Skip directly to site content Skip directly to search


Español  Other Languages


An official website of the United States government Here's how you know
 Official icon**Official websites use .gov**
				A .gov website belongs to an official government organization in the United States. Lock icon**Secure .gov websites use HTTPS**
				A lock (  ) or https:// means you've safely connected to the .gov website. Share sensitive information only on official, secure websites.


Centers for Disease Control and Prevention. CDC twenty four seven. Saving Lives, Protecting People
	CDC LogoCDC Logo
 Search

		×
			NIOSH
			NIOSH

				All CDC
				Submit


The National Institute for Occupational Safety and Health (NIOSH)





Current Intelligence Bulletin 13: Explosive Azide Hazard						Print
																Related Pages
										August 16, 1976
DHHS (NIOSH) Publication Number 78-127
The National Institute for Occupational Safety and Health (NIOSH) is alerting you to an explosive hazard which may exist in hospital and clinical laboratory plumbing systems due to sodium azide formulated into diluents used in conjunction with automatic blood cell counters. These counters are found in over 15,000 hospitals and clinical laboratories throughout the United States.
NIOSH has recently learned of violent sodium azide-related explosions associated with automatic blood cell counters at a number of hospitals in the United States and Canada. In addition, we are aware of a violent azide explosion occurring while a constant temperature water bath in which sodium azide had been used as a preservative was being repaired. These explosions have the propensity to propel metallic fragments over a wide area and the potential for causing serious injury to exposed workers and others in the vicinity.
Sodium azide is a common preservative in many *in vitro* diagnostic products and is found in concentrations up to 0.1% in diluents used with automatic blood cell counters. After completion of the blood count procedure, the waste (containing azide) is commonly discharged into a drain used solely for this purpose, thus bathing the drain pipeline with solutions of sodium azide. Over a period of time, the azide reacts with copper, lead, brass, or solder in the plumbing system to form an accumulation of lead and/or copper azide. Lead azide is a more sensitive primary explosive than nitroglycerine and a more effective detonating agent than mercury fulminate; in comparison with lead azide, copper azide is even more explosive and too sensitive to be used commercially.
Future accumulation of lead and copper azides in plumbing systems can be retarded by thoroughly flushing any drain known to receive azides with copious amounts of water several times a day. The use of copper-and lead-free lines between the point of discharge of azide and the nearest pipe in which there is a good stream of water, or the use of azide-free reagents, may prevent future accumulation of explosive azides in plumbing. HOWEVER, THESE MEASURES WILL NOT DECONTAMINATE PLUMBING ALREADY CONTAINING EXPLOSIVE AZIDES. Procedures for the decontamination of plumbing systems containing copper and/or lead azide are enclosed.
Laboratory maintenance workers, especially plumbers, should be alerted to the azide hazard so that proper precautions can be taken. Violent explosions have resulted when plumbers have attempted to penetrate blocked azide-contaminated drainage systems with a flexible metal probe (snake) or to cut or saw azide-contaminated drain lines.
NIOSH would appreciate being advised of the details of any azide-related explosions and welcome your comments regarding the effectiveness of the enclosed decontamination procedures.
*[signature]*
John F. Finklea, M.D.
Director

Decontamination Procedures for Azide Contaminated Plumbing
The following procedure 1 has been suggested by the Center for Disease Control, U.S. Public Health Service, for use in its laboratories:
Prepare 1 to 2 liters of 10% sodium hydroxide solution (100 g NaOH per liter of water).
Syphon all liquid from the trap and drain using a soft rubber or plastic hose. Use proper precautions against any hazardous chemicals which may be present.
Slowly pour the sodium hydroxide solution into the trap.
Tape to the sink a warning sign reading “Do Not Use Sink...Contains Caustic Material.”
Allow the solution to remain in the trap for a minimum of 16 hours.
Flush the drain with water for a minimum of 15 minutes.If the drain will not flow, the sodium hydroxide should be removed by syphoning, if possible, then diluted with water. Maintenance personnel should be advised that the drain is potentially contaminated with explosive agents and caustic material.The above procedure is designed to decontaminate a drain trap. Longer lengths of drain lines can be decontaminated with a similar procedure after plugging the drain below the point at which any azide contamination is likely to have occurred and then filling the entire length of pipe with 10% sodium hydroxide solution.
Where it is not possible for a drain line to remain filled with sodium hydroxide solution for at least 16 hours, Coulter Electronics, Inc. has suggested the following:  2
Pour five gallons of sodium hydroxide solution into the piping rapidly enough to simulate the flushing, action of a water closet. CAUTION: solution is caustic!
Allow the pipe to remain undisturbed by water or other effluents for at least 16 hours.
Flush with copious amounts of water.
Repeat steps 1, 2, and 3 two more times at intervals of a week or so.Descriptions of several other procedures which have been suggested for the decontamination of azides are listed in the *Journal of Chemical Education*. 3

Precautions
Because the possibility of residual sodium hydroxide will always exist, personnel should wear gloves and face shields when breaking the drain line or trap for maintenance. (This equipment should be worn when breaking any laboratory drain, as the presence of hazardous chemicals should always be suspected.)
Extreme caution should be exercised when plugging a drain line potentially contaminated with heavy metal azides.
The decontamination of plumbing systems containing copper or its alloys (e.g., brass) should include a supplemental treatment with nitrous acid, since the sodium hydroxide procedure may not adequately remove accumulations of copper azides. The following nitrous acid decontamination procedure 4 has been employed with success:
Close the exit of the drain beyond the point of potential azide accumulation.
Fill the drain line with nitrous acid, prepared immediately before use by mixing equal volumes of a 20% solution of acetic acid with a 20% solution of sodium nitrite.CAUTION! The area should be well ventilated, as toxic vapors (oxides of nitrogen) may be released when azide reacts with nitrous acid.
Allow the nitrous acid solution to remain in the drain for twenty-four hours.
Open the exit of the drain.
Immediately repeat procedure once.
Note
The decontamination of plumbing systems is complicated by a number of factors, including the possible coating of heavy metal azides by impervious materials as well as the possible accumulation of heavy metal azides in cracks and threads of plumbing. Although the decontamination procedures do reduce the risk of explosion, even a “decontaminated” system should be treated with respect in recognition of the possibility of its being explosive. Maintenance people should be alerted so that proper precautions can be taken before working on plumbing potentially contaminated with heavy metal azides. Good work practices include shielding the person working on the plumbing, maximizing the distance between the person and the plumbing, and keeping all unnecessary personnel out of the area.

References
Manual Guide–Safety Management No. CDC-22, Decontamination of Laboratory Sink Drains to Remove Azide Salts, Center for Disease Control, Atlanta, Georgia, April 30, 1976.
Personal Communication from Coulter Diagnostics, Inc., June 3, 1976.
James O. Wear, “Azide Hazards with Automatic Blood Cell Counters,” Journal of Chemical Education, 52, A23-A25, January 1975.
This procedure has been recommended and used by the Department of Health and Social Security, Government of the United Kingdom.
							
					
							Last Reviewed:  June 6, 2014						Source: 	National Institute for Occupational Safety and Health
					Facebook
        Twitter
        LinkedIn
        Syndicate homeNIOSH Publications & Products
NIOSH-Issued Publications
Search NIOSHTIC-2 Research Database
Docket
Documents for Public Review
Peer Review Agenda
Databases
eNews
Federal Register Noticesplus icon
FRN Archive 2007-2010
FRN Archive 2005-2006
FRN Archive 2001-2004
FRN Archive 1996-2000
Press Releases and Updatesplus icon
Press Release Archives 2010-2013
Press Release Archives 2006-2009
Press Release Archives 2001-2005
Press Release Archives 1998-2000
Press Release Archives 1992-1997
Regulations
Regulatory Agenda
Science Blog
Software
Video Error processing SSI file  Follow NIOSH Facebook
Pinterest
Twitter
YouTube
 NIOSH Homepage NIOSH A-Z
Workplace Safety & Health Topics
Publications and Products
Programs
Contact NIOSH


About CDC
					Contact Us
					800-232-4636
 Facebook Twitter Instagram LinkedIn Youtube Pinterest Snapchat
 CONTACT CDC
				Contact Us
							Call 800-232-4636
							Email Us
			ABOUT CDC
				About CDC
							Jobs
							Funding
			POLICIES
				Accessibility
							External Links
							Privacy
							Policies
							No Fear Act
							FOIA
							OIG
							Nondiscrimination
							Vulnerability Disclosure Policy
			CONNECT WITH US
				Facebook
								Twitter
								Instagram
								LinkedIn
							Youtube
								Pinterest
								Snapchat
								Email
			LANGUAGES
				Español
							繁體中文
							Tiếng Việt
							한국어
							Tagalog
							Русский
							العربية
							Kreyòl Ayisyen
							Français
							Polski
							Português
							Italiano
							Deutsch
							日本語
							فارسی
							English


U.S. Department of Health & Human Services
			Accessibility
			External Links
			Privacy
			Policies
			No Fear Act
			FOIA
			Nondiscrimination
			OIG
			Vulnerability Disclosure Policy
			USA.gov


Exit Notification / Disclaimer Policy Close
 Links with this icon  indicate that you are leaving the CDC website. The Centers for Disease Control and Prevention (CDC) cannot attest to the accuracy of a non-federal website.
				Linking to a non-federal website does not constitute an endorsement by CDC or any of its employees of the sponsors or the information and products presented on the website.
				You will be subject to the destination website's privacy policy when you follow the link.
				CDC is not responsible for Section 508 compliance (accessibility) on other federal or private website. Website Disclaimers
 Cancel Continue