Skip to main content
          Skip to FDA Search
          Skip to in this section menu
          Skip to footer links


U.S. flagAn official website of the United States government

					Here’s how you know
          			Dot gov**The .gov means it’s official.**Federal government websites often end in .gov or .mil. Before sharing sensitive information, make sure you're on a federal government site.
							      SSL**The site is secure.** The **https://** ensures that you are connecting to the official website and that any information you provide is encrypted and transmitted securely. 
			U.S. Food and Drug Administration
				
        Search
            Menu
							
						Search FDA
									
									Submit search 
		Featured
							Contact FDA
	                FDA Guidance Documents
	                Recalls, Market Withdrawals and Safety Alerts
	                Press Announcements
	                Warning Letters
	                Advisory Committees
	                En Español

					Products
            Food
							Drugs
							Medical Devices
							Radiation-Emitting Products
							Vaccines, Blood, and Biologics
							Animal and Veterinary
							Cosmetics
							Tobacco ProductsTopics
            About FDA
							Combination Products
							Regulatory Information
							Safety
							Emergency Preparedness
							International Programs
							News and Events
							Training and Continuing Education
							Inspections and Compliance
							Science and ResearchInformation For
            Consumers
	            Patients
	            Industry
	            Health Professionals
	            Federal, State and Local Officials 
 In this section:

                Search for FDA Guidance Documents

Search for FDA Guidance Documents

                        Search General and Cross-Cutting Topics Guidance Documents
          Advisory Committee Guidance Documents
          Import and Export Guidance Documents
          Cross-cutting Guidance Documents Home
          Regulatory Information
          Search for FDA Guidance Documents
          Radiation Safety Considerations for X-Ray Equipment Designed for Hand-Held UseSearch for FDA Guidance Documents GUIDANCE DOCUMENT
	Radiation Safety Considerations for X-Ray Equipment Designed for Hand-Held Use
					
		
		December 2008


	Download the  Final Guidance Document 

    Final

	  
	  
	Share
    Post
    Linkedin

    Email
    PrintDocket Number:
		    FDA-2020-D-0957
	    	    		    Issued by:
		    Guidance Issuing Office
          Center for Devices and Radiological Health, Office of Product Evaluation and Quality, Office of In Vitro Diagnostics and Radiological Health


PDF Printer Version
(164 KB)

**Document issued on: December 24, 2008**

For questions regarding this document contact CDR Sean Boyd at 301-796-5895.

CDRH Logo

**U.S. Department of Health and Human Services
Food and Drug Administration
Center for Devices and Radiological Health**

**Diagnostic Devices Branch
Division of Mammography Quality and Radiation Programs
Office of Communication, Education, and Radiation Programs**

***Contains Nonbinding Recommendations***

Preface

Public Comment

Written comments and suggestions may be submitted at any time for Agency consideration to the Division of Dockets Management, Food and Drug Administration, 5630 Fishers Lane, Room 1061, (HFA-305), Rockville, MD, 20852. When submitting comments, please refer to the exact title of this guidance document. Comments may not be acted upon by the Agency until the document is next revised or updated.

Additional Copies

Additional copies are available from the Internet. You may also send an e-mail request to CDRH-Guidance@fda.hhs.gov to receive a copy of the guidance. Please use the document number (1680) to identify the guidance you are requesting.

Guidance for Industry and FDA Staff

Radiation Safety Considerations for X-Ray Equipment Designed for Hand-Held Use

Introduction

The intent of this guidance is to advise manufacturers and FDA staff about safety procedures and recommendations that should be provided to the end user to promote safe use of hand-held x-ray equipment. In general, there are a number of regulations, consensus safety standards, and radiation protection guidelines governing the performance and use of diagnostic x-ray equipment. Diagnostic x-ray equipment manufactured for the US market must meet the performance requirements described in Title 21 of the Code of Federal Regulations, sections 1020.30 through 1020.33, which require manufacturers to provide users with operational and safety information as well as to meet standards for equipment performance.1

Internationally, radiation safety standards are provided by the International Electrotechnical Commission (IEC) in its collateral standard 60601-1-3.2 Voluntary guidelines, particularly relevant to handheld equipment, have been provided by the National Council on Radiation Protection and Measurement (NCRP) for radiation protection in dentistry.3 Many states regulate the use of x-ray equipment under their own regulations based on the *Suggested State Regulations for the Control of Radiation*, published by the Conference of Radiation Control Program Directors (CRCPD).4

FDA's guidance documents, including this guidance, do not establish legally enforceable responsibilities. Instead, guidances describe the Agency's current thinking on a topic and should be viewed only as recommendations, unless specific regulatory or statutory requirements are cited. The use of the word *should* in Agency guidances means that something is suggested or recommended, but not required.

**The Least Burdensome Approach**

We believe we should consider the least burdensome approach in all areas of medical device and electronic product regulation. This guidance reflects our careful review of the relevant scientific and legal requirements and what we believe is the least burdensome way for you to comply with those requirements. However, if you believe that an alternative approach would be less burdensome, please contact us so we can consider your point of view. You may send your written comments to the contact person listed in the preface to this guidance or to the CDRH Ombudsman. Comprehensive information on CDRH's Ombudsman, including ways to contact him, can be found on the Internet.

1. Issue

There are two basic sources of potential x-ray exposure to operators that should be addressed during the use of hand-held x-ray diagnostic equipment:

Leakage radiation transmitted through the equipment housing and shielding that surrounds the source assembly
	Backscatter radiation from the patient and nearby structures.2. Guidance

A primary consideration in any radiographic procedure is to reduce the dose to the patient as well as to the operator as much as possible while still achieving the diagnostic goals. This guidance briefly discusses the hazards unique to hand-held medical x-ray equipment, clarifies applicable requirements, and makes recommendations for safe use of the equipment.

**What are the exposure concerns with handheld x-ray systems?**

The federal performance standard for diagnostic x-ray systems and their major components did not anticipate hand-held x-ray systems at the time it was written. It therefore does not address system performance attributes or protective measures in light of the operator’s proximity to the source assembly and the patient. Because of such proximity, hand-held x-ray systems pose increased operator exposure concerns due to leakage radiation and backscatter radiation. The radiation exposure to the operator will be the sum of any radiation leaking from the x-ray tube source assembly (leakage radiation) and any radiation that scatters from the patient or any objects in the room that are in the x-ray field (backscattered radiation.)

**What limits are placed on leakage and backscatter radiation?**

The applicable performance standard for a diagnostic source assembly, under 21 CFR 1020.30(k), limits radiation leakage from the x-ray source assembly to an air kerma of 0.88 mGy (corresponding to an exposure of 100 mR) in one hour at a distance of one meter from the x-ray source when measured as specified in the standard.

**How can leakage and backscatter radiation to the operator be reduced?**

***Shield the unit housing as required by the federal standard.***
	Leakage radiation is of particular concern because the operator may be holding the x-ray source assembly. Increasing the distance from the source may not be possible. Hand-held x-ray equipment must incorporate into its design physical means to protect the operator from leakage radiation with sufficient shielding surrounding the source assembly to ensure compliance with 21 CFR 1020.30(k).
	***Provide external shielding or a means to increase distance between the operator and the unit.***
	To reduce operator exposure to backscatter radiation, manufacturers of hand-held x-ray equipment should incorporate physical means to provide operator shielding or allow operation of the unit at a distance. For example, an external, physical shield may be supplied with the equipment to be placed between the operator and the patient to attenuate backscatter radiation. Additionally, you may provide a means to initiate x-ray exposure using an equipment stand and remote switch from a distance to increase distance between the operator and the unit during exposure.  
	Any product features designed into the equipment to protect the operator from radiation exposure should be described in user information with instructions for their proper use.
	***Measure typical exposures near and around the unit.***
	Because operators will typically be closer than 1 m to the unit, exposure measurements at distances closer than those required by the Federal standard become important. Manufacturers of hand-held x-ray units should make and report exposure measurements at distances near the unit, especially representative of positioning where the operator will actually be in contact with or in close proximity to the unit during use. For example, exposures at the location of handgrips and control devices are of particular interest to the operator of the unit.  
	These data may be used to estimate an operator’s whole-body or partial-body radiation dose for a single exposure or over an extended period of typical use. The data may be compared to occupational exposure limits to ensure limits are not exceeded and allow users to implement appropriate control measures based on an actual exposure assessment.
	***Identify necessary safety precautions as required by the Federal standard.***
	Under 21 CFR 1020.30(h)(1)(i) manufacturers must provide “adequate instructions concerning radiological safety procedures and precautions which may be necessary because of unique features of the equipment.” A unit that is hand-held constitutes a unique feature that requires manufacturers to describe any safety procedures and precautions necessary to reduce operator exposure. Safety precautions may include designating a significant zone of occupancy and providing exposure information near the unit as described in the IEC standard.2  
	Information on exposure levels near and around a hand-held unit should be included with operator instructions to identify the unique exposure hazards posed by the equipment. Providing exposure information will support any manufacturer’s precautions and allow the operator to develop additional safe working practices to limit unnecessary exposure during equipment use. Precautions may include wearing appropriate personnel monitoring and protective equipment, such as a personnel exposure monitoring device or lead-lined gloves and gowns. Procedural controls may be recommended to limit the number of exposures initiated by a single operator over a period or time, or to limit use of the hand-held unit without an equipment stand or remote switch in such circumstances only as necessary.

	Manufacturers may also include information describing exposure hazards and radiation safety precautions appropriate for patients and other individuals present in the area during examination (e.g., a person holding a child or other personnel assisting with procedures).3. Getting More Information

For additional information on state requirements applicable to hand-held x-ray systems, refer to your state’s radiation control program regulations. The CRCPD website has a list of contacts for each state (). Please contact your state officials on this subject. Another resource to consider is the *Suggested State Regulations for the Control of Radiation*. Some state programs do not allow the use of hand-held x-ray equipment.

If you have any questions about this guidance, contact CDR Sean Boyd in the Division of Mammography Quality and Radiation Programs, Office of Communication, Education and Radiation Programs, Center for Devices and Radiological Health, Food and Drug Administration,

10903 New Hampshire Avenue, Silver Spring, MD 20993 at 301-796-5895 or sean.boyd@fda.hhs.gov.

4. References

*Code of Federal Regulations* , Title 21 (Food and Drugs), Part 1020 (Performance Standards for Ionizing Radiation Emitting Products), Sections 1020.30-1020.33.
	*International Standard IEC 60601-1-3, Medical electrical equipment – Part 1-3: General requirements for basic safety and essential performance – Collateral Standard: Radiation protection in diagnostic X-ray equipment,* Edition 2.0, (International Electrotechnical Commission, Geneva, 2008).
	*Radiation Protection in Dentistry,* NCRP Report no. 145, (National Council on Radiation Protection and Measurements, Bethesda, MD, 2003).
	*Suggested State Regulations for the Control of Radiation,*(Conference of Radiation Control Program Directors, 2008).


 
	Submit Comments
	   		  			    		Submit Comments Online
					You can submit online or written comments on any guidance at any time (see 21 CFR 10.115(g)(5))
						If unable to submit comments online, please mail written comments to:
						Dockets Management
							Food and Drug Administration
							5630 Fishers Lane, Rm 1061
							Rockville, MD 20852
						All written comments should be identified with this document's docket number: FDA-2020-D-0957.

                                                                                  Content current as of:
      09/18/2015
         
                  
Regulated Product(s)
		 Medical Devices
			  			     			  			     Radiation-Emitting ProductsTopic(s)
			  PostmarketSearch for FDA Guidance Documents

        Search for FDA Guidance Documents

          
                        Search General and Cross-Cutting Topics Guidance Documents
          Advisory Committee Guidance Documents
          Import and Export Guidance Documents
          Cross-cutting Guidance Documents
 
 Footer Links
    FDA Archive
          About FDA
          Accessibility
      Visitor Information
          Website Policies / Privacy
          No FEAR Act
          Vulnerability Disclosure Policy
      FOIA
          HHS.gov
          USA.gov
    Contact FDA
      Follow FDA on Facebook
      Follow FDA on X
      Follow FDA on Instagram
      Follow FDA on LinkedIn
      View FDA videos on YouTube
      Subscribe to FDA RSS feeds
    FDA Homepage
    Contact Number
      1-888-INFO-FDA (1-888-463-6332) Back toTop