Evaluation of Decontaminated N95 Respirators 

Date Tested: 7/10/2020 – 7/16/2020

Respirator Model(s): 3M 9205+ 

Tests: Filtration with NaCl (modified version of STP-0059), Manikin Fit Factor with Static Advanced Headform, and Strap 

Integrity with Tensile Testing 

Decontamination Method: VPHP 

Decontamination Cycles: 5, 10 & 20 cycles 

While decontamination and reuse of FFRs are not consistent with standard and approved usage, these options 

may need to be considered when FFR shortages exist. This assessment was developed to quantify the filtration 

efficiency and manikin fit factor1 of an N95 respirator that has been decontaminated. This assessment is not to 

determine the effectiveness of the decontamination procedure at killing pathogenic microorganisms. The results 

provided in this report are specific to the subset of samples that were provided to NPPTL for evaluation. These 

results may be used to update the CDC guidance for Crisis Capacity Strategies (during known shortages). 

53 respirators that were unworn and not subjected to any pathogenic microorganisms were submitted for 

evaluation. This included 15 respirators that were subjected to 5 cycles of the VPHP decontamination process, 

15 respirators subjected to 10 cycles, 15 respirators subjected to 20 cycles, and an additional 8 respirators that 

served as controls. Figure 1 photos document the procedures used. The samples were tested using a modified 

version of the NIOSH Standard Test Procedure (STP) TEB-APR-STP-0059 to determine particulate filtration 

efficiency. The TSI, Inc. model 8130 using sodium chloride aerosol was used for the filtration evaluation. For the 

laboratory fit evaluation, a static manikin headform was used to quantify changes in manikin fit factor. The TSI, 

Inc. PortaCount® PRO+ 8038 in “N95 Enabled” mode was used for this evaluation. Additionally, tensile strength 

testing of the straps was performed to determine changes in strap integrity. The Instron® 5943 Tensile Tester 

was used for this evaluation. The full assessment plan can be found here. 

Filtration Efficiency Results: All respirators measured more than 95%. See Table 1. 

Manikin Fit Factor Results: The manikin fit factor showed passing fit factors (greater than 100) for all respirators 

evaluated. See Table 2. 

Strap Integrity Results: No visual degradation of the straps was observed. Decreases in recorded force were 

shown for both the top and bottom straps at 5, 10, and 20 cycles. See Table 3. 

1The American Industrial Hygiene Association defines the Manikin Fit Factor as “An expression related to the amount of leakage measured through the 

face or neck seal of a respirator mounted to a manikin under specified airflow and environmental conditions. If the challenge to the seal is an airborne 

substance, it is the ratio of its airborne concentration outside the respirator divided by the concentration that enters the respirator through the seal. If the 

challenge is airflow or air pressure, conditions and assumptions for quantifying leakage must be specified. Leakage from other sources (e.g., air purifying 

elements) must be essentially zero. The respirator may be mounted to the manikin without sealants; be partially sealed to the manikin; or be sealed to the 

manikin with artificially induced leaks.” 
Figure 1. Laboratory Test Photos
Table 1. Filter Efficiency Evaluation

Notes:

• The test method utilized in this assessment is not the NIOSH standard test procedure that is used for certification 

of respirators. Respirators assessed to this modified test plan do not necessarily meet the requirements of STP0059, and therefore cannot be considered equivalent to N95 respirators that were tested to STP-0059. 

Respirator Model, 

Decon Method, # of 

cycles

Treated 

Sample #

Flow Rate

(Lpm)

Initial Filter 

Resistance

(mmH2O)

Initial Percent 

Leakage (%)

Maximum 

Percent 

Leakage (%)

Filter 

Efficiency 

(%)

3M 9205+, VPHP,

5 cycles

Min Fil Eff: 98.11%

Max Fil Eff: 99.86%

1 85 7.0 0.920 1.02 98.98

2 85 6.8 1.83 1.89 98.11

3 85 6.7 0.124 0.209 99.79

4 85 6.7 0.066 0.144 99.86

5 85 6.6 0.214 0.265 99.74

6 85 7.1 0.546 0.615 99.39

7 85 6.6 0.211 0.274 99.73

8 85 6.9 0.436 0.494 99.51

9 85 6.9 1.72 1.89 98.11

10 85 6.8 0.435 0.552 99.45

Control 1 85 6.9 0.063 0.201 99.80

3M 9205+, VPHP,

10 cycles

Min Fil Eff: 97.77%

Max Fil Eff: 99.78%

1 85 6.8 0.105 0.236 99.76

2 85 7.1 0.723 0.726 99.27

3 85 6.7 0.990 1.07 98.93

4 85 6.4 2.09 2.23 97.77

5 85 6.5 0.969 1.02 98.98

6 85 6.9 0.573 0.704 99.30

7 85 6.7 0.444 0.599 99.40

8 85 6.9 0.428 0.520 99.48

9 85 7.0 0.170 0.224 99.78

10 85 6.5 1.37 1.42 98.58

Control 2 85 7.3 1.26 1.37 98.63

3M 9205+, VPHP,

20 cycles

Min Fil Eff: 96.13%

Max Fil Eff: 99.69%

1 85 6.3 2.76 3.08 96.92

2 85 5.8 2.50 2.59 97.41

3 85 6.5 1.50 1.57 98.43

4 85 6.3 1.52 1.63 98.37

5 85 6.1 1.31 1.38 98.62

6 85 6.8 0.277 0.315 99.69

7 85 6.3 1.61 1.71 98.29

8 85 5.9 1.71 1.71 98.29

9 85 6.6 1.69 1.83 98.17

10 85 5.8 3.49 3.87 96.13

Control 1 85 6.5 0.795 0.905 99.10

Control 2 85 6.7 0.065 0.179 99.82
Table 2. Manikin Fit Evaluation

Manikin Fit Factor of Decontaminated N95s

Respirator 

Model, Decon 

Method, # of 

cycles

Treated

Sample #

mFF Normal 

Breathing 1

mFF Deep 

Breathing

mFF Normal 

Breathing 2

Overall 

Manikin Fit 

Factor

3M 9205+, VPHP,

5 cycles 

Static Advanced 

Medium 

Headform 

(Hanson Robotics)

11 200+ 200+ 200+ 200+

12 200+ 169 200+ 189

13 200+ 200+ 200+ 200+

14 200+ 200+ 200+ 200+

15 200+ 200+ 200+ 200+

Control 3

200+ 200+ 200+ 200+

3M 9205+, VPHP,

10 cycles

Static Advanced 

Medium 

Headform 

(Hanson Robotics)

11 200+ 200+ 200+ 200+

12 200+ 200+ 200+ 200+

13 200+ 200+ 200+ 200+

14 200+ 200+ 200+ 200+

15 200+ 200+ 200+ 200+

Control 4

200+ 200+ 200+ 200+

3M 9205+, VPHP,

20 cycles

Static Advanced 

Medium 

Headform 

(Hanson Robotics)

11 175 166 165 169

12 200+ 200+ 200+ 200+

13 200+ 200+ 200+ 200+

14 200+ 200+ 200+ 200+

15 200+ 200+ 200+ 200+

Control 3

200+ 200+ 200+ 200+

Control 4 200+ 200+ 200+ 200+

Notes:

• Per OSHA 1910.134(f)(7), if the fit factor as determined through an OSHA-accepted quantitative fit testing protocol 

is equal to or greater than 100 for tight-fitting half facepieces, then the fit test has been passed for that respirator.

• This assessment does not include fit testing of people and only uses two exercises (normal and deep breathing) on 

a manikin headform. 

• This assessment is a laboratory evaluation using a manikin headform and varies greatly from the OSHA individual 

fit test. This headform testing only includes normal breathing and deep breathing on a stationary (non-moving) 

headform; therefore, fit results from this assessment cannot be directly translated to using the standard OSHAaccepted test. Instead, this testing provides an indication of the change in fit performance (if any) associated with 

the decontamination of respirators. 
Table 3. Strap Integrity Evaluation

Tensile Force in Respirator Straps of Decontaminated N95s

(recorded force values are at 150% strain)

Respirator Model, Decon 

Method, # of cycles Straps from Treated Sample # Force in Top Strap (N)

Force in Bottom 

Strap (N)

3M 9205+, VPHP,

5 cycles 

1 1.534 1.511

2 1.541 1.613

3 1.582 1.579

Decontaminated Strap 

Average 1.552 1.568

Control 1 1.668 1.995

Control Strap Average 1.731 1.948

% Change

((Deconned - Controls) / 

Controls)

-10.34% -19.51%

3M 9205+, VPHP,

10 cycles 

1 1.492 1.678

2 1.534 1.620

3 1.438 1.479

Decontaminated Strap 

Average 1.488 1.592

Control 2 1.793 1.900

Control Strap Average 1.731 1.948

% Change

((Deconned - Controls) / 

Controls)

-14.04% -18.28%

3M 9205+, VPHP,

20 cycles 

1 1.377 1.611

2 1.520 1.496

3 1.433 1.590

Decontaminated Strap 

Average 1.443 1.566

Control 1 1.744 1.919

Control 2 1.807 1.961

Control Strap Average 1.776 1.94

% Change

((Deconned - Controls) / 

Controls)

-18.75% -19.28%