Skip directly to site content Skip directly to page options Skip directly to A-Z link Skip directly to A-Z link Skip directly to A-Z link


Centers for Disease Control and Prevention. CDC twenty four seven. Saving Lives, Protecting People
Search
            ×
                        Mining
                        Mining
                            NIOSH
                            All CDC
                    Submit
                For a full list of topics:A-Z Index
                Advanced Search
            Advanced Search


The National Institute for Occupational Safety and Health (NIOSH)





Mining Contract: Continuous Float Dust Monitor
Keywords: Float dust
This page is archived for historical purposes and is no longer being maintained or updated.
Contract #
200-2013-57232Start Date
9/23/2013End Date
12/23/2014Research Concept
This research proposes the design, fabrication, and testing of a continuous float coal dust monitor (CFDM) in both the laboratory and in a test coal mine setting, with the goal of optimizing the design to provide the necessary sensitivity, dynamic range, and battery lifetime. This instrument will measure the concentration of in-mine, airborne float coal dust in near real time. This measurement is intended to be used as an input to an automated rock duster whereby the rock duster will vary the amount of inerting dust with respect to the concentration of explosive coal dust.Topic Areas
Explosion Prevention
Rock Dusting
Contract Status & Impact
This contract is complete. To receive a copy of the final report, send a request to mining@cdc.gov.
The airborne presence and surface accumulation of float coal dust in underground coal mines poses a serious safety hazard to miners. Unless inerted properly, float coal dust can be re-entrained and aerosolized into the airstream by the pressure wave of a methane explosion. To control float coal dust, mines apply inerting rock dust to achieve mixtures of at least 80% inerting material and 20% coal dust. Compliance with 30 CFR 75.403 requires that samples be taken from the mine floor, roof, ribs, and suspended items and sent off for analysis, which results in the percentage of incombustible material. After receiving these results, a mine will then know whether more or less rock dust is required.
A necessary input to an automated, programmable rock duster capable of distributing rock dust concentrations “on demand” is the concentration of float coal dust in real time; however, no sensor technology for this purpose currently exists. While instruments are available to measure coal dust in real time, these units are not intended to be used under the same conditions as a float dust monitor. Specifically, a float dust monitor must be able to measure both very low and very high (0–1,000 milligrams per cubic meter) concentrations of float dust. Because float coal dust has a larger size range than respirable mine dust (sub-75 microns compared to sub-10 microns), a sensor must be able to avoid dust losses within a unit’s internal components. It must also be robust against excessive dust loading associated with float coal and rock dust that will inevitably occur in a permanent sensor installation.
Under this contract, Akita Innovations designed, fabricated, and tested a continuous float dust monitor in the OMSHR longwall gallery at the Pittsburgh Research Laboratory in Pittsburgh, PA, with the goal of optimizing the design to provide the necessary sensitivity, dynamic range, battery lifetime, and real-time responsiveness. After investigating more than five different measurement techniques, Akita chose to pursue a light transmission/extinction method using a multi-pass optical absorbance cell. A benchtop working prototype was built and tested. It exhibited excellent sensitivity and linearity when exposed to glass slides with various representative concentrations of float coal dust.
A second prototype was constructed and tested at OMSHR’s longwall gallery using various dust types. Results of this testing demonstrated that the second prototype did not have the ability to distinguish between coal and rock dust. However, it did differentiate between dust particles and water spray droplets. By far, the most significant issue with the current CFDM is that dust rapidly accumulates on the mirrors, reducing their reflectivity to the point where the instrument no longer generates valid data. This must be resolved prior to any further advancement of this research. Additionally, before the instrument can be used as an input to an automated, programmable rock duster, an algorithm which relates the CFDM output to an actual dust concentration must be developed. Finally, although designed to consider intrinsically safe requirements, the current prototype was not constructed to be intrinsically safe. An intrinsically safe prototype will need to be constructed before testing in an underground coal mine is possible.
See Also
Characterization of Airborne Float Coal Dust Emitted During Continuous Mining, Longwall Mining, and Belt Transport
Effect of Ground Control Mesh on Dust Sampling and Explosion Mitigation
Float Dust Control in Underground Coal Mines
Full-Scale Testing of the Float Dust Deposition Meter
LiDAR Technology Adaptation to Underground Coal Mines for Float Coal Dust and Rock Dust Mapping Applications
Open-air Sprays for Capturing and Controlling Airborne Float Coal Dust on Longwall Faces
Reducing Float Coal Dust: Field Evaluation of an Inline Auxiliary Fan Scrubber
Sampling and Analysis Method for Measuring Airborne Coal Dust Mass in Mixtures with Limestone (Rock) Dust
Sensors for Automated Control of Coal Dust (SACCD)
Technology News 515 - Float Coal Dust Explosion Hazards
Page last reviewed: July 18, 2016
Page last updated: July 18, 2016
Content source: National Institute for Occupational Safety and Health, Mining Program
Facebook
        Twitter
        LinkedIn
        Syndicate Mining
        Site BrowserSafety and Health TopicsData & StatisticsMethodologyMine DisastersMSHA Data FilesMine & Mineworker ChartsTools & PublicationsTools You Can UsePublicationsNews & ArticlesResearch ProgramNORA Mining CouncilStrategic PlanMining Program, 2019-2023Miner Health Program, 2020-2030ProjectsContractsFunding OpportunitiesMining LinksAbout UsContact NIOSH MiningEmploymentFacilitiesHistoryVisitor Information Explosion Prevention
Rock Dusting
 NIOSH Homepage
 NIOSH A-Z
Workplace Safety & Health Topics
Publications and Products
Programs


About CDC
					Contact Us
					800-232-4636
 Facebook Twitter Instagram LinkedIn Youtube Pinterest Snapchat
 CONTACT CDC
				Contact Us
							Call 800-232-4636
							Email Us
			ABOUT CDC
				About CDC
							Jobs
							Funding
			POLICIES
				Accessibility
							External Links
							Privacy
							Policies
							No Fear Act
							FOIA
							OIG
							Nondiscrimination
							Vulnerability Disclosure Policy
			CONNECT WITH US
				Facebook
								Twitter
								Instagram
								LinkedIn
							Youtube
								Pinterest
								Snapchat
								Email
			LANGUAGES
				Español
							繁體中文
							Tiếng Việt
							한국어
							Tagalog
							Русский
							العربية
							Kreyòl Ayisyen
							Français
							Polski
							Português
							Italiano
							Deutsch
							日本語
							فارسی
							English


U.S. Department of Health & Human Services
			Accessibility
			External Links
			Privacy
			Policies
			No Fear Act
			FOIA
			Nondiscrimination
			OIG
			Vulnerability Disclosure Policy
			USA.gov


Exit Notification / Disclaimer Policy Close
 Links with this icon  indicate that you are leaving the CDC website. The Centers for Disease Control and Prevention (CDC) cannot attest to the accuracy of a non-federal website.
				Linking to a non-federal website does not constitute an endorsement by CDC or any of its employees of the sponsors or the information and products presented on the website.
				You will be subject to the destination website's privacy policy when you follow the link.
				CDC is not responsible for Section 508 compliance (accessibility) on other federal or private website. Website Disclaimers
 Cancel Continue