Skip directly to site content Skip directly to search


Español  Other Languages


Centers for Disease Control and Prevention. CDC twenty four seven. Saving Lives, Protecting People
 Search
		×
			NIOSH
			NIOSH
				All CDC
				Submit
			For a full list of topics:A-Z Index
			Advanced Search
		Advanced Search


The National Personal Protective Technology Laboratory (NPPTL)








NIOSH Conformity Assessment Interpretation Notice										Related Pages
					**Subject:** NIOSH Respirator Approval Program’s use of the NIOSH Bivariate Panel during Laboratory Respirator Protection Level Testing of Chemical, Biological, Radiological, and Nuclear (CBRN) Self-Contained Breathing Apparatus, CBRN Air-Purifying Respirators, and CBRN Powered Air-Purifying Respirators beginning November 15, 2021NIOSH CA 2021-1039
November 2021
1 – SUMMARY
In 2019, NIOSH began phasing out the use of the Los Alamos National Laboratory (LANL) Panels with the publication of NIOSH CA 2019-1011, NIOSH Respirator Approval Program’s Use of the NIOSH Bivariate Panel during Isoamyl Acetate Fit Testing, and NIOSH has further updated the following Standard Testing Procedures (STPs) to include use of the NIOSH Bivariate Panel (NIOSH Panel):
TEB-CBRN-APR-STP-0352: Determination of Laboratory Respirator Protection Level (LRPL) Values for CBRN Self-Contained Breathing Apparatus (SCBA) Facepieces or CBRN Air-Purifying Respirator (APR).
TEB-CBRN-APR-STP-0552: Determination of Laboratory Respirator Protection Level (LRPL) Values for CBRN Tight-Fitting Powered Air-Purifying Respirator (PAPR).
TEB-CBRN-APR-STP-0553: Determination of Laboratory Respiratory Protection Level (LRPL) Values for CBRN Loose-Fitting Powered Air-Purifying Respirator (PAPR).NIOSH informed approval holders regarding implementation of the NIOSH Panel during a Manufacturers Meeting on October 17, 2018. The implementation was delayed due to program workload, the NFPA 1981:2018 update, and the COVID-19 Response.
Implementation of the NIOSH Panel for CBRN air-purifying escape respirators (APERs) will be forthcoming.
The NIOSH Panel’s anthropometric limits and individual panel cells (1-10) are shown in Figure 1. The population cell distribution of the NIOSH Panel is given in Table 1, based on a 25-member panel. The updated STPs document how the NIOSH Panel will be implemented for use with 25- , 29- , or 38-member panels based on the number of facepiece sizes. This information is included in Tables 2, 3, and 4, respectively. If an approval holder or applicant submits a respirator design to fit a sub-population of users, the approval holder or applicant shall provide NIOSH with information on the expected size distribution based on the NIOSH Panel (e.g., panel cell numbers, face dimensions). This information will be considered during initial engineering review for assigning and conducting the NIOSH LRPL testing.
Beginning November 15, 2021, NIOSH will implement use of the NIOSH Panel during evaluations of all new approvals of CBRN SCBAs, CBRN APRs, and CBRN PAPRs (tight- and loose-fitting). These types of approval projects, accepted in the NIOSH data management system on or after November 15, 2021, will be evaluated using the NIOSH Panel and the relevant, updated STP. A new approval may include a respirator facepiece configuration previously approved as part of a complete and different respirator approval configuration and will be tested using the NIOSH Panel. Applicants may continue to use the LANL panels to generate pre-submission test data; however, NIOSH recommends using the procedure outlined in the updated STP.
Respirators previously approved following testing with LANL Panels continue to be NIOSH-approved respirators. When these approvals are submitted for an extension of approval and the application requires human subject testing, the respirator will be evaluated using the NIOSH Panel.
Certified product audit testing including the use of the anthropometric panels will allow for the use of the panel used during the original testing for the approval until June 30, 2022. For example, if the full-facepiece LANL panel was used during the original testing, the full-facepiece LANL panel will be used for certified product audit testing until June 30, 2022.
Figure 1. The NIOSH Bivariate Panel (NIOSH Panel) includes ten individual member cells defined by face width and face length (mm).**Figure 1. The NIOSH Bivariate Panel (NIOSH Panel) includes ten individual member cells defined by face width and face length (mm).**
**Table 1: NIOSH Panel cell designations, population distribution, and number of test subjects per cell, based on a 25-member panel [Zhuang 2007].**
Table 1: NIOSH Panel cell designations, population distribution, and number of test subjects per cell, based on an 18-member panel
NIOSH Panel – cell number
Population distribution (%)
Number of test subjects based
on the population distribution1
5.5
22
5.3
23
10.5
24
25.0
55
7.1
26
5.7
27
21.3
48
8.7
29
5.2
210
3.5
2Total number of subjects
25
**Table 2. CBRN APR, SCBA, or PAPR 25-member panel for LRPL testing of a respirator designed and manufactured in one unique size.**
Table 1: NIOSH Panel cell designations, population distribution, and number of test subjects per cell, based on an 18-member panel
25-Member PanelNIOSH Panel – cell number
Number of test subjects1
2 subjects
An additional **six subjects** will be selected from cells **2, 3, 4, 7, 8**
with the intent to consider the 25-member population
distribution (as given in the Table 1, above). No more
than four subjects are permitted from any
one panel cell.2
2 subjects3
2 subjects4
2 subjects5
2 subjects6
2 subjects7
2 subjects8
2 subjects9
2 subjects10
1 subject
**Table 3. CBRN APR, SCBA, or PAPR 29-member panel for LRPL testing of a respirator designed and manufactured in two unique sizes.**
Table 1: NIOSH Panel cell designations, population distribution, and number of test subjects per cell, based on an 18-member panel
29-Member PanelNIOSH Panel – cell number
Number of test subjects1
**14-15 subjects** from cells **1-6** (at least
one from each cell and no more
than four from any one cell for
a total of 14-15 subjects)**Smaller size**2345
**14-15 subjects** from cells **5-10** (at least
one from each cell and no more
than four from any one cell for
a total of 14-15 subjects)**Larger size**678910
**Table 4. CBRN APR, SCBA, or PAPR 38-member panel for LRPL testing of a respirator designed and manufactured in three unique sizes. The STPs include information for testing respirator designs that are designed and manufactured in more than three unique sizes.**
Table 1: NIOSH Panel cell designations, population distribution, and number of test subjects per cell, based on an 18-member panel
38-Member PanelNIOSH Panel – cell number
Number of test subjects1
**10 subjects** from cells **1-4**
(at least one from each cell
and no more than four
from any one cell for a total
of 10 subjects)**Smaller Size**23
**17 subjects** from cells **3-8**
(at least one from each cell
and no more than four
from any one cell for a total
of 17 subjects)**Medium Size**4567
**11 subjects** from cells **7-10**
(at least one from each cell
and no more than four
from any one cell for a
total of 11 subjects)**Larger Size**89102 – AUTHORITY
42 C.F.R. Part 84, Approval of Respiratory Protective Devices
Statement of Standard for Chemical, Biological, Radiological, and Nuclear (CBRN) Full Facepiece Air Purifying Respirator (APR) (cdc.gov)
Statement of Standard for Self-Contained Breathing Apparatus (SCBA) with Chemical, Biological, Radiological, and Nuclear (CBRN) Protection (cdc.gov)
Statement of Standard for Chemical, Biological, Radiological, and Nuclear (CBRN) Powered Air-Purifying Respirators (PAPR) (cdc.gov)
NIOSH Standard Testing Procedures
Revised TEB-CBRN-APR-STP-0352 (Revision 2.0, Dated 28 September 2021)
Revised TEB-CBRN-APR-STP-0552 (Revision 2.0, Dated 20 October 2021)
Revised TEB-CBRN-APR-STP-0553 (Revision 2.0, Dated 20 October 2021)3 – REFERENCES
Approval of Respiratory Protective Devices, 42 C.F.R. Part 84
Hack, A., McConville, J.: Respirator Protection Factors: Part 1-Development of an Anthropometric Test Panel. Am. Ind. Hyg. Assoc. J (39): 970-975 (1978).
Zhuang, Z., Bradmiller, B., Shaffer, R.: New Respirator Fit Test Panels Representing the Current U.S. Civilian Work Force. J. Occup. Environ. Hyg. 4: 647-659 (2007).
Federal Register: Development of Inward Leakage Standards for Half-Mask Air-Purifying Particulate Respirators. Request for comment and notice of public meeting.
Transitioning to the NIOSH Panel presented at the October 17, 2018 Manufacturers Meeting: NIOSH Conformity Assessment Notice (CA 2018-1007) | NPPTL | NIOSH | CDC
NIOSH CA 2019 -1011 NIOSH Respirator Approval Program’s use of the NIOSH Bivariate Panel during Isoamyl Acetate Fit Testing beginning February 1, 2019.Appendix A – Background and Supplemental Information
Prior to November 1, 2021, the NIOSH Respirator Approval Program used human test subjects, identified by the LANL panels (Figures A1 and A2), to assess fit in accordance with 42 Code of Federal Regulations Part 84 (42 CFR Part 84).
Figure A1. Los Alamos Natio nal Laboratory Full Facepiece Panel, with 10 cell boxes identified.**Figure A1. Los Alamos Natio nal Laboratory Full Facepiece Panel, with 10 cell boxes identified.**

Figure A2. Los Alamos National Laboratory Half Facepiece Panel, with 10 cell boxes identified.**Figure A2. Los Alamos National Laboratory Half Facepiece Panel, with 10 cell boxes identified.**

**Note:** While working to provide this notice and update STP-0352, NIOSH marked the historical reference to Attachment C of the SCBA Statement of Standard (provided to stakeholders on November 17, 2001) as superseded. Attachment C was superseded when STP-0352 was finalized as Revision 1.0 on June 10, 2008.NIOSH Respirator Approval Program’s use of the NIOSH Bivariate Panel during Laboratory Respirator Protection Level Testing of Chemical, Biological, Radiological, and Nuclear (CBRN) Self-Contained Breathing Apparatus, CBRN Air-Purifying Respirators, and CBRN Powered Air-Purifying Respirators beginning November 15, 2021 [PDF – 436 KB]
							
					
							Last Reviewed:  November 1, 2021						Source: 	National Institute for Occupational Safety and Health
					Facebook
        Twitter
        LinkedIn
        Syndicate homeNPPTL
What’s New on the NPPTL Websiteplus icon
NPPTL What’s New Archives
A to Z Index
Respiratory Protection Videos
Resources for Mine Workers
Resources for Emergency Responders
Respirator Approval Programplus icon
Respirator Approval Information
Respirator Standards
Respirator User Notices
Resources and Technical Support
Post-market Evaluationsplus icon
PPE CASE Reports
PPE CASE Notes
Respirator Assessments to Support COVID-19 Responseplus icon
Beyond Shelf Life/Stockpiled Assessment Results
International Assessment Results
Decontaminated Assessment Results
Non-NIOSH Approved Innovative Filtering Facepiece Respirator Assessment Results
Protective Clothing and Ensembles
Personal Protective Technology Program
NIOSH Respiratory Protection Week 2023
NPPTL Infographics
NPPTL Fact Sheets
Healthcare Respiratory Protection Resourceplus icon
Policies and Procedures
Program Administrator
Evaluation and Selection
Medical Evaluation
Fit Testing
Proper Use and Storage
Training
Program Evaluation
Recordkeeping
Support for Firefighter Personal Protective Equipmentplus icon
Self-Contained Breathing Apparatusplus icon
Program Overview
Evaluation Process
Completed Reports
Resources
Firefighter Turnout Gearplus icon
Current Projects
Completed Projects
Participation on Standards Setting Organizations
Resources
Considerations for Selecting Protective Clothing
NPPTL Participation in Standard Development Committees
Respiratory Protection Information Trusted Sourceplus icon
Types of Respiratory Protection
Use of NIOSH-approved Respirators
Frequently Asked Questionsplus icon
Respirator FAQs
Fit Test FAQs
NIOSH Approval FAQs
Healthcare Setting Specific FAQs
Approved Particulate Filtering Facepiece Respirators
Certified Equipment List (CEL)
PPE CASE Reports
Respirator User Notices
Meetings, Webinars, & Conferencesplus icon
Archived Meetings and Conferences
Conformity Assessment Notices and Letters to Respirator Manufacturers and Interested Parties
About NPPTL
Contact NPPTL Connect with NPPTL Subscribe to Listserv
PPE concerns? Contact Us
 Follow NIOSH Facebook
Pinterest
Twitter
YouTube
 NIOSH Homepage NIOSH A-Z
Workplace Safety & Health Topics
Publications and Products
Programs
Contact NIOSH


About CDC
					Contact Us
					800-232-4636
 Facebook Twitter Instagram LinkedIn Youtube Pinterest Snapchat
 CONTACT CDC
				Contact Us
							Call 800-232-4636
							Email Us
			ABOUT CDC
				About CDC
							Jobs
							Funding
			POLICIES
				Accessibility
							External Links
							Privacy
							Policies
							No Fear Act
							FOIA
							OIG
							Nondiscrimination
							Vulnerability Disclosure Policy
			CONNECT WITH US
				Facebook
								Twitter
								Instagram
								LinkedIn
							Youtube
								Pinterest
								Snapchat
								Email
			LANGUAGES
				Español
							繁體中文
							Tiếng Việt
							한국어
							Tagalog
							Русский
							العربية
							Kreyòl Ayisyen
							Français
							Polski
							Português
							Italiano
							Deutsch
							日本語
							فارسی
							English


U.S. Department of Health & Human Services
			Accessibility
			External Links
			Privacy
			Policies
			No Fear Act
			FOIA
			Nondiscrimination
			OIG
			Vulnerability Disclosure Policy
			USA.gov


Exit Notification / Disclaimer Policy Close
 Links with this icon  indicate that you are leaving the CDC website. The Centers for Disease Control and Prevention (CDC) cannot attest to the accuracy of a non-federal website.
				Linking to a non-federal website does not constitute an endorsement by CDC or any of its employees of the sponsors or the information and products presented on the website.
				You will be subject to the destination website's privacy policy when you follow the link.
				CDC is not responsible for Section 508 compliance (accessibility) on other federal or private website. Website Disclaimers
 Cancel Continue