Skip directly to site content Skip directly to page options Skip directly to A-Z link Skip directly to A-Z link Skip directly to A-Z link


Centers for Disease Control and Prevention. CDC twenty four seven. Saving Lives, Protecting People
Search
            ×
                        Mining
                        Mining
                            NIOSH
                            All CDC
                    Submit
                For a full list of topics:A-Z Index
                Advanced Search
            Advanced Search


The National Institute for Occupational Safety and Health (NIOSH)





Mining Topic: Atmospheric Monitoring
Keywords: Airflow Anemometers Methane monitors Ventilation Ventilation monitoring systems
Fast Find
Related Contracts
Related Projects
All atmospheric monitoring content
What is the health and safety problem?
The proper control and distribution of ventilation air in working areas of underground mines is crucial to the health and safety of mine workers. Many underground coal mines cover vast areas where workers are not located. Continuous knowledge of the ventilation system status, along with information on the presence of methane or products of combustion from fires, is critical for detecting and correcting problems in the mine atmosphere in their earliest stage.
To this end, various sensors can be used in targeted areas of the mine to collect environmental data on levels of combustible gas or products of combustion that may provide an early indication that heating is in progress. In addition to levels of combustible gas, monitoring airflow velocity and direction continuously can improve the safety of the underground workforce. However, continuous measurements of air velocity may not always provide accurate assessments of ventilation conditions, as these values can change rapidly depending on measurement location and airflow conditions.
What is the extent of the problem?
Multiple gas sensors installed in a mine to collect mine atmospheric dataMultiple gas sensors installed in a mine to collect mine atmospheric data.
Under MSHA regulations for atmospheric monitoring systems, monitoring in active mining areas usually involves methane and products of combustion around electrical and belt installations. Monitoring of ventilation airflow velocity is less common, with little information gathered on airflow direction. Large portions of many underground operations may not be continuously monitored for airflow velocity and methane accumulations.
How is the NIOSH Mining Program addressing this problem?
NIOSH Mining investigated the use of ultrasonic velocity instrumentation in the underground mine environment. Ultrasonic anemometers are known for their accuracy, capability to measure low airflows, continuous measurement characteristics, and ability to provide a directional sign to the air velocity. This can be especially beneficial in situations with low air velocity in that airflow reversals may occur.
NIOSH research efforts focused on the installation of integrated airflow and methane monitors in active areas of underground coal mining operations. This work identified specific sampling protocols for these devices and assessed the behavior of these instruments in relation to the presence of airway obstructions such as personnel and equipment. NIOSH also identified potential applications and limitations of integrated air velocity and methane monitoring systems and developed guidelines to interpret air velocity and methane monitor outputs.
NIOSH Mining conducted full-scale experiments in the Safety Research Coal Mine (SRCM) to evaluate different sensors that can be used for early fire detection in underground battery charging stations, underground diesel fuel storage areas, and belt entry with a low air velocity. The appropriate sensor deployment strategies for those locations were developed based on test results. For underground battery charging stations, the focus was on the cross-interference for different carbon monoxide (CO) sensors caused by hydrogen (H2) gas released during the battery charging. For underground diesel fuel storage areas, various CO sensors, smoke sensors, and flame sensor were examined with different fire scenarios. For the belt entry with a low air flow velocity (less than 50 fpm), the responses of CO and smoke sensors located at eight sensors stations were studied with different fires at low air velocities.
What are the significant findings?
NIOSH Mining laboratory testing showed that air velocity readings from continuous recording instruments correlated well with those obtained using standard measurement techniques. Obstructions upstream of a continuous recording anemometer led to noticeable impacts on air velocity readings, suggesting that movement of these obstructions in the ventilation airstream could be correlated to changes in instrument readings. In conditions with steady airflow, recorded velocities showed little variation over time, whereas turbulent conditions produced greater variability. This work also identified the minimum number of instrument readings needed to obtain an accurate assessment of ventilation air velocity in both steady and turbulent airflow conditions.
NIOSH tests results for underground battery charging stations show that only one out of seven tested CO sensors was minimally affected by hydrogen, and H2-compensated CO sensors did not perform any better than non-H2-compensated CO sensors. The cross-interference from H2 was also affected by the ventilation airflow and the sensor location. Test results for underground diesel fuel storage areas indicate that smoke sensor responded in a shorter time compared to the CO sensors in majority of tests.  The flame sensor produced the shortest response times in all tests compared to the CO sensors and the smoke sensors. The flame sensor performance was not affected by the fire location. For the belt entry with low air flow velocity, experimental results demonstrate that the low airflow velocity could cause smoke unevenly distributed over the mine entry and delay the alarm times for CO and smoke sensors for the small fires. In majority of fire tests, smoke sensors had shorter alarm times than the CO sensors located at the same place. The smoke sensors tested in this study were significantly affected by the high humidity in the tests and could not function properly. The results indicate that reduced sensor spacing is needed for CO sensors at low air flow velocities (less than 50 fpm).
What are the next steps?
No future work in this area is planned at NIOSH now.
See Also
Composition Change Model for Sealed Atmosphere in Coal Mines
Continuous Monitoring of Airflow and Methane in Coal Mines
Experimental and Modeling Investigation of the Effect of Ventilation on Smoke Rollback in a Mine Entry
Investigation into the Practical Use of Belt Air at US Longwall Operations
Measurement of Airflow in a Simulated Underground Mine Environment Using an Ultrasonic Anemometer
Methods to Determine the Status of Mine Atmospheres - an Overview
Methods to Improve Efficiency of Mine Ventilation Systems
Methods to Improve Mine Ventilation System Efficiency
Novel Stopping Designs for Large-Opening Metal/Nonmetal Mines
Sonic Anemometer Airflow Monitoring Technique for Use in Underground Mines
Using Ultrasonic Anemometers to Evaluate Face Ventilation Conditions
Using Ultrasonic Anemometers to Evaluate Factors Affecting Face Ventilation Effectiveness
Page last reviewed: September 27, 2022
Page last updated: September 1, 2020
Content source: National Institute for Occupational Safety and Health, Mining Program
Facebook
        Twitter
        LinkedIn
        Syndicate Mining
        Site BrowserSafety and Health TopicsData & StatisticsMethodologyMine DisastersMSHA Data FilesMine & Mineworker ChartsTools & PublicationsTools You Can UsePublicationsNews & ArticlesResearch ProgramNORA Mining CouncilStrategic PlanMining Program, 2019-2023Miner Health Program, 2020-2030ProjectsContractsFunding OpportunitiesMining LinksAbout UsContact NIOSH MiningEmploymentFacilitiesHistoryVisitor Information Explosion Prevention
Fire Detection
Ventilation Overview
 NIOSH Homepage
 NIOSH A-Z
Workplace Safety & Health Topics
Publications and Products
Programs


About CDC
					Contact Us
					800-232-4636
 Facebook Twitter Instagram LinkedIn Youtube Pinterest Snapchat
 CONTACT CDC
				Contact Us
							Call 800-232-4636
							Email Us
			ABOUT CDC
				About CDC
							Jobs
							Funding
			POLICIES
				Accessibility
							External Links
							Privacy
							Policies
							No Fear Act
							FOIA
							OIG
							Nondiscrimination
							Vulnerability Disclosure Policy
			CONNECT WITH US
				Facebook
								Twitter
								Instagram
								LinkedIn
							Youtube
								Pinterest
								Snapchat
								Email
			LANGUAGES
				Español
							繁體中文
							Tiếng Việt
							한국어
							Tagalog
							Русский
							العربية
							Kreyòl Ayisyen
							Français
							Polski
							Português
							Italiano
							Deutsch
							日本語
							فارسی
							English


U.S. Department of Health & Human Services
			Accessibility
			External Links
			Privacy
			Policies
			No Fear Act
			FOIA
			Nondiscrimination
			OIG
			Vulnerability Disclosure Policy
			USA.gov


Exit Notification / Disclaimer Policy Close
 Links with this icon  indicate that you are leaving the CDC website. The Centers for Disease Control and Prevention (CDC) cannot attest to the accuracy of a non-federal website.
				Linking to a non-federal website does not constitute an endorsement by CDC or any of its employees of the sponsors or the information and products presented on the website.
				You will be subject to the destination website's privacy policy when you follow the link.
				CDC is not responsible for Section 508 compliance (accessibility) on other federal or private website. Website Disclaimers
 Cancel Continue