Skip directly to site content Skip directly to page options Skip directly to A-Z link


Centers for Disease Control and Prevention. CDC twenty four seven. Saving Lives, Protecting People
															Search
		×
			NIOSH
			NIOSH
				All CDC
				Submit
			For a full list of topics:A-Z Index
			Advanced Search
		Advanced Search


The National Institute for Occupational Safety and Health (NIOSH)


Section Navigation
			
			CDC Home


Facebook
        Twitter
        LinkedIn
        Syndicate





Control of Nitrous Oxide During Cryosurgery										Related Pages
										January 1999
DHHS (NIOSH) Publication Number 99-105
cover page - Control of Nitrous Oxide During Cryosurgery
Hazard
Compressed gases such as nitrous oxide (N2O) are often used to obtain the cold temperatures needed for cryosurgery. Cryosurgical instruments which use compressed gas are designed to allow the gas to expand through a valve inside the metal tip of the cryosurgical probe, causing the tip to reach extremely low temperatures. If the exhaust gas from the probe is improperly vented, N2O concentrations in air can reach several thousand parts per million during a cryosurgical procedure; and depending on the room ventilation rate, levels may remain elevated for long periods of time following the procedure. Exposures should be minimized to prevent short-term behavioral and long-term reproductive health effects that can be caused by N2O.
Controls
NIOSH studies have shown that airborne concentrations of N2O from cryosurgical units can be effectively controlled by ventilation and equipment maintenance, and if this is not possible, through substitution with a less hazardous cryogenic gas.
Ventilation
Most modern N2O cryosurgical units are equipped with a scavenging port where the gas is exhausted (see Figure 1). Some older units not equipped with scavenging can be retrofitted by the manufacturer to include a scavenging port. A length of plastic tubing available from the manufacturer can be connected directly to this port, with the other end connected to a discharge location. There are various options for how to discharge the N2O. You should consult with the manufacturer to determine which scavenging methods they recommend for their equipment. Some of the more common discharge methods used include the following:
**Exhaust Directly to Outdoors**: Since the gas is under considerable pressure, the discharge end of the tubing can be exhausted directly outdoors. This can be accomplished by placing the end of the hose out of a window or attached to a pipe placed through a wall or ceiling to the outdoors. Care should be taken to ensure that the discharge location is not an area where people are normally present or in close proximity to an outside air intake or window where N2O could re-enter the building.
**Exhaust into a Nonrecirculating General Ventilation System**: If a nonrecirculating general ventilation system is present, the hose can be placed up in an exhaust air vent and the N2O will be discharged out of the building. Care should be taken to ensure that the hose is placed far enough into the duct system to prevent N2O from re-entering the room. To make sure that it is not, air monitoring should be conducted. Discharge hoses should not be placed in recirculating general ventilation duct work since this could distribute the N2O to other areas of the building.
**Exhaust to Vacuum Scavenging System**: In some instances, the discharge hose can be hooked to a vacuum system for removal of the N2O. A qualified individual should determine if an existing waste anesthetic gas disposal system or a medical-surgical vacuum source can be used for this purpose. The high exhaust flowrates from some cyrosurgical units (up to 90 liters per minute) may be greater than the capacity of the vacuum system or leave minimal vacuum capacity for other parts of the system. Due to the high gas pressure, a specially designed interface may be needed between the vacuum system and the cryogenic scavenging tube to prevent back pressurization. Since N2O is an oxidizer, its corrosive effect on vacuum system components and compatibility with materials and lubricants used in the vacuum system should also be considered.
image of vacuum systemEquipment Maintenance
Even with a proper scavenging unit, leaks can occur that can cause high levels of N2O in the work area. The equipment manufacturer should be consulted to determine what routine maintenance is necessary for the particular cryogenic unit. Periodic leak testing of all fittings and connections should be conducted to ensure this source of exposure is minimized. A direct reading N2O monitor or a soap solution can be used for this purpose.
Substitution
If scavenging is not possible, the user should consider substitution of carbon dioxide (CO2), which can also be used as a cryogenic gas on many of the cryosurgical units. CO2 is considered less hazardous and has a much higher airborne exposure limit than N2O. However, since it does not cool as low as N2O (-79 °C for CO2 vs. -89 °C for N2O), it will likely require longer freeze times. The cryosurgical unit manufacturer should be consulted for guidance regarding the use of CO2 with their equipment. Additionally, experts in the medical field should be consulted concerning the implications of changing to CO2 as the freeze agent as this may not be a medically viable option in all cases.
Acknowledgments
The principal contributors to this publication are William J. Daniels, Charles S. McCammon, and Jerome P. Flesch of NIOSH and Lee Shands, Indian Health Service. The drawing is by Dick Carlson. Desktop publishing was provided by Kristina Dragon and Susan Kaelin.
This document is in the public domain and may be freely copied or reprinted. NIOSH encourages all readers of this *HAZARD CONTROLS* to make it available to all interested employers and workers.
Control of Nitrous Oxide During Cryosurgerypdf icon [PDF – 98.45 KB]
							
					
						Page last reviewed:  June 6, 2014					Content source: 	National Institute for Occupational Safety and Health homeNIOSH Publications & Products
NIOSH-Issued Publications
Search NIOSHTIC-2 Research Database
Docket
Documents for Public Review
Peer Review Agenda
Databases
eNews
Federal Register Noticesplus icon
FRN Archive 2007-2010
FRN Archive 2005-2006
FRN Archive 2001-2004
FRN Archive 1996-2000
Press Releases and Updatesplus icon
Press Release Archives 2010-2013
Press Release Archives 2006-2009
Press Release Archives 2001-2005
Press Release Archives 1998-2000
Press Release Archives 1992-1997
Regulations
Regulatory Agenda
Science Blog
Software
Video Follow NIOSH Facebook
Pinterest
Twitter
YouTube
 NIOSH Homepage NIOSH A-Z
Workplace Safety & Health Topics
Publications and Products
Programs
Contact NIOSH


About CDC
					Contact Us
					800-232-4636
 Facebook Twitter Instagram LinkedIn Youtube Pinterest Snapchat
 CONTACT CDC
				Contact Us
							Call 800-232-4636
							Email Us
			ABOUT CDC
				About CDC
							Jobs
							Funding
			POLICIES
				Accessibility
							External Links
							Privacy
							Policies
							No Fear Act
							FOIA
							OIG
							Nondiscrimination
							Vulnerability Disclosure Policy
			CONNECT WITH US
				Facebook
								Twitter
								Instagram
								LinkedIn
							Youtube
								Pinterest
								Snapchat
								Email
			LANGUAGES
				Español
							繁體中文
							Tiếng Việt
							한국어
							Tagalog
							Русский
							العربية
							Kreyòl Ayisyen
							Français
							Polski
							Português
							Italiano
							Deutsch
							日本語
							فارسی
							English


U.S. Department of Health & Human Services
			Accessibility
			External Links
			Privacy
			Policies
			No Fear Act
			FOIA
			Nondiscrimination
			OIG
			Vulnerability Disclosure Policy
			USA.gov


Exit Notification / Disclaimer Policy Close
 Links with this icon  indicate that you are leaving the CDC website. The Centers for Disease Control and Prevention (CDC) cannot attest to the accuracy of a non-federal website.
				Linking to a non-federal website does not constitute an endorsement by CDC or any of its employees of the sponsors or the information and products presented on the website.
				You will be subject to the destination website's privacy policy when you follow the link.
				CDC is not responsible for Section 508 compliance (accessibility) on other federal or private website. Website Disclaimers
 Cancel Continue