Approvals: 1st Level 2nd Level 3rd Level 

National Institute for Occupational Safety and Health 

National Personal Protective Technology Laboratory 

P.O. Box 18070 

Pittsburgh, PA 15236 

Procedure No. RCT-ASR-STP-0148A Revision: 1.1 Date: 12 September 2005

DETERMINATION OF REMOTE GAUGE LEAK FLOW TEST FOR CLOSED-CIRCUIT, DEMAND 

AND PRESSURE-DEMAND, SELF-CONTAINED BREATHING APPARATUS 

STANDARD TESTING PROCEDURE (STP) 

1. PURPOSE

This test establishes the standard procedure for ensuring that the level of protection provided by 

the remote gauge leak flow test requirements on Closed-Circuit, Demand and Pressure-Demand, 

Self-Contained Breathing Apparatus (SCBA) submitted for Approval, Extension of Approval, or 

examined during Post-Certification Product Audits meet the certification requirements set forth in 

42 CFR, Part 84, Subpart G, Section 84.63(a)(c)(d), and Subpart H, 84.82(f)(g); Volume 60, 

Number 110, June 8, 1995. 

2. GENERAL

This STP describes the Determination of Remote Gauge Leak Flow Test for Closed-Circuit, 

Demand and Pressure-Demand, Self-Contained Breathing Apparatus test in sufficient detail that a 

person knowledgeable in the appropriate technical field can select equipment with the necessary 

resolution, conduct the test, and determine whether or not the product passes the test. 

3. EQUIPMENT/MATERIALS

3.1. The list of necessary test equipment and materials follows: 

3.1.1. ISI Anthropometric Test heads with tube for measuring breathing resistance and 

air flows - Model SR-085 or equivalent.

Uncontrolled Copy 

 Uncontrolled Copy 

Uncontrolled Copy 

 Uncontrolled Copy 

Uncontrolled Copy 

 Uncontrolled Copy 

Uncontrolled Copy 

 Uncontrolled Copy 
Procedure No. RCT-ASR-STP-0148A Revision: 1.1 Date: 12 September 2005 Page 2 of 7 

3.1.2. Electric timer, calibrated to hundredths of a minute (Precision Scientific 

Company) or equivalent. 

3.1.3. Singer Dry Test Meter (Model 806) one cubic foot per revolution or equivalent. 

4. TESTING REQUIREMENTS AND CONDITIONS

4.1. Prior to beginning any testing, all measuring equipment to be used must have been 

calibrated in accordance with the manufacturer's calibration procedure and schedule. At 

a minimum, all measuring equipment utilized for this testing must have been calibrated 

within the preceding 12 months using a method traceable to the National Institute of 

Standards and Technology (NIST). 

4.2. The compressed gas cylinder must meet all applicable Department of Transportation 

requirements for cylinder approval as well as for retesting/requalification. 

4.3. Normal laboratory safety practices must be observed. This includes all safety precautions 

described in the current ALOSH Facility Laboratory Safety Manual. 

4.3.1. Safety glasses, lab coats, and hard-toe shoes must be worn during all testing. 

4.3.2. Work benches must be maintained free of clutter and non-essential test 

equipment. 

4.3.3. When handling any glass laboratory equipment, lab technicians and personnel 

must wear special gloves which protect against lacerations or punctures. 

5. PROCEDURE

Note: Reference Section 3 for equipment, model numbers and manufacturers. For calibration 

purposes, use those described in the manufacturer's operation and maintenance manuals. 

Uncontrolled Copy 

 Uncontrolled Copy 

Uncontrolled Copy 

 Uncontrolled Copy 

Uncontrolled Copy 

 Uncontrolled Copy 

Uncontrolled Copy 

 Uncontrolled Copy 
Procedure No. RCT-ASR-STP-0148A Revision: 1.1 Date: 12 September 2005 Page 3 of 7 

5.1. If unit has a shut off valve for gauge and gauge line, charge cylinder to 1000 pounds per 

square inch. If unit does not have a shut off valve, charge cylinder to 25% of its rated 

service pressure. 

5.2. Remove remote pressure gauge from unit. 

5.3. Attach gauge line to dry test meter. 

5.4. Mount facepiece on anthropometric head and make sure facepiece is sealed. 

5.5. Block off all ports exiting anthropometric test head. 

5.6. Place unit in use mode (i.e., demand, Pressure demand). 

5.7. Open cylinder valve and simultaneously start timer. 

5.8. When cylinder is completely drained stop timer. 

5.9. Record number of revolutions on dry test meter and time on timer. 

5.10. Refer to data analysis. 

5.11. Data Analysis 

Note: One revolution on dry test meter is equal to 28.32 liters per minute. Take number of 

revolutions recorded in Step 5.9. and multiply by 28.32. Divide this number by the time 

also recorded in Step 5.9. 

Note: This test should be done on a minimum of two respirators or more if additional testing is 

required (42 CFR, Part 84, Sections 84.12, 84.30, and 84.60). 

6. PASS\FAIL CRITERIA

6.1. The criterion for passing this test is set forth in 42 CFR, Part 84, Subpart G, Section 

84.63(a)(c)(d), and Subpart H, 84.82(f)(g); Volume 60, Number 110, June 8, 1995. 

6.2. This test establishes the standard procedure for ensuring that: 

 84.63 Test requirements; general. 

(a) Each respirator and respirator component shall when tested by the applicant and by 

the Institute, meet the applicable requirements set forth in subparts H through L of this 

part. 

(c) In addition to the minimum requirements set forth in subparts H through L of this 

part, the Institute reserves the right to require, as a further condition of approval, any 

additional requirements deemed necessary to establish the quality, effectiveness, and 

Uncontrolled Copy 

 Uncontrolled Copy 

Uncontrolled Copy 

 Uncontrolled Copy 

Uncontrolled Copy 

 Uncontrolled Copy 

Uncontrolled Copy 

 Uncontrolled Copy 
Procedure No. RCT-ASR-STP-0148A Revision: 1.1 Date: 12 September 2005 Page 4 of 7 

safety of any respirator used as protection against hazardous atmospheres. 

(d) Where it is determined after receipt of an application that additional requirements will 

be required for approval, the Institute will notify the applicant in writing of these 

additional requirements, and necessary examinations, inspections, or tests, stating 

generally the reasons for such requirements, examinations, inspections, or tests. 

84.82 Gas pressure gages; minimum requirements. 

(f) The loss of gas through a broken gage or severed gage connection shall not exceed 70 

liters per minute when the cylinder pressure is 6,900 kN/m.2

 (1,000 pounds per square 

inch gage) or when the liquid level is at one-half. 

(g) Where gages are connected to the apparatus through a gage line, the gage and line 

shall be capable of being isolated from the apparatus except where the failure of the gage 

or line would not impair the performance or service life of the apparatus. 

6.3. To determine impairment of approved self-contained breathing apparatus performance 

and service life, as the result of failure of a gage or gage line without shut-off valve, as 

required by 84.82(g), MSHA and NIOSH require that the remaining service life indicator 

of the apparatus be set to operate at 25 percent of rated service time and that the 

following formula be used to determine the maximum allowable leakage of gas through 

the gauge or gauge line. 

Liters of O2 in Cylinder at full pressure: - (1.75 Liters x rated service time in minutes) = 

X Liters. X/20% of rated service time = Lpm Allowable Leakage 

7. RECORDS\TEST SHEETS

7.1. All test data will be recorded on the LOSS OF GAS THROUGH GAUGE, CLOSEDCIRCUIT, SELF-CONTAINED BREATHING APPARATUS test data sheet. 

7.2. All videotapes and photographs of the actual test being performed, or of the tested 

equipment shall be maintained in the task file as part of the permanent record. 

7.3. All equipment failing any portion of this test will be handled as follows; 

7.3.1. If the failure occurs on a new certification application, or extension of approval 

application, send a test report to the RCT Leader and prepare the hardware for 

return to the manufacturer. 

7.3.2. If the failure occurs on hardware examined under an Off-the-Shelf Audit the 

hardware will be examined by a technician and the RCT Leader for cause. All 

equipment failing any portion of this test may be sent to the manufacturer for 

examination and then returned to NIOSH. However, the hardware tested shall be 

held at the testing laboratory until authorized for release by the RCT Leader, or 

his designee, following the standard operating procedures outlined in Procedure 

Uncontrolled Copy 

 Uncontrolled Copy 

Uncontrolled Copy 

 Uncontrolled Copy 

Uncontrolled Copy 

 Uncontrolled Copy 

Uncontrolled Copy 

 Uncontrolled Copy 
Procedure No. RCT-ASR-STP-0148A Revision: 1.1 Date: 12 September 2005 Page 5 of 7 

for Scheduling, and Processing Post-Certification Product Audits, RB-SOP0005-00. Uncontrolled Copy 

 Uncontrolled Copy 

Uncontrolled Copy 

 Uncontrolled Copy 

Uncontrolled Copy 

 Uncontrolled Copy 

Uncontrolled Copy 

 Uncontrolled Copy 
Procedure No. RCT-ASR-STP-0148A Revision: 1.1 Date: 12 September 2005 Page 6 of 7 

LOSS OF GAS THROUGH GAUGE, CLOSED-CIRCUIT, 

SELF-CONTAINED BREATHING APPARATUS

Project No : Date: __________ 

Company : ___________________________________________ 

Respirator Type : ___________________________________________ 

Reference: 42 CFR, Part 84, Subpart H, 84.82(f)(g). 

Requirement: The loss of gas through a broken gage or severed gage connection shall not exceed 70 

liters per minute when the cylinder pressure is 6,900 kN/m.2

 (1,000 pounds per square 

inch gage) or when the liquid level is at one-half. 

Where gages are connected to the apparatus through a gage line, the gage and line shall 

be capable of being isolated from the apparatus except where the failure of the gage or 

line would not impair the performance or service life of the apparatus. 

To determine impairment of approved self-contained breathing apparatus performance 

and service life, as the result of failure of a gage or gage line without shut-off valve, as 

required by 84.82(g), MSHA and NIOSH require that the remaining service life indicator 

of the apparatus be set to operate at 25 percent of rated service time and that the 

following formula be used to determine the maximum allowable leakage of gas through 

the gauge or gauge line. 

 Liters of O2 in 

 Cylinder at full pressure: - (1.75 Liters x rated service time in minutes) = X Liters. 

 X/20% of rated service time = Lpm Allowable Leakage Shall not exceed 70 lpm. 

Results: 

Event No. Cylinder Pressure-psig Flow/LPM 

 1. ___________________ __________ 

 2. ___________________ __________ 

 3. ___________________ __________ 

 4. ___________________ __________ 

 5. ___________________ __________ 

Comments: 

_____________________________________________________________________________________

_____________________________________________________________________________________ 

Test Engineer: PASS FAIL________ 

Uncontrolled Copy 

 Uncontrolled Copy 

Uncontrolled Copy 

 Uncontrolled Copy 

Uncontrolled Copy 

 Uncontrolled Copy 

Uncontrolled Copy 

 Uncontrolled Copy 
Procedure No. RCT-ASR-STP-0148A Revision: 1.1 Date: 12 September 2005 Page 7 of 7 

Revision History 

Revision Date Reason for Revision 

1.0 2 February 2001 Historic document 

1.1 12 September 

2005 

Update header and format to reflect lab move from Morgantown, WV 

No changes to method 

 

Uncontrolled Copy 

 Uncontrolled Copy 

Uncontrolled Copy 

 Uncontrolled Copy 

Uncontrolled Copy 

 Uncontrolled Copy 

Uncontrolled Copy 

 Uncontrolled Copy