### Revised

- Features of Approved Amalgam Separators ( March 200 9 )

The following listing of Features of Approved Amalgam Separators should be used as a replacement for page

14 of your "re: amalgam recovery" booklet. The listing will change occasionally throughout the course of the

voluntary amalgam recovery program. Before making purchasing decisions, please reference this Web site or

call the MDA office at 612

- 767

- 8400.

Man ufacturer or distributor of amalgam separator: Air Techniques

Model name or number: Guardian Amalgam Collector

Removal technology used: Sedimentation

Capacity: Variable, as determined by the volume of the air

- water separator tank

Flow rate factors: 3 liter s per minute

Location: Single central location at the outlet of an air

- water separator

Physical dimensions: Height: 7.5", Width: 10.5", Depth: 6.25"

Cost: $1,500 initial purchase (which includes 1 replacement box, $750 for additional replacement boxes)

Other information: System designed for gravity flow, so the amalgam collection container must be installed

below the air

- water separator. An air

- water separator must be purchased if current vacuum system does not

include one. Cost includes amalgam recycli ng fees. Replacement boxes typically needed every 6

- 12 months.

Manufacturer or distributor of amalgam separator: R&D Services

Model name or number: The Amalgam Collector CH9 or CH12

Removal technology used: Sedimentation

Capacity: 1 per chair

Flow rate fa ctors: Working capacity is 2.3 liters for the CH9 and 3.7 liters for the CH 12

Location: Chairside

Physical dimensions: Height: 9" or 12" (plus 4" for fittings), Diameter: 8"

Cost: $495

Other information: Select height of a unit to be purchased based on yo ur flow rate. Dental office regularly

adjusts outlet tube height, decants liquid and occasionally empties the sludge waste. Canisters may be

replaced or reused. Dental office manages their amalgam waste solids.

Manufacturer or distributor of amalgam separ ator: R&D Services

Model name or number: The Amalgam Collector CE18 or CE24

Removal technology used: Sedimentation

Capacity: Variable

Flow rate factors: Working capacity is 6.5 liters for the CE 18 and 16.4 liters for the CE 24

Physical dimensions: Height: 18" or 24" (plus 4" or 6" for fittings), Diameter: 8" or 10"

Cost: $695 for the CE1 8 and $1,250 for the CE 24

Other information: Select height of a unit to be purchased based on your flow rate. Dental office regularly

adjusts outlet tube height, decants liquid and occasionally empties the sludge waste. Canisters may be

replaced or reused. Dental office manages their amalgam waste solids.

Manufacturer or distributor of amalgam separator: American Dental Accessories

Model name or number: Asdex AS

- 9 , Super Premium

Removal technology used: Filtration

Capacity: One filter unit per chair

Flow rate factors: 1 L

/minute

Location: Chairside

Physical dimensions: Height: 11.5", Diameter: 3"

Cost: $2 03 for the initial unit; $79 per filter ($300 for four filters)

Other information: Dental office replaces a used filter with a new filter every 4

- 6 weeks. Dental office to

manage waste filters.Manufacturer or distributor of amalgam separator: American Dental Accessories

Model name or number: A D

- 1000

Removal technology used: Filtration

Capacity: 12 chairs

Flow rate factors: 2 L

/minute

Location: Single central location at outlet of air/water separator for either a wet or dry vacuum pump system

Physical dimensions: Height: 8", Width 14”, Depth 10. 5”

Cost: $796 for the initial unit; $496 per replacement filter box

Other information: Filter lasts 18 months based on one full time dentist. Cost of replacement filter includes pre

paid mailer and disposal.

Manufacturer or distributor of amalgam separator: Dental Recycling North America (DRNA)

Model name or number: BullfroHg 1 0

Removal technology used: Sedimentation

Capacity: 6 chairs

Flow rate factors: Holding tank volume is 10 liters

Physical dimensions: Height: 20.5", Diameter: 8.5"

Cost: $50/ month service agreement (2 year minimum contract)

Other information: 120 VAC power supply needed to pump settled effluent from unit. Unit utilizes an automatic

timer. Service fee includes recycling of amalgam waste. Unit is available in larger sizes and pr iced accordingly.

Manufacturer or distributor of amalgam separator: Dental Recycling North America (DRNA)

Model name or number: MRU

Removal technology used: Sedimentation, filtration and ion exchange

Capacity: 6 chairs

Flow rate factors: Holding tank volu me is 1 0 liters

Location: Single central location just upstream of the vacuum pump

Physical dimensions: Height: 24", Width 16”, Depth 12"

Cost: Call DRNA for pricing information

Other information: 120 VAC power supply needed to pump settled effluent from unit. Unit utilizes an automatic

timer. Service fee includes recycling of amalgam waste. Unit is available in larger sizes and priced accordingly.

Manufacturer or distributor of amalgam separator: SolmeteX

Model name or number: Hg5

– Rev K

Removal technol ogy used: Sedimentation

Capacity: 10 chairs

Flow rate factors: 1 L

/minute, Surge tank capacity is 4.6 liters

Location: Single central location just upstream of the vacuum pump.

Physical dimensions: Height: 29", Width: 14", Depth: 8"

Cost: $750 for the initial unit, $170 per replacement collection container, $285 per replacement collection

container and recycling

Other information: Filter replaced 1 2 months or when sediment reaches full line.

Manufacturer or distributor of amalgam se parator: SolmeteX

Model name or number: Hg 10

Removal technology used: Sedimentation, filtration, and chemical binding

Capacity: 10

- 25 chairs

Flow rate factors: Not applicable since it uses a 95 liter (25 gallon) holding tank and batch treatment

Location: Single central location downstream of the vacuum pump.

Physical dimensions: Tank Dimensions: Height: 48", Diameter: 25", plus 6 filters mounted on the wall

Cost: $7,450 for the initial unit and filters, $1,050 for a 6

- filter replacement kit. Individual fil ter units are also

available.

Other information: Units are custom made to accommodate an office's flow rate. Filter kit would be replaced

from quarterly to annually. 120V AC power supply needed to pump effluent through the treatment.Manufacturer or distributor of amalgam separator: Maximum Separation Systems

Model name or number: MSS 2000

Removal technology used: Sedimentation and filtration

Capacity: 16 chairs

Flow rate factors:

- Surge tank capacity: 22 liters

- Flow through rate: 2. 5liters/minute

Location: Single central location just upstream of the vacuum pump.

Physical dimensions: Height: 28", Width: 18.5", Depth: 15"

Cost: $1,395 per unit

- $165 per settling tank plus $185 recycling fee.

Other information: Initial cost includes two initial settling tanks, both of which would last one dentist

approximately two years or two dentists approximately one year. 120V AC power supply needed.

Manufacturer or distributor of amalgam separator: Maximu m Separation Systems

Model name or number: MSS 1000

Removal technology used: Sedimentation and filtration

Capacity: 6 chairs

Flow rate factors:

- Surge tank capacity: 11 liters

- Flow through rate: 1 liter/minute

Location: Single central location just upst ream of the vacuum pump.

Physical dimensions: Height: 24", Width: 18.5" , Depth: 15"

Cost: $968 per unit, $165 per settling tank plus $185 recycling fee.

Other information: The MSS 1000 has one settling tank which will last one dentist approximately one ye ar.

1 20V AC power supply needed.

Manufacturer of distributor of amalgam separator: AB Dental Trends

Model name or number: Rasch 890 Single

- Stage series

Removal technology used: Sedimentation, filtration, and ion exchange

Capacity: 12 chairs

Flow rate factors: 2 liters/minute

Location: Single central location at the vacuum pump

Physical dimensions: Height: 9.0", Width: 9", Depth: 12.25"

Note that the dimensions will increase if an air/water separator is necessary to complete the installation.

Cost: $736 for canister and connectors

If an air

- water separator is also necessary, the overall cost will be $865

- $ 1070

Other information: Cost of a replacement canister is $649. The canister is typically replaced approximately

every 18 months per doctor. Cani ster replacement cost includes management of the waste.

Manufacturer or distributor of amalgam separator: AB Dental Trends

Model name or number: Rasch 890

- 4000 Two

- Stage series

Removal technology used: Sedimentation, filtration, and ion exchange.

Capacity: 12 chairs

Flow rate factors: 4 liters/minute

Location: Single central location at the vacuum pump

Physical dimensions: Height: 6", Width: 9", Depth: 12"

Note that the dimensions will increase if an air/water separator is necessary.

Cost: Appro x. $1330, or $1900 overall if an air

- water separator is necessary

Other information: The 4000 series models (4000 & 4500) each use two canisters in series. The cost of each

replacement canister is $649. Each primary stage canister is typically replaced app roximately every 18 months.

Canister cost includes management of the waste. The secondary canister is replaced approximately every 5

years.Manufacturer or distributor of amalgam separator: Rebec LLC

Model name or number: Catch 1000 (formerly called RME 2000

- CatchHg)

Removal technology used: Sedimentation and filtration

Capacity: 8 chairs

Flow rate factors: 1 liter/minute. Surge tank capacity: 19 liters

Location: Single central location just upstream of the vacuum pump.

Physical dimensions: Height: 23.5" , Width: 20", Depth: 8"

Cost: $1,895 per unit

- $395 per replaced settling tank.

Other information: Cost includes processing of the waste. For 8 operatories, the settle tank would be changed

out annually. Larger systems are available.

Manufacturer or distributor of amalgam separator: Rebec LLC

Model name or number: Catch 400

Removal technology used: Sedimentation and filtration

Capacity: 4 chairs

Flow rate factors: 1 L

/minute. Surge tank capacity: 8 liters

Location: Single central location just upstream of the vacuum pump.

Physical dimensions: Height: 18", Width: 14", Depth: 8"

Cost: $985 per unit

- $395 per replaced settling tank.

Other information: Cost includes processing of the waste. For 4 operatories, the settle tank would be changed

out annually. Larger systems are available.

Manufacturer or distributor of amalgam separator: Rebec LLC

Model name or number: Catch 9000

Removal technology used: Sedimentation

Capacity: 30 chairs

Flow rate factors: 1 L /minu te. Surge tank capacity: 33 liters

Location: Single central location just upstream of the vacuum pump.

Physical dimensions: Height: 48", Width: 24", Depth: 22"

Cost: $3,920 per unit. $795 annual collectors/buffer replacement

Other information: Unit comes w ith a metal stand and c an accommodate pipe sizes to 4”. Annual collector

replacement includes shippin g, recycling, and documentation

### The following units are are no longer available in their original configuration

. Both are modified

and are listed above with the appropriate specifications.

Manufacturer or distributor of amalga m separator: SolmeteX

Model name or number: Hg 5

Removal technology used: Sedimentation, f iltration, and chemical binding

Capacity: 10 chairs

Flow rate factors: 50 ml/minute, Surge tank capacity is 4.6 liters

Location: Single central location just upstream of the vacuum pump.

Physical dimensions: Height: 29", Width: 14", Depth: 8"

Cost: $695 for the initial unit; $250 per filter

Other information: Filter replaced approximately every 6 months or when full. The $250 filter purchase includes

recycling.

Manufa cturer or distributor of amalgam separator: American Dental Accessories

Model name or number: Asdex AS

- 9

Removal technology used: Filtration

Capacity: One filter unit per chair

Flow rate factors: 500 ml/minute

Location: Chairside

Physical dimensions: Heigh t: 11.5", Diameter: 3"

Cost: $229 for the initial unit; $79 per filter

Other information: Dental office replaces a used filter with a new filter every 4

- 6 weeks. Dental office to

manage waste filters.