Skip to main content
          Skip to FDA Search
          Skip to in this section menu
          Skip to footer links


U.S. flagAn official website of the United States government

					Here’s how you know
          			Dot gov**The .gov means it’s official.**Federal government websites often end in .gov or .mil. Before sharing sensitive information, make sure you're on a federal government site.
							      SSL**The site is secure.** The **https://** ensures that you are connecting to the official website and that any information you provide is encrypted and transmitted securely. 
			U.S. Food and Drug Administration
				
        Search
            Menu
							
						Search FDA
									
									Submit search 
		Featured
							Contact FDA
	                FDA Guidance Documents
	                Recalls, Market Withdrawals and Safety Alerts
	                Press Announcements
	                Warning Letters
	                Advisory Committees
	                En Español

					Products
            Food
							Drugs
							Medical Devices
							Radiation-Emitting Products
							Vaccines, Blood, and Biologics
							Animal and Veterinary
							Cosmetics
							Tobacco ProductsTopics
            About FDA
							Combination Products
							Regulatory Information
							Safety
							Emergency Preparedness
							International Programs
							News and Events
							Training and Continuing Education
							Inspections and Compliance
							Science and ResearchInformation For
            Consumers
	            Patients
	            Industry
	            Health Professionals
	            Federal, State and Local Officials 
 In this section:

                Search for FDA Guidance Documents

Search for FDA Guidance Documents

                        Search General and Cross-Cutting Topics Guidance Documents
          Advisory Committee Guidance Documents
          Import and Export Guidance Documents
          Cross-cutting Guidance Documents Home
          Regulatory Information
          Search for FDA Guidance Documents
          Laser Products - Conformance with IEC 60825-1 and IEC 60601-2-22; (Laser Notice No. 50)Search for FDA Guidance Documents GUIDANCE DOCUMENT
	Laser Products - Conformance with IEC 60825-1 and IEC 60601-2-22; (Laser Notice No. 50)
					
		
		June 2007


	Download the  Final Guidance Document 

    Final

	  
	  
	Share
    Post
    Linkedin

    Email
    PrintDocket Number:
		    FDA-2020-D-0957
	    	    		    Issued by:
		    Guidance Issuing Office
          Center for Devices and Radiological Health


PDF Printer Version
(126 KB)

**Document issued on: June 24, 2007**

UPDATE: This document will be withdrawn on December 31, 2024. Please use Laser Products - Conformance with IEC 60825-1 Ed. 3 and IEC 60601-2-22 Ed. 3.1 (Laser Notice No. 56), which will supersede this document on January 1, 2025. See Getting a Radiation Emitting Product to Market: Frequently Asked Questions for more information.


This document supersedes Laser Products - Conformance with IEC 60825-1, Am. 2 and IEC 60601-2-22; Final Guidance for Industry and FDA (Laser Notice No. 50), July 26, 2001.

For questions regarding this document send an email message to RadHealth@fda.hhs.gov.

CDRH Logo

**U.S. Department of Health and Human Services
Food and Drug Administration
Center for Devices and Radiological Health**

**Electronic Products Branch
Division of Mammography Quality and Radiation Programs
Office of Communication, Education, and Radiation Programs**

**Preface**

Public Comment

Written comments and suggestions may be submitted at any time for Agency consideration to the Division of Dockets Management, Food and Drug Administration, 5630 Fishers Lane, Room 1061, (HFA-305), Rockville, MD, 20852. When submitting comments, please refer to the exact title of this guidance document. Comments may not be acted upon by the Agency until the document is next revised or updated.

**Additional Copies**

You may obtain additional copies by sending an e-mail request to CDRH-Guidance@fda.hhs.gov to receive an electronic copy of the guidance. Please use the document number (1346) to identify the guidance you are requesting.

Guidance for Industry and FDA Staff

Guidance on Laser Products - Conformance with IEC 60825-1 and IEC 60601-2-22
(Laser Notice No. 50)

***This guidance represents the Food and Drug Administration's (FDA's) current thinking on this topic. It does not create or confer any rights for or on any person and does not operate to bind FDA or the public. You can use an alternative approach if the approach satisfies the requirements of the applicable statutes and regulations. If you want to discuss an alternative approach, contact the FDA staff responsible for implementing this guidance. If you cannot identify the appropriate FDA staff, call the appropriate number listed on the title page of this guidance.***

Introduction

This guidance describes the conditions under which laser product manufacturers may introduce into United States commerce laser products that comply with the IEC standards 60825-1, as amended, and 60601-2-22. This guidance also describes additional requirements of the CDRH standard and alternate certification statements to be used with such products.

Laser products for introduction into commerce in or imported into the United States must:

Comply with 21 CFR §1040.10 and §1040.11 as applicable,
	Be certified and identified in accordance with 21 CFR §1010.2 and §1010.3, and
	Be reported in accordance with 21 CFR §1002.10.CDRH has issued notices to laser product manufacturers and importers stating non-objection to:

Lack of emission indicators or beam attenuators on Class II and Class IIIa systems, and
	Hazard warning labels as specified in IEC 60825-1.CDRH will not object to conformance with many sections of IEC 60825-1, as amended, and IEC 60601-2-22 as alternatives to comparable sections of 21 CFR §1040.10 and §1040.11. CDRH plans to amend federal regulations for laser products to reflect those sections of the IEC standards. CDRH is also listing sections of its standard that contain requirements to which manufacturers must conform. This action is appropriate because of the Center's intent to harmonize its requirements with many of those of the IEC standards.

FDA's guidance documents, including this guidance, do not establish legally enforceable responsibilities. Instead, guidances describe the Agency's current thinking on a topic and should be viewed only as recommendations, unless specific regulatory or statutory requirements are cited. The use of the word *should* in Agency guidances means that something is suggested or recommended, but not required.

The Least Burdensome Approach

We believe we should consider the least burdensome approach in all areas of medical device and electronic product regulation. This guidance reflects our careful review of the relevant scientific and legal requirements and what we believe is the least burdensome way for you to comply with those requirements. However, if you believe that an alternative approach would be less burdensome, please contact us so we can consider your point of view. You may send your written comments to the contact person listed in the preface to this guidance or to the CDRH Ombudsman. Comprehensive information on CDRH's Ombudsman, including ways to contact him, can be found on the Internet.

**1. Issue**

The Food and Drug Administration’s (FDA) Center for Devices and Radiological Health (CDRH) intends to amend its standards for laser products at 21 CFR §1040.10 and §1040.11 to harmonize many of its requirements with those of the IEC 60825-1 and 60601-2-22 standards. Although CDRH began its amendment process in anticipation of the amendment of IEC 60825-1, it is not yet ready to publish an amendment. CDRH has acknowledged the advantages of one set of criteria and requirements worldwide. Edition 1.2 of IEC 60825-1 was published in August, 2001. Edition 2 was published in March, 2007. Edition 2 of IEC 60601-2-22 was published in November, 1995. Edition 3 was published in May, 2007. As a result, manufacturers distributing products in both the U.S. and countries that require conformance with, or that recognize IEC 60825-1, will have to evaluate the conformance of their products with this standard. This requires them to often change the hazard classification of their products. These manufacturers are requesting relief from CDRH so that they will have to comply with only one laser product radiation safety standard.

2. Guidance

**For what sections of 21 CFR can conformance with sections of IEC 60825-1 and 60601-2-22 be substituted?**

Effective immediately, and until the effective date(s) of any amendments of the Federal regulations affecting laser products, CDRH will not object to conformance with the comparable sections of IEC 60825-1, Editions 1.2 or 2 or 60601-2-22 in lieu of conformance with the following sections of 21 CFR §1040:

1040.10(b) Definitions
1040.10(c)(1) Classification
1040.10(d) Accessible emission limits
1040.10(e) Tests for determination of compliance
1040.10(f)(1) Protective housing
1040.10(f)(2) Safety interlocks
1040.10(f)(3) Remote Interlock connector
1040.10(f)(4) Key control
1040.10(f)(5) Laser radiation emission indicator
1040.10(f)(6) Beam attenuator
1040.10(f)(7) Location of controls
1040.10(f)(8) Viewing optics
1040.10(f)(9) Scanning safeguard
1040.10(g) Labeling requirements
1040.10(h)(1) User information
1040.11(a) Medical laser products

CDRH intends to harmonize the requirements of these sections with those of the IEC standards.

**Which applicable sections of 21 CFR are unaffected by this guidance?**

The following sections are unaffected. Laser products must conform to the following sections of the CDRH standards:

1010.2 Certification
1010.3 Identification
1010.4 Variances
1040.10(a) Applicability
1040.10(c)(2) Removable laser systems
1040.10(f)(10) Manual reset mechanism
1040.10(h)(2) Purchasing and servicing information
1040.10(i) Modification of a certified product
1040.11(b) Surveying, leveling and alignment laser products
1040.11(c) Demonstration laser products

CDRH intends to retain these requirements even if they differ from the IEC. They are either beyond the scope of the IEC standard, are sufficiently different, or are not normative and included as recommendations in the User’s Guide section of the IEC standard.

**What should manufacturers do in using this guidance?**

Use the following modified statement of compliance on the certification label: “Complies with FDA performance standards for laser products except for deviations pursuant to Laser Notice No. 50, dated (Insert date of this guidance.)” or “Complies with 21 CFR 1040.10 and 1040.11 except for deviations pursuant to Laser Notice No. 50, dated June 24, 2007”; and
	Submit product reports or supplemental reports to describe changes to products in accordance with this guidance.The effectiveness of this guidance will end on the effective date(s) of any amendments to the Federal regulations applicable to laser products under Chapter 1, Subchapter J of Title 21 of the Code of Federal Regulations.

3. Getting More Information

You can get more information about our requirements for lasers from our electronic product radiation control web page.

If you have questions about this guidance, contact Dan Hewett, Office of Communication, Education, and Radiation Programs, Center for Devices and Radiological Health, 10903 New Hampshire Avenue, Silver Spring, MD 20993 Fax: (301) 796-5864, or e-mail daniel.hewett@fda.hhs.gov.


 
	Submit Comments
	   		  			    		Submit Comments Online
					You can submit online or written comments on any guidance at any time (see 21 CFR 10.115(g)(5))
						If unable to submit comments online, please mail written comments to:
						Dockets Management
							Food and Drug Administration
							5630 Fishers Lane, Rm 1061
							Rockville, MD 20852
						All written comments should be identified with this document's docket number: FDA-2020-D-0957.

                                                                                  Content current as of:
      08/29/2023
         
                  
Regulated Product(s)
		 Medical Devices
			  			     			  			     Radiation-Emitting Products
			     			  						              
			
			
			
			
			
			
			
			  
			  Laser NoticeTopic(s)
			  Postmarket
				    				       				    					
				  
				  
				  
				  
				     
				  Laser NoticeSearch for FDA Guidance Documents

        Search for FDA Guidance Documents

          
                        Search General and Cross-Cutting Topics Guidance Documents
          Advisory Committee Guidance Documents
          Import and Export Guidance Documents
          Cross-cutting Guidance Documents
 
 Footer Links
    FDA Archive
          About FDA
          Accessibility
      Visitor Information
          Website Policies / Privacy
          No FEAR Act
          Vulnerability Disclosure Policy
      FOIA
          HHS.gov
          USA.gov
    Contact FDA
      Follow FDA on Facebook
      Follow FDA on X
      Follow FDA on Instagram
      Follow FDA on LinkedIn
      View FDA videos on YouTube
      Subscribe to FDA RSS feeds
    FDA Homepage
    Contact Number
      1-888-INFO-FDA (1-888-463-6332) Back toTop