Skip to main content
          Skip to FDA Search
          Skip to in this section menu
          Skip to footer links


U.S. flagAn official website of the United States government

					Here’s how you know
          			Dot gov**The .gov means it’s official.**Federal government websites often end in .gov or .mil. Before sharing sensitive information, make sure you're on a federal government site.
							      SSL**The site is secure.** The **https://** ensures that you are connecting to the official website and that any information you provide is encrypted and transmitted securely. 
			U.S. Food and Drug Administration
				
        Search
            Menu
							
						Search FDA
									
									Submit search 
		Featured
							Contact FDA
	                FDA Guidance Documents
	                Recalls, Market Withdrawals and Safety Alerts
	                Press Announcements
	                Warning Letters
	                Advisory Committees
	                En Español

					Products
            Food
							Drugs
							Medical Devices
							Radiation-Emitting Products
							Vaccines, Blood, and Biologics
							Animal and Veterinary
							Cosmetics
							Tobacco ProductsTopics
            About FDA
							Combination Products
							Regulatory Information
							Safety
							Emergency Preparedness
							International Programs
							News and Events
							Training and Continuing Education
							Inspections and Compliance
							Science and ResearchInformation For
            Consumers
	            Patients
	            Industry
	            Health Professionals
	            Federal, State and Local Officials 
 In this section:

                Letters to Health Care Providers

Letters to Health Care Providers

                        2024 Letters to Health Care Providers
          2023 Letters to Health Care Providers
          2022 Letters to Health Care Providers
          2021 Letters to Health Care Providers Home
          Medical Devices
          Medical Device Safety
          Letters to Health Care Providers
          Do Not Use Cardinal Health Monoject Syringes with Syringe Pumps and PCA Pumps - Letter to Health Care ProvidersLetters to Health Care Providers Do Not Use Cardinal Health Monoject Syringes with Syringe Pumps and PCA Pumps - Letter to Health Care Providers
  
  	
    Subscribe to Email Updates
    Share
    Post
    Linkedin

    Email
    PrintNovember 20, 2023

The U.S. Food and Drug Administration (FDA) is alerting health care providers and facilities not to use Cardinal Health Monoject syringes with syringe pumps and patient-controlled analgesia (PCA) pumps while the FDA further evaluates this issue.

In June 2023, Cardinal Health began distributing Monoject syringes branded as “Cardinal Health” Monoject syringes. These new syringes differ from the previously branded “Covidien” Monoject syringes as they have different dimensions (example shown in image below) and are made by a different contract manufacturer. The dimensional changes made to the Cardinal Health Monoject syringes, when used with syringe pumps or PCA pumps, may result in recognition, compatibility, and pump performance issues such as overdose, underdose, delay in therapy, and delays in occlusion alarms.

On September 20, 2023, Cardinal Health initiated a recall for certain Cardinal Health Monoject single use Luer-lock tip syringes to not be used with syringe pumps. This recall is a correction, not a product removal. The FDA classified this as a Class I recall, the most serious type of recall. However, the FDA has concerns that the information provided by Cardinal Health has not sufficiently mitigated the risk of incompatibility when Cardinal Health Monoject syringes are used with other pumps, specifically PCA pumps. In addition, the FDA has concerns that the risk of incompatibility could apply to additional sizes of Cardinal Health Monoject syringes. The FDA is working with the manufacturer to address these questions.

Example showing the 20mL Covidien Monoject syringe on the left has different dimensions than the 20mL Cardinal Health Monoject syringe on the right.
Example showing the 20mL Covidien Monoject syringe on the left has different dimensions than the 20mL Cardinal Health Monoject syringe on the right.Recommendations for Health Care Providers

At this time, the FDA recommends that health care providers and facilities:

Do not use **Cardinal Health** Monoject syringes with syringe pumps or PCA pumps.
	You may continue to use **Covidien** Monoject syringes with syringe pumps or PCA pumps.
	Be aware that both brands of syringes state only “monoject” on the syringe itself and do not include the company name (see example in the image above).
		Keep the outer packaging with the **Covidien** Monoject syringe prior to using it with a syringe pump or PCA pump in order for end-users to verify that the syringe can be used.
	Stay alert for further updates and recommendations from the FDA or manufacturers.
	Report any problems with Cardinal Health Monoject syringes to the FDA.Background

Cardinal Health Monoject disposable syringes are used to inject fluid into or withdraw fluids from the body. When used with syringe pumps and PCA pumps, disposable syringes are loaded with fluid or medications and placed into the pump.

Syringe pumps deliver solutions such as fluids, medications, and blood products to patients. PCA pumps typically deliver pain medications.

Currently, the FDA has received over a dozen Medical Device Reports (MDRs) involving delay in therapy as well as inaccurate therapy (overdose or underdose) associated with the use of Cardinal Health Monoject syringes with a syringe pump or a PCA pump. Currently, the FDA is not aware of any deaths associated with this issue.

FDA Actions

The FDA will continue to:

Work with Cardinal Health and other applicable manufacturers to further evaluate the issue and identify additional mitigation strategies as needed.
	Monitor reports of adverse events related to the issue.
	Keep health care providers and the public informed if significant, new information or recommendations become available.Other Affected Devices

The following list of Medical Device Recall Database entries are from infusion pump manufacturers that use Cardinal Health Monoject syringes. The Unique Device Identifier (UDI) information provided by the manufacturer can be found in the "Code Information" section of each entry. The FDA continues to work with syringe pump and PCA pump manufacturers, and this list of recalls associated with Cardinal Health Monoject syringes will be updated as additional recalls are initiated.

Manufacturer Name
			Product Name
			Version or Model
			Date Recall Initiated by the ManufacturerCarefusion 303, Inc.
			Alaris
			BD Alaris
			BD Alaris
			Alaris PCA Module 8120BD Alaris PCU REF 8015BD Alaris Syringe Module, REF 8110
			September 15, 2023

Reporting Problems to the FDA

The FDA encourages health care providers to report any adverse events or suspected adverse events experienced with the Cardinal Health Monoject syringes, or any medical device.

Health care personnel employed by facilities that are subject to the FDA's user facility reporting requirements should follow the reporting procedures established by their facilities.
	Voluntary reports can be submitted through MedWatch, the FDA Safety Information and Adverse Event Reporting program.
	Device manufacturers and user facilities must comply with the applicable Medical Device Reporting (MDR) regulations.By promptly reporting adverse events, you can help the FDA identify and better understand the risks associated with medical devices.

Contact Information

If you have questions about this letter, contact the Division of Industry and Consumer Education (DICE).

                                                                                  Content current as of:
      11/20/2023Letters to Health Care Providers

          
                        2024 Letters to Health Care Providers
          2023 Letters to Health Care Providers
          2022 Letters to Health Care Providers
          2021 Letters to Health Care Providers
 Subscribe to Medical Device Safety and Recalls
                        Sign up to receive email updates on medical device recalls, safety communications, and other safety information.
    Enter your email address to subscribe:
 Footer Links
    FDA Archive
          About FDA
          Accessibility
      Visitor Information
          Website Policies / Privacy
          No FEAR Act
          Vulnerability Disclosure Policy
      FOIA
          HHS.gov
          USA.gov
    Contact FDA
      Follow FDA on Facebook
      Follow FDA on X
      Follow FDA on Instagram
      Follow FDA on LinkedIn
      View FDA videos on YouTube
      Subscribe to FDA RSS feeds
    FDA Homepage
    Contact Number
      1-888-INFO-FDA (1-888-463-6332) Back toTop