uncertainty. To process a claim efficiently as noncompensable, an uncertainty of 1.3 can be applied 

as maximizing to the measured dose. For a reasonable estimate, an uncertainty based on the GSD in 

Table 6-1 could be applied to the measured dose.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 61 of 135

Neutron-Photon Ratio. The workplace neutron fields for specific types of nuclear weapons 

components are classified. Unclassified information on neutron spectra from nuclear weapons 

components is not available, but there are two sources, both with significant components above 

2 MeV. Before 1960, it is, therefore, recommended that a neutron-photon dose equivalent ratio of 1.7 

be used for NTS workers who were involved in nuclear device assembly or nuclear explosive 

assembly using Special Nuclear Material for neutron dose estimation during the periods when these 

operations were conducted based on NTS records (based on ORAUT-TKBS-0013-6, Pantex Plant –

Occupational External Dosimetry; ORAUT 2007). Assembly occurred in Area 27, normally performed 

by the scientists and technicians from the laboratory facilities.

For exposures after 1960, if neutron dose information is not specifically available for those involved 

with final assembly and arming operations, photon exposure records, together with neutron-photon

dose ratios, can be used. The neutron-photon ratios can be derived from experience at the Pantex 

Plant, where weapons assembly operations were conducted. Analysis of dose records for each 

Pantex worker with a positive neutron dose greater than 50 mrem for 1993 to 2003 yields a geometric 

mean (GM) of 0.8 and GSD of 1.6. An upper 95th-percentile value of 1.7 should be used for the 

neutron-photon dose ratio (ORAUT 2007).

Special Situations

Reactor Operations. Reactor and linear accelerator tests were conducted by various agencies at NTS 

(e.g., Operations BREN and HENRE and the Super Kukla reactor operations):

• For claimants with a record of participation in Operation BREN, specific information for the 

project is available in Auxier et al. (1962, 1963), Cheka et al. (1965), Sanders et al. (1962), 

Sims and Ragan (1987), Thorngate and Loy (1965), and Thorngate et al. (1967). 

• For claimants with a record of participation in Operation HENRE, specific information for the 

project is available in Burson (1971), Butler and Haywood (1971), French and Mooney (1971), 

Haywood, Provenzano, and Auxier (1965), Haywood and Auxier (1965), Provenzano et al. 

(1966), Sanna et al. 1969, and Thorngate et al. (1969). 

• For claimants with a record of participation at the Super Kukla reactor facility, specific 

information for the reactor is available in Wimett (1965). 

There were many individual and unique tests at NTS and this TBD does not include a detailed 

account of all operations and activities. Additional information and references will be researched as 

appropriate as claims requiring this information are submitted.

6.6 ATTRIBUTIONS AND ANNOTATIONS

Where appropriate in this document, bracketed callouts have been inserted to indicate information,

conclusions, and recommendations provided to assist in the process of worker dose reconstruction. 

These callouts are listed here in the Attributions and Annotations section, with information to identify 

the source and justification for each associated item. Conventional References, which are provided in 

the next section of this document, link data, quotations, and other information to documents available 

for review on the Project’s Site Research Database.

[1] DeMarre, Martha E. Manager, Nuclear Testing Archive. October 2003.

Ms. DeMarre, directly and through a large number of background documents, has provided a 

chronicle of the NTS radiation protection program for external and internal exposure. Ms. 

DeMarre gained knowledge about the radiation protection programs for each of the test series 

from documents such as Defense Nuclear Agency reports DNA-6000f through DNA-6040f.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 62 of 135

[2] DeMarre, Martha E., Manager, Nuclear Testing Archive, and Griffith, Richard V., Principal 

Consultant, ORAU Team. January 2004.

Beta doses are calibrated and measured in terms of absorbed dose. “Beta-particle fields shall 

be calibrated in terms of absorbed dose at a tissue depth of 7 mg/cm2 ...” (DOE 1986a). The 

special unit of absorbed dose is the rad. Dose equivalent, H, is defined as the product of 

absorbed dose and quality factor, Q, or more recently, radiation weighting factor, WR. The 

quality factor and radiation weighting factor for both Q and WR have been set by ICRP equal 

to 1. Therefore, the beta dose equivalent and absorbed dose are numerically equal. 

Therefore, beta doses can also be reported in dose equivalent with the special unit of rem.

[3] Griffith, Richard V. ORAU Team. Principal Consultant. April 2007.

After review of the dosimetry practices used to determine the values to be entered in the 

dosimetry records, the bias factor represents the under- or overestimate of the measurement 

quantity by the recorded dose. By the convention adopted by the NTS dose reconstruction 

project, a bias factor less than 1.0 is used if the recorded value is believed to be an 

underestimate of the measurement quantity, while a bias factor greater than 1.0 is used if the 

recorded value is an overestimate of the measurement quantity. Simply put, bias factor = 

recorded value/measurement quantity.

[4] Griffith, Richard V. ORAU Team. Principal Consultant. April 2007.

OCAS-IG-001 (NIOSH 2007b) specifies that the value of the measurement quantity should be 

multiplied by the dose conversion factors in Appendix B to obtain the organ dose.

[5] DeMarre, Martha E. Manager, Nuclear Testing Archive. October 2003.

The dosimetry processing procedures, including the use of control films, has been outlined in 

REECo Environmental Sciences Standard Procedures for 1952 through 1985. These 

procedures have been compiled by Ms. DeMarre and are available. However, because of the 

number of procedures involved, they are not referenced individually.

[6] DeMarre, Martha E. Manager, Nuclear Testing Archive. October 2003.

This is a matter of NTS record.

[7] DeMarre, Martha E., Manager, Nuclear Testing Archive, and Griffith, Richard V., Principal 

Consultant, ORAU Team. January 2004.

The temporary assignment of major laboratory and contractor employees at NTS for up to 

several weeks, perhaps punctuated by a return home over a weekend or for a few days, was a 

common practice and well known in the AEC/DOE contractor community.

[8] DeMarre, Martha E. Manager, Nuclear Testing Archive. January 2004.

Ms. DeMarre has substantiated the practice of contractor employees exchanging NTS-issued 

dosimeters for contractor badges on completion of their assignments.

[9] Griffith, Richard V. ORAU Team. Principal Consultant. February 2007.

This is based on the experience of data entry technicians during data entry. According to 

Cheryl Moore, Dade Moeller and Associates, NIOSH Project Data Entry Supervisor, often the 

LLNL or LANL records identify NTS results as “NTS” or “visitor.”

[10] DeMarre, Martha E. Manager, Nuclear Testing Archive. February 2006.

Sandia National Laboratories handled the dosimetry (if there was any) for TTR. REECo had a 

Human Resources Office outside NTS and one in Las Vegas. There was a REECo medical 

clinic in Mercury and one in Las Vegas (leased space until 1994).
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 63 of 135

When a REECo employee “processed in,” the employee would usually start the process in Las 

Vegas. If the employee worked at NTS or Tonopah, they would usually report to the REECo 

Human Resources Office outside the gate. When they were ready to go to Medical (which 

was inside the gate at Mercury), they would have to get a dosimeter at Building 1000 (guard 

building outside the gate). Once they had their badge, they would go to REECo Medical 

(Mercury Building 650) and have their physical. They would physically be on NTS for a very 

short time (hours). 

They are listed as NTS visitors if there is a place for a visitor indicator in the record. They do 

not have a permanent NTS security badge for TTR employment.

TTR employment was considered “Work for Others” (in support of Sandia and the U.S. Air 

Force).

[11] Griffith, Richard V. ORAU Team. Principal Consultant. February 2007.

A hot particle is essentially a point source of radiation. If that particle is not deposited directly 

on the dosimeter, the radiation field from the particle spreads such that the dosimeter cannot 

distinguish between the radiation field from the particle and that from a distributed source. 

Therefore, a hot particle deposited on the body at even a small distance from the dosimeter 

could not be identified.

[12] Griffith, Richard V. ORAU Team. Principal Consultant. February 2007.

Hot particles will be removed in a short time (usually less than a day) after deposition on the 

body or clothing by washing or removal of the clothing. Once the particle has been removed, 

there is no way of knowing that it had been deposited unless associated radiation damage to 

the skin and underlying tissue manifested itself as an area of physical damage or a lesion.

[13] Griffith, Richard V. ORAU Team. Principal Consultant. February 2007.

During interviews with people who were involved in NTS operations, including [name redacted]

(NTS radiation protection specialist), there were a number of references to occasions when 

workers had intentionally left dosimeters in locations where they knew there would be little or 

no dose. One of the reasons commonly given was concern about reduction in pay that might 

be associated with work restrictions if dose limits were exceeded. Reports of this practice 

were not limited to NTS activities.

[14] DeMarre, Martha E., Manager, Nuclear Testing Archive, and Griffith, Richard V., Principal 

Consultant, ORAU Team. February, 2004.

The development of dosimetry technology, not only at NTS but elsewhere in the AEC/DOE 

community, is documented in many reports by REECo, the national laboratories, the military, 

and other contractors. This development is reflected in several of the reports in the reference 

list, and is not necessary here.

[15] Kathren, Ronald L. ORAU Team. Consultant. January 2004.

The potential large uncertainty (a factor of 2) is based on professional experience. The 

recommendation that “dose reconstructors should double the reported value to ensure 

favorability to claimants and to account for uncertainties” follows from this experience.

[16] Griffith, Richard V. ORAU Team. Principal Consultant. February 2007.

The sensitive element of film and TLD dosimeters used to detect incident beta particles is 

covered with a thin layer of protective material (paper, plastic, etc.). The range of a 0.015 MeV 

beta particle in unit density material is 4 × 10-3 mm, so the beta particles with energies 

≤0.015 MeV would be absorbed in the inert protective material and not detected.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 64 of 135

[17] DeMarre, Martha E., Manager, Nuclear Testing Archive, and Griffith, Richard V., Principal 

Consultant, ORAU Team. February 2004.

Mechanisms for production of neutrons are limited and well defined:

• Nuclear fission

• Alpha,neutron reactions with a small number of light nuclides (lithium, boron, beryllium, 

etc.)

• Accelerator production

The facilities or operations at NTS that were capable of producing neutrons are identified in 

ORAUT-TKBS-0008-2 (ORAUT 2008) and summarized briefly in this section. Because of the 

nature of these operations and the associated hazards, access of individuals has been 

carefully controlled.

NTA film, albedo TLDs, and TEDs were not issued to all personnel at NTS. They were issued 

only to individuals who might have a potential for exposure to neutrons.

The brief reactor runs had a very small potential for neutron exposure. The higher potential 

was from sources (such as Pu-Be and 252Cf) to an extremely small number of people. The 

tasks included instrument calibration and well logging. The number of workers with a potential 

for exposure would be extremely low.

[18] Griffith, Richard V. ORAU Team. Principal Consultant. February 2004.

The use of field calibrations is and has been standard practice in radiation protection for many 

years. It allows direct comparison of the response of dosimeters and reference or survey 

instruments in the occupational environment and eliminates, among other things, uncertainties 

about the effects of energy response because the operational field becomes the calibration 

field as well.

[19] Griffith, Richard V. ORAU Team. Principal Consultant. February 2004.

Neutron calibration can be performed with isotopic sources, reactor beams, or accelerators. 

However, use of facilities such as reactors and accelerators for calibration of dosimeters and 

survey instruments involves a significant commitment of resources and labor. They are 

normally used only for special calibration requirements such as establishing the energy 

response of instruments and dosimeter designs. Once that has been done, simpler, less 

labor-intensive methods are used to confirm proper operation of neutron detection equipment. 

Isotopic sources such as 252Cf, 238Pu-Be, etc., are much more readily available to the normal 

radiation protection organization, and are used much more widely.

[20] DeMarre, Martha E., Manager, Nuclear Testing Archive, and Griffith, Richard V., Principal 

Consultant, ORAU Team. March 2004.

Operational field calibrations for neutrons require levels of a few tenths of a millirem so the 

measurements can be conducted in a reasonable time. From discussions with NTS radiation 

protection specialists such as [name redacted] and [name redacted], or in going through NTS

records, there is no indication that such field calibrations were attempted.

[21] Griffith, Richard V. ORAU Team. Principal Consultant. March 2007.

Because of variations in TED materials, etc., the practical detection level for fast neutrons 

(from 252Cf, 238Pu-Be, etc.) is higher than that achieved under highly controlled conditions –

typically about 25 mrem. However, for moderated spectra, a significant fraction of the 

neutrons will have energies below the energy detection threshold for TEDs. In addition, 

because the factor to be applied to NTS neutron dose in the 0.1-to-2-MeV region is 2 as a 
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 65 of 135

result of the difference between NCRP and ICRP dose conversion factors (Table 6-10), the 

recommended MDL is 30 mrem (see Section 6.3.4.2.3).

[22] DeMarre, Martha E., Manager, Nuclear Testing Archive, and Griffith, Richard V., Principal 

Consultant, ORAU Team. March 2004.

Because of the scale and design of operations, and the time to reentry, exposure to 

radioiodines and noble gases during tunnel reentries was limited to those actually involved in 

the reentry – miners and radiation technicians.

[23] Griffith, Richard V. ORAU Team. Principal Consultant. March 2007.

Fission products are formed directly, so there are at least two fission product atoms formed 

from each fission. Activation products are formed as a secondary process through interaction 

of the fission product neutrons with adjacent materials (soil, structures, etc.). Activation 

product formation is geometry-dependent, so fewer activation product atoms are formed from 

an air burst than a surface burst. Moreover, the activation products are distributed with depth 

in soil because they originate primarily by interactions with neutrons that penetrate the soil 

rather than in deposition from the atmosphere. Thus, beta doses from activation products are 

reduced because most of the activation products are deeper in the soil than the range of the 

emitted beta particles.

[24] Griffith, Richard V. ORAU Team. Principal Consultant. March 2007.

Attachment D shows, using two different methods, that personnel doses at distances greater 

than 6 km would be less than 1 mrem.

[25] DeMarre, Martha E. Manager, Nuclear Testing Archive. April 2006.

The device was assembled by specialists from the laboratories (LANL, LLNL, etc.).

[26] DeMarre, Martha E., Manager, Nuclear Testing Archive, and Griffith, Richard V., Principal 

Consultant, ORAU Team. February 2004.

Neutron sources were used for specific jobs such as instrument calibration and well logging. 

These jobs required specialized training, including the potential hazards associated with use of 

the sources, and involved only a few individuals such as radiation technicians. When the 

sources were in use, individuals who were not directly involved in the task were kept away 

from them and at a safe distance.

[27] DeMarre, Martha E. Nuclear Testing Archive. Manager. October 2003.

Ms. DeMarre has provided information about personnel dosimeter issuance.

[28] Griffith, Richard V. ORAU Team. Principal Consultant. March 2007.

The error introduced in using a 1-to-1 rad-to-roentgen relationship and ignoring the conversion 

of 100 rad = 114 R is 14 %. However, the uncertainties caused by the significant beta 

geometry and energy dependence of the beta-photon ratios are much larger and can exceed a 

factor of 2 or more. Therefore, 14% becomes acceptable in the context of other much larger 

errors.

[29] Griffith, Richard V. ORAU Team. Principal Consultant. March 2007.

Beta-photon ratios are highly dependent on geometry, including any intervening material (air, 

etc.), beta and photon energies, and other factors. Although calculations can be made for 

well-defined situations, application of the results can result in significant errors if the actual 

exposure conditions deviate significantly from the assumptions used for the calculations.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 66 of 135

[30] Griffith, Richard V. ORAU Team. Principal Consultant. March 2007.

The range of a 70-keV beta particle is about 7 mg/cm2 in tissue, and it cannot penetrate the 

dead or cornified outer layer of the skin. Therefore, there is no beta dose to the sensitive layer 

from beta particles with energies below 70 keV.

[31] Griffith, Richard V. ORAU Team. Principal Consultant. March 2007.

Unmonitored workers are highly unlikely to have been in areas where significant beta fields 

would have existed. Moreover, in view of the short range of beta particles in air (generally less 

than 100 cm), (1) workers would have had to be quite close to these sources and (2) the 

sources would have been much more intense than contamination background fields. 

Therefore, it is quite reasonable to assign no unmonitored electron dose for unmonitored 

workers.

[32] Arana, Joel. ORAU Team. Senior Health Physicist. March 2007.

Explanation of the thermal neutron component in the NTS film badge as recommended by Tim 

Taulbee of NIOSH. In addition, the thermal neutron component is not used for accident 

monitoring.

[33] DeMarre, Martha E., Manager, Nuclear Testing Archive, and Griffith, Richard V., Principal 

Consultant, ORAU Team. March 2004.

The personnel with potential for noble gas exposure during a postshot drilling operation were 

those whose job assignments called for them to be directly involved in the drilling operation –

drilling crew members and radiological technicians. Because of the training and skills that 

were required for these jobs, the number of people who were involved was small.

[34] Rollins, Eugene M. ORAU Team. Division Manager. August 2007.

ORAU Team resolution of comments provided by Sanford Cohen & Associates. 

[35] Griffith, Richard V. ORAU Team. Principal Consultant. April 2007.

Allen and Schoengold (1995) states, “The Panasonic UD-802 TLDs have well-known and 

documented energy responses.” As a large commercial supplier of dosimeters, this is true of 

all Panasonic TLDs, including the UD-809, which is not specifically addressed in the REECo 

document.

[36] Griffith, Richard V. ORAU Team. Principal Consultant. April 2007.

This information is provided in detail in the Characteristics of the Panasonic UD-8xx series 

Thermoluminescent Dosimeters in Panasonic (2012) (also available from the Panasonic Web 

site at http://www.panasonic.com/industrial/other/pdf/dosimeter_types.pdf).

[37] Griffith, Richard V. ORAU Team. Principal Consultant. April 2007.

The neutron response of NTA film at 4 MeV is approximately 2.1 times that at 1 MeV (IAEA 

1990). This means that 4-MeV neutrons, when corrected for a personal dose equivalent 

difference of less that 10%, will create twice as many tracks as 1-MeV neutrons per unit 

Hp(10). As a result, if a dosimeter has been calibrated with an α,n source such as plutonium 

or AmBe and that dosimeter is used in a fission neutron environment, the number of track 

density is interpreted as a dose that is half the actual fission neutron dose. This underestimate 

is worse if the field has a significant number of neutrons below the 0.5-MeV NTA threshold.

[38] Griffith, Richard V. ORAU Team. Principal Consultant. March 2007.

Immersion in a cloud was generally associated with radioactive gases, which are normally 

dispersed quite rapidly in an open environment. Under normal conditions, this would not be 

common. Most exposures would have resulted from surfaces (ground, etc.) that were 
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 67 of 135

contaminated by postshot fallout. The exception would be exposure following a release of 

gases during postshot drilling.

[39] Griffith, Richard V. ORAU Team. Principal Consultant. March 2007.

Using data from Keith Eckerman of the EPA and the EPA Federal Guidance Report-13 code 

(Eckerman et al. 1999), it is possible to calculate beta-photon ratios for the fallout inventories 

generated by Hicks (1981a,b,c,d,e,f,g,h,i). These ratios can be applied, with care, to the 

photon doses from personal dosimetry records.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 68 of 135

REFERENCES

AEC (U.S. Atomic Energy Commission), 1961, Radiological Safety, Chapter 0524, NTSO-0524-1, 

Nevada Test Site Organization, Las Vegas, Nevada, June 26. [SRDB Ref ID: 12468]

Allen, T. E., and C. R. Schoengold, 1995, External Dosimetry Technical Basis Manual for the Nevada 

Test Site and Other Designated Locations, (548)D-6.4.1, Reynolds Electrical & Engineering 

Company, Las Vegas, Nevada, December. [SRDB Ref ID: 14099]

Author unknown, undated, Nevada Test Site External Dosimetry Fact Sheet, Monitoring of Personnel 

Exposures in Association with the Nuclear Testing Program. [SRDB Ref ID: 1904]

Auxier, J. A, F. F. Haywood, and L. W. Gilley, 1963, General Correlative Studies – Operation BREN, 

CEX-62.03, U.S. Atomic Energy Commission, Civil Effects Test Operations, Oak Ridge, 

Tennessee, September. [SRDB Ref ID: 37782]

Auxier, J. A., F. W. Sanders, F. F. Haywood, J. H. Thorngate, and J. S. Cheka, 1962, Technical 

Concept - Operation BREN, CEX-62.01, U.S. Atomic Energy Commission, Civil Effects Test 

Operations, Oak Ridge, Tennessee, January. [SRDB Ref ID: 37779]

Barss, N. M., compiler, 2000, Methods and Applications for Dose Assessment of Beta Particle 

Radiation, SAIC-001/2024, Science Applications International Corporation, January. [SRDB 

Ref ID: 26379]

Barss, N. M., and R. L. Weitz, 2006, “Reconstruction of External Dose from Beta Radiation Sources of 

Nuclear Weapon Origin,” Health Physics, volume 91, number 4, pp. 379–389. [SRDB Ref ID: 

26776]

Beck, H. L., 1980, Exposure Rate Conversion Factors for Radionuclides Deposited on the Ground, 

EML-378, U.S. Department of Energy, Environmental Measurements Laboratory, New York, 

New York, July. [SRDB Ref ID: 35519]

Becker, K., 1966, Photographic Film Dosimetry, The Focal Press, London, England.

Black, R. H., 1962, “Some Factors Influencing the β-Dosage to Troops,” Health Physics, volume 8, pp. 

131–141.

BN (Bechtel Nevada), 2001, Dosimetry Issue/Return, Organization Instruction OI-0441.024, Rev. 3, 

Las Vegas, Nevada, October 24. [SRDB Ref ID: 2062]

Boone, F. W., E. D. Bennett, and S. B. Adams, 1970, Progress Report, NRDS Personnel Dosimetry 

Program, Statistical Comparison of TLD and Film Dosimeter Data, 33-37, Pan American World 

Airways, Nuclear Rocket Development Station Project, Jackass Flats, Nevada, July 24. 

[SRDB Ref ID: 13988]

Brady, W. J., and G. K. Iverson, 1968, Combination Personnel Dosimeter and Security Credential 

Holder, NVO-162-27 (revised), Reynolds Electrical & Engineering Company, Nevada Test 

Site, Mercury, Nevada, September. [SRDB Ref ID: 1477]

Brady, W. J., and A. G. Nelson, 1985, Radiac Instruments and Film Badges Used at Atmospheric 

Nuclear Tests, DNA-TR-84-338, Jaycor, Alexandria, Virginia, September 26. [SRDB Ref ID: 

1510]
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 69 of 135

Broido, A., and J. D. Teresi, 1961, “Analysis of the Hazards Associated with Radioactive Fallout 

Material – I. Estimation of γ- and β-Doses,” Health Physics, volume 5, pp. 63–69.

Burson, Z. G., 1971, Radiation Output Description from the T(d,n)4

He Reaction in a Large Target 

(1000 cm2

) — Operation HENRE, CEX-65.04, U.S. Atomic Energy Commission, Civil Effects 

Test Operations, Oak Ridge, Tennessee, March. [SRDB Ref ID: 39858]

Butler, H.M. and F. F. Haywood, 1971, Radiation Safety Practices—Operation HENRE, CEX-65.05, 

U.S. Atomic Energy Commission, Civil Effects Test Operations, Oak Ridge, Tennessee,

January. [SRDB Ref ID: 39861]

Cardis, E., M. Vrijheid, M. Blettner, E. Gilbert, M. Hakama, C. Hill, G. Howe, J. Kaldor, C. R. 

Muirhead, M. Schubauer-Berigan, T. Yoshimura, F. Bermann, G. Cowper, J. Fix, C. Hacker, B. 

Heinmiller, M. Marshall, I. Thierry-Chef, D. Utterback, Y-O. Ahn, E. Amoros, P. Ashmore, A. 

Auvinen, J-M. Bae, J. B. Solano, A. Biau, E. Combalot, P. Deboodt, A. Diez Sacristan, M. 

Eklof, H. Engels, G. Engholm, G. Gulis, R. Habib, K. Holan, H. Hyvonen, A. Kerekes, J. 

Kurtinaitis, H. Malker, M. Martuzzi, A. Mastauskas, A. Monnet, M. Moser, M. S. Pearce, D. B. 

Richardson, F. Rodriguez-Artalejo, A. Rogel, H. Tardy, M. Telle-Lamberton, I. Turai, M. Usel, 

and K. Veress, 2005, “Risk of Cancer After Low Doses of Ionising Radiation: Retrospective 

Cohort Study in 15 Countries,” British Medical Journal, volume 331, p. 77. [SRDB Ref ID: 

64512]

Cheka, J. S., 1954, “Recent Developments in Film Monitoring of Fast Neutrons,” Nucleonics, volume 

12, number 6, pp. 40–43. [SRDB Ref ID: 25005]

Cheka, J. S., F. W. Sanders, T. D. Jones, and W. H. Shinpaugh, 1965, Distribution of Weapons 

Radiation in Japanese Residential Structures, CEX-62.11, U.S. Atomic Energy Commission, 

Civil Effects Test Operations, Oak Ridge, Tennessee, August. [SRDB Ref ID: 37784]

Collison, T. D., 1953, Report to the Test Director, Radiological Safety Operation, WT-702(REF.), 

Armed Forces Special Weapons Project, Field Command, Albuquerque, New Mexico, June. 

[SRDB Ref ID: 2480]

Collison, T. D., 1955, Report to the Test Director, Radiological Safety, WT-1166, Armed Forces 

Special Weapons Project, Field Command, Albuquerque, New Mexico, May. [SRDB Ref ID: 

2498]

Coryell, C. D., and N. Sugarman, editors, 1951, The Fission Products, National Nuclear Energy 

Series, Book I, Division IV, Part IV, McGraw-Hill, New York, New York.

Cross, W. G., H. Ing, N. O. Freedman, and J. Mainville, 1982, Tables of Beta-ray Dose Distributions in 

Water, Air and Other Media, AECL-7617, Atomic Energy of Canada Limited, Chalk River 

Nuclear Laboratories, Chalk River, Ontario, September. [SRDB Ref ID: 40277]

DeMarre, M. E., 2003, “RE: Information Request,” e-mail to L. J. Arent (Oak Ridge Associated 

Universities Team), U.S. Department of Energy, Nevada Operations Office, Las Vegas, 

Nevada, September 17. [SRDB Ref ID: 14053]

DeMarre, M. E., 2006, “Exposure Matrix for NTS and PPG Associated Persons Based on Query of 

Historical Dosimetry (1945-1983) Database,” e-mail to J. Fix (Oak Ridge Associated 

Universities Team), U.S. Department of Energy, Nevada Operations Office, Las Vegas, 

Nevada, August 3. [SRDB Ref ID: 27080]
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 70 of 135

Demarre, M. E., 2007a, “RE: Neutron Calibration Sources at NTS,” e-mail to D. Griffith, (Oak Ridge 

Associated Universities), U.S. Department of Energy, Nevada Operations Office, Las Vegas, 

Nevada, November 6. [SRDB Ref ID: 42676]

DeMarre, M. E., 2007b, “RE: Summary Dosimetry Statistics,” e-mail to C. Smith (Oak Ridge 

Associated Universities), U.S. Department of Energy, Nevada Operations Office, Las Vegas, 

Nevada, August 20. [SRDB Ref ID: 34464]

Goetz, J., J. Klemm, C. Thomas, and R. Weitz, 1985, Neutron Exposure for DoD Nuclear Test 

Personnel, DNA-TR-84-405, Defense Nuclear Agency, Washington, D.C., August 15. [SRDB 

Ref ID: 40286]

Griffith, R. V., 2004, 1951-1973-NTS Instrument Summary, Private Communication, documented 

communication with M. E. DeMarre (U.S. Department of Energy, Nevada Operations Office), 

Oak Ridge Associated Universities Team, Oak Ridge, Tennessee. [SRDB Ref ID: 2080]

Griffith, R. V., 2006a, Use of Pocket Ion Chambers at the NTS,” documented communication with M. 

E. DeMarre (U.S. Department of Energy, Nevada Operations Office), Oak Ridge Associated 

Universities Team, Oak Ridge, Tennessee. [SRDB Ref ID: 40282]

Griffith, R. V., 2006b, “NTPR Data Definitions,” documented communication with M. E. DeMarre (U.S. 

Department of Energy, Nevada Operations Office), Oak Ridge Associated Universities Team, 

Oak Ridge, Tennessee. [SRDB Ref ID: 40283]

Griffith, R. V., 2006c, “Extremity Monitoring,” documented communication with M. E. DeMarre (U.S. 

Department of Energy, Nevada Operations Office), Oak Ridge Associated Universities Team, 

Oak Ridge, Tennessee. [SRDB Ref ID: 40284]

DOE (U.S. Department of Energy), 1982, Radiological Safety, Chapter 0524, NTS-SOP-0524, 

Nevada Operations Office, Las Vegas, Nevada, July 23. [SRDB Ref ID: 13991]

DOE (U. S. Department of Energy), 1986a, Department of Energy Standard for the Performance 

Testing of Personnel Dosimetry Systems, DOE/EH-0027, Assistant Secretary for Environment, 

Safety, and Health, Washington, D.C., December. [SRDB Ref ID: 12294]

DOE (U.S. Department of Energy), 1986b, Handbook for the Department of Energy Laboratory 

Accreditation Program for Personnel Dosimetry Systems, DOE/EH-0026, Assistant Secretary 

for Environment, Safety, and Health, Washington, D.C., December. [SRDB Ref ID: 12293]

DOE (U.S. Department of Energy), 1988, Radiation Protection for Occupational Workers, Order 

5480.11, Office of Environment, Safety and Health, Washington D.C., December 21. [SRDB 

Ref ID: 8073]

DOE (U.S. Department of Energy), 1994, U.S. Department of Energy Radiological Control Manual, 

DOE/EH-0256T, Rev. 01, Office of Assistant Secretary for Environment, Safety and Health, 

Washington, D.C., April. [SRDB Ref ID: 14856]

DOE (U.S. Department of Energy), 1995, Department of Energy Laboratory Accreditation Program for 

Personnel Dosimetry Systems, DOE-STD-1095-95, U.S. Department of Energy, Washington, 

D.C., December. [SRDB Ref ID: 40288]
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 71 of 135

DOE (U.S. Department of Energy), 1997, Linking Legacies: Connecting the Cold War Nuclear 

Weapons Production Processes to Their Environmental Consequences, DOE/EM-0319, Office 

of Environmental Management, Washington, D.C., January. [SRDB Ref ID: 11930]

DOE (U.S. Department of Energy), 2000, United States Nuclear Tests, July 1945 through September 

1992, DOE/NV--209-REV 15, Nevada Operations Office, Las Vegas, Nevada, December. 

[SRDB Ref ID: 2433]

DOE (U.S. Department of Energy), 2007a, Closure Report for Corrective Action Unit 118: Area 27 

Super Kukla Facility, Nevada Test Site, Nevada, DOE/NV-1223, National Nuclear Security 

Administration, Nevada Site Office, Las Vegas, Nevada, September. [SRDB Ref ID: 42679]

DOE (U.S. Department of Energy), 2007b, Nuclear Rocket Development Station Tested Nuclear 

Rocket for Mars Mission, National Nuclear Security Administration, Nevada Site Office, Las 

Vegas, Nevada, November 7. [SRDB Ref ID: 40274]

Dudley, R. A., 1966, “Dosimetry with Photographic Emulsions,” Radiation Dosimetry, Volume 2, F. H. 

Attix and W. C. Roesch, editors, Academic Press, New York, New York.

Durham, J. S., 1992, VARSKIN Mod 2 and SADDE Mod 2: Computer Codes for Assessing Skin Dose 

from Skin Contamination, NUREG/CR-5873, PNL-79132, U.S. Nuclear Regulatory 

Commission, Washington, D.C., November. [SRDB Ref ID: 22725]

Eberhart, C. F., 1994, “Final Hazard Analysis for Drillback Operations at the Nevada Test Site,” 

memorandum to Distribution, Los Alamos National Laboratory, Los Alamos, New Mexico, 

February 14. [SRDB Ref ID: 40275]

Eckerman, K. F., 2006a, “Fwd: Re: More Questions,” e-mail to R. V. Griffith (Oak Ridge Associated 

Universities), Oak Ridge National Laboratory, Oak Ridge, Tennessee, October 30. [SRDB Ref 

ID: 40280,l p. 4]

Eckerman, K. F., 2006b, “Re: Beta-Photon Dose-Rate Conversion Factors,” e-mail to R. V. Griffith 

(Oak Ridge Associated Universities), Oak Ridge National Laboratory, Oak Ridge, Tennessee, 

August 3. [SRDB Ref ID: 40281, p. 3]

Eckerman, K. F., R. W. Leggett, C. B. Nelson, J. S. Puskin, and A. C. B. Richardson, 1999, Cancer 

Risk Coefficients for Environmental Exposure to Radionuclides, Radionuclide-Specific Lifetime 

Radiogenic Cancer Risk Coefficients for the U.S. Population, Based on Age-Dependent 

Intake, Dosimetry, and Risk Models, Federal Guidance Report No. 13, EPA 402-R-99-001, 

U.S. Environmental Protection Agency, Office of Radiation and Indoor Air, Washington, D.C., 

September. [SRDB Ref ID: 13719]

Ehrlich, M., 1954, Photographic Dosimetry of X- and Gamma Rays, Handbook 57, U.S. Department of 

Commerce, National Bureau of Standards, Washington, D.C., August 20. [SRDB Ref ID: 

11309]

Ehrlich, M., 1962, The Use of Film Badges for Personnel Monitoring, Safety Series No. 8, STI-PUB43, International Atomic Energy Agency, Vienna, Austria, June 28.

Firestone, R. B., and L. P. Ekström, 2004, LBNL Isotopes Project - LUNDS Universitet, WWW Table

of Radioactive Isotopes, Lawrence Berkeley National Laboratory and Lund University, 

Sweden, Berkeley, California. [SRDB Ref ID: 42379]
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 72 of 135

Fix, J. J., E. S. Gilbert, and W. V. Baumgartner, 1994, An Assessment of Bias and Uncertainty in 

Recorded Dose from External Sources of Radiation for Workers at the Hanford Site, PNL10066, Battelle Memorial Institute, Pacific Northwest Laboratory, Richland, Washington, 

August. [SRDB Ref ID: 305]

Fix, J. J., L. Salmon, G. Cowper, and E. Cardis, 1997, “A Retrospective Evaluation of the Dosimetry 

Employed in an International Combined Epidemiologic Study,” Radiation Protection Dosimetry, 

volume 74, numbers 1/2, pp. 39–53. [SRDB Ref ID: 7769]

Fix, J. J., R. H. Wilson, and W. V. Baumgartner, 1997, Retrospective Assessment of Personnel 

Neutron Dosimetry for Workers at the Hanford Site, PNNL-11196, Battelle, Pacific Northwest 

National Laboratory, Richland, Washington, February. [SRDB Ref ID: 5275]

French, R. L. and L. G. Mooney, 1971, Differential Measurements of Fast-Neutron Air-Ground 

Interface Effects, Project 9.2 – Operation HENRE, CEX-65.92, U.S. Atomic Energy 

Commission, Civil Effects Test Operations, Oak Ridge, Tennessee, July. [SRDB Ref ID: 

39867]

Friesen, H. N., 1995, Radiological Effluents Released from Nuclear Rocket and Ramjet Engine Tests 

at the Nevada Test Site, 1959 through 1969, Fact Book, DOE/NV 401, U.S. Department of 

Energy, Nevada Operations Office, Las Vegas, Nevada, June. [SRDB Ref ID: 2594]

Glasstone, S., and P. J. Dolan, editors, 1977, The Effects of Nuclear Weapons, Third Edition, U.S. 

Department of Defense and U.S. Department of Energy, Washington, D.C.

Griffith, R. V., 2004, Resolution of Question Regarding Quantities Used for Reporting the 

Measurement of Personnel Exposures to Photons at the NTS, 1962 to 1987, Woodbridge, 

California, August 4. [SRDB Ref ID: 13989]

Griffith, R. V., 2007, Correction Factors for External Environmental Dose, Oak Ridge Associated 

Universities, Oak Ridge, Tennessee, January 16. [SRDB Ref ID: 29310]

Griffith, R. V., 2008, Estimation of Fractional Photon Contribution by NTS Work Area and Operation, 

Oak Ridge Associated Universities, Oak Ridge, Tennessee. [SRDB Ref ID: 41175]

Griffith, R.V. and Tommasino, L. 1990, “Etch Track Detectors in Radiation Dosimetry,” The Dosimetry 

of Ionizing Radiation, Volume III, Academic Press, pp. 323–426.

Hadlock, D. E., L. W. Brackenbush, R. V. Griffith, D. E. Hankins, M. A. Parkhurst, C. M. Stroud, L. G. 

Faust, and E. J. Vallario, 1988, Personnel Neutron Dose Assessment Upgrade, Volume I: 

Personnel Neutron Dosimetry Assessment, PNL-6620 Vol. 1, Battelle Memorial Institute, 

Pacific Northwest Laboratory, Richland, Washington, July. [SRDB Ref ID: 14858]

Hankins, D. E., 1977, “Energy Dependence Measurements of Remmeters and Albedo Neutron 

Dosimeters at Neutron Energies of Thermal and Between 2 keV and 5.67 MeV,” Proceedings 

of the Fourth International Congress of the International Radiation Protection Association, 

volume 2, pp. 553–556. [SRDB Ref ID: 13994]

Hankins, D. E., S. G. Homann, and J. Westermark, 1987, Personnel Neutron Dosimetry Using 

Electrochemically Etched CR-39 Foils, UCRL-53833, University of California, Lawrence 

Livermore National Laboratory, Livermore, California, November 20. [SRDB Ref ID: 13995]
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 73 of 135

Haywood, F. F., and J. A. Auxier, 1965, Technical Concept—Operation HENRE, CEX-65.02, U.S. 

Atomic Energy Commission, Civil Effects Test Operations, Oak Ridge, Tennessee, March. 

[SRDB Ref ID: 40287]

Haywood, F. F., T. G. Provenzano, and J. A. Auxier, editors, 1965, Operations Plan—Operation 

HENRE, CEX-65.03, U.S. Atomic Energy Commission, Civil Effects Test Operations, Oak 

Ridge, Tennessee, September. [SRDB Ref ID: 41177]

Hicks, H. G., 1981a, Calculation Concentrations of any Radionuclide Deposited on the Ground by 

Release from Underground Nuclear Detonations, Tests of Nuclear Rockets, and Tests of 

Nuclear Ramjet Engines, UCRL-53228, University of California, Lawrence Livermore National 

Laboratory, Livermore, California, November. [SRDB Ref ID: 1828]

Hicks, H. G., 1981b, Results of Calculations of External Gamma Radiation Exposure Rates from 

Fallout and the Related Radionuclide Compositions, Operation Ranger, 1951, UCRL-53152-

Pt. 1, University of California, Lawrence Livermore National Laboratory, Livermore, California, 

July. [SRDB Ref ID: 1854]

Hicks, H. G., 1981c, Results of Calculations of External Gamma Radiation Exposure Rates from 

Fallout and the Related Radionuclide Compositions, Operation Buster-Jangle, 1951, UCRL53152 Pt. 2, University of California, Lawrence Livermore National Laboratory, Livermore, 

California, July. [SRDB Ref ID: 1831]

Hicks, H. G., 1981d, Results of Calculations of External Gamma Radiation Exposure Rates from 

Fallout and the Related Radionuclide Compositions, Operation Tumbler-Snapper, 1952, 

UCRL-53152 Pt. 3, University of California, Lawrence Livermore National Laboratory, 

Livermore, California, July. [SRDB Ref ID: 1832]

Hicks, H. G., 1981e, Results of Calculations of External Gamma Radiation Exposure Rates from 

Fallout and the Related Radionuclide Compositions, Operation Upshot-Knothole, 1953, UCRL53152-Pt.4, University of California, Lawrence Livermore National Laboratory, Livermore, 

California, July. [SRDB Ref ID: 1835]

Hicks, H. G., 1981f, Results of Calculations of External Gamma Radiation Exposure Rates from 

Fallout and the Related Radionuclide Compositions, Operation Teapot, 1955, UCRL-53152 Pt. 

5, University of California, Lawrence Livermore National Laboratory, Livermore, California, 

July. [SRDB Ref ID: 1838]

Hicks, H. G., 1981g, Results of Calculations of External Gamma Radiation Exposure Rates from 

Fallout and the Related Radionuclide Compositions, Operation Plumbob, 1957, UCRL-53152 

Pt. 6, University of California, Lawrence Livermore National Laboratory, Livermore, California,

July. [SRDB Ref ID: 1839]

Hicks, H. G., 1981h, Results of Calculations of External Gamma Radiation Exposure Rates from 

Fallout and the Related Radionuclide Compositions, Operation Hardtack II, UCRL-53152 Pt. 7, 

University of California, Lawrence Livermore National Laboratory, Livermore, California, July. 

[SRDB Ref ID: 1846]

Hicks, H. G., 1981i, Results of Calculations of External Gamma Radiation Exposure Rates from 

Fallout and the Related Radionuclide Compositions, Operations Nougat through Bowline, 

1962–1968, UCRL-53152 Pt. 8, University of California, Lawrence Livermore National 

Laboratory, Livermore, California, July. [SRDB Ref ID: 1851]
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 74 of 135

Hicks, H. G., 1982, “Calculation of the Concentration of any Radionuclide Deposited on the Ground by 

Offsite Fallout from a Nuclear Detonation”, Health Physics, volume 42, number 5, pp. 585–

600. [SRDB Ref ID: 35516]

Hicks, H. G., 1984, Results of Calculations of External Gamma Radiation Exposure Rates from Local 

Fallout and the Related Radionuclide Compositions of Selected U.S. Pacific Events, UCRL53505, University of California, Lawrence Livermore National Laboratory, Livermore, 

California, February. [SRDB Ref ID: 1856]

IAEA (International Atomic Energy Agency), 1979, Radiological Safety Aspects of the Operation of 

Electron Linear Accelerators, Technical Reports Series No. 188, Vienna, Austria, March.

IAEA (International Atomic Energy Agency), 1990, Compendium of Neutron Spectra and Detector 

Responses for Radiation Protection Purposes, Technical Reports Series No. 318, Vienna, 

Austria, July 26.

IAEA (International Atomic Energy Agency), 2001, Compendium of Neutron Spectra and Detector 

Responses for Radiation Protection Purposes, Supplement to Technical Reports Series No. 

318, Technical Reports Series 403, Vienna, Austria.

ICRP (International Commission on Radiological Protection), 1987, Data for Use in Protection Against 

External Radiation, Publication 51, Pergamon Press, Oxford, England.

ICRP (International Commission on Radiological Protection), 1991, 1990 Recommendations of the 

International Commission on Radiological Protection, Publication 60, Pergamon Press, Oxford, 

England. [SRDB Ref ID: 11517]

ICRU (International Commission on Radiation Units and Measurements), 1962, Radiation Quantities 

and Units, Report 10a, Bethesda, Maryland.

ICRU (International Commission on Radiation Units and Measurements), 1988, Determination of 

Dose Equivalents from External Radiation Sources–Part 2, Report 43, Bethesda, Maryland, 

December 15. [SRDB Ref ID: 8774]

ICRU (International Commission on Radiation Units and Measurements), 1992, Measurement of Dose 

Equivalents from External Photon and Electron Radiations, Report 47, Bethesda, Maryland.

ICRU (International Commission on Radiation Units and Measurements), 1993, Quantities and Units 

in Radiation Protection Dosimetry, Report 51, Bethesda, Maryland.

ICRU (International Commission on Radiation Units and Measurements), 1998, Conversion 

Coefficients for Use in Radiological Protection Against External Radiation, Report 57, 

Bethesda, Maryland.

ICRU (International Commission on Radiation Units and Measurements), 2001, Determination of 

Operational Dose Equivalent Quantities for Neutrons, Report 66, Bethesda, Maryland, 

October. [SRDB Ref ID: 100949]

Ing, H., and S. Makra, 1978, Compendium of Neutron Spectra in Criticality Accident Dosimetry, 

Technical Reports Series No. 180, International Atomic Energy Agency, Vienna, Austria.

ISO (International Organization for Standardization), 2001, Reference Neutron Radiations – Part 1: 

Characteristics and Methods of Production, ISO/CD 8529-1:2001, Geneva, Switzerland.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 75 of 135

Johnson, W. H., J. Cramer, B. Thompson, and W. Sitko, 1966, Postshot Drilling Handbook, UCID15034, University of California, Lawrence Radiation Laboratory, Livermore, California, 

October. [SRDB Ref ID: 39413]

Kathren, R. L., 1967, “Dosimetry of 14-MeV Neutrons with Eastman NTA Personnel Monitoring Film,” 

Health Physics, volume 13, number 9, pp. 1039–1040. [SRDB Ref ID: 25004]

Kathren, R. L., 1987, “External Beta-Photon Dosimetry for Radiation Protection,” The Dosimetry of 

Ionizing Radiation, Volume II, K. R. Kase and F. H. Attix, editors, pp. 321–370, Academic 

Press, New York, New York.

Kathren, R. L., 2004, Photon Doses and Energies at Selected Times following a Nuclear Detonation, 

The Kathren Group, Richland, Washington, July 28. [SRDB Ref ID: 13997]

Kathren, R. L., and H. V. Larson, 1969, “Radiological Calibration and Standardization for Health 

Physics: A Program, a Plea, and a Proposal,” Health Physics, volume 16, pp. 778–782. 

[SRDB Ref ID: 13356]

Kathren, R. L., C. T. Prevo, and S. Block, 1965, “Angular Dependence of Eastman Type A (NTA) 

Personnel Monitoring Film,” Health Physics, volume 11, number 10, pp. 1067–1069. [SRDB 

Ref ID: 6180]

Kathren, R. L., P. R. Zurakowski, and M. Covell, 1966, “Effect of Humidity and Dose on Personnel 

Monitoring Film,” American Industrial Hygiene Association Journal, volume 27, pp. 388–395.

Kean, W. B., 1951, Exercise Desert Rock 1, Headquarters, III Corps, Fort MacArthur, California. 

[SRDB Ref ID: 41176]

Kocher, D. C., and K. F. Eckerman, 1981, “Electron Dose-Rate Conversion Factors for External 

Exposure of the Skin,” Health Physics, volume 40, number 4, pp. 467–475.

Kocher, D. C., and K. F. Eckerman, 1987, “Electron Dose-Rate Conversion Factors for External 

Exposure of the Skin from Uniformly Deposited Activity on the Body Surface,” Health Physics, 

volume 53, number 2, pp. 135–141.

Kulp, J. L., and J. L. Dick, 1960, “The Radiation Hazard from Contaminated Aircraft,” Health Physics, 

volume 4, pp. 133–156. [SRDB Ref ID: 29206, p. 4]

Lalos, G., editor, 1989, Film Badge Dosimetry in Atmospheric Nuclear Tests, National Academy of 

Sciences, National Research Council, National Academy Press, Washington, D.C. [SRDB Ref 

ID: 1905]

LASL (Los Alamos Scientific Laboratory), 1953, Operation Buster-Jangle Radiological Safety, WT425(EX), University of California, Los Alamos, New Mexico, July (Extracted Version, October 

1, 1979). [SRDB Ref ID: 2458]

Leavitt, M. O., 2006, HHS Designation of Additional Members of the Special Exposure Cohort under 

the Energy Employees Occupational Illness Compensation Program Act, Designating a Class 

of Employees from Nevada Test Site, Mercury, Nevada, U.S. Department of Health and 

Human Services, Office of the Secretary, Washington, D.C., June 26. [SRDB Ref ID: 39411]
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 76 of 135

Lehman, R. L., 1961, Energy Response and Physical Properties of NTA* Personnel Neutron 

Dosimeter Nuclear Track Film, UCRL-9513, University of California, Lawrence Radiation 

Laboratory, Berkeley, California, January 13. [SRDB Ref ID: 14860]

LRL (Lawrence Radiation Laboratory), 1964, LRL Standard Operating Guide, Postshot Drilling, 

University of California, Livermore, California, August 13. [SRDB Ref ID: 40276]

Mayer, S., T. Otto, and N. Golnik, 2004, “Determination of the Photon-Contribution of a 238Pu-Be 

Source,” Nuclear Instruments and Methods in Physics Research, volume B213, pp. 214-217.

NCRP (National Council on Radiation Protection and Measurements), 1971, Protection Against 

Neutron Radiation, NCRP Report 38, Bethesda, Maryland, January 4. [SRDB Ref ID: 11520]

NCRP (National Council on Radiation Protection and Measurements), 1978, A Handbook of 

Radioactivity Measurements Procedures, 2nd Edition, Report 58, Bethesda, Maryland.

NCRP (National Council on Radiation Protection and Measurements), 1990, Limit for Exposure to 

“Hot Particles” on the Skin, Report 106, Washington, D.C.

Nelms, A. T., and J. W. Cooper, 1959, “U-235 Fission Product Decay Spectra at Various Times After 

Fission,” Health Physics, volume 1, number 4, pp. 427–441.

NIOSH (National Institute for Occupational Safety and Health), 2006, Petition Evaluation Report, 

Petition SEC-00055, Office of Compensation Analysis and Support, Cincinnati, Ohio, April 14.

NIOSH (National Institute for Occupational Safety and Health), 2007a, Petition Evaluation Report 

Petition SEC-00084, Office of Compensation Analysis and Support, Cincinnati, Ohio, 

November.

NIOSH (National Institute for Occupational Safety and Health), 2007b, External Dose Reconstruction 

Implementation Guideline, OCAS-IG-001, Rev. 3, Office of Compensation Analysis and 

Support, Cincinnati, Ohio, November 21.

NIOSH (National Institute for Occupational Safety and Health), 2010a, Radiation Exposures Covered 

for Dose Reconstructions Under Part B of the Energy Employees Occupational Illness 

Compensation Program Act, DCAS-IG-003, Rev. 1, Division of Compensation Analysis and 

Support, Cincinnati, Ohio, October 5.

NIOSH (National Institute for Occupational Safety and Health), 2010b, Petition Evaluation Report, 

Petition SEC-00084, Rev. 1, Office of Compensation Analysis and Support, Cincinnati, Ohio, 

January 25.

Norton, J. F., 2006, “Transuranic Waste Sub-Project Overview and Update,” U.S. Department of 

Energy, Nevada Site Office, Las Vegas, Nevada, November 8. [SRDB Ref ID: 42681]

NRDL (Naval Radiological Defense Laboratory), 1968, Hazards to Personnel Re-entering the Nevada 

Test Site Following Nuclear Reactor Tests, NRDL-TR-68-149, San Francisco, California, 

December 19. [SRDB Ref ID: 8033]

ORAUT (Oak Ridge Associated Universities Team), 2005, Interpretation of Dosimetry Data for 

Assignment of Shallow Dose, ORAUT-OTIB-0017, Rev. 01, Oak Ridge, Tennessee, October 

11.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 77 of 135

ORAUT (Oak Ridge Associated Universities Team), 2006, External Dose Reconstruction, ORAUTPROC-0006, Rev. 01, Oak Ridge, Tennessee, June 5.

ORAUT (Oak Ridge Associated Universities Team), 2007, Pantex Plant – Occupational External 

Dosimetry, ORAUT-TKBS-0013-6, Rev. 01, Oak Ridge, Tennessee, June 22.

ORAUT (Oak Ridge Associated Universities Team), 2008, Nevada Test Site – Site Description, 

ORAUT-TKBS-0008-2, Rev. 01 PC-1, Oak Ridge, Tennessee, May 27.

ORAUT (Oak Ridge Associated Universities Team), 2010a, Nevada Test Site – Occupational 

Environmental Dose, ORAUT-TKBS-0008-4, Rev. 02, Oak Ridge, Tennessee, April 15.

ORAUT (Oak Ridge Associated Universities Team), 2010b, Nevada Test Site – Occupational Internal 

Dose, ORAUT-TKBS-0008-5, Rev. 02, Oak Ridge, Tennessee, April 15.

ORAUT (Oak Ridge Associated Universities Team), 2010c, Rocky Flats Plant – Occupational External 

Dose, ORAUT-TKBS-0011-6, Rev. 02 PC-1, Oak Ridge, Tennessee, October 20.

Provenzano, T. G., E. J. Story, F. F. Haywood, and H. T. Miller, 1966, Feasibility Study: Intense 14-

Mev Neutron Source For Operation HENRE, CEX-65.01, U.S. Atomic Energy Commission, 

Civil Effects Test Operations, Oak Ridge, Tennessee, May. [SRDB Ref ID: 40270]

Pan Am (Pan American World Airways), 1967, Final Proposal for TLD System at NRDS for the Space 

Nuclear Propulsion Office, Nevada Extension, PAA 33-14, September 15. [SRDB Ref ID: 

13998]

Panasonic (Panasonic Industrial Company), 2012, “TLD Badges/Dosimeters,” Secaucus, New Jersey, 

May 8. [SRDB Ref ID: 113030]

REECo (Reynolds Electrical & Engineering Company), 1961, Standard Operating Procedures for the 

Radiological Safety Division, RRS-61-3, Nevada Test Site, Mercury, Nevada, January. [SRDB 

Ref ID: 1922]

REECo (Reynolds Electrical & Engineering Company), 1962, Basic Radiological Safety Training 

Manual, REECo-559:62-0140 (revision), Las Vegas, Nevada, January. [SRDB Ref ID: 1568]

REECo (Reynolds Electrical & Engineering Company), 1968, “Section 29, Personnel Dosimetry and 

Records,” RadSafe Department Standard Operating Procedures, Las Vegas, Nevada, July 24. 

[SRDB Ref ID: 2007]

REECo (Reynolds Electrical & Engineering Company), 1990, Procurement and Acceptance Criteria, 

Standard Operating Procedure AABAC.D.02.40, Rev. 0, March 30. [SRDB Ref ID: 2054, p. 

131]

REECo (Reynolds Electrical & Engineering Company), 1995a, Nevada Test Site Radiation Protection 

Program, DOE/NV/11432-203, Rev. 3, Las Vegas, Nevada, December. [SRDB Ref ID: 

14002]

REECo (Reynolds Electrical & Engineering Company), 1995b, Monthly Calibration of Gamma 

Radiation Fields at Inner and Outer Dosimeter Support Rings, C34-6209-086.A, Las Vegas, 

Nevada, November 30. [SRDB Ref ID: 14006]

REECo (Reynolds Electrical & Engineering Company), 1995c, Thermoluminescent Dosimetry Quality 

Assessment, C34-6209-039.A, Las Vegas, Nevada, November 30. [SRDB Ref ID: 14007]
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 78 of 135

Roberts, R.B., 1937, “Investigation of the Deuteron-Deuteron Reaction”, Physical Review, volume 51, 

number 10, pp. 810–817.

Rollins, G., 2007, “Interview with Burce W. Church Regarding NTS Radiation Protection Practices -

1966 to 1995,” documented communication with B. Church (U.S. Department of Energy, 

retired), Oak Ridge, Tennessee, November 30. [SRDB Ref ID: 36653]

Sanders, F. W., J. A. Auxier, and J. S. Cheka, 1959, “A Simple Method of Minimizing the Energy 

Dependence of Pocket Ionization Chambers,” Health Physics, volume 2, pp. 308–309.

Sanders, F. W., F. F. Haywood, M. I. Lundin, L. W. Gilley, J. S. Cheka, and D. R. Ward, 1962, 

Operation Plan and Hazards Report—Operation BREN, CEX-62.02, U.S. Atomic Energy 

Commission, Civil Effects Test Operations, Oak Ridge, Tennessee, April. [SRDB Ref ID: 

37781]

Sanna, R. S., J. E. McLaughlin, A. Lazanoff, and K. O’Brien, 1969, Neutron Spectrometry–Operation 

HENRE (Program 6), CEX-65.60, U.S. Atomic Energy Commission, New York Operations 

Office, New York, New York, April. [SRDB Ref ID: 39864]

Schwartz, R. B., and C. M. Eisenhauer, 1980, The Design and Construction of a D2O-Moderated 252Cf 

Source for Calibrating Neutron Personnel Dosimeters Used at Nuclear Power Reactors, 

NUREG/CR-1204, U.S. Nuclear Regulatory Commission, Washington, D.C. [SRDB Ref ID: 

40273]

Sebelius, K., 2010, HHS Designation of Additional Members of the Special Exposure Cohort under 

the Energy Employees Occupational Illness Compensation Program Act, Designating a Class 

of Employees from Nevada Test Site, Mercury, Nevada, U.S. Department of Health and 

Human Services, Office of the Secretary, Washington, D.C., April 5. 

Shipler, D. B., 2004, “Verification of Telephone Conversation with Dillard Shipler and Jerome Martin 

8/17/04,” letter to R. Carr, Battelle, Pacific Northwest National Laboratory, Richland, 

Washington, August 18. [SRDB Ref ID: 13481]

Shipman, T. L., T. N. White, S. Shlaer, J. T. Brennan, E. Pinson, L. A. Eddy, G. W. Taylor, W. R. 

Kennedy, L. C. Anderson, J. D. Shumway, J. Z. Morse, J. Campbell, and W. E. de Alva, 1951, 

Log of JM-4 Operations, Rad-Safe Group, Operation Ranger, 23 January to 11 February 1951, 

Nellis Air Force Base, Las Vegas, Nevada, July 12. [SRDB Ref ID: 40278]

Shleien, B., L. A. Slaback, Jr., and B. K. Birky, editors, 1998, Handbook of Health Physics and 

Radiological Health, Third Edition, Williams & Wilkins, Baltimore, Maryland, January.

Sims, C. S., and G. E. Ragan, 1987, Health Physics Research, Reactor Reference Dosimetry, ORNL6240, Martin Marietta Energy Systems, Oak Ridge National Laboratory, Tennessee, June. 

[SRDB Ref ID: 40368]

Sondhaus, C. A., and V. P. Bond, 1955, Operation Castle, Addendum Report for Project 4.1, Physical 

Factors and Dosimetry in the Marshall Island Radiation Exposures, WT-939, U.S. Naval 

Radiological Defense Laboratory, San Francisco, California, December. [SRDB Ref ID: 

14876]

Storm, E., 1951, untitled memorandum to W. G. Kratz, U.S. Army Corps of Engineers, Engineer 

Research & Development Laboratories, Fort Belvoir, Virginia, July 12. [SRDB Ref ID: 40279]
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 79 of 135

Thierry-Chef, I., F. Pernicka, M. Marshall, E. Cardis, and P. Andreo, 2002, “Study of a Selection of 10 

Historical Types of Dosemeter: Variation of the Response to Hp(10) with Photon Energy and 

Geometry of Exposure,” Radiation Protection Dosimetry, volume 102, number 2, pp. 101–113. 

[SRDB Ref ID: 11296]

Thorngate, J. H., and E. T. Loy, 1965, Post Pulse Gamma-Radiation Spectrum–Operation BREN, 

CEX-62.13, U.S. Civil Effects Test Operations, AEC, Oak Ridge, Tennessee, December. 

[SRDB Ref ID: 39846]

Thorngate, J. H., J. A. Auxier, F. F. Haywood, and S. Helf, 1967, Energy and Angular Distribution of 

Neutrons and Gamma Rays–Operation BREN, CEX-62.12, U.S. Atomic Energy Commission, 

Civil Effects Test Operations, Oak Ridge, Tennessee, January. [SRDB Ref ID: 39836]

Thorngate, J. H., D. R. Johnson, P. T. Perdue, 1969, Energy and Angular Distribution of Neutrons and 

Gamma Rays: – Operation HENRE, CEX-65.11, U.S. Atomic Energy Commission, Civil 

Effects Test Operations, Oak Ridge, Tennessee, April. [SRDB Ref ID: 39863]

Unger, L. M., and D. K. Trubey, 1982, Specific Gamma-Ray Dose Constants for Nuclides Important to 

Dosimetry and Radiological Assessment, ORNL/RSIC-45/R1, Union Carbide Corporation, Oak 

Ridge National Laboratory, Oak Ridge, Tennessee, May. [SRDB Ref ID: 40289]

Wimett, T. F., 1965, Fast Burst Reactors in the U.S.A., LA-DC-6786, University of California, Los 

Alamos Scientific Laboratory, Los Alamos, New Mexico, January. [SRDB Ref ID: 40339]
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 80 of 135

GLOSSARY

absorbed dose, D

Amount of energy (ergs or joules) deposited in a substance by ionizing radiation per unit mass 

(grams or kilograms) of the substance and measured in units of rads or grays. See dose.

accreditation

For external dosimetry, the assessment of whether or not a personnel dosimetry system meets 

specific criteria. The assessment includes dosimeter performance and the associated quality 

assurance and calibration programs.

accuracy

The characteristics of an analysis or determination that ensures that both the bias and 

precision of the resultant quantity will remain within the specified limits.

albedo dosimeter

Thermoluminescent dosimeter that measures the thermal, intermediate, and fast neutrons 

scattered and moderated by the body or a phantom from an incident fast neutron flux.

algorithm

Set of rules or steps for solving a problem, especially for calculating a value.

alpha particle (α)

See alpha radiation.

alpha radiation

Positively charged particle emitted from the nuclei of some radioactive elements. An alpha 

particle consists of two neutrons and two protons (a helium nucleus) and has an electrostatic 

charge of +2.

backscatter

Reflection or refraction of radiation at angles over 90 degrees from its original direction.

beta particle (β)

See beta radiation.

beta radiation

Charged particle emitted from some radioactive elements with a mass equal to 1/1,837 that of 

a proton. A negatively charged beta particle is identical to an electron. A positively charged 

beta particle is a positron.

bremsstrahlung

Electromagnetic radiation released as a result of inelastic scattering of a moving charged 

particle within the nucleus of an atom. X-rays produced in a typical medical X-ray tube 

frequently originate from inelastic scattering of accelerated electrons in the anode material.

buildup

Increase in flux or dose due to scattering in the medium.

calibration

Adjustment or determination of the response or reading of an instrument relative to a standard 

or a series of conventionally true values.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 81 of 135

calibration blank

Dosimeter not exposed to radiation. This dosimeter establishes the dosimetry system base 

line or zero dose value.

collective dose equivalent

Sum of the dose equivalents of all individuals in an exposed population in units of person-rem 

or person-sievert. See dose.

confidence level

The interval about an estimate of a stated quantity within which the value of the quantity is 

expected to be with a specified probability. See uncertainty.

control dosimeter

Dosimeter used to establish the dosimetry system response to radiation dose by exposing it to 

a known amount of radiation. See calibration.

curie (Ci)

Traditional unit of radioactivity equal to 37 billion (3.7 × 1010) becquerels, which is 

approximately equal to the activity of 1 gram of pure 226Ra.

deep absorbed dose (Dd)

Absorbed dose in units of rem or sievert at a depth of 1 centimeter (1,000 milligrams per 

square centimeter). See dose.

deep dose equivalent (Hd)

Dose equivalent in units of rem or sievert for a 1-centimeter depth in tissue (1,000 milligrams 

per square centimeter). See dose.

densitometer

Instrument that uses a photoelectric cell to measure the transition of light through developed 

photographic film to determine the optical density.

density reading

See optical density.

DOE Laboratory Accreditation Program (DOELAP)

Program for accreditation by DOE of DOE site personnel dosimetry and radiobioassay 

programs based on performance testing and the evaluation of associated quality assurance, 

records, and calibration programs.

dose

In general, the specific amount of energy from ionizing radiation that is absorbed per unit of 

mass. Effective and equivalent doses are in units of rem or sievert; other types of dose are in 

units of roentgens, rads, reps, or grays.

dose equivalent (H)

In units of rem or sievert, product of absorbed dose in tissue multiplied by a weighting factor 

and sometimes by other modifying factors to account for the potential for a biological effect 

from the absorbed dose. See dose.

dosimeter

Device that measures the quantity of received radiation, usually a holder with radiationabsorbing filters and radiation-sensitive inserts packaged to provide a record of absorbed dose 
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 82 of 135

received by an individual. See albedo dosimeter, film dosimeter, neutron film dosimeter, 

pocket ionization chamber, thermoluminescent dosimeter, and track-etch dosimeter.

dosimetry system

System for assessment of received radiation dose. This includes the fabrication, assignment, 

and processing of external dosimeters, and/or the collection and analysis of bioassay samples, 

and the interpretation and documentation of the results.

DuPont 552 film packet

Film packet containing DuPont 502 sensitive film and DuPont 510 insensitive film.

DuPont 558 film packet

Film packet containing DuPont 508 film with sensitive and insensitive emulsions on either side.

element

One of the known chemical substances in which the atoms have the same number of protons. 

Elements cannot be broken down further without changing their chemical properties.

error

Difference between the correct, true, or conventionally accepted value and the measured or 

estimated value. Sometimes used to mean estimated uncertainty. See accuracy and 

uncertainty.

exchange period (frequency)

Period (weekly, biweekly, monthly, quarterly, etc.) for routine exchange of dosimeters.

exposure

(1) In general, the act of being exposed to ionizing radiation. (2) Measure of the ionization 

produced by X- and gamma-ray photons in air in units of roentgens.

fast neutron

Neutron with energy equal to or greater than 10 kiloelectron-volts. This type of neutron causes 

fission in some isotopes (e.g., 238U, 239Pu). See intermediate neutron and slow neutron.

favorable to claimants

In relation to dose reconstruction for probability of causation analysis, having the property of 

ensuring that there is no underestimation of potential dose, which often means the assumption 

of a value that indicates a higher dose than is likely to have actually occurred in the absence of 

more accurate information. See probability of causation.

film

In the context of external dosimetry, radiation-sensitive photographic film in a light-tight 

wrapping. See film dosimeter.

film density

See optical density.

film dosimeter

Package of film for measurement of ionizing radiation exposure for personnel monitoring 

purposes. A film dosimeter can contain two or three films of different sensitivities, and it can 

contain one or more filters that shield parts of the film from certain types of radiation. When 

developed, the film has an image caused by radiation measurable with an optical 

densitometer. Also called film badge.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 83 of 135

filter

Material used in a dosimeter to adjust radiation response to provide an improved tissue 

equivalent or dose response.

first collision dose

Measure for neutron radiation that relates dose to neutron flux through a thin layer of tissue. A 

graph referred to as the first collision curve derives from the assumption that the probability of 

two or more interactions per neutron is negligible. Because the charged secondary radiation 

from fast neutrons is short range, the first collision dose in irradiated material is nearly the 

same as the absorbed dose.

fluence

Either particle fluence (the number of photons entering a sphere of unit cross-sectional area) 

or energy fluence (the sum of the energies of the photons passing through a unit area).

gamma radiation

Electromagnetic radiation (photons) of short wavelength and high energy (10 kiloelectron-volts 

to 9 megaelectron-volts) that originates in atomic nuclei and accompanies many nuclear 

reactions (e.g., fission, radioactive decay, and neutron capture). Gamma photons are identical 

to X-ray photons of high energy; the difference is that X-rays do not originate in the nucleus.

gamma ray, particle, or photon (γ)

See gamma radiation.

gamma-ray interactions

Interaction of gamma rays with matter occurs through three primary processes as follows:

• Photoelectric absorption in which a gamma-ray (or X-ray) photon with energy somewhat 

greater than that of the binding energy of an electron transfers all its energy to the electron, 

which consequently separates from the atom.

• Compton scattering in which X- or gamma-ray radiation is attenuated when an incident 

photon interacts with an orbital electron of an atom to produce a recoil electron and a 

scattered photon of energy less that the incident photon.

• Pair production in which X- or gamma-ray radiation is absorbed when an incident photon is 

annihilated in the vicinity of the nucleus of the absorbing atom with subsequent production 

of an electron and positron pair. This reaction only occurs for incident photon energies that 

exceed 1.02 megaelectron-volts.

gray

International System unit of absorbed radiation dose, which is the amount of energy from any 

type of ionizing radiation deposited in any medium; 1 Gy equals 1 joule per kilogram or 

100 rads.

intermediate neutron

Neutron with energy between 0.5 electron-volts and 10 kiloelectron-volts. See fast neutron

and slow neutron.

ionizing radiation

Radiation of high enough energy to remove an electron from a struck atom and leave behind a 

positively charged ion. High enough doses of ionizing radiation can cause cellular damage. 

Ionizing particles include alpha particles, beta particles, gamma rays, X-rays, neutrons, 
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 84 of 135

high-speed electrons, high-speed protons, photoelectrons, Compton electrons, 

positron/negatron pairs from photon radiation, and scattered nuclei from fast neutrons. See 

alpha radiation, beta radiation, gamma radiation, neutron radiation, photon radiation, and X-ray 

radiation.

isotope

One of two or more atoms of a particular element that have the same number of protons 

(atomic number) but different numbers of neutrons in their nuclei (e.g., 234U, 235U, and 238U). 

Isotopes have very nearly the same chemical properties. See element.

kiloelectron-volt (keV)

Unit of particle energy equal to 1,000 (1 × 103

) electron-volts.

luminescence

Emission of light from a material as a result of some excitation. See thermoluminescence.

Manhattan Engineer District (MED)

Subdivision of the U.S. Army Corps of Engineers that administered the World War II 

Manhattan Project to develop the first nuclear bomb. The word Manhattan was chosen to 

divert attention from the Project’s real purpose. The U.S. Atomic Energy Commission 

assumed control of MED facilities and activities in 1946.

megaelectron-volt (MeV)

Unit of particle energy equal to 1 million (1 × 106

) electron-volts.

minimum detectable activity or amount

Smallest amount (activity or mass) of an analyte in a sample that can be detected with a 

probability β of nondetection (Type II error) while accepting a probability α of erroneously 

deciding that a positive (nonzero) quantity of analyte is present in an appropriate blank sample 

(Type I error).

minimum detection level (MDL)

See minimum detectable activity.

monitoring

Periodic or continuous determination of the presence or amount of ionizing radiation or 

radioactive contamination in air, surface water, groundwater, soil, sediment, equipment 

surfaces, or personnel (for example, bioassay or alpha scans). In relation to personnel, 

monitoring includes internal and external dosimetry including interpretation of the 

measurements.

multiple-collision neutron dose

Dose in relation to the neutron flux through tissue based on the assumption that two or more 

interactions per neutron occurs and results in greater energy deposition.

neutron

Basic nucleic particle that is electrically neutral with mass slightly greater than that of a proton. 

There are neutrons in the nuclei of every atom heavier than normal hydrogen.

neutron film dosimeter

Film dosimeter with a nuclear track emulsion, type A, film packet.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 85 of 135

neutron radiation

Radiation that consists of free neutrons unattached to other subatomic particles emitted from a 

decaying radionuclide. Neutron radiation can cause further fission in fissionable material such 

as the chain reactions in nuclear reactors, and nonradioactive nuclides can become 

radioactive by absorbing free neutrons. See neutron.

nuclear emulsion

Thick photographic coating in which the tracks of various fundamental particles show as black 

traces after development. The number of tracks in a given area is a measure of the dose from 

that radiation. See nuclear track emulsion, type A.

nuclear track emulsion, type A (NTA)

A film that is sensitive to fast neutrons. The developed image has tracks caused by neutrons 

that can be seen by using oil immersion and 1,000-power microscope.

nuclide

Stable or unstable isotope of any element. Nuclide relates to the atomic mass, which is the 

sum of the number of protons and neutrons in the nucleus of an atom. A radionuclide is an 

unstable nuclide.

nonpenetrating dose (NP or NPen)

Dose from beta and lower energy photon (X-ray and gamma) radiation that does not penetrate 

the skin. It is often determined from the open-window dose minus the shielded window dose. 

See dose.

open window (OW)

Area of a film dosimeter that has little to no radiation shielding (e.g., only a holder and visible 

light protection). See film dosimeter.

optical density

Measure of the degree of opacity of photographic or radiographic film defined as OD = log10

(I0/I), the base-10 logarithm of the ratio of the reference light intensity I0 (without film) to the 

transmitted light intensity (through the film) Also called film density and density reading.

pencil dosimeters

See pocket ionization chamber.

penetrating dose (P or Pen)

Dose from moderate to higher energy photons and neutrons that penetrates the outer layers of 

the skin. See dose.

personal dose equivalent, Hp(d)

Dose equivalent in units of rem or sievert in soft tissue below a specified point on the body at 

an appropriate depth d. The depths selected for personal dosimetry are 0.07 millimeters 

(7 milligrams per square centimeter) and 10 millimeters (1,000 milligrams per square 

centimeter), respectively, for the skin (shallow) and whole-body (deep) doses. These are 

noted as Hp(0.07) and Hp(10), respectively. The International Commission on Radiological 

Measurement and Units recommended Hp(d) in 1993 as dose quantity for radiological 

protection.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 86 of 135

photon

Quantum of electromagnetic energy generally regarded as a discrete particle having zero rest 

mass, no electric charge, and an indefinitely long lifetime. The entire range of electromagnetic 

radiation that extends in frequency from 1023 cycles per second (hertz) to 0 hertz.

photon radiation

Electromagnetic radiation that consists of quanta of energy (photons) from radiofrequency 

waves to gamma rays.

pocket ionization chamber (PIC)

Cylindrical monitoring device commonly clipped to the outer clothing of an individual to 

measure ionizing radiation. A PIC may be self-reading or require the use of a outside device 

to be able to read the dosimeter. Also called pencil, pocket pencil, pencil dosimeter, and 

pocket dosimeter.

probability of causation (POC)

For purposes of dose reconstruction for the Energy Employees Occupational Illness 

Compensation Program Act, the percent likelihood, at the 99th percentile, that a worker 

incurred a particular cancer from occupational exposure to radiation.

quality factor (Q)

Principal modifying factor (which depends on the collision stopping power for charged 

particles) that is employed to derive dose equivalent from absorbed dose. The quality factor 

multiplied by the absorbed dose yields the dose equivalent. See dose, relative biological 

effectiveness, and weighting factor.

rad

Traditional unit for expressing absorbed radiation dose, which is the amount of energy from 

any type of ionizing radiation deposited in any medium. A dose of 1 rad is equivalent to the 

absorption of 100 ergs per gram (0.01 joules per kilogram) of absorbing tissue. The rad has 

been replaced by the gray in the International System of Units (100 rads = 1 gray). The word 

derives from radiation absorbed dose.

radiation

Subatomic particles and electromagnetic rays (photons) with kinetic energy that interact with 

matter through various mechanisms that involve energy transfer. See ionizing radiation.

radioactive

Of, caused by, or exhibiting radioactivity.

radioactivity

Property possessed by some elements (e.g., uranium) or isotopes (e.g., 14C) of spontaneously 

emitting energetic particles (electrons or alpha particles) by the disintegration of their atomic 

nuclei. See radionuclide.

radionuclide

Radioactive nuclide. See radioactive and nuclide.

random errors

When a given measurement is repeated and the values do not agree exactly. The causes of 

the disagreement between the values must also be the causes of their differences from the 

true value.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 87 of 135

relative biological effectiveness (RBE)

Ratio of the absorbed dose of a reference radiation to the absorbed dose of a test radiation 

that produces the same biological effects, other conditions being equal. A factor applied to 

account for differences between the amount of cancer effect produced by different forms of 

radiation.

rem

Traditional unit of radiation dose equivalent that indicates the biological damage caused by 

radiation equivalent to that caused by 1 rad of high-penetration X-rays multiplied by a quality 

factor. The sievert is the International System unit; 1 rem equals 0.01 sievert. The word 

derives from roentgen equivalent in man; rem is also the plural.

rep

Historical quantity of radiation (usually other than X-ray or gamma radiation) originally defined 

as 93 ergs absorbed per gram in the body and redefined in the 1940s or early 1950s as the 

amount that would liberate the same amount of energy (93 ergs per gram) as 1 roentgen of Xor gamma rays. Replaced by the gray in the International System of Units; 1 rep is 

approximately equal to 9.3 milligray. The word derives from roentgen equivalent physical.

roentgen

Unit of photon (gamma or X-ray) exposure for which the resultant ionization liberates a positive 

or negative charge equal to 2.58 × 10-4 coulombs per kilogram (or 1 electrostatic unit of 

electricity per cubic centimeter) of dry air at 0°C and standard atmospheric pressure. An 

exposure of 1 R is approximately equivalent to an absorbed dose of 1 rad in soft tissue for 

higher energy photons (generally greater than 100 kiloelectron-volts).

scattering

Change in direction of radiation by refraction or reflection, often accompanied by a decrease in 

radiation due to absorption by the refracting or reflecting material.

shallow absorbed dose (Ds)

Absorbed dose at a depth of 0.07 millimeter (7 milligrams per square centimeter) in a material 

of specified geometry and composition.

shallow dose equivalent (Hs)

Dose equivalent in units of rem or sievert at a depth of 0.07 millimeter (7 milligrams per square 

centimeter) in tissue equal to the sum of the penetrating and nonpenetrating doses.

shielding

Material or obstruction that absorbs ionizing radiation and tends to protect personnel or 

materials from its effects.

sievert (Sv)

International System unit for dose equivalent, which indicates the biological damage caused 

by radiation. The unit is the radiation value in gray (equal to 1 joule per kilogram) multiplied by 

a weighting factor for the type of radiation and a weighting factor for the tissue; 1 Sv equals 

100 rem.

skin dose

See shallow dose equivalent.

slow neutron

Neutrons with energy less than 1 electron-volts. See fast neutron and intermediate neutron.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 88 of 135

thermal neutron

Neutron in thermal equilibrium with its surroundings having an average energy of 

0.025 electron-volt.

tissue equivalent

Substance with response to radiation equivalent to tissue. A tissue-equivalent response is an 

important consideration in the design and fabrication of radiation measuring instruments and 

dosimeters.

thermoluminescence

Property that causes a material to emit light as a result of heat.

thermoluminescent dosimeter (TLD)

Device for measuring radiation dose that consists of a holder containing solid chips of material 

that, when heated, release the stored energy as light. The measurement of this light provides 

a measurement of absorbed dose.

thermoluminescent dosimeter chip

Small block or crystal of lithium fluoride in a thermoluminescent dosimeter. A TLD-600 

dosimeter contains a chip made from more than 95% 6

Li for neutron radiation detection, and a 

TLD-700 dosimeter contains a chip made from more than 99.9% 7

Li for photon and beta 

radiation detection. Also called crystals.

track-etch dosimeter (TED)

Device for evaluation of fast neutron dose through examination of traces left by the neutrons 

on the Columbia Resin Number 39 emulsion.

uncertainty

Standard deviation of the mean of a set of measurements. The standard error reduces to the 

standard deviation of the measurement when there is only one determination. See accuracy, 

confidence level, and error. Also called standard error.

U.S. Atomic Energy Commission (AEC)

Federal agency created in 1946 to assume the responsibilities of the Manhattan Engineer 

District (nuclear weapons) and to manage the development, use, and control of nuclear energy 

for military and civilian applications. The U.S. Energy Research and Development 

Administration and the U.S. Nuclear Regulatory Commission assumed separate duties from 

the AEC in 1974. The U.S. Department of Energy succeeded the U.S. Energy Research and 

Development Administration in 1979.

weighting factor (wT)

The ratio of the stochastic risk arising from tissue T to the total risk when the whole body is 

irradiated uniformly.

whole-body (WB) dose

Dose to the entire body excluding the contents of the gastrointestinal tract, urinary bladder, 

and gall bladder and commonly defined as the absorbed dose at a tissue depth of 

10 millimeters (1,000 milligrams per square centimeter). Also called penetrating dose. 

See dose.

X-ray

Electromagnetic radiation (photons) produced by bombardment of atoms by accelerated 

particles. X-rays are produced by various mechanisms including bremsstrahlung and electron 
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 89 of 135

shell transitions within atoms (characteristic X-rays). Once formed, there is no difference 

between X-rays and gamma rays, but gamma photons originate inside the nucleus of an atom.
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 90 of 135

TABLE OF CONTENTS

SECTION TITLE PAGE

A.1 Beta/Gamma Dosimetry......................................................................................................... 91

A.1.1 Photographic Film Dosimeters....................................................................................91

A.1.2 Thermoluminescent Dosimeters .................................................................................93

A.1.2.1 NRDS Thermoluminescent Dosimeters........................................................ 93

A.1.2.2 REECo Thermoluminescent Dosimeters ...................................................... 95

A.1.2.3 Energy Response......................................................................................... 96

A.1.2.4 Dose Measurement...................................................................................... 97

A.1.2.5 Calibration.................................................................................................... 98

A.2 Neutron Dosimetry ................................................................................................................. 98

A.2.1 NTA Film ....................................................................................................................98

A.2.2 TLD Albedo Detectors ..............................................................................................100

A.2.3 Track-etch Detectors ................................................................................................101

A.2.3.1 Dose Algorithms......................................................................................... 102

A.3 Quality Assurance................................................................................................................ 103

A.4 Beta Particle Field Characteristics........................................................................................ 105

A.5 Neutron Field Characteristics ............................................................................................... 106

LIST OF TABLES

TABLE TITLE PAGE

A-1 Panasonic UD-802 dosimeter configuration ........................................................................... 96

A-2 Photon exposure to dose correction factors for NIST reference radiations ............................. 98

A-3 Bias values and uncertainties for NTS DOELAP accreditation categories ............................ 105

A-4 Fraction of Hp(10,0) for neutrons with energies less than En................................................. 110

LIST OF FIGURES

FIGURES TITLE PAGE

A-1 Typical photon energy response curves for Li2B4O7 and CaSO4 TLD elements...................... 96

A-2 Panasonic UD-802 energy responses.................................................................................... 97

A-3 NTA film neutron energy response......................................................................................... 99

A-4 Neutron energy response of NTS albedo dosimeters ........................................................... 101

A-5 Neutron energy response of NTS TED................................................................................. 102

A-6 Electrochemically etched CR-39 dose response .................................................................. 103

A-7 Lethargy neutron spectra for 252Cf calibration source with and without room scatter............. 107

A-8 Lethargy neutron spectra for 252Cf with various thickness moderators.................................. 108

ATTACHMENT A

DOSIMETRY TECHNOLOGY

Page 1 of 21
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 91 of 135

A-9 Lethargy neutron spectra for an 241AmBe calibration source with and without room 

scatter.................................................................................................................................. 108

A-10 Fraction of Hp(10,0) due to neutrons with energies less than En ........................................... 109

A.1 BETA/GAMMA DOSIMETRY

A.1.1 Photographic Film Dosimeters

Photographic emulsions of various types and in various holders (i.e., film badges) were used at NTS 

for personnel monitoring from the start of operations in January 1951 until 1987 (Author unknown 

undated; Allen and Schoengold 1995). The basic principles, theory, and practice of photographic film 

dosimetry for beta and photon radiation are well known and have been described in a number of 

standard texts and references (Becker 1966; Dudley 1966; Ehrlich 1954, 1962; Kathren 1987; Lalos

1989). The following paragraphs discuss factors related to film dosimetry that provide a general 

background to the interpretation and reconstruction of dosimetry results at NTS.

The term film badge, or film badge dosimeter, as used in this document, refers to the entire dosimeter 

issued to personnel, which typically consisted of a dental-size film packet housed in a holder of 

varying sophistication designed to improve the response characteristics and measurement 

capabilities. The film consisted of a plastic base covered on one or both sides with a layer of suitable 

photographic emulsion. One or two pieces of film were wrapped in a light-tight paper package to 

comprise the packet, which in turn was placed in a holder containing metallic filter(s) to compensate 

for the photon energy dependence of the film. The sensitivity of the film to ionizing radiation was 

largely a function of the size of the AgBr grains in the emulsion. Because the typical photographic 

emulsion for dosimetry has an effective range of about 3 orders of magnitude, the packet typically 

contained two pieces of film with different sensitivities. The effective ranges of these films 

overlapped, which permitted an effective response range of about 5 orders of magnitude for 

penetrating photon radiations. 

In principle, film dosimetry is simple; it consists of the determination of optical density or degree of 

blackening produced by exposure to radiation. The degree of blackening produced by the radiation 

incident on the film is determined in terms of the net optical density, the logarithm of which is typically 

plotted against the logarithm of exposure or dose to produce a calibration or response curve, or an 

algorithm with which the exposure registered by the film can be determined. The response of a film 

emulsion to photon radiation is not linear with dose or exposure. Near the lower (less than 50 mR) 

and upper (about 1.5 to 5 R) limits of the range of the film, small increments of density represent 

relatively large changes in dose. Therefore, the uncertainty at the low and high ends of the dose 

response curve (i.e., net optical density versus exposure), where the slope is quite shallow, is 

relatively large. However, the total uncertainty for high-energy exposures (more than about 150 keV 

to 3 MeV) for these dose ranges (1.5 to 5 R) should not exceed a factor of 2. For high-energy photon 

exposures in the region of 50 to 1,500 mR (the steep portion of the dose response curve), the 

uncertainty is much less and should be within 20% to 30%. Dose reconstructors should consider 

these uncertainty values to be broad estimates, not exact values.

Although a number of different film types were used for dosimetry at NTS, they generally had similar 

characteristics and responses to beta and photon radiations. Uncertainties in dosimetry with these 

films are largely, if not exclusively, the result of external factors rather than differences in the films.

ATTACHMENT A

DOSIMETRY TECHNOLOGY

Page 2 of 21
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 92 of 135

The response of film to photon radiation is affected by a number of interrelated variables. From the 

standpoint of dose reconstruction at NTS, the most important of these are energy dependence, 

angular dependence, and effects of temperature and humidity. Because of the high relative atomic 

number (Z) of the AgBr in the film emulsion in relation to soft tissue, the response or energy absorbed 

per unit exposure of the film is a strong function of the energy of the exposing photons, rising steeply 

at energies below about 200 keV to a peak at about 30 keV, and then falling off steeply. At about 

30 keV, the energy of maximum response, the degree of blackening per unit exposure is about 30-fold 

greater than that for photons with energies of few hundred thousand to a few million electron-volts. 

Because the relationship between exposure and soft tissue dose is approximately constant over a 

wide energy range, it was essential to compensate or correct for the energy dependence of the 

photographic response. This was accomplished by placing metallic filters over portions of the film to 

alter or flatten the energy dependence characteristics such that the film responded more like soft 

tissue over a wide energy range.

Although energy dependence can be a large source of error in film badge dosimetry, reasonably good 

results can be obtained by a four-element badge: one with an OW or no filter over a portion of the film 

packet, and high-Z, medium-Z, and low-Z filters over other portions of the packet. By observing the 

relationships of the degree of darkening under each filter area, and comparing these with exposures 

to known energies and doses of photon radiations, a reasonably accurate assessment of photon dose 

can be made over a wide range of energies. This is particularly true at NTS, where exposures were 

largely to photons with energies greater than a few hundred thousand electron-volts at angles close to 

normal to the plane of the film. In general, errors in dose interpretation that result from energy 

dependence will result in overestimates of dose and are, therefore, conservative and favorable to 

claimants. However, given the single-element badges in use until 1966, some dose from lower 

energy photons, specifically those with energies below about 80 keV, could have been missed 

because of attenuation in the filter; therefore, the dose interpretation could be low. Given the photon 

energy spectra at NTS, any such loss of dose is likely to have been relatively low.

Photon calibration of film dosimeters was typically performed in terms of exposure in free air, with the 

plane of the film oriented normally to the direction of the incident photon beam. Initial calibrations 

were made with a high-energy photon source (e.g., 60Co or 137Cs) with an energy response by film 

similar to the energy spectra to which personnel were exposed in the field. After introduction of the 

advanced multielement film badge in January 1966, calibration was conducted with X-ray sources as 

well as high-energy photons. This enabled interpretation of exposure to the film with a higher degree 

of accuracy over a broader spectrum of photon energies.

The response of film, whether in a bare film packet or in a holder, is a function of the angle of 

incidence of the exposing radiation on the plane of the film. Because the film is in a small flat sheet, 

an edge-on exposure (i.e., an exposure with the incident radiation parallel to the plane of the film) 

produces a different effect or optical density pattern than an exposure normal (at right angles) to the 

plane of the film. The effect is strongly dependent on both the energy of the exposing photons and 

the type of film holder or film badge, but is generally minimal at photon energies above a few hundred 

thousand electron-volts and for angles of incidence ranging from about 30 to 150 degrees in relation 

to the plane of the film. However, for angles of incidence approaching parallelism (i.e., 0 degrees) 

with the plane of the film, the effect can be pronounced, and can lead to significant underestimates in 

dose. The problem, however, should be more or less minimal for exposures at NTS because these 

were (1) typically to high-energy photons and at angles close to normal with the plane of the film and 

(2) probably largely accounted for in the interpretation of the badge.

ATTACHMENT A

DOSIMETRY TECHNOLOGY

Page 3 of 21
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 93 of 135

Environmental conditions can significantly affect film badge results and can result in large 

uncertainties and errors in dose estimation. Temperatures greater than 130o

F (50oC) can induce 

fogging, but several days at this temperature are necessary before fogging or increased density 

occurs. At higher temperatures, which could have occurred in closed vehicles or buildings at NTS, 

fogging can result after shorter exposure times (Kathren, Zurakowski, and Covell 1966). If not 

corrected, high-temperature fogging always produces an overestimate and, therefore, a dose 

interpretation favorable to claimants. 

High relative humidity results in a fading of the latent image before development and in a decrease in 

the measured density of the film for a given dose. Latent image fading results in a low estimate of 

dose. However, studies have shown that latent image fading is not a problem until films have been 

worn for intervals exceeding 4 to 6 weeks, and can be largely overcome by encasing the film packet in 

a polyethylene pouch, as was done for hot and humid conditions encountered during nuclear tests in 

the Pacific (Kathren, Zurakowski, and Covell 1966). Given the low-humidity conditions at NTS, latent 

image fading is unlikely to be of significance to film dosimetry.

Films were exchanged at varying intervals and, once collected, were developed under controlled 

conditions. Optical densities were determined with a densitometer, and doses were determined from 

calibration curves, which were log-log plots of net optical density versus dose obtained from a series 

of films exposed to known levels of photons. The use of control films at sites such as the badge 

house, where there was no expectation of a radiation field other than from background, permitted the 

inherent density of the film and any density attributable to background radiation (the so-called 

background fog) to be subtracted from the reading, thereby providing the net optical density 

attributable to occupational exposure. In practice, background films were used to zero the 

densitometer, which then gave a net optical density reading. 

A.1.2 Thermoluminescent Dosimeters

A.1.2.1 NRDS Thermoluminescent Dosimeters

The first routine use of TLDs at NTS began in 1970. Starting in February 1966, Pan Am used TLDs at 

the NRDS as part of the site effluent monitoring program. These dosimeters contained a calciumfluoride phosphor bound to a helically wound wire in an evacuated glass tube and were ideal for the 

intended purpose but unsuitable for personnel dosimetry (Boone, Bennett, and Adams 1970).

Other installations had successfully demonstrated the applicability of LiF TLDs for personnel 

dosimetry (Boone, Bennett, and Adams 1970). A brief investigation was performed and proposals 

submitted for institution of a LiF TLD program at NRDS in 1967 to complement the REECo-supplied 

personnel film dosimetry service.

On July 1, 1968, Pan Am initiated a TLD program at the NRDS with the cooperation of the Health 

Services Laboratory of the AEC Idaho Operations Office (IDO). This laboratory supplied LiF TLD 

chips and provided readout services and considerable consultation based on its extensive LiF TLD 

experience at the National Reactor Testing Station. From July 1, 1968, until July 1970, NRDS 

personnel who were not likely to receive a significant exposure received TLDs. All reported 

exposures were in the background range (0 to 30 mrem). Routine use of TLDs began in July 1970 

and continued until the NRDS ceased operation at the end of 1972 (Boone, Bennett, and Adams 

1970; DOE 2007).

ATTACHMENT A

DOSIMETRY TECHNOLOGY

Page 4 of 21
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 94 of 135

Beginning in July 1970 (and repeated every quarter thereafter), a minimum of one set of dosimeters 

was exposed on the Pan Am Calibration Range to a total dose(s) of from 100 mR to 5 R. Each set of 

dosimeters consisted of 1 Pan Am R-chamber, 6 REECo film, and 12 IDO TLD chips (6 to be read out 

at Pan Am and 6 at IDO). The Pan Am Calibration Range was crosschecked against Pan Am 

R-chambers, which were calibrated by NBS (one set was sent to NBS annually). If the difference 

between the range and the R-chamber exceeded ±6%, the data points were retaken (Boone, Bennett, 

and Adams 1970).

A calibration factor was calculated for each type of dosimeter as follows:

D D x 0 CF

R

− = (A-1)

where:

CF = calibration factor

Dx = mean indicated dose from the set of six exposed dosimeters

D0 = mean indicated dose from the set of six unexposed dosimeters

R = R-chamber indicated dose

As long as the average calibration factor over the dose range of interest fell in the range of 0.9 to 1.1 

(±10%), the results were documented but no further action was taken because this level of accuracy 

was considered adequate. If the average calibration factor fell outside this range, the parties 

concerned were notified and another set of data was taken. In general, the relative accuracy of both 

measurement systems, as indicated by the body of data considered, is adequate for personnel 

dosimetry purposes.

In the case of Pan Am-read TLDs, a calibration factor was always necessary due to the relative 

response of the reader. The appropriate calibration factor was employed as follows:

D CF D D c i =×− ( 0 ) (A-2)

where:

Dc = calibration factor

D0 = actual dose

Di = indicated dose

Boone, Bennett, and Adams (1970) contains additional information on calibrations and calibration 

comparisons between Pan Am, IDO, and REECo.

Figure 6-2 shows the security badge and enclosed TLD insert. It was a plastic holder in which two LiF 

dosimeters were placed. One dosimeter was shielded by the tantalum-cadmium filter area of the 

badge. The other dosimeter was unshielded. The two dosimeters were held in place by retainers 

built into the insert (for pliable LiF-Teflon discs) or hinged retainers (for extruded LiF chips). The 

insert was punched with a binary code that identified security badges for issuance and collection. 

Finally, the bottom left-hand portion of the dosimeter insert was coded to enable security guards to 

determine if dosimeters were being worn for the proper period.

ATTACHMENT A

DOSIMETRY TECHNOLOGY

Page 5 of 21
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 95 of 135

A.1.2.2 REECo Thermoluminescent Dosimeters

With the advent of DOE requirements to restrict personnel exposures to ALARA and with emphasis 

on accurate dosimetry at low doses, REECo Environmental Sciences Department personnel began 

evaluating TLD systems and neutron dosimeters in the early 1980s to replace the film badge and 

neutron TLD (Allen and Schoengold 1995).

After evaluating several dosimetry systems, the Environmental Sciences Department determined that 

the Panasonic 802 TLD and the neutron TED were the best combination for NTS exposure conditions. 

These were put into use January 1, 1987. The security credential holder was redesigned to 

accommodate both dosimeters (Figures 6-3 and 6-4).

DOE Order 5480.11, “Radiation Protection for Occupational Workers” (DOE 1988), established the 

radiation protection standards and program for DOE, its contractor personnel, and other occupational 

personnel to protect workers from ionizing radiation. It defined policies and procedures required to 

operate DOE facilities and conduct activities to keep personnel exposures well below the limits set by 

Order 5480.11 to meet ALARA goals. U.S. Department of Energy Radiological Control Manual

(known as the RadCon Manual; DOE 1994) describes radiation protection standards and program 

requirements as they relate to the NTS and Yucca Mountain Project organizations. Nevada Test Site 

Radiation Protection Program, published in May 1995 (REECo 1995a), demonstrated compliance with 

DOE rules for protecting individuals from ionizing radiation.

The four-element Panasonic UD-802 TLD was the primary dosimeter for routine use issued to all 

monitored personnel until 2001, when it was replaced by the Panasonic 809 dosimeter, which also 

contained four elements (Table 6-1). The Panasonic UD-807 TLD is used to monitor personnel 

working in situations where the likelihood of exposure to an extremity is significantly greater than 

exposure to the whole body.

In most instances, dosimeters (i.e., TLDs) were processed quarterly. Some personnel working in 

locations where high exposures were more likely to occur exchanged their dosimeters monthly. The 

NTS monitoring program was designed to ensure that personnel exposures were kept below the 

annual limit of a total effective dose equivalent of 5 rem.

The Panasonic UD-802 TLD was designed to identify the type and energy of detected radiation, so 

the prescribed tissue depth dose equivalents could be determined accurately. In specific, the fourelement Panasonic 802 was intended to measure the following:

• External photon radiation from 0.010 to 1,000 rem

• Gamma energy range of 0.010 to 10 MeV

• Beta radiation from 0.030 to 1,000 rad

• Beta energy range from approximately 0.30 to 10 MeV

Table A-1 lists TLD element composition, filtration, and radiation type.

The dosimeters were calibrated against known exposures to provide an accurate transition from 

measured exposure to dose equivalent. Dosimeter calibration factors were normalized to the 

corrected readings from run calibration factor (RCF) dosimeters processed with the field dosimeters. 

The computer program separated the calibration dosimeters, used the known RCF exposure value, 

and calculated RCFs that were applied to the remainder of the dosimeters in the run. The RCFs kept 

ATTACHMENT A

DOSIMETRY TECHNOLOGY

Page 6 of 21
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 96 of 135

a reader in calibration over long periods and maintained consistency among different readers (Allen 

and Schoengold 1995).

Table A-1. Panasonic UD-802 dosimeter configuration (Allen and Schoengold 1995).

Element Phosphor Filtration

Radiation type and 

dose equivalent

E1 Li2B4O7:Cu Teflon/polyester, 18 mg/cm2 Gamma, beta, shallow dose

E2 Li2B4O7:Cu Teflon/polyester and plastics, 70 mg/cm2 Gamma, beta, eye dose

E3 CaSO4:Tm Teflon/polyester, ABS plastics, and black ceramic, 

645 mg/cm2

Gamma

E4 CaSO4:Tm Teflon/polyester, ABS and lead, 1,042 mg/cm2 Gamma, deep dose

A.1.2.3 Energy Response

Panasonic UD-802 and UD-809 TLD elements have well-known energy responses (Allen and 

Schoengold 1995) [35]. Like many radiation detection devices, there is an energy (about 20 keV) 

below which radiation does not deliver sufficient energy to the TLD element for detection. As photon 

radiation energy increases, the Li2B4O7:Cu elements have a relatively flat response, while the 

CaSO4:Tm elements over-respond between about 50 and 200 keV. The higher CaSO4:Tm response 

is due primarily to the effective Z of 14.4 for CaSO4, compared to 7.3 for Li2B407:Cu. The lithium 

borate effective Z is much closer to the effective Z for tissue (7.4), and this relationship is important to 

photoelectric effect interactions. The response of both types of elements is relatively flat to about 

5 MeV, where the CaSO4 response increases slightly due to its higher effective Z and, in this case, 

pair production interactions. Figure A-1 shows typical photon energy response curves for Li2B4O7 and 

CaSO4 TLD elements. Figure A-2 shows the response for the UD-802 TLD, including filters. The first 

element in the UD-802 TLD has a total filtration density depth of about 18 mg/cm2

, making it sensitive 

to beta particle radiation with energies of 100 keV or more. The second element responds to beta 

radiation with energies above about 300 keV. Because beta particles have a low linear energy 

transfer and the Li2B4O7 is nearly equivalent to tissue, the beta energy response is relatively flat and 

similar to the photon response shown in Figure A-2 for photon radiation (starting at 0.1 MeV for 

element E1 and 0.3 MeV for element E2). 

Figure A-1. Typical photon energy response curves for 

Li2B4O7 and CaSO4 TLD elements (Allen and 

Schoengold 1995).

ATTACHMENT A

DOSIMETRY TECHNOLOGY

Page 7 of 21
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 97 of 135

Figure A-2. Panasonic UD-802 energy 

responses (Allen and Schoengold 

1995).

NTS algorithms identify radiation types and energies and then apply factors necessary to obtain a 

correct dose equivalent from a TLD reading. Thus, the energy response characteristics of the UD-802 

TLD, over the useable range of the TLD, are automatically factored into the dose equivalent 

determination. The validity of the algorithms for energy response correction is verified through the 

satisfactory completion of DOELAP accreditation for photon and beta particle radiation in several 

radiation energy categories (Allen and Schoengold 1995).

Unlike the Panasonic UD-802, the UD-809 introduced in 2001 has four lithium borate [Li2B4O7(Cu)]

chips. One uses lithium, enriched to 99.99% in 7

Li and 11B, both of which have a negligible neutron 

response. The other three employ neutron-sensitive 6

Li2

10B4O7(Cu) chips, with 6

Li enriched to 95.33% 

and 10B enriched to 94.64%. These two isotopes have high n,α cross-sections. The elements are 

shielded with tin and cadmium on front and back, in various combinations [36].

A.1.2.4 Dose Measurement

The background, element correction factor (ECF), and RCF corrected measurements from TLD 

elements provide an indication of the dose received by the person wearing the dosimeter. However, 

because TLDs are not truly tissue equivalent in relation to dose equivalent, algorithms are used to 

convert the dosimeter response to a dose equivalent value at the specific depth in tissue (Allen and 

Schoengold 1995).

The NTS algorithms were developed by irradiating TLDs mounted on a phantom of tissue substitute to 

specific radiation quantities and types (Allen and Schoengold 1995). The dosimeter measurement 

was compared to the dose equivalent calculated by knowing the type and energy of the radiation. 

This process was repeated many times until enough data were collected to develop equations and 

relationships between a dosimetric reading and the dose equivalent at the specified depth in tissue. 

Published factors for converting radiation exposure to dose equivalent were used in the development 

of the algorithms. Table A-2 lists exposure to dose equivalent conversion factors specified in the 

DOELAP Standard for NIST reference photon fields.

ATTACHMENT A

DOSIMETRY TECHNOLOGY

Page 8 of 21
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 98 of 135

Table A-2. Photon exposure to dose conversion 

factors for NIST reference radiations.

a

NIST reference 

radiation

Conversion factor (mrem/R)

Shallow (0.07 mm) Deep (10 mm)

Filtered X-rays

M30 1.08 0.45

S60 1.15 1.07

M150 1.41 1.47

H150 1.41 1.41

K-fluorescence X-ray (keV)

16 1.08 0.38

24 1.07 0.47

34 1.07 0.99

43 1.28 1.30

58 1.47 1.54

78 1.61 1.72

100 1.59 1.74

Cs-137 photon (keV)

662 1.03 1.03

a. Source: Allen and Schoengold (1995).

A.1.2.5 Calibration

Several Quality Assurance (QA) operations are performed on new dosimeters before they are 

approved for field use. The numerical coding on each TLD label is verified to ensure accurate 

tracking. Before use, and annually thereafter, dosimeter phosphor elements are calibrated by 

determining an ECF for each element. In accordance with procedures, a dosimeter with an ECF out 

of tolerance is removed from use.

TLDs are calibrated with the NTS 137Cs gamma source calibration range. The 137Cs source is 

traceable to NIST through a secondary standard calibrated ion chamber. An integrating electrometer, 

calibrated annually for NIST traceability, is used to verify the accuracy of each TLD and TLD reader 

calibration exposure. All temperature and barometric pressure measuring instruments used for the 

calibration range are calibrated annually to NIST traceable standards.

Monthly Calibration of Gamma Radiation Fields at Inner and Outer Dosimeter Support Rings (REECo 

1995b) outlines procedures for performing gamma source exposures. The results of the exposures 

are used to ensure proper performance of the gamma sources, provide required exposure rate 

information to upgrade source calibration, and document exposure trends for quality control (QC).

A.2 NEUTRON DOSIMETRY

A.2.1 NTA Film

Fast neutrons interact with the hydrogen in the NTA emulsion and film, producing recoil protons that, 

having kinetic energy and being charged, travel through the emulsion, creating a string of exposed 

individual grains of AgBr along their path (Lehman 1961). On development, these show up as tracks 

of grains of AgBr. Track length is a function of the energy of the recoil proton and the angle of travel 

with respect to the plane of the film. Fast neutron dose is determined by direct visual counting, 

usually with a microscope, of the number of proton recoil tracks in a predetermined (for statistical 

ATTACHMENT A

DOSIMETRY TECHNOLOGY

Page 9 of 21
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 99 of 135

purposes) number of microscope fields, usually 100. A countable track must have a length of at least 

three grains (i.e., there must be three grains in a row for a track to be registered).

Tracks can be produced in NTA film emulsion either by protons produced by the 14N(n,p) reaction with 

low-energy neutrons (less than 10 eV) or by direct recoils from energetic neutron interactions with 

hydrogen in the film. The relatively low thermal neutron fluences, the low (n,p) reaction cross-section, 

and the large fluence of thermal neutrons per millirem compared to fast neutrons (2,200 compared 

with 7.5) rendered thermal neutron dosimetry impractical.

Based on theoretical considerations, a minimum neutron energy of about 450 keV is needed to 

produce a proton recoil track, although in practice the minimum detectable energy or threshold energy 

was about 800 keV (Figure A-3). Thus, NTA film is essentially insensitive to neutrons below 500 to 

800 keV, depending on the quality of processing. Therefore, it is suitable only for occupational 

environments in which the majority of the dose comes from neutrons with energies above 1 MeV, or in 

which the fraction of neutrons below 1 MeV is reasonably well known. It is important to calibrate NTA 

film dosimeters with a source that has a neutron energy spectrum similar to that in which the 

individual could be exposed. Initially, a PuBe neutron spectrum was used for calibration. The PuBe 

neutron spectrum has an average energy of about 4 MeV and is considerably richer in fast neutrons 

than the fission spectrum (with average energy of about 1 MeV), and so is likely to result in a 

calibration factor (i.e., tracks/cm2

-n) that would underestimate the fast neutron dose [37]. Any 

moderation of the fission spectrum encountered under field conditions would further exacerbate the 

underestimate.

Figure A-3. NTA film neutron energy response (IAEA 1990).

NTA film has a number of severe limitations that lead to large uncertainties in neutron dosimetry. As 

noted, the response of NTA film was highly dependent on neutron energy and angle of incidence with 

respect to the plane of the film (Cheka 1954; Lehman 1961; Kathren, Prevo, and Block 1965; Kathren 

1967). Another limitation of NTA film is its use dose range. For fast neutron fluences corresponding 

to a few rem, the track density becomes so great that accurate dosimetry becomes difficult if not 

impossible.

NTA films respond to other radiation qualities (photons, betas, etc.). Concomitant photon dose, which 

results in film blackening, can render track counting difficult, introduce errors and, if the optical density 

ATTACHMENT A

DOSIMETRY TECHNOLOGY

Page 10 of 21
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 100 of 135

produced by the photon exposure is sufficiently great, obscure proton recoil tracks altogether. Track 

counting itself is questionable; different persons given identical sections of an exposed NTA film to 

count often produce highly variable track counts. It has been long recognized that, in the dosimetry 

laboratory, human factors associated with reading large numbers of neutron films under a microscope 

can significantly affect neutron dosimetry results. Research at the Rocky Flats Plant in 1994 

reevaluated neutron doses for selected plutonium workers, as described in ORAUT-TKBS-0011-6, 

Rocky Flats Plant – Occupational External Dose (ORAUT 2010c). This research indicated that the 

original evaluations of films could have contained significant errors and that the resulting neutron 

doses could be significantly higher or lower than the doses actually received. The degree of variation 

is a function of the track length and the personal style of the individual doing the counting. 

Uncertainty from this source alone could easily be as great as a factor of 2 (ORAUT 2010c).

The latent image produced by proton recoil tracks in NTA film is highly susceptible to fading before 

development. Fading is a function of time after exposure and is particularly severe in high humidity 

(as much as 75% per week) but can be minimized and largely controlled if the films are sealed in a 

moisture-proof pouch before use. Humidity-induced latent image fading is probably not a significant 

source of uncertainty or dose underestimation at NTS because of the low humidity, although some 

latent image fading is likely if the wearing interval exceeds a month.

Given these limitations, neutron dose estimates made with NTA film are likely to have a high degree 

of uncertainty and generally will underestimate the actual dose from fast neutrons. As noted in 

Table 6-1, this leads to the application of a bias for neutron data obtained with NTA film when detailed 

information about the neutron fields is not available (see Section 6.3.4.3).

A.2.2 TLD Albedo Detectors

The primary advantages of TLD albedo dosimeters are the high sensitivity compared with NTA film 

and the availability of automated TLD readers for rapid reading. The primary disadvantage is that the 

energy response does not match the personal dose equivalent response, so they are highly energydependent (Figure A-4). The energy response can be improved slightly by dosimeter encapsulation, 

as used in the NTS dosimeter design. However, the response is still highly dependent on the neutron 

spectrum.

For albedo dosimeters, neutron fields can be put in four categories based on their relative spectra 

characteristics: (1) reactors, linear accelerators, and accelerators for medical therapy; (2) nuclear fuel 

fabrication areas; (3) radioactive neutron sources; and (4) high-energy accelerators with little or no 

shielding. Within a neutron spectral class, neutron response relative to Hp(10) does not vary by more 

than a factor of 2. The large energy dependence is still a big disadvantage. The advantage to TLD 

albedo detectors, compared to film and to a lesser degree CR-39, is that they detect neutrons of all 

energies and have simple automatic TLD readouts.

Calibration curves have been established for working areas that can reduce workplace-dependent 

changes of albedo response within ±30%. Depending on the neutron field, the lowest detectable dose 

using albedo TLDs varies from 5 to 20 mrem. Albedo dosimeters can be combined with TEDs for 

separate measurement of fast neutrons. In a combination detector, the albedo detector serves as the 

basic neutron detector for screening.

ATTACHMENT A

DOSIMETRY TECHNOLOGY

Page 11 of 21
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 101 of 135

Figure A-4. Neutron energy response of NTS albedo dosimeters

(IAEA 1990).

A.2.3 Track-Etch Dosimeters

The TED is a dosimetry-grade polymer called CR-39. When fast neutrons interact with the plastic, 

submicroscopic damage trails are created. These trails can be enlarged using chemical etching 

techniques to form tracks visible under a 400-power microscope. The tracks can be made more 

visible using electrochemical etching that causes breakdown to form subsurface trees that are easily 

counted under low magnification (Griffith and Tommasino 1990).

The NTS TED consists of three CR-39 foils (i.e., individual dosimeter pieces) cut from 0.025-in. 

(0.0635-cm)-thick sheets covered on both sides with 0.005-in.-thick polyethylene film. They are heat 

sealed under an opaque blister to a plasticized card. A bar code label is applied and (for personnel 

service) the assembly is placed in a cavity in the NTS Personnel Dosimeter.

A three-step etching process is used to develop the damage tracks that result from neutron 

interactions. After removal of the polyethylene film, the track detectors are first etched for 45 min at 

60°C in 6.5N KOH. This is followed by the electrochemical etching step. An alternating potential of 

3 kV is applied across the track detector in the etch bath at a frequency of 60 Hz for 3 hours. The 

resultant tracks are amplified, forming “trees” under the track detector surface. A third step using 3 kV 

but at 2,000 Hz produces tracks that are more uniform and easily recognized as tracks. The final step 

is referred to as the blow-up stage of the process. This produces well-defined round or elliptical holes 

large enough to be seen and counted with low (4-power) magnification. The tracks are recorded with 

a television camera interfaced to a commercial bacteria colony counter that can be used to count 

several standard 0.09-cm2 fields. The dose equivalent is determined by a software program that 

converts the number of foil net tracks to a dose equivalent.

The TEDs were initially calibrated with an unmoderated 252Cf source at Pacific Northwest National 

Laboratory. Secondary calibration is provided by the NTS 238PuBe neutron source. Secondary 

calibration-check TED foils are processed with each batch of personnel monitoring foils to determine 

the RCF for the processing run. Primary and secondary calibration data are recorded in units of 

millirem per track per square centimeter in the TED program. The tracks from the calibration check 

foils processed with each batch of foils are compared to the primary and secondary calibration data to 

ATTACHMENT A

DOSIMETRY TECHNOLOGY

Page 12 of 21
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 102 of 135

determine the dose equivalent conversion factor for the process batch. The energy response of NTS 

TEDs is shown in Figure A-5.

Figure A-5. Neutron energy response of NTS TED (Allen and Schoengold 1995).

Background exposure is subtracted from TED results to eliminate the TED exposure that is not part of 

the individual’s occupational dose. The natural or background neutron radiation level is extremely low 

and, therefore, NTS uses the standard practice of using control TEDs. Control TEDs are prepared 

along with batches of TEDs for issue and are retained in the low-background dosimetry operations 

facility until the corresponding personnel TEDs are processed. Control, calibration-check, and 

personnel foils are processed together, and the number of tracks in control foils is subtracted from the 

number of tracks in personnel and calibration-check foils. The calibration-check results are used to 

determine the correct dose equivalent calibration factor.

A.2.3.1 Dose Algorithms

The CR-39 TED is known to have a nonlinear response, with the observed or measured dose 

equivalent being a slight over-response in comparison with the delivered dose equivalent for low 

doses, then changing to an under-response as the dose increases. Because the nonlinearity can vary 

with CR-39 processing methods, specific data for the NTS TED system were collected. A series of 

TED exposures was made using an unmoderated PuBe source and appropriate conversion factors 

relating the source to an equivalent unmoderated 252Cf source. Two TEDs (six foils) were exposed to 

12 dose equivalents between 130 and 6,505 mrem. The results ranged from 126 to 2,930 mrem, as 

shown in Figure A-6.

2 HH H tl l = = 0.77 0.000459 (A-3)

ATTACHMENT A

DOSIMETRY TECHNOLOGY

Page 13 of 21
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 103 of 135

Figure A-6. Electrochemically etched CR-39 dose response

(Allen and Schoengold 1995).

TED algorithms were developed by exposing many TEDs to calibrated neutron exposures and 

comparing processed results with calculated and known dose equivalents. Irradiations ranging from 

about 100 to 7,000 mrem were used to determine the shape of the curve relating true dose equivalent 

to observed dose. Linear regression analysis of the data pairs (true and observed dose) showed that 

a second-degree polynomial fit the relationship between true dose equivalent and indicated for 

uncorrected dose equivalent. That polynomial, shown below, is the algorithm used by the TED 

program to calculate the neutron dose equivalent HT from the TED uncorrected dose equivalent HI:HI

in the above equation is calculated by the TED program from the number of net tracks and the 

millirem to track calibration factors.

Because the Panasonic UD-802 TLD has a very slight response to neutrons, two algorithms are 

necessary to determine dose equivalent when exposure includes photons and neutrons. One 

algorithm corrects TLD response for neutron interference; the other converts TED results to dose 

equivalent. The deep dose equivalent is the sum of the deep dose from photons and the dose from 

neutrons. Therefore, in all neutron exposure cases, a TLD evaluation is performed to determine the 

deep dose from photons.

A.3 QUALITY ASSURANCE

The QA program was established and has been maintained through adherence to QC procedures and 

practices. As required by DOE rules for protecting individuals from ionizing radiation, the program 

includes internal audits at intervals of 3 years or less. The QC personnel for the external dosimetry 

program monitor and test program operations, data records, and performance. The effectiveness of 

the QA program is demonstrated through satisfactory completion and maintenance of DOELAP 

accreditation (Allen and Schoengold 1995).

Dosimeters are carefully tested before being put in use, and their performance is routinely monitored. 

Acceptable processing and recording equipment calibration and operation is verified by internal QA 

reviews and by participation in external assessments. For example, REECo Health Protection 

Department procedure, Thermoluminescent Dosimeter Quality Assessment (REECo 1995c), outlined 

ATTACHMENT A

DOSIMETRY TECHNOLOGY

Page 14 of 21
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 104 of 135

the methods for assessment of the performance and adequacy of TLD issuing, processing, and 

reporting techniques.

The NTS external dosimetry program was one of the early programs accredited by DOE. 

Accreditation was initially requested in 1989 and updated in 1990. The onsite assessment was 

conducted in 1991, and accreditation was granted in 1992. DOELAP accreditation requires that 

dosimetry operations satisfy specific standards for accuracy of measurements, records, reports, and 

QA activities.

The NTS external dosimetry program has maintained accreditation in the following DOELAP 

categories:

I. Low-energy photon (high dose)

II. High-energy photon (high dose)

IIIA. Low-energy photon

IIIB. Low-energy photon (plutonium)

IV. High-energy photon

VA. Beta

VI. Neutrons (unmoderated, 252Cf)

VII. Mixtures

III + IV

III + VA

IV + VA

III + VI

IV + VI

The accredited categories are based on possible accident scenarios and probable operational 

exposure conditions at NTS and include all DOELAP categories except two, which were omitted 

because:

• Category VB, “Beta Particles – Special” – uranium exposure environment does not exist at 

NTS. Therefore, TLDs have not been calibrated for dose measurements for beta particle 

radiation from natural or depleted uranium slabs.

• Category VI, “Neutron 252Cf (moderated)” – the unmoderated neutron category more closely 

approximates the neutron energy spectra in the NTS occupational environment.

The bias values and uncertainties in Table A-3 are based on DOELAP performance testing for TLD

systems within 30% at an approximate 95% confidence level (Allen and Schoengold 1995).

ATTACHMENT A

DOSIMETRY TECHNOLOGY

Page 15 of 21
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 105 of 135

Table A-3. Bias values and uncertainties for NTS DOELAP accreditation categories (Allen and 

Schoengold 1995).

Category

Bias(B) and standard deviation (S) (rem)

Deep dose Shallow dose

B S B S

I. Low-energy photons (X-ray) – high dose 0.036 0.065 N/Aa N/A

II. High-energy photons – high dose 0.046 0.070 N/A N/A

IIIa. Low-energy photons (X-ray) – general 0.001 0.117 0.03 0.097

IIIb. Low-energy photons (X-ray) – plutonium environments 0.022 0.142 0.011 0.134

IV. High-energy photons 0.099 0.048 0.039 0.058

V. Beta particles – general N/A N/A 0.091 0.036

a. N/A = not applicable.

A.4 BETA PARTICLE FIELD CHARACTERISTICS

The range of a beta particle of known energy can be rigorously calculated by integration of the –dE/dx

equation, which shows that the rate of energy loss is a complex function directly proportional to the 

number of atoms per cubic centimeter of absorber and to the Z of the absorber, and is exponentially 

related to the kinetic energy of the beta particle (Cross et al. 1982). However, a reasonably good 

approximation of the range R of a beta particle of energy E (in megavolts-electron) in any medium can 

be calculated in terms of density thickness (in milligrams per square centimeter) from the following 

empirical equation:

(1.265 0.0954ln ) 412 E R E − = (A-4)

which holds over the 0.01- to 2.5-MeV energy range. For energies above 2.5 MeV, the appropriate 

empirical relationship is:

R E = − 530 106 (A-5)

A somewhat less exact but still reasonable approximation for the range of a beta particle in units of 

grams per square centimeter is to divide the beta particle energy by 2: a 3-MeV particle would have a 

range of 1.5 g/cm2

, which corresponds to 36 ft, approximately the same value (within about 3%) 

obtained using either of the above equations.

Because of attenuation of beta radiation in air, the external dose associated with a fallout field is 

strongly dependent on distance. As one moves away from the beta field, more and more of the lower 

energy particles in the spectrum reach the end of their range and no longer contribute to the dose; 

1 m of air is sufficient to attenuate all betas with energies below about 400 keV. As noted above, 36 ft 

of air is sufficient to absorb virtually all beta radiation associated with a fallout field, so beyond that 

distance there is zero beta dose. In contrast, as one moves closer to the fallout beta field, the 

external beta dose rate rises more rapidly than the concomitant gamma dose rate. Therefore, the 

beta-gamma dose rate ratio is a function of distance from the fallout field; at distances greater than 

36 ft, the ratio is zero because there is no beta dose, as noted above. Calculations by Sondhaus and 

Bond (1955) on persons highly exposed to Pacific Proving Ground weapons fallout contamination 

from the Bravo event indicate that, in relation to the average whole-body gamma dose from a fallout 

field, the corresponding beta dose would range from about a factor of 2 at the head to about a factor 

of 10 at the bottom of the feet (bare) (standing on the contamination). At 1 m above the ground, the 

beta-gamma dose rate ratio was 3. This is consistent with the data obtained onboard a contaminated 

ATTACHMENT A

DOSIMETRY TECHNOLOGY

Page 16 of 21
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 106 of 135

ship that showed a beta-gamma ratio of 10 at the deck level, 8 at knee level, and about 5 at film 

badge level (about 4 ft above the deck).

The above results appear inconsistent with the observations of Kulp and Dick (1960), who reported 

beta dose rates at 1 in. from a contaminated aircraft surface to be typically 10-fold greater than 

gamma dose rates at 1 ft from the surface. When a correction is made for the distance by using 1/d

relationship for an infinite plane source to obtain the gamma dose rate at 1 in., the beta-gamma dose 

rate ratio falls to less than 1. However, the Kulp and Dick measurements involved vastly different 

geometries with opportunities for beta shielding and, therefore, might not be applicable to the type of 

exposures expected in the field at NTS. A more rigorous approach was taken by Broido and Teresi 

(1961), who evaluated the surface dose rate from beta radiation from a fallout field in relation to the 

gamma dose, and found the beta dose to be 13 times that of the gamma dose. Comparing the beta 

surface dose with the gamma dose at 1 m, Broido and Teresi observed a ratio of 40. In a study by 

Black (1962) in which actual measurements were made of doses to troops crawling through a 

contaminated fallout field, the beta-gamma dose ratio was about 7.

In these studies, the concern was the dose from contamination deposited on the skin, a situation 

different from the external exposure situation that could be encountered in the field at NTS and that 

would be likely to produce a higher beta-gamma dose ratio. Although numerous theoretical and 

empirical studies have been made of gamma radiation doses above fallout-contaminated ground, 

there have been few such studies of beta radiation doses. Review of operational monitoring logs from 

several Plumbob detonations indicate, on the basis of survey meter readings, a beta-gamma dose 

rate ratio in the range of slightly greater than 1 to a maximum of about 3, regardless of distances 

above the field (ranging to 3 ft), and for times ranging from 1 to 4 weeks after the detonation. 

Provisional dose ratios take no credit for shielding or attenuation of beta particles by clothing (which 

could reduce them by a factor of 2 or more) and assume a 1-m distance from the fission product field 

with no overburden, leaching, vegetation, or other material or weathering action that would reduce the 

ratio, however likely this could be. 

Specific information on beta-gamma ratios for specific radionuclides, including the special case of 

immersion in iodine and noble gas clouds is in Attachment C.

A.5 NEUTRON FIELD CHARACTERISTICS

As indicated in Section 6.3.5.3, primary NTS operations with neutron exposure potential have been:

• Low-level waste

• Nuclear device assembly

• NRDS and BREN tower calibrations and operation (Operations BREN and HENRE)

• PLUTO reactor (nuclear-powered ramjet engine) 

• Super Kukla reactor

• Nuclear explosive assembly using special nuclear material

• Down-hole well logging

• Neutron detection instrument calibration facilities

The neutron source spectra from these operations were either from fission of uranium and TRU 

nuclides or isotopic sources involving n,α reactions, such as 238PuBe or 241AmBe. Moderation or 

scattering of the source neutrons results in a “softening” of the spectrum, with an increase in the 

ATTACHMENT A

DOSIMETRY TECHNOLOGY

Page 17 of 21
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 107 of 135

fraction of low-energy neutrons present. This, in turn, results in (1) lower RBEs, quality factors, or 

radiation weighting factors, and (2) changes in the effective calibration factors for the energydependent dosimeters and instruments used for personal neutron dose assessment.

There is no indication that neutron spectral measurements were conducted at NTS for radiation 

protection purposes. However, the International Atomic Energy Agency has compiled extensive 

neutron spectral functions, instrument and dosimeter response functions, and dosimetric quantity 

response functions (Ing and Makra 1978; IAEA 1990, 2001). Because the source spectra for NTS 

operations are limited to those discussed above, adequate simulation can be obtained by selecting 

the proper neutron production mechanism and moderation and scatter conditions from the spectral 

catalogs that best simulate those for NTS operations. 

Figures A-7 to A-9 show the effect of scattering and moderation on the neutron spectra. Figure A-10 

and Table A-4 further indicate the impact of spectral softening from scatter and moderation; that is, 

the fraction of Hp(10,0) due to neutrons with energies less than the indicated value. For example, with 

the “hard,” unmoderated 241AmBe source spectrum, about 74% of the dose is due to neutrons above 

2 MeV. The PuBe spectrum at 1 m is probably more characteristic of an unmoderated calibration 

source spectrum, with about 63% of the Hp(10) due to neutrons above 2 MeV. In contrast, consider 

the much softer 252Cf source spectrum in a 15-cm diameter D2O sphere where 9% of the dose comes 

from neutrons below 0.1 MeV and only 31% of the dose comes from neutrons with energies greater 

than 2 MeV. 

1.E-05

1.E-04

1.E-03

1.E-02

1.E-01

1.E+00

1.E-09 1.E-08 1.E-07 1.E-06 1.E-05 1.E-04 1.E-03 1.E-02 1.E-01 1.E+00 1.E+01

Neutron energy - MeV

Fluence/unit log energy (lethargy)

Bare 252Cf

Bare 252Cf

w/room

scatter

Figure A-7. Lethargy neutron spectra for 252Cf calibration source with and without room scatter

(IAEA 2001).

ATTACHMENT A

DOSIMETRY TECHNOLOGY

Page 18 of 21
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 108 of 135

1.E-03

1.E-02

1.E-01

1.E+00

1.E-07 1.E-06 1.E-05 1.E-04 1.E-03 1.E-02 1.E-01 1.E+00 1.E+01

Neutron energy - meV

Fluence/unit energ

y (letharg

y)

252Cf - ISO

252Cf in

10.16 cm

CH2

252Cf in

22.86 cm

CH2

252Cf in

22.86 cm

CH2 w/10.2

cm Fe cover

Figure A-8. Lethargy neutron spectra for 252Cf with various thickness moderators (IAEA 2001).

1.E-04

1.E-03

1.E-02

1.E-01

1.E+00

1.E-08 1.E-07 1.E-06 1.E-05 1.E-04 1.E-03 1.E-02 1.E-01 1.E+00 1.E+01

Neutron energy - MeV

Fluence/unit energy (lethargy)

241AmBe

w/o room

scatter

241AmBe

w/room

scatter

Figure A-9. Lethargy neutron spectra for an 241AmBe calibration source with and without room 

scatter (IAEA 2001).

ATTACHMENT A

DOSIMETRY TECHNOLOGY

Page 19 of 21
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 109 of 135

0

0.2

0.4

0.6

0.8

1

1.2

1.E-06 1.E-05 1.E-04 1.E-03 1.E-02 1.E-01 1.E+00 1.E+01

Neutron energy, En - MeV

Fraction of Hp(0,10) <En

252Cf - ISO 252Cf D2O Mod. - ISO

AmBe - ISO TRU Plant, Pu Repro. Plant, Heavily shielded

Bare 238PuO2 at 100 cm Godiva Spectrum

Figure A-10. Fraction of Hp(10,0) due to neutrons with energies less than En.

ATTACHMENT A

DOSIMETRY TECHNOLOGY

Page 20 of 21
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 110 of 135

Table A-4. Fraction of Hp(10,0) for neutrons with energies less than En.

Energy

En – MeV

Cf-252

ISO

Bare 

Cf-252 

w/o 

shadow 

cone

Cf-252 D2O 

Mod. - ISO

Cf-252 in 

D2O 

w/o shadow 

cone

241AmBe 

ISO

AmBe 

w/o 

shadow 

cone

PuBe 

at 1m

TRU Plant, Pu 

Repro. Plant, 

heavily 

shielded

Bare 

238PuO2

at 100 

cm

BREN

HPRRa

HENRE 

linear 

acceleratorb

Super 

Kukla 

reactorc

Godiva 

spectrum

1.00E-02 0.000 0.004 0.018 0.030 0.000 0.003 0.002 0.028 0.012 0.098 0.000 0.000 0.000

1.99E-02 0.000 0.005 0.027 0.039 0.000 0.003 0.002 0.032 0.013 0.103 0.000 0.000 0.000

5.01E-02 0.001 0.008 0.050 0.064 0.000 0.005 0.003 0.046 0.014 0.123 0.000 0.000 0.001

1.00E-01 0.003 0.017 0.089 0.106 0.001 0.010 0.005 0.080 0.021 0.150 0.000 0.100 0.008

1.99E-01 0.017 0.042 0.155 0.181 0.009 0.028 0.027 0.162 0.058 0.236 0.000 0.100 0.043

3.98E-01 0.072 0.114 0.228 0.283 0.037 0.070 0.066 0.321 0.225 0.3271 0.011 0.540 0.158

5.01E-01 0.110 0.158 0.269 0.328 0.053 0.099 0.088 0.392 0.331 0.374 0.011 0.540 0.226

7.94E-01 0.236 0.292 0.361 0.462 0.099 0.169 0.158 0.549 0.544 0.516 0.011 0.540 0.394

1.00E+00 0.325 0.382 0.407 0.516 0.127 0.212 0.242 0.625 0.646 0.565 0.022 0.794 0.492

1.25E+00 0.430 0.486 0.476 0.579 0.158 0.260 0.310 0.694 0.744 0.619 0.038 0.794 0.591

1.99E+00 0.662 0.708 0.668 0.775 0.262 0.376 0.373 0.814 0.842 0.772 0.074 0.875 0.781

3.98E+00 0.927 0.939 0.916 0.961 0.660 0.683 0.744 0.931 0.911 0.937 0.143 0.958 0.958

5.01E+00 0.968 0.969 0.963 0.969 0.813 0.806 0.871 0.956 0.930 0.967 0.168 1.000 0.983

7.94E+00 0.999 0.988 0.998 0.992 0.995 0.969 0.997 0.989 0.977 0.993 0.201 1.000 1.000

1.00E+01 1.000 0.993 1.000 0.994 1.000 0.991 1.000 0.999 0.987 0.996 0.217 1.000 1.000

1.58E+01 1.000 1.000 1.000 1.000 1.000 1.000 1.000 1.000 1.000 1.000 1.000 1.000 1.000

a. Source: Sims and Ragan (1987).

b. Source: Burson (1971).

c. Source: Wimett (1965).

ATTACHMENT A

DOSIMETRY TECHNOLOGY

Page 21 of 21
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 111 of 135

TABLE OF CONTENTS

SECTION TITLE PAGE

B.1 Introduction .......................................................................................................................... 111

B.2 Work Area and Operation-Dependent Energy Group Photon Fractions................................ 111

LIST OF TABLES

TABLE TITLE PAGE

B-1 Work area and operation-dependent photon fractions.......................................................... 112

B.1 INTRODUCTION

Radionuclides with photon energies greater than 250 keV contribute a significant fraction of the 

penetrating (Eγ ≥ 30 keV) photon exposures at NTS. If the fraction of photons with energies above 

250 keV in a given area and operation is known, it is possible to make a more realistic, yet favorable 

to claimants, estimate of the fraction of exposure or personal dose equivalent in this higher energy 

group.

B.2 WORK AREA AND OPERATION-DEPENDENT ENERGY GROUP PHOTON 

FRACTIONS

ORAUT-TKBS-0008-2, Nevada Test Site – Site Description (ORAUT 2008, Table 2-2) presents an 

inventory of the radionuclides at NTS by area and operation. This table is a basis for estimating the 

likely exposure and dose equivalent fraction in the 30-to-250-keV and >250-keV energy groups for 

these areas and operations. The photon (gamma and X-ray) decay properties for each nuclide were

obtained from the WWW Table of Radioactive Isotopes on the Internet at 

http://ie.lbl.gov/toi/nucSearch.asp (Firestone and Ekström 2002). The fraction of photons per decay 

have been determined in the <30-keV, 30-to-250-keV, and >250-keV three energy groups for each 

nuclide (Griffith 2008). A lower energy cutoff of 10 keV was used for photons below 30 keV because 

they are highly unlikely to contribute a significant fraction to the external exposure.

Using the radionuclide inventory in Table 2-2 and assuming equal total photon contribution for each 

nuclide in any area and operation, the photon fraction production has also been estimated for each 

area and operation. However, two adjustments have been made to these data. First, it is assumed 

that, because of the low penetrating power of photons less than 30 keV, 50% of these will be 

attenuated in surrounding material before they can contribute to individual exposures. Second, 20% 

of the photons with energies greater than 250 keV are assumed to be scattered, and result in an 

increased contribution to the 30-to-250-keV energy group, thus reducing the contribution from the 

higher energy photons accordingly. The results are listed in Table B-1.

ATTACHMENT B

ENERGY GROUP ALLOCATION OF PHOTON EXPOSURE AND PERSONAL DOSE 

EQUIVALENT BY NTS WORK AREA AND OPERATION

Page 1 of 2
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 112 of 135

Table B-1. Work area and operation-dependent photon fractions.

Operation Areas

Photon fraction Adjusted photon fractiona

<30 

keV 

30–250 

keV

>250 

keV

<30 

keV 

30–250 

keV

>250 

keV

Drillback operations 1–10 and 18–20 0.06 0.36 0.58 0.03 0.50 0.47

Reentry and mineback 

operations

1, 12, 15, and 16 0.08 0.22 0.70 0.04 0.38 0.58

Routine tunnel operations 1, 12, 15, and 16 0.00 0.03 0.97 0.00 0.22 0.78

Decontamination facility 6 0.24 0.28 0.48 0.14 0.43 0.44

Treatability test facility 25 0.22 0.32 0.47 0.12 0.46 0.42

Atmospheric safety test areas 5 and 11 None indicated None indicated

Atmospheric weapons test areas 1–5, 7–11, and 18 0.18 0.31 0.51 0.10 0.45 0.45

Low-level waste site 3 0.47 0.25 0.28 0.31 0.40 0.29

Low-level waste site 5 0.34 0.31 0.36 0.20 0.45 0.34

Radiation instrument calibration 6 and 23 0.54 0.25 0.22 0.36 0.40 0.24

Radiograph operations 23 0.01 0.04 0.96 0.00 0.23 0.76

Well logging operations 1–10 and 18–20 0.49 0.23 0.29 0.32 0.38 0.30

Nuclear explosive/device 

assembly

6 and 27 0.73 0.27 0.00 0.57 0.43 0.00

Nuclear rocket development 25 and 26 0.04 0.29 0.67 0.02 0.43 0.55

Radioactive source storage 6 and 23 0.40 0.25 0.32 0.26 0.41 0.33

Radiochemistry and counting 

laboratories

6 and 23 0.16 0.25 0.59 0.09 0.40 0.51

a. Adjustment factors:

<30-keV contribution = 50% of relative photon contribution to account for attenuation of low-energy photons.

30 to 250 KeV = photon contribution in that energy range +20% of >250-keV contribution to account for scatter of highenergy photons.

>250 keV = 0.8 × high-energy photon contribution to account for loss due to scatter.

ATTACHMENT B

ENERGY GROUP ALLOCATION OF PHOTON EXPOSURE AND PERSONAL DOSE 

EQUIVALENT BY NTS WORK AREA AND OPERATION

Page 2 of 2
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 113 of 135

TABLE OF CONTENTS 

SECTION TITLE PAGE

C.1 Introduction .......................................................................................................................... 114

C.2 Beta-Photon Ratios from Standing on a Contaminated Surface After Nuclear Test or 

Reactor Operation................................................................................................................ 114

C.3 Beta-Photon Ratios from Immersion in a Contaminated Cloud After Nuclear Test or 

Reactor Operation................................................................................................................ 117

C.4 Beta-Photon Ratios for Other Geometries............................................................................ 120

C.4.1 Photon Dose Constants............................................................................................ 120

C.4.2 Beta Dose Rates ...................................................................................................... 126

C.5 Summary of Beta-Photon Ratios.......................................................................................... 128

LIST OF TABLES

TABLE TITLE PAGE

C-1 Calculated beta-photon ratios for mixed fission and activation products from Sedan 

and tests from Operations RANGER, BUSTER-JANGLE, TUMBLER-SNAPPER, 

AND UPSHOT-KNOTHOLE-contaminated surfaces ............................................................ 118

C-2 Beta-photon ratios for exposure from surface contamination................................................ 120

C-3 Calculated beta-photon ratios for mixed fission and activation products from 

Operations RANGER, BUSTER-JANGLE, TUMBLER-SNAPPER, and UPSHOTKNOTHOLE tests – immersion in a contaminated semi-infinite cloud................................... 121

C-4 Photon dose constants, beta dose rates, and beta-photon ratios for NTS 

radionuclides........................................................................................................................ 124

C-5 Beta-photon ratios for exposure from immersion in an infinite cloud of mixed fission 

and activation products ........................................................................................................ 126

C-6 Geometry-dependent beta-photon ratios for NTS radionuclides........................................... 128

LIST OF FIGURES

FIGURES TITLE PAGE

C-1 Beta-gamma dose ratios for contaminated surfaces used for Pacific tests ........................... 115

C-2 Calculated beta-photon ratios for skin for 11 atmospheric tests from Operation 

UPSHOT-KNOTHOLE ......................................................................................................... 116

C-3 Calculated beta-photon ratios for skin for nuclear rockets and ramjet test effluents ............. 117

C-4 Calculated beta-photon ratios for immersion in a contaminated semi-infinite cloud –

maximum of values from 11 atmospheric test radionuclide inventories ................................ 123

ATTACHMENT C

BETA-PHOTON RATIO ESTIMATES FOR NTS WORK AREAS AND OPERATIONS

Page 1 of 13

ATTACHMENT C

BETA-PHOTON RATIO ESTIMATES FOR NTS WORK AREAS AND OPERATIONS

Page 1 of 17
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 114 of 135

C.1 INTRODUCTION

As beta particles from sources outside the body enter tissue, the dose falls off rapidly with depth, and 

tissues and organs lying deeper than 10 mm in the body are unaffected. Therefore, beta particles are 

appropriately ignored in considering external dose to most tissues and organs, which lie deeper than 

10 mm; for them the appropriate quantities are gamma dose and neutron dose. The two exceptions 

are the skin, with its sensitive component (basal cells) at a depth of 0.07 mm, and the eye, with its 

sensitive component (lens) at a depth of 3 mm. The potential contribution from beta particles should 

be considered when the dose to skin or the lens of the eye is assessed.

As a result of a range of operations at NTS, a range of radionuclides can be found across the Site. 

An inventory of these nuclides is in Table 2-2. However, nuclear testing and reactor operations, 

including nuclear rocket and ramjet tests, result in a much larger inventory in areas where these 

operations were conducted. The situation is made much more complex by the time dependence of 

the radionuclide inventory after detonation or reactor operation. This time dependence results in a 

variation in the radiological properties of the inventory that varies with time. A set of publications by 

Harry Hicks at LLNL (Hicks 1981a,b,c,d,e,f,g,h,i, 1982, 1984) produced a vast set of data covering 

177 fission and activation products for periods from 1 hour to 50 years after detonation or reactor 

operation. Surface roughness effects are simulated by using values of (mR/hr)/(μCi/m2

) for a 

relaxation length of 0.16 g/cm2 (Beck 1980). According to Beck, the concentration of fallout varies 

exponentially with soil depth, Z, according to the relation C = Coe-αZ

, and he defines relaxation length 

as 1/α. The Hicks publications represent a detailed source term for beta-photon ratio calculations.

C.2 BETA-PHOTON RATIOS FROM STANDING ON A CONTAMINATED SURFACE AFTER 

NUCLEAR TEST OR REACTOR OPERATION 

The fundamentals of the method for estimating external dose to the skin and eye from standing on 

contaminated ground are summarized as follows:

Beta dose to the skin or lens of the eye from external sources is accrued with the gamma dose from 

radioactive fallout, contamination, or neutron-induced radionuclides. As a result, the beta dose is 

proportional to the gamma dose, and its relative magnitude can be expressed by a beta-gamma dose 

ratio.

1. The beta-gamma dose ratio depends on radionuclide decay and distribution and on geometric 

relationships between the exposed individual and the radiation source. Gamma and beta 

energy spectra are interdependent functions of time. As a consequence, the beta-gamma

dose ratio depends on time since detonation and distance from the detonation site.

2. Because of the attenuation characteristics of electrons in matter, beta dose assessments 

depend more critically than gamma dose assessments on geometry and the shielding material 

between the radioactive source and the exposed individual. As a consequence, the nature of 

specific job- or task-related activities and their associated protective measures entails special 

attention and evaluation in determining a beta dose component.

3. Beta doses to skin are evaluated at the anatomic location where a skin cancer has been 

diagnosed. The depth for evaluation is 0.07 mm, the conventional depth of the basal-cell layer 

of the skin, which is assumed to be the tissue at risk for skin cancer. Beta doses to the lens of 

ATTACHMENT C

BETA-PHOTON RATIO ESTIMATES FOR NTS WORK AREAS AND OPERATIONS

Page 2 of 17
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 115 of 135

the eye are assessed at a depth of 3 mm below the front surface of the eye, where the tissue 

at risk for posterior subcapsular cataract development is assumed to be.

4. A beta energy greater than 0.07 MeV is required to penetrate the dead epidermal layer, so 

beta particles with energies less than that are not included in dose assessments.

5. Skin and eye doses are assessed as the sum of the applicable “high-sided” beta and “highsided” gamma doses (neutron doses presumably are included if they are significant).

6. Fallout deposited on a surface is a semi-infinite plane isotropic source, and decontamination 

activities are considered in evaluating beta doses. Assessments of skin doses are simplified 

by ignoring attenuation of electrons by large fallout particles that contain volume-distributed 

activity, particle and photon scattering due to surface roughness, particle and photon 

attenuation due to penetration into a radioactive surface, and radioactive-source depletion due 

to weathering, chemical dissociation, or environmental transport (concentration or dispersion). 

Those simplifications have the effect of making the calculated doses overestimates.

Barss (2000) provides methods and tables useful for assessing beta dose. They include separate 

tables of beta-gamma dose ratios from exposure to fission products on the ground as a function of 

time since detonation for NTS and Pacific tests (in the Pacific, one table applies to Operation 

CASTLE, Test BRAVO fallout, and a second table applies to all other tests). A separate table of betagamma dose ratios is provided for activation products in soil. The beta-gamma dose ratios are 

calculated from published beta-particle and gamma-ray spectra. The beta-gamma dose ratios from 

the table for Pacific tests are shown in Figure C-1. The figure illustrates the substantial variation of 

beta-gamma dose ratios with time after detonation and height above ground. The beta-gamma dose 

ratios for the lens of the eye are much smaller because the greater depth of the sensitive tissue 

(3 mm for lens versus 0.07 mm for skin) leads to much more attenuation.

Figure C-1. Beta-gamma dose ratios for contaminated surfaces used for 

Pacific tests (Barss 2000).

Activation products are distributed with depth in soil because they originate primarily by interactions 

with neutrons that penetrate into the soil rather than in deposition from the atmosphere. Thus, betaATTACHMENT C

BETA-PHOTON RATIO ESTIMATES FOR NTS WORK AREAS AND OPERATIONS

Page 3 of 17
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 116 of 135

gamma dose ratios of activation products are small because most of the activation products are 

deeper in the soil than the range of the emitted beta particles.

Extensive calculations of beta and photon dose conversion factors for skin from (1) surface 

contamination and (2) air immersion are available for 825 radionuclides (Eckerman et al. 1999), based 

on the work of Kocher and Eckerman (1981). The geometry involves the use of a reference adult 

phantom placed (1) on a contaminated ground plane and (2) immersed in a semi-infinite cloud of 

contaminated air. Eckerman et al. does not provide separate conversion factors for beta and photons. 

However, they are available for skin only from Eckerman (2006b).

Hicks (1981a,b,c,d,e,f,g,h,i, 1982, 1984) presents the surface contamination after each event in three 

tables of radionuclide concentration in terms of microcuries per cubic meter: zero to 21 hours after 

detonation (177 radionuclides); zero to 300 days after detonation (128 radionuclides); and zero to 50 

years after detonation (37 radionuclides). These data have been used with the beta and photon dose 

conversion factors (Eckerman 2006b) to calculate radionuclide inventory-weighted beta-photon ratios. 

Examples of the resulting calculations for selected atmospheric tests are shown in Figure C-2. These 

are consistent with the values shown in Figure C-1 for a height of about 120 cm. Both figures show a 

significant increase at later times, 100 to 200 days. The maximum time for Figure C-1 is 2 years, so 

the decrease after that in Figure C-2 out to 50 years is not shown. Figure C-3 shows the results of 

similar calculations for the effluents from nuclear rocket and ramjet tests. Tabulated beta-photon 

ratios for the fallout from 35 tests are listed in Table C-1. These values can be modified with 

appropriate factors for shielding and distance (Barss and Weitz 2006).

1

10

100

1.E-03 1.E-02 1.E-01 1.E+00 1.E+01 1.E+02 1.E+03 1.E+04 1.E+05

Elapsed Time - Days

Beta/Photon Sv/Sv

Annie Nancy Ruth Dixie Ray Badger

Simon Encore Harry Grable Climax

Figure C-2. Calculated beta-photon ratios (Sv/Sv) for skin for 11 atmospheric tests from 

Operation UPSHOT-KNOTHOLE.

ATTACHMENT C

BETA-PHOTON RATIO ESTIMATES FOR NTS WORK AREAS AND OPERATIONS

Page 4 of 17
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 117 of 135

1

10

100

1.E-03 1.E-02 1.E-01 1.E+00 1.E+01 1.E+02 1.E+03 1.E+04 1.E+05

Elapsed Time - Days

Beta/Photon Sv/Sv

Figure C-3. Calculated beta-photon ratios (Sv/Sv) for skin for nuclear rockets and ramjet test 

effluents.

External beta doses from standing on contaminated ground or other surfaces can be estimated by 

applying a beta-gamma dose ratio to an upper bound gamma dose, which is determined from film 

badge data or dose reconstruction. Table C-1 lists time-dependent beta-photon ratios for fission and 

activation products from detonation or time of operation to 50 years. Based on these data, the betaphoton ratios can be used if the exposure is presumed to have occurred for an extended 

contaminated surface (e.g., ground contamination from dispersal of fallout). The beta-photon ratios 

would be reduced by attenuation of clothing [anticontamination clothing (e.g., coveralls), shoes, 

gloves, etc.], increased separation between the contaminated surface and the individual (e.g., working 

on an elevated structure), or any intervening material (surface coatings, floorings, etc.) and stay times. 

The dose reconstructor should determine if the exposure resulted from the Energy Employee walking 

directly on the contaminated surface, or if the job site characteristics involved nonradioactive, 

intervening surfaces of materials.

Information in the claim files might support the appropriateness of using the calculated values listed in 

Table C-2. The ratio would be applied on a dosimeter cycle-by-cycle basis if there is indication in the 

claim file that the worker was in radiation controlled areas for a particular badging cycle.

C.3 BETA-PHOTON RATIOS FROM IMMERSION IN A CONTAMINATED CLOUD AFTER 

NUCLEAR TEST OR REACTOR OPERATION

The special situation exists in which a worker might have been exposed to photons and beta particles 

by immersion in a contaminated cloud. This situation is known to have occurred due to release of 

radioiodines and noble gases during post-test drilling operations. Because of the exposure geometry, 

beta-gamma ratios under such circumstances are likely to have been different than those addressed 

above.

The beta dose rate factors due to immersion in semi-infinite contaminated clouds of radionuclides in 

air have been estimated by Kocher and Eckerman (1981) and implemented in Eckerman et al. (1999). 

ATTACHMENT C

BETA-PHOTON RATIO ESTIMATES FOR NTS WORK AREAS AND OPERATIONS

Page 5 of 17
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 118 of 135

Table C-1. Calculated beta-photon ratios for mixed fission and activation products from Sedan and tests from Operations RANGER,

BUSTER-JANGLE, TUMBLER-SNAPPER, and UPSHOT-KNOTHOLE-contaminated surfaces.

Event Year

Beta-photon ratio

Hours Days Years

Overall 

averagea

1-day 

reentrya

Onsite 

during 

yeara 0. 1.0 2. 12. 1.0 10. 50. 200. 1.0 2.0 5.0 10. 20. 50.

RANGER Able 1951 10.4 7.5 6.2 8.1 6.0 2.8 5.0 7.2 15.6 47.0 18.0 13.2 14.9 15.6 12.7 8.0 24.3

RANGER Baker 9.6 7.1 5.9 7.4 5.7 2.7 4.3 7.8 19.2 43.9 17.0 8.8 9.6 10.1 11.4 7.4 21.6

RANGER Easy 12.5 8.3 6.4 6.8 4.8 2.3 4.3 7.8 19.2 43.9 17.0 8.8 9.6 10.0 11.6 7.1 21.6

RANGER Baker 2 9.3 7.0 5.9 7.5 5.7 2.7 4.3 7.8 19.2 43.9 17.0 8.8 9.6 10.1 11.3 7.4 21.6

RANGER Fox 8.2 6.7 5.9 8.3 6.5 3.0 4.8 7.3 17.0 45.9 18.0 11.9 13.4 14.0 12.2 8.0 23.5

BUSTER Baker 1951 5.8 5.8 5.4 7.8 6.6 3.0 3.7 7.3 18.5 41.5 16.1 6.9 7.3 7.7 10.2 7.4 18.9

BUSTER Charlie 8.2 6.6 5.9 8.4 6.6 3.0 4.5 6.4 14.8 45.0 17.7 12.3 13.9 14.5 12.0 8.1 21.9

BUSTER Dog 8.8 6.9 5.9 8.1 6.3 2.8 4.5 6.5 15.4 44.4 17.5 11.4 12.9 13.4 11.8 7.9 21.9

BUSTER Easy 7.0 6.3 5.7 8.6 6.9 3.1 4.4 6.5 15.2 44.5 17.6 11.6 13.1 13.7 11.7 8.2 21.5

JANGLE Sugar 1951 7.2 6.1 5.6 6.6 5.2 2.8 6.5 12.3 22.8 45.7 21.9 16.9 16.5 15.8 13.7 6.5 32.9

JANGLE Uncle 7.2 6.1 5.6 6.6 5.2 2.8 6.5 12.3 22.8 45.8 21.9 16.9 16.5 15.8 13.7 6.5 32.9

TUMBLER-SNAPPER

Able

1952 11.6 7.9 6.4 7.8 5.6 2.7 5.5 8.3 18.7 59.0 24.4 16.4 16.4 15.8 14.8 7.8 26.9

TUMBLER-SNAPPER 

Baker

11.5 7.8 6.3 7.9 5.7 2.7 5.5 8.3 18.7 59.0 24.4 16.4 16.3 15.8 14.7 7.9 26.9

TUMBLER-SNAPPER 

Charlie

9.0 6.9 6.0 8.2 6.3 2.9 5.2 8.6 20.3 55.4 23.3 14.1 13.9 13.5 13.8 8.0 25.8

TUMBLER-SNAPPER

Dog

7.6 6.4 5.7 8.4 6.7 3.1 5.2 8.6 20.3 55.4 23.3 14.1 13.9 13.5 13.7 8.1 25.8

TUMBLER-SNAPPER 

Easy

6.0 5.8 5.4 7.0 5.8 3.0 5.9 11.6 24.1 47.3 21.8 14.4 14.0 13.4 13.3 6.8 30.0

TUMBLER-SNAPPER 

Fox

7.1 6.1 5.6 6.8 5.4 2.9 5.9 11.6 24.1 47.5 21.9 14.5 14.1 13.5 13.4 6.7 30.0

TUMBLER-SNAPPER 

George

7.0 6.2 5.6 6.8 5.4 2.9 5.9 11.6 24.1 47.5 21.9 14.5 14.1 13.5 13.4 6.7 30.0

TUMBLER-SNAPPER 

How

5.6 5.7 5.4 7.1 5.9 3.1 5.9 11.6 24.0 47.3 21.8 14.5 14.1 13.5 13.3 6.8 30.0

UPSHOT-KNOTHOLE

Annie

1953 6.9 6.1 5.3 6.9 5.5 2.9 5.8 11.5 23.5 45.1 20.4 13.8 13.8 13.5 12.9 6.7 29.6

ATTACHMENT C

BETA-PHOTON RATIO ESTIMATES FOR NTS WORK AREAS AND OPERATIONS

Page 6 of 17
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 119 of 135

Event Year

Beta-photon ratio

Hours Days Years

Overall 

averagea

1-day 

reentrya

Onsite 

during 

yeara 0. 1.0 2. 12. 1.0 10. 50. 200. 1.0 2.0 5.0 10. 20. 50.

UPSHOT-KNOTHOLE 

Nancy

6.9 6.0 5.5 6.9 5.5 2.9 5.9 11.6 23.8 44.1 20.5 14.0 13.9 13.5 12.9 6.7 30.0

UPSHOT-KNOTHOLE 

Ruth

4.7 5.3 5.3 6.8 5.1 2.7 6.3 11.0 21.1 49.0 22.6 16.9 16.5 15.8 13.5 6.5 31.4

UPSHOT-KNOTHOLE 

Dixie

7.6 6.6 5.6 8.4 6.6 3.0 4.4 6.6 15.7 50.8 23.0 14.2 13.9 13.5 12.9 8.1 21.5

UPSHOT-KNOTHOLE 

Ray

8.5 6.6 5.8 6.8 5.2 2.8 6.3 11.0 21.6 48.9 22.5 16.9 16.5 15.8 13.9 6.8 31.4

UPSHOT-KNOTHOLE 

Badger

8.0 6.4 5.7 6.6 5.2 2.7 5.8 11.7 24.2 43.0 19.8 13.2 13.1 12.9 12.7 6.6 29.6

UPSHOT-KNOTHOLE 

Simon

6.7 6.0 5.6 7.1 5.7 2.9 6.2 11.3 22.4 46.9 22.1 16.1 15.7 15.1 13.6 6.9 31.2

UPSHOT-KNOTHOLE 

Encore

8.3 6.7 5.9 8.6 6.8 3.1 5.4 8.5 22.4 46.9 22.1 16.1 15.7 15.1 13.7 8.3 26.6

UPSHOT-KNOTHOLE 

Harry

7.1 6.1 5.6 6.9 5.5 2.9 6.0 11.4 23.3 46.4 20.8 14.7 14.5 14.0 13.2 6.8 30.4

UPSHOT-KNOTHOLE 

Grable

6.2 6.0 5.6 8.7 7.0 3.5 5.5 8.0 18.0 57.7 25.1 17.1 16.7 16.0 14.4 8.2 26.8

UPSHOT-KNOTHOLE 

Climax

6.7 6.3 5.7 8.9 7.2 3.3 4.6 6.4 14.9 54.6 24.1 15.8 15.7 15.1 13.5 8.4 22.3

Sedan 1962 5.3 5.4 4.8 4.0 3.3 2.5 2.6 2.9 4.6 11.6 10.1 5.3 6.5 7.8 5.5 4.2 12.3

Maximum 12.5 8.3 6.4 8.9 7.2 3.5 6.5 12.3 24.2 59.0 25.1 17.1 16.7 16.0 16.0 8.8 32.9

a. The average values can be used if a reasonable approach is required or if the employee is not directly identified with an event.

ATTACHMENT C

BETA-PHOTON RATIO ESTIMATES FOR NTS WORK AREAS AND OPERATIONS

Page 7 of 17
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 120 of 135

Table C-2. Beta-photon ratios for exposure from 

surface contamination.

Elapsed time after

production event

Beta-photon

(Sv/Sv)

0 to 50 d 10

50 to 365 d 25

1 to 5 yr 60

>5 yr 25

As for the surface contamination geometry, the separate unpublished beta and photon dose 

conversion factors for each radionuclide are available (Eckerman 2006b). Table C-3 lists the timedependent beta-photon ratios for the radionuclide inventories in Hicks (1981a,b,c,d,e,f,g,h,i, 1982, 

1984), but in a semi-infinite cloud. The ratios are lower than those from a contaminated surface 

because the betas are absorbed by the cloud to a greater extent than the photons. The exposure of 

post-test drilling crews to clouds of fission product noble gases (krypton and xenon), and halogens 

(bromine and iodine) has been noted. A separate calculation was made for cloud immersion betaphoton ratios after 11 tests in the RANGER and BUSTER-JANGLE series using only the isotopes of 

those elements. There is very little variation in the results from one test to another, because there is 

little variation and fractionation for those nuclides. The maximum ratios are shown in Figure C-4. The 

ratios are consistently below 1.0 until the 10.7-year 85Kr begins to dominate after about 100 days.

C.4 BETA-PHOTON RATIOS FOR OTHER GEOMETRIES

In addition to activation and fission products from nuclear tests and reactor operation, radionuclides 

have been used or handled in various areas on NTS. An inventory of these radionuclides is in 

ORAUT-TKBS-0008-2, Nevada Test Site – Site Description (ORAUT 2008, Table 2-2). For the most 

part, the geometries for exposure to these radionuclides are difficult to specify with accuracy. The 

beta-photon ratios are also difficult to specify with certainty. In many cases, such as the use of 

encapsulated radioactive sources, although the radionuclides involved might have an inherently high 

beta-photon ratio, the encapsulation is likely to absorb the associated beta particles. To estimate

beta-photon ratios for a geometry other than extended contaminated surfaces or cloud immersion, it is 

necessary to select an arbitrary, but realistic, geometry as a basis for the ratio estimates. The 

calculations in this attachment are based on a distance of 1 m for a point source.

C.4.1 Photon Dose Constants

The values of the specific gamma ray dose constant, Γ, in mSv/hr/MBq at 1 m were established for 

each radionuclide in the inventory listed in Table 2-2. The values of Γ for several of those nuclides 

are listed in Table 6.2.2 of the Handbook of Health Physics and Radiological Health (Shleien, 

Slaback, and Birky 1998). Those values are based on the work of Unger and Trubey (1982), and 

appear in Table C-4. 

ATTACHMENT C

BETA-PHOTON RATIO ESTIMATES FOR NTS WORK AREAS AND OPERATIONS

Page 8 of 17
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 121 of 135

Table C-3. Calculated beta-photon ratios for mixed fission and activation products from Operations RANGER, BUSTER-JANGLE, 

TUMBLER-SNAPPER, and UPSHOT-KNOTHOLE tests – immersion in a contaminated semi-infinite cloud.

Event Year

Beta-photon ratio

Hours Days Years

Overall 

averagea

1-day 

reentrya

Onsite 

during 

yeara 0. 1.0 2. 12. 1.0 10. 50. 200. 1.0 2.0 5.0 10. 20. 50.

RANGER Able 1951 1.4 1.0 0.8 1.2 1.0 0.5 0.7 1.0 2.2 6.6 2.6 2.0 2.3 2.4 1.8 1.2 3.4

RANGER Baker 1.3 0.9 0.8 1.1 0.9 0.5 0.6 1.1 2.7 6.2 2.4 1.3 1.5 1.6 1.6 1.1 3.0

RANGER Easy 1.7 1.1 0.8 1.1 0.9 0.5 0.6 1.1 2.7 6.2 2.4 1.3 1.5 1.6 1.7 1.1 3.0

RANGER Baker 2 1.2 0.9 0.8 1.1 0.9 0.5 0.6 1.1 2.7 6.2 2.4 1.3 1.5 1.6 1.6 1.1 3.0

RANGER Fox 1.1 0.9 0.8 1.2 1.0 0.5 0.7 1.0 2.4 6.5 2.6 1.8 2.0 2.1 1.8 1.2 3.4

RANGER Baker 1951 0.8 0.8 0.7 1.0 0.9 0.5 0.5 1.0 2.6 5.9 2.3 1.0 1.2 1.2 1.5 1.0 2.6

BUSTER Charlie 1.1 0.9 0.8 1.2 1.0 0.5 0.6 0.9 2.1 6.4 2.5 1.8 2.1 2.2 1.7 1.2 2.9

BUSTER Dog 0.8 0.8 0.7 1.0 1.0 0.5 0.6 0.9 2.2 6.3 2.5 1.7 2.0 2.1 1.7 1.0 2.9

BUSTER Easy 0.9 0.8 0.7 1.2 1.0 0.5 0.6 0.9 2.1 6.3 2.5 1.7 2.0 2.1 1.7 1.1 2.9

JANGLE Sugar 1951 0.9 0.8 0.7 0.9 0.8 0.5 0.8 1.6 3.2 6.6 3.2 2.5 2.5 2.4 2.0 0.9 4.1

JANGLE Uncle 0.9 0.8 0.7 0.9 0.8 0.5 0.8 1.6 3.2 6.6 3.2 2.6 2.5 2.4 2.0 0.9 4.1

TUMBLER-SNAPPER Able 1952 1.6 1.0 0.8 1.2 1.0 0.6 0.8 1.1 2.6 8.4 3.6 2.5 2.5 2.4 2.2 1.2 3.9

TUMBLER-SNAPPER 

Baker

1.6 1.0 0.8 1.2 1.0 0.6 0.8 1.1 2.6 8.4 3.6 2.5 2.5 2.4 2.2 1.2 3.9

TUMBLER-SNAPPER 

Charlie

1.2 0.9 0.8 1.2 1.0 0.5 0.7 1.2 2.8 7.9 3.4 2.1 2.1 2.1 2.0 1.2 3.5

TUMBLER-SNAPPER Dog 1.0 0.8 0.7 1.2 1.0 0.5 0.7 1.2 2.8 7.9 3.4 2.1 2.1 2.1 2.0 1.1 3.5

TUMBLER-SNAPPER 

Easy

0.8 0.8 0.7 1.0 0.8 0.5 0.8 1.6 3.4 6.8 3.2 2.2 2.2 2.1 1.9 1.0 4.1

TUMBLER-SNAPPER Fox 0.9 0.8 0.7 1.0 0.8 0.5 0.8 1.6 3.4 6.8 3.2 2.2 2.2 2.1 1.9 1.0 4.1

TUMBLER-SNAPPER 

George

0.9 0.8 0.7 1.0 0.8 0.5 0.8 1.6 3.4 6.8 3.2 2.2 2.2 2.1 1.9 1.0 4.1

TUMBLER-SNAPPER How 0.7 0.7 0.7 1.0 0.8 0.5 0.8 1.6 3.4 6.8 3.2 2.2 2.2 2.1 1.9 1.0 4.1

UPSHOT-KNOTHOLE 

Annie

1953 0.9 0.8 0.7 1.0 0.8 0.5 0.8 1.6 3.3 6.5 3.0 2.1 2.1 2.1 1.9 1.0 4.1

UPSHOT-KNOTHOLE 

Nancy

0.9 0.8 0.7 1.0 0.8 0.5 0.8 1.6 3.3 6.1 2.8 2.0 2.1 2.1 1.8 1.0 4.1

ATTACHMENT C

BETA-PHOTON RATIO ESTIMATES FOR NTS WORK AREAS AND OPERATIONS

Page 9 of 17
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date: 11/09/2012 Page 122 of 135

Event Year

Beta-photon ratio

Hours Days Years

Overall 

averagea

1-day 

reentrya

Onsite 

during 

yeara 0. 1.0 2. 12. 1.0 10. 50. 200. 1.0 2.0 5.0 10. 20. 50.

UPSHOT-KNOTHOLE 

Ruth

0.7 0.7 0.7 1.0 0.8 0.5 0.8 1.5 3.6 7.6 3.6 2.8 2.8 2.6 2.1 1.0 4.0

UPSHOT-KNOTHOLE 

Dixie

1.0 0.9 0.7 1.2 1.0 0.5 0.6 0.9 2.2 7.1 3.2 2.1 2.1 2.1 1.8 1.1 2.9

UPSHOT-KNOTHOLE Ray 1.1 0.9 0.8 1.0 0.8 0.5 0.8 1.5 3.0 7.0 3.3 2.6 2.5 2.4 2.0 1.0 4.0

UPSHOT-KNOTHOLE 

Badger

1.0 0.8 0.7 1.0 0.8 0.5 0.8 1.6 3.4 6.0 2.7 1.9 2.0 2.0 1.8 1.0 4.1

UPSHOT-KNOTHOLE 

Simon

0.9 0.8 0.7 1.0 0.8 0.5 0.8 1.5 3.1 6.5 3.1 2.4 2.4 2.3 1.9 1.0 4.0

UPSHOT-KNOTHOLE 

Encore

1.1 0.9 0.8 1.2 1.0 0.5 0.7 1.2 3.1 6.5 3.1 2.4 2.4 2.3 1.9 1.2 3.5

UPSHOT-KNOTHOLE 

Harry

0.9 0.8 0.7 1.0 0.8 0.5 0.8 1.6 3.3 6.7 3.0 2.2 2.2 2.1 1.9 1.0 4.1

UPSHOT-KNOTHOLE 

Grable

0.8 0.8 0.7 1.2 1.0 0.5 0.8 1.1 2.5 8.3 3.7 2.6 2.5 2.4 2.1 1.1 3.9

UPSHOT-KNOTHOLE 

Climax

0.9 0.8 0.7 1.2 1.0 0.5 0.6 0.9 2.1 7.8 3.5 2.4 2.4 2.3 1.9 1.1 2.9

Maximum 1.7 1.1 0.8 1.2 1.0 0.6 0.8 1.6 3.6 8.4 3.7 2.8 2.8 2.6 2.3 1.2 4.1

a. The average values can be used if a reasonable approach is required or if the employee is not directly identified with an event.

ATTACHMENT C

BETA-PHOTON RATIO ESTIMATES FOR NTS WORK AREAS AND OPERATIONS

Page 10 of 17
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date 11/09/2012 Page 123 of 135

0.1

1

10

100

0.001 0.01 0.1 1 10 100 1000

Elapsed Time - Days

Beta-Photon Ratio

Figure C-4. Calculated beta-photon ratios for immersion in a contaminated 

semi-infinite cloud – maximum of values from 11 atmospheric test 

radionuclide inventories.

For Table 2-2 nuclides not listed in Shleien, Slaback, and Birky (1998, Table 6.2.2), the values of Γ

were calculated using the procedure of Unger and Trubey (1982), as in Table C-4:

1. The photon (gamma and X-ray) decay properties for each nuclide are from the WWW Table of 

Radioactive Isotopes on the Internet at http://ie.lbl.gov/toi/nucSearch.asp (Firestone and 

Ekström 2002).

2. The values of the specific gamma ray dose constant, Γ, were calculated using the following 

equation:

7

2

1 Γ (1 10 )( ) ( ) mSv/hr per MBq 4π

SD E i i R= × ∑ (C-1)

where

R = 100 cm

Si = emission probability of each photon

Ei = energy of each photon (MeV)

D(Ei) = dose rate per unit flux density

The dose rate per unit flux density is determined as follows:

2 3 ln ( ) (ln ) (ln ) (ln ) DE A B E C E F E =+ + + (C-2)

where

If photon energy (MeV) is: = 0.01 to 0.03 0.03 to 0.5 0.5 to 5.0

then A = –20.477 –13.626 –13.133

ATTACHMENT C

BETA-PHOTON RATIO ESTIMATES FOR NTS WORK AREAS AND OPERATIONS

Page 11 of 17
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date 11/09/2012 Page 124 of 135

B = –1.7454 –0.57117 0.72008

C = –1.0954 –0.033603

F = –0.24897

Values of Γ for nuclides not presented in Table 6.2.2 of Shleien, Slaback, and Birky (1998) were 

calculated as above and are listed in Table C-4 in bold.

Table C-4. Photon dose constants, beta dose rates, and beta-photon 

ratios for NTS radionuclides.

Radionuclide 

of concern

Photon dose 

constant, Γa

(mSv/hr per MBq)

Beta dose rate

at 1 mb D(0.07)

(mGy/hr per MBq)

Beta-photon

(Gy/Sv)

Ac-227 2.36E-06 0.0E+00 0.0E+00

Am-241 8.48E-05 0.0E+00 0.0E+00

Am-243 8.46E-05 0.0E+00 0.0E+00

AmBe-241 8.48E-05 N/A N/A

Ba-133 1.23E-04 0.0E+00 0.0E+00

Ba-140 4.45E-05 1.2E-02 2.7E+02

Cd-109 4.98E-05 0.0E+00 0.0E+00

Ce-139 5.55E-05 0.0E+00 0.0E+00

Ce-141 1.98E-05 3.1E-04 1.5E+01

Ce-143 6.89E-05 3.9E-03 5.6E+01

Ce-144 6.30E-06 D(0.07) = 9.2E-03 1.5E+03

D(10) = 1.5E-05 2.4E+00

Cf-252 1.13E-05 0.0E+00 0.0E+00

Cm-244 1.74E-05 0.0E+00 0.0E+00

Co-57 4.09E-05 4.0E-03 9.8E+01

Co-60 3.70E-04 0.0E+00 0.0E+00

Cs-134 2.70E-04 1.4E-03 5.2E+00

Cs-137c 1.03E-04 2.1E-03 2.1E+01

Eu-152 2.01E-04 3.0E-03 1.5E+01

Eu-154 2.04E-04 2.8E-03 1.4E+01

Eu-155 1.80E-05 0.0E+00 0.0E+00

Fe-59 1.79E-04 5.0E-05 2.8E-01

H-3 0.00E+00 0.0E+00 N/A

Hg-203 6.84E-05 0.0E+00 0.0E+00

I-131 7.65E-05 1.2E-03 1.6E+01

I-132 3.86E-04 6.9E-03 1.8E+01

I-133 1.11E-04 6.8E-03 6.1E+01

I-135 2.33E-04 5.7E-03 2.4E+01

Ir-192 1.60E-04 1.2E-03 7.5E+00

Kr-85 4.23E-07 3.5E-03 8.3E+03

Lu-174 3.06E-05 5.6E-03 1.8E+02

Mn-54 1.38E-04 2.8E-03 2.0E+01

Mo-99 3.05E-05 6.3E-03 2.1E+02

Na-22 3.62E-04 8.5E-04 2.3E+00

Na-24 5.24E-04 8.9E-03 1.7E+01

Ni-63 0.00E+00 0.0E+00 N/A

Np-237 1.25E-04 0.0E+00 0.0E+00

Pm-147 7.23E-10 0.0E+00 0.0E+00

ATTACHMENT C

BETA-PHOTON RATIO ESTIMATES FOR NTS WORK AREAS AND OPERATIONS

Page 12 of 17
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date 11/09/2012 Page 125 of 135

Radionuclide 

of concern

Photon dose 

constant, Γa

(mSv/hr per MBq)

Beta dose rate

at 1 mb D(0.07)

(mGy/hr per MBq)

Beta-photon

(Gy/Sv)

Pu-238 2.14E-05 0.0E+00 0.0E+00

Pu-239 8.15E-06 0.0E+00 0.0E+00

Pu-240 2.03E-05 0.0E+00 0.0E+00

Pu-241 0.00E+00 0.0E+00 0.0E+00

Pu-242 1.68E-05 0.0E+00 0.0E+00

PuBe-238 N/A N/A N/A

Ra-226 3.27E-06

RaBe-226 N/A N/A N/A

Rh-100 4.55E-04 4.5E-03 9.9E+00

Rh-101 1.32E-04 2.6E-04 2.0E+00

Rh-102m 4.37E-04 1.1E-03 2.6E+00

Rh-103m 6.91E-06 0.0E+00 0.0E+00

Sb-124 2.88E-04 4.1E-03 1.4E+01

Sb-125 1.03E-04 6.0E-04 5.8E+00

Sn-113 4.84E-05 3.6E-03 7.4E+01

Sr-85 2.05E-04 3.2E-03 1.6E+01

Sr-89 2.21E-08 8.3E-03 3.8E+05

Sr-90/Y-90 0.00E+00 9.7E-03 N/A

Ta-182 2.09E-04 6.6E-04 3.2E+00

Tc-99m 3.32E-05 0.0E+00 0.0E+00

Te-132 7.55E-05 0.0E+00 0.0E+00

Th-228 2.14E-05 0.0E+00 0.0E+00

Th-230 1.86E-05 0.0E+00 0.0E+00

Th-232 1.85E-05 0.0E+00 0.0E+00

Th-234 2.04E-05 0.0E+00 0.0E+00

U-233 7.87E-06 0.0E+00 0.0E+00

U-234 2.10E-05 0.0E+00 0.0E+00

U-235 9.16E-05 0.0E+00 0.0E+00

U-235 chainc 2.39E-04 1.8E-07 7.5E-04

U-238 1.76E-05 0.0E+00 0.0E+00

U-238 chaind 6.24E-05 8.5E-03 1.4E+02

Xe-133 2.78E-05 0.0E+00 0.0E+00

Xe-133m 3.03E-05 0.0E+00 0.0E+00

Xe-135 5.12E-05 4.8E-03 9.4E+01

Y-88 4.82E-04 0.0E+00 0.0E+00

Y-91 5.40E-07 8.3E-03 1.5E+04

Yb-169 8.84E-05 2.6E-03 2.9E+01

Zr-95e 1.26E-4

+

1.30E-4

6.3E-05 2.5E-01

Zr-97e 2.92E-5

+

1.26E-4

1.2E-02 7.7E+01

a. Values of the photon dose constant, Γ, that were calculated using Equation C-1 are listed in

bold.

b. Values of beta dose rate that were calculated using the approximation in Equation C-4 are 

listed in bold.

c. In equilibrium with Th-231. mSv/hr per MBq U = 235.

d. In equilibrium with Th-234, Pa-234m, and U-234. mSv/hr per MBq U = 238.

e. In equilibrium with daughter.

ATTACHMENT C

BETA-PHOTON RATIO ESTIMATES FOR NTS WORK AREAS AND OPERATIONS

Page 13 of 17
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date 11/09/2012 Page 126 of 135

Immersion in a contaminated fallout debris cloud was a less frequent circumstance than exposure to 

fallout after deposition on the ground or other surface [38]. However, exposure of workers to plumes 

of radioactive gas frequently occurred following releases from underground tests such as those during 

post-test drilling before gas blocking or during accidental venting (Johnson et al. 1966; Eberhart 

1994), or could have occurred on tunnel reentry. External beta doses from immersion in a 

contaminated cloud or plume can be calculated by applying a beta-gamma dose ratio to an estimated 

upper bound gamma dose, which is determined from cycle film badge data or dose reconstruction 

[39].

Table C-3 lists time-dependent beta-photon ratios for fission and activation products from detonation 

or time of operation to 50 years. Based on these data, the beta-photon ratios listed in Table C-5 are 

recommended if the exposure occurred from immersion in a contaminated gas cloud (e.g., a 

radioactive gas release during postshot drilling). Because the release of gases is primarily associated 

with weapons detonations at NTS, this geometry is recommended for drillback, reentry, and mineback 

operations. These ratios could be applied on a dosimeter cycle-by-cycle basis if there is indication in 

the claim records that the worker was involved in tunnel reentry or a venting incident.

Table C-5. Beta-photon ratios for exposure from 

immersion in an infinite cloud of mixed fission and 

activation products. 

Elapsed time after

production event

Beta-photon

Sv/Sv

0 to 50 d 1.5

50 to 365 d 5

1 to 5 yr 10

>5 yr 5

If the exposure is known to have resulted from immersion in a plume consisting only of noble gases 

(krypton and xenon) and halides (bromine and iodine), the beta-photon ratios can be determined 

using the information in this attachment.

C.4.2 Beta Dose Rates 

The beta dose rates determined in this section are considered to be those at 0.07 mm. The beta 

decay data for the nuclides listed in Table C-4 are from WWW Table of Radioactive Isotopes on the 

Internet at http://ie.lbl.gov/toi/nucSearch.asp (Firestone and Ekström 2002). For comparison, beta 

dose rates in mGy/hr/MBq were estimated at 1 m. The range of a 350-keV electron in air is about 

1 m. The beta contributions from nuclides with maximum beta energies ≤ 350 keV were ignored and 

the beta dose rate at 1 m was set to 0.

Cross et al. (1982) presents values of dose rates in air for a number of radionuclides of interest at 

NTS. The dose rate values (mGy-cm2 per MBq-hr) are given at various distances, R, from the source 

out to the beta range in air. Where a value of R = 100 is presented, the dose rate at that distance, 

divided by R2 = 104

, is used. In some cases dose rates are given for distances <100 cm and >100 

cm. In those cases, the dose rate at 100 cm was determined by interpolation using the values for 

distances that span 100 cm. The interpolation method that provides the best fit depends on the beta 

spectral characteristics of the particular nuclide. However, for most nuclides, a linear interpolation 

yields the best fit, and, considering the overall errors involved, a linear interpolation can be considered 

adequate.

ATTACHMENT C

BETA-PHOTON RATIO ESTIMATES FOR NTS WORK AREAS AND OPERATIONS

Page 14 of 17
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date 11/09/2012 Page 127 of 135

As an example, in the case of 22Na, dose rates are given at d1 = 97.5 cm and d3 = 105 cm. The dose 

rate at 100 cm is determined as follows:

1 3

1

2 2

2 2 1 () ()

(100) ( )

3 1

2 2

2 (97.5) (105)

(97.5)

2

[(100 )( )]

( )

[(100 97.5)( )]

(105 97.5)

[(2.5)(9.7 6.0)] 9.7

(7.5)

8.5 mGy-cm per MBq-hr

d d

d

d RJ RJ

RJ RJ

d d

RJ RJ

R J

− − ′ ′ ′ ′ = − −

− − ′ ′ = − ′ −

− = −

=

(C-3)

Therefore,

2 2

(100) (100)

4

100

8.5 10 mGy/hr per MBq

D RJ

−

= ÷ ′

= ×

(C-4)

There are several radionuclides listed in Table 2-2 of the site description (ORAUT 2008) and 

Table C-4 for which Cross et al. (1982) does not include dose rate values. In those cases, the 

following approximation (IAEA 1979) was used:

2 D(100) 0.008 mGy/hr nCd− = ×× × (C-5)

where

n = fraction of disintegrations that emit a beta

C = activity in becquerels

d = distance in meters

Because betas with energies ≥350 keV are absorbed in air, the value of D(100) is further reduced by 

the fraction F of emitted betas that are so absorbed. As an approximation, F is determined by:

β

β

β

E 350

F

E

− = (C-6)

where Eβ is the beta energy in keV.

For each radionuclide emitting betas with n different maximum energies, 

β

1 β

n i 350

i i

E

F = E

− = ∑ (C-7)

The nuclides in Table C-4 for which 1-m beta dose rates have been estimated using this 

approximation are indicated in bold.

ATTACHMENT C

BETA-PHOTON RATIO ESTIMATES FOR NTS WORK AREAS AND OPERATIONS

Page 15 of 17
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date 11/09/2012 Page 128 of 135

C.5 SUMMARY OF BETA-PHOTON RATIOS

A summary and comparison of beta-photon ratios for the radionuclide inventory in Table 2-2 of 

ORAUT (2008) is listed in Table C-6. This indicates the sensitivity of the ratios to the assumptions 

made in selecting the exposure geometry.

Table C-6. Geometry-dependent beta-photon ratios for NTS radionuclides 

(Table 2-2). 

Radionuclide 

of concern

Beta-photon ratio (Sv/Sv)

Surface contamination Immersion Point source at 1 m

Ac-227 0.0E+00 2.5E-01 0.0E+00

Am-241 4.0E-04 1.2E-03 0.0E+00

Am-243 9.6E-06 7.9E-04 0.0E+00

AmBe-241 4.0E-04 1.2E-03 N/A

Ba-133 5.9E-05 2.4E-02 0.0E+00

Ba-140 7.8E+00 1.5E+00 2.7E+02

Cd-109 0.0E+00 0.0E+00 0.0E+00

Ce-139 0.0E+00 8.8E-02 0.0E+00

Ce-141 4.7E-01 1.5E+00 1.5E+01

Ce-143 1.0E+01 1.6E+00 5.6E+01

Ce-144a 1.4E+02 1.9E+01 1.5E+03

Cf-252 0.0E+00 3.6E-05 2.4E+00

Cm-244 0.0E+00 0.0E+00 0.0E+00

Co-57 5.0E-05 1.5E-02 0.0E+00

Co-60 3.7E-03 1.8E-02 9.8E+01

Cs-134 2.0E-01 8.9E-02 0.0E+00

Cs-137a 2.1E+02 3.9E+02 5.2E+00

Eu-152 3.3E-01 6.9E-02

Eu-154 1.1E+00 1.9E-01 1.5E+01

Eu-155 0.0E+00 1.1E-01 1.4E+01

Fe-59 1.5E-02 5.9E-02 0.0E+00

H-3 - - 2.8E-01

Hg-203 3.7E-06 1.9E-01 N/A

I-131 4.3E-01 4.1E-01 0.0E+00

I-132 1.9E+00 2.4E-01

I-133 5.4E+00 7.2E-01 1.8E+01

I-135 1.8E+00 2.4E-01

Ir-192 2.6E-01 2.2E-01

Kr-85 2.2E+02 9.4E+01 7.5E+00

Lu-174 6.3E-03 1.8E-03 8.3E+03

Mn-54 3.1E-03 2.8E-04 1.8E+02

Mo-99 2.0E+01 2.7E+00 2.0E+01

Na-22 4.6E-02 7.5E-02 2.3E+00

Na-24 1.5E+00 1.4E-01 1.7E+01

Ni-63 N/A N/A N/A

Np-237 0.0E+00 5.2E-02 0.0E+00

Pm-147 0.0E+00 6.6E+02 0.0E+00

Pu-238 8.8E-04 1.2E-02 0.0E+00

ATTACHMENT C

BETA-PHOTON RATIO ESTIMATES FOR NTS WORK AREAS AND OPERATIONS

Page 16 of 17
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date 11/09/2012 Page 129 of 135

Radionuclide 

of concern

Beta-photon ratio (Sv/Sv)

Surface contamination Immersion Point source at 1 m

Pu-239 3.2E-04 3.5E-02 0.0E+00

Pu-240 1.2E-06 1.2E-02 0.0E+00

Pu-241 0.0E+00 1.2E-01 0.0E+00

Pu-242 0.0E+00 1.5E-02 0.0E+00

PuBe-238 8.8E-04 1.2E-02 N/A

Ra-226 7.9E-05 3.1E-01 -

RaBe-226 7.9E-05 3.1E-01 N/A

Rh-100 2.4E-01 2.7E-02 9.9E+00

Rh-101 5.2E-05 4.2E-02 2.0E+00

Rh-102m 2.8E+00 3.7E-01 2.6E+00

Rh-103m 0.0E+00 0.0E+00 0.0E+00

Rh-106 5.5E+01 8.2E+00 2.2E+02

Ru-103 1.1E-01 6.4E-02 1.3E+00

Ru-106a 5.5E+01 8.2E+00 2.2E+02

Sb-124 1.6E+00 2.2E-01 1.4E+01

Sb-125 1.3E-01 1.2E-01 5.8E+00

Sn-113 0.0E+00 1.2E-02 7.4E+01

Sr-85 5.2E-02 7.7E-03 1.6E+01

Sr-89 1.8E+03 3.8E+02 3.8E+05

Sr-90 2.1E+02 8.4E+02 -

Ta-182 1.3E-01 8.2E-02 3.2E+00

Tc-99m 0.0E+00 4.4E-02 0.0E+00

Te-132 0.0E+00 1.2E-01 0.0E+00

Th-228 0.0E+00 1.4E-01 0.0E+00

Th-230 0.0E+00 6.6E-02 0.0E+00

Th-232 0.0E+00 9.0E-02 0.0E+00

Th-234 0.0E+00 7.1E-01 0.0E+00

U-233 2.6E-05 2.0E-01 0.0E+00

U-234 1.1E-04 7.5E-02 0.0E+00

U-235 3.6E-05 3.0E-02 0.0E+00

U-235 chaina 4.1E-01 2.0E-01 7.5E-04

U-238 0.0E+00 0.0E+00 0.0E+00

U-238 chainb 1.7E+02 3.1E+01 1.4E+02

Xe-133 1.1E-07 1.3E+00 0.0E+00

Xe-133m 0.0E+00 4.5E+00 0.0E+00

Xe-135 6.2E+00 1.3E+00 9.4E+01

Y-88 3.6E-03 5.2E-04 0.0E+00

Y-90 1.3E+03 2.7E+02 1.5E+04

Y-91 8.8E+02 1.3E+02

Yb-169 6.0E-05 1.0E-01 2.5E-01

Zr-95c 2.7E-02 5.0E-02

Zr-97c 2.3E+01 2.6E+00

a. In equilibrium with Th-231. mSv/hr per MBq U-235.

b. In equilibrium with Th-234, Pa-234m, and U-234. mSv/hr per MBq U-238.

c. In equilibrium with daughter.

ATTACHMENT C

BETA-PHOTON RATIO ESTIMATES FOR NTS WORK AREAS AND OPERATIONS

Page 17 of 17
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date 11/09/2012 Page 130 of 135

TABLE OF CONTENTS 

SECTION TITLE PAGE

D.1 Introduction .......................................................................................................................... 130

D.2 Neutron Dose as a Function of Distance.............................................................................. 130

LIST OF FIGURES

FIGURES TITLE PAGE

D-1 Neutron spectra for fission weapon and thermonuclear weapon per kiloton total 

energy yield.......................................................................................................................... 131

D-2 Initial neutron dose per kiloton as a function of slant range from fission and 

thermonuclear air bursts, based on 0.9 normal sea-level air density .................................... 132

D-3 Comparison of the initial neutron dose per kiloton as a function of slant range from 

fission and thermonuclear air bursts..................................................................................... 133

D-4 Neutron dose equivalent as a function of slant range for atmospheric detonations 

with a range of weapon yields .............................................................................................. 133

D-5 1-mrem neutron isodose curves for continental detonations................................................. 135

D.1 INTRODUCTION

One hundred atmospheric nuclear tests were conducted at NTS from January 27, 1951, to July 17, 

1962 (DOE 2000). Essentially all the neutrons accompanying a nuclear explosion are released either 

in the fission or fusion process. All the neutrons from the fusion and more than 99% of the fission 

neutrons are produced almost immediately, within less than a millionth of a second of the initiation of 

the explosion (Glasstone and Dolan 1977). These are referred to as the “prompt” neutrons. In 

addition, somewhat less than 1% of the fission neutrons, called the “delayed” neutrons, are emitted 

subsequently. The majority of these delayed neutrons are released within the first minute, and so 

constitute part of the initial nuclear radiation. While the shielding afforded by even a few feet of soil 

attenuates the neutrons from underground tests by several orders of magnitude, only distance and 

minor attenuation by air contribute to the reduction of neutron dose from atmospheric tests. 

Therefore, in principle, significant occupational neutron exposure from an atmospheric test was 

possible if the person was close enough to the detonation point.

D.2 NEUTRON DOSE AS A FUNCTION OF DISTANCE

The spectra of neutrons produced by fission and thermonuclear weapons are shown in Figure D-1 

(Glasstone and Dolan 1977). The presentation is in terms of total neutrons per energy group per 

kiloton weapon yield. Based on these spectra, the neutron yield for a fission weapon would be about 

7.7 × 1022 neutrons per kT, and for a thermonuclear weapon, the yield would be about 1.4 × 1023

neutrons per kT. The fission neutron spectrum does not change very much to a distance of at least 

1,600 yd. In contrast, for thermonuclear spectra, the relative contribution for neutrons above about 

8 MeV decreases markedly for 400 to 1,600 yd (Glasstone and Dolan 1977).

ATTACHMENT D

NEUTRON DOSES FROM ATMOSPHERIC NUCLEAR TESTS

Page 1 of 6
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date 11/09/2012 Page 131 of 135

Neutron dose as a function of slant range can be determined from the neutron spectra as a function of 

distance (Figure D-2; Glasstone and Dolan 1977) for (1) fission and (2) thermonuclear air bursts. In 

the case of a fission weapon detonation, two curves are presented – offensive and defensive. The 

difference is the result of differences in weapons parameters. The defensive curves were used in 

prediction of slant range doses. The neutron dose as a function of slant range from Figure D-2 is 

compared in Figure D-3, which shows little difference between fission and thermonuclear weapon 

doses per kiloton. Using a radiation weighting factor of 20, the dose equivalent as a function of slant 

range is shown in Figure D-4 for a range of thermonuclear yields from 1 kT to 1 MT. Although the 

yield range is 3 orders of magnitude, the predicted 1-mrem dose equivalent slant range varies from 

about 3.8 km for 1 kT to 5.2 km for 1 MT.

Figure D-1. Neutron spectra for (a) fission weapon and (b) thermonuclear weapon 

per kiloton total energy yield (Glasstone and Dolan 1977). 

ATTACHMENT D

NEUTRON DOSES FROM ATMOSPHERIC NUCLEAR TESTS

Page 2 of 6
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date 11/09/2012 Page 132 of 135

Figure D-2. Initial neutron dose per kiloton as a function of slant range from 

(a) fission and (b) thermonuclear air bursts, based on 0.9 normal sea-level air 

density (Glasstone and Dolan 1977).

The Defense Nuclear Agency published a report on neutron exposure to U.S. Department of Defense 

(DOD) personnel during nuclear tests (Goetz et al. 1985). The results were based on radiation 

transport calculations using the ATR (Air Transport of Radiation) computer code. 

Calculations were made for NTS and Pacific Proving Ground tests, taking into account differences in 

test conditions such as altitude (air density) and humidity. The results of these calculations are 

presented in terms of dose equivalent using a quality factor of 13.

The isodose contours for the continental detonations are shown in Figure D-5 (Goetz et al. 1985). 

Two tests were used to bound the range of possible results:

ATTACHMENT D

NEUTRON DOSES FROM ATMOSPHERIC NUCLEAR TESTS

Page 3 of 5

ATTACHMENT D

NEUTRON DOSES FROM ATMOSPHERIC NUCLEAR TESTS

Page 3 of 6
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date 11/09/2012 Page 133 of 135

Figure D-3. Comparison of the initial neutron dose per kiloton as a function of slant 

range from fission and thermonuclear air bursts (after Glasstone and Dolan 1977).

Figure D-4. Neutron dose equivalent as a function of slant range for atmospheric 

detonations with a range of weapon yields.

• Operation RANGER, Test EASY: The neutron spectrum was that of a pure fission device, 

typical of earlier weapons. The mean air density between the surface and burst height 

(330 m) is 1.18 × 10-3 g/cm3

, a very high value for NTS tests. The higher density and resulting 

increase in attenuation results in a lower neutron dose at a given range.

ATTACHMENT D

NEUTRON DOSES FROM ATMOSPHERIC NUCLEAR TESTS

Page 4 of 6
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date 11/09/2012 Page 134 of 135

• Operation PLUMBOB, Test HOOD: The neutron source spectrum was that of a boosted 

fission device, typical of later weapons. The mean air density between the surface and burst 

height (460 m) is 0.992 × 10-3 g/cm3, the lowest of any low-altitude NTS detonation. The low 

air density and enhanced neutron output result in a higher neutron dose at a given range.

Using the data in Figure D-5, the 1-mrem slant range at the surface for 1- to 80-kT detonations from 

about 4.1 km to 5 km are consistent with the data from Glasstone and Dolan (1977). Therefore, it can 

reasonably be concluded that personnel working on the surface beyond a range of 6 km would 

receive neutron doses less than 1 mrem for the NTS atmospheric test series.

The isodose curves in Figure D-5 provide information on neutron doses at altitudes as high as 9 km 

(29,500 ft). This information is useful in assessing potential neutron exposures to air crews 

associated with the testing program.

ATTACHMENT D

NEUTRON DOSES FROM ATMOSPHERIC NUCLEAR TESTS

Page 5 of 6
Document No. ORAUT-TKBS-0008-6 Revision No. 03 Effective Date 11/09/2012 Page 135 of 135

Figure D-5. 1-mrem neutron isodose curves for continental detonations (Goetz et al. 1985).

ATTACHMENT D

NEUTRON DOSES FROM ATMOSPHERIC NUCLEAR TESTS

Page 6 of 6