Hu-25

Comes complete with operator’s manual.

Ordering Example: HX15-W, wall-mount RH/temp transmitter wtih 4 to 20 mA outputs, $510.

Specifications

Accuracy:

Relative Humidity: ±2% RH,

3% to 95% @ 25°C (77°F);

±0.05% RH/°C for -40 to 150°C

(-40 to 302°F);

Time Constant: 30 s, 90%

response, 1 ms moving air

Temperature: ±0.5°C (1°F),

-40 to 180°C (-40 to 356°F); time

constant 4 s, 60% response, 1 ms;

moving air

Sensors: RH, thin polymer capacitor;

temp, 1000 Ω Platinum RTD

Power: 7 to 30 Vdc

Dimensions:

Electronics:

55 H x 80 W x 120 mm L

(2.16 x 3.14 x 4.72"); wall mount

Probe: 79 x 16 mm D (3 x 0.625"),

1 m (40") PFA cable, metal wall

mounting clip

Duct Probe: 229 x 16 mm D

(9 x 0.625"), 1 m (40") PFA cable,

removable and adjustable duct flange

 -40 to 180°C (-40 to 356°F)

Operating Range

 Remote Stainless

Steel Probe (Included)

 79 mm (3") Long

Wall-Mounted Probe

 229 mm (9") Long

Duct-Mounted Probe

 Accuracy: 2% RH, 0.5°C

HX15-W, $510.

The HX15 Series 2-wire remote

transmitters measure relative

humidity and temperature over

a large temperature range

-40 to 180°C (-40 to 356oF).

The HX15-W 79 mm (3")

stainless steel probe is used

for wall mounting. The HX15-D

229 mm (9") stainless steel duct

probe has a removable flange.

The probes are connected to

housings with 1 m (40") PFA

cables. Probes are not field

replaceable.

Outputs: 4 to 20 mA for 0 to 100% RH

and 4 to 20 mA for -40 to 180°C

(-40 to 356°F)

Operating Temperature Range:

Electronics: ABS enclosure

NEMA 4X (IP66), -20 to 70°C

(-4 to 158°F)

Probes: -40 to 180°C

(-40 to 356°F)

To Order (Specify Model Number)

Model No. Price Description

HX15-W $510 Wall-mounted RH/temp transmitter [79 mm (3") probe]

HX15-D 559 Duct-mounted RH/temp transmitter

[229 mm (9") probe]

PSU-93 40 Power supply, unregulated 16 to 23 Vdc, 300 mA max

High-Temperature Relative

Humidity/Temperature

Transmitters

HX15 Series

Starts at

$

510

AVAILABLE FOR FAST DELIVERY!
One Omega Drive | Stamford, CT 06907 | 1-888-TC-OMEGA (1-888-826-6342) | info@ome

www.omega.com

More than 100,000 Products Availab

CANADA

www.omega.ca

Laval(Quebec)

1-800-TC-OMEGA

UNITED KINGDOM

www. omega.co.uk

Manchester, England

0800-488-488

GERMANY

www.omega.de

Deckenpfronn, Germany

0800-8266342

FRANCE

www.omega.fr

Guyancourt, France

088-466-342

BENELUX

www.omega.nl

Amstelveen, NL

0800-099-33-44

UNITED STATES

www.omega.com

1-800-TC-OMEGA

Stamford, CT.

CZECH REPUBLIC

www.omegaeng.cz

Karviná, Czech Republic

596-311-899

Temperature

Calibrators,Connectors,General Test and Measurement

Instruments,Glass Bulb Thermometers,Handheld Instruments

for Temperature Measurement, Ice Point References,

Indicating Labels,Crayons,Cements and Lacquers, Infrared

Temperature Measurement Instruments, Recorders Relative

Humidity Measurement Instruments, RTD Probes, Elements

and Assemblies, Temperature & Process Meters,Timers and

Counters, Temperature and Process Controllers and Power

Switching Devices,Thermistor Elements, Probes and

Assemblies,Thermocouples Thermowells and Head and Well

Assemblies,Transmitters,Wire

Pressure, Strain and Force

Displacement Transducers, Dynamic Measureme

Force Sensors, Instrumentation for Pressure and S

Measurements, Load Cells, Pressure Gauges, Pres

Reference Section, Pressure Switches, Pressure Tra

Proximity Transducers, Regulators,

Strain Gages, Torque Transducers, Valves

pH and Conductivity

Heaters

Band Heaters, Cartridge Heaters, Circulation He

Comfort Heaters, Controllers, Meters and Switch

lbll d 

Flow and Level

Air Velocity Indicators, Doppler Flowmeters, Level

Measurement, Magnetic Flowmeters, Mass Flowmeters,

Pitot Tubes, Pumps, Rotameters,Turbine and Paddle Wheel

Flowmeters, Ultrasonic Flowmeters, Valves,Variable Area

Flowmeters, Vortex Shedding Flowmeters

Data Acquisition

Auto-Dialers and Alarm Monitoring Systems,

Communication Products and Converters, Data

Acquisition and Analysis Software, Data Logge

Plug-in Cards, Signal Conditioners, USB, RS232

and Parallel Port Data Acquisition Systems, Wir

Transmitters and Receivers