# Microsoft Corporation

1516

SAR Evaluation Report #: MCSO1607

Report Prepared By Northwest EMC Inc.



## NORTHWEST EMC – (888) 364-2378 – www.nwemc.com

California

– Minnesota

– Oregon

– New York

– WashingtonWTD 12.5.23 

# CERTIFICATE OF EVALUATION

Last Date of Test: July 27, 2012 Microsoft Corporation Model:1516

Applicable Standard Test Description Specification Test Method Pass/Fail FCC OET 65C:2001 IEEE Std 1528:2003 FCC KDB 447498 D01 v04 FCC KDB 248227 D01 v01r02 FCC 2.1093:2012 FCC 15.247:2012 FCC 15.407:2012 FCC KDB 616217 D03 v01 Pass

FCC KDB 865664 SAR Evaluation

Health Safety Code 6:2009 RSS-102, Issue 4:2010 Pass


### Highest SAR Values Frequency Bands (GHz) Head 1g (W/kg) Body 1g (W/kg) Limit 1g (W/kg) Exposure Environment

2.4 N/A 1.44

5.2, 5.3, 5.6, 5.8 N/A 1.5 1.6 General Population Uncontrolled



### Deviations From Test Standards
None



### Approved By:

Don Facteau, IS Manager NVLAP Lab Code: 200630-0

Test Facility

The measurement facility used to collect the data is located at:

Northwest EMC, Inc. 22975 NW Evergreen Parkway, Suite 400 Hillsboro, OR 97124

Phone: (503) 844-4066 Fax: 844-3826

This report must not be used to claim product certification, appr oval, or endorsement by NVLAP, NIST, or any agency of the fede ral government of the United States of America.

Product compliance is the responsibility of the client, therefore the tests and equipment modes of operation represented in thi s report were agreed upon by the client, prior to testing. This Report may only be duplicated in its entirety. The results of this test pertain only to the sample(s) tested. The specific description is noted in each of the individual sections of the test report s upporting this certificate of test.REV 2012.08.02 

# REVISION HISTORY

Revision Number Description Date Page Number

00 None

Barometric Pressure The recorded barometric pressure has been normalized to sea level.

# Accreditations and Authorizations Revision 02/21/2012



### United States

FCC - Designated by the FCC as a Telecommunications Certificati on Body (TCB). Certification chambers, Open Area Test Sites, and conducted measurement facilities are listed with the FCC.
A2LA - Accredited by A2LA to ISO / IEC Guide 65 as a product certifie r. This allows Northwest EMC to certify transmitters to FCC and IC specifications.
NVLAP - Each laboratory is accredited by NVLAP to ISO 17025. The scope includes radio, ITE, and medical standards from around the world. See: http://www.nwemc.com/accreditations/



### Canada

IC - Recognized by Industry Canada as a Certification Body (CB) . Certification chambers and Open Area Test Sites are filed with IC.



### European Union
European Commission – Validated by the European Commission as a Conformity Assessment Body (CAB) under the EMC directive and as a Notified Bo dy under the R&TTE Directive.



### Australia/New Zealand

ACMA - Recognized by ACMA as a CAB for the acceptance of test data.



### Korea

KCC / RRA - Recognized by KCC’s RRA as a CAB for the acceptance of test data.



### Japan

VCCI - Associate Member of the VCCI. Conducted and radiated measurement facilities are registered.



### Taiwan

BSMI – Recognized by BSMI as a CAB for the acceptance of test data.
NCC - Recognized by NCC as a CAB for the acceptance of test data.



### Singapore

IDA – Recognized by IDA as a CAB for the acceptance of test data.



### Hong Kong

OFTA – Recognized by OFTA as a CAB for the acceptance of test data.



### Vietnam

MIC – Recognized by MIC as a CAB for the acceptance of test data.



### Russia

GOST – Accredited by Certinform VNIINMASH, CERTINFO, SAMTES, and Federal CHEC to perform EMC and Hygienic testing for Information Tec hnology products to GOST standards

.REV 2012.08.03 

# LOCATIONS

Oregon Labs EV01-EV12 22975 NW Evergreen Pkwy, #400 Hillsboro, OR 97124 (503) 844-4066 California Labs OC01-OC13 41 Tesla Irvine, CA 92618 (949) 861-8918 New York Labs WA01-WA04 4939 Jordan Rd. Elbridge, NY 13060 (315) 685-0796 Minnesota Labs MN01-MN08 9349 W Broadway Ave. Brooklyn Park, MN 55445 (763) 425-2281 Washington Labs SU01-SU07 14128 339 th Ave. SE Sultan, WA 98294 (360) 793-8675

VCCI

A-0108 A-0029 A-0109 A-0110
Industry Canada
2834D-1, 2834D- 2 2834B-1, 2834B-2, 2834B-3 2834E-1 2834C-1WTD 12.5.23 

# PRODUCT DESCRIPTION

Client and Equipment Unde r Test (EUT) Information

Company Name: Microsoft Corporation Address: One Microsoft Way City, State, Zip: Redmond, WA 98052-6399 Test Requested By: Mike Boucher Model: 1516 First Date of Test: July 17, 2012 Last Date of Test: July 27, 2012 Receipt Date of Samples: July 17, 2012 Equipment Design Stage: Production Equipment Condition: No Damage


### Information Provided by the Party Requesting the Test

Functional Description of the EUT (Equipment Under Test):

The EUT is the Model 1516 tablet computer containing a combination WLAN - Bluetooth radio module. The tablet contains two dual band (2.4 and 5GHz) ant ennas. One antenna (MAIN) is used for both WLAN and Bluetooth, while the other (MIMO) antenna is used only for WLAN operation. The pea k gain in the 2.4 GHz band is 5.2 dBi and the peak gain in the 5 GHz band is 2.2 dBi.. The WLAN and Bluetooth radios can transmit simultaneously.

The WLAN radio is an 802.11a/b/g/n radio module wi th 2x2 MIMO and both 20 MHz and 40 MHz channel bandwidths. The two WLAN antennas transmit simultaneously only in “n” (MIMO) modes. The frequency bands of the 802.11 a/b/g/n radio:

• 2400 – 2483.5 MHz

• 5150 – 5350 MHz

• 5470 – 5600 MHz

• 5650 – 5725 MHz

• 5725 – 5850 MHz

The Bluetooth radio is capable of both basic and extended data rates as well as low energy operation. It operates in the 2400 – 2483.5 MHz frequency band.

The closest spacing between the WLAN antennas is 14 cm. The antennas are near the top edge of the tablet. The closest spacing of the antennas to the user is 3mm.

The diagonal screen size is greater than 20cm (7.9) in ches therefore KDB 941225 is not applicable; instead, KDB 616217 is applicable.

In normal operation, the Model 1516 table is held in the hands. There is no usage model for operation near the head. There are no authorized accessories to wear the t ablet on the body. Only t he tablet configurations anticipated by KDB 616217 are applicable.WTD 12.5.23 

# PRODUCT DESCRIPTION

Testing Objective:

To demonstrate compliance with the SAR requirement s of FCC 2.1093 and Canada’s Health Safety Code 6. This evaluation will be used to support an original Gr ant of Certification for FCC ID: C3K1516, and IC: 3048A1516.

Test Locations

The FCC’s starting point for SAR guidance is KDB 447498 D01 Mobile Portable RF Exposure v04

4) b) The following procedures are applicable to tablet computers with antennas installed along the tablet edges while operating in Tablet Mode. (Foot note 21) When the output power of an antenna is > 60/f(GHz) mW, SAR is required for both bo ttom face and edge exposure conditions.

i) Each antenna is evaluated for bottom face expo sure with the base/bottom of the tablet in direct contact with a flat phantom. Converti ble tablets must be tested in normal use conditions with the display folded on top of the keyboard section. The simultaneous transmission test requirements in item 3) b) ii) (1) may be applied to tablet computers in this operating mode.

ii) Antennas installed along the edges of a tabl et are each evaluated with the corresponding edge in direct contact with a flat phantom. The applicable edge configurations include: (A) one fixed display orientation in either portrait or landscape configuration; (B) two fixed display orientations with one in portrait and one in landscape configurations; and (C) multiple display orientations supporting both portrait and landscape configurations.

(1) For edge configuration (A): SAR is required for each antenna located within 5 cm of the tablet edge closet to the user for the applicabl e display orientation. For antenna(s) located
≥ 5 cm from this edge, the test reduction and exclusion procedures for laptop computers in KDB 616217 are applied. (Footnote 22)

(2) For edge configurations (B) and (C): The procedures for edge configuration (A) are applied to each antenna, for the applicable display orientations where the corresponding edge is closest to the user. For each antenna, SAR is required only for the edge with the most conservative exposure condition.

The bottom face (referred to as “back” in this report ) and the top edge were tested. The antennas are located closest to the top edge.

Simultaneous Transmission

During testing, a KDB analysis was done to determine whether a SAR evaluation is required for simultaneous transmission. SAR KDB 616217 D03 SAR Supplement is the FCC’s Policy for SAR evaluation of Notebooks, Netbooks, Laptops, and Tablet Computers. Whether a SAR evaluation is required for simultaneous transmission is determined by the output power, antenna spacing, and SAR distributions of each antenna.

Since the sum of the highest SAR from each of the indivi dual antennas is greater than 1.6 W/kg, the condition of Item 4(b) of KDB 616217 was applied.

b) “for antennas included in the simultaneous tran smission configuration that require SAR evaluation, when the separation distance between each antenna pair is

i) greater than 5 • [(SAR1 + SAR2) / 1.6] 1.5 cm, rounded to the nearest cm, andWTD 12.5.23 

# PRODUCT DESCRIPTION

ii) the ∑ of [(the highest MPE for each mobile transmitter/antenna included in the simultaneous transmission configuration) / (the corresponding MPE limit)] < 1

where: ∑ in a) excludes antennas that do not r equire SAR evaluation, and MPE does not apply to displays < 10” diagonal for both a) and b)”

Assuming a worst case SAR value of 1.19 W/kg from eac h antenna in MIMO mode, the equation from 4(b)(i) becomes:

5*[(1.19 + 1.19)/1.6] 1.5 = 9 cm

Since the antenna spacing of 14cm is greater than 9 cm, and there are no MPE exposure conditions to consider, simultaneous SAR is not required.

MIMO Evaluation

However, the FCC’s Guidance for SAR testing of 802.11 a/b/g device is found in KDB 248227. It states:

“SAR for MIMO is measured with all antennas transmitting simultaneously.

For many low-power devices, when the peak SAR loca tions are more than 5 cm apart, the 1-g SAR can usually be treated independently with little or no not iceable impact. Therefore spatial summing could be optional”

Although the highest conducted output power modes we re not MIMO, MIMO SAR evaluations were conducted in the 2.4 and 5 GHz bands to show that with a 14 cm antenna spacing, there were no overlapping SAR regions. The zoom scans of each hot spot were centered on the individual antennas. The maximum SAR measured for each MIMO mode was significantly lower than ot her modes reported in this SAR evaluation.

Scope

Per KDB 447498 D01 Mobile Portable RF Exposure v04:

4) b) The following procedures are applicable to tablet computers with antennas installed along the tablet edges while operating in Tablet Mode. (Foot note 21) When the output power of an antenna is > 60/f(GHz) mW, SAR is required for both bottom face and edge exposure conditions.

Since the maximum conducted output power of the Bluet ooth radio is 2.2 mW, which is far below the 25mW threshold specified above, the Bluetooth radio does not require SAR evaluation.

The SAR evaluation documented in this report is for the 802.11 ab/g/n portion of the EUT.WTD 12.5.23 

# CONFIGURATIONS

Configuration MCSO1607- 1

Software/Firmware Running during test Description Version WiFi Commands 1.0.8.18

EUT Description Manufacturer Model/Part Number Serial Number Tablet Computer Microsoft Corporation 1516 EV3

Peripherals in test setup boundary Description Manufacturer Model/Part Number Serial Number USB Ethernet Adapter CISCO USB300M CU901L700750 Power Adapter Unknown 1512 NA2171111

Remote Equipment Outside of Test Setup Boundary Description Manufacturer Model/Part Number Serial Number Computer Lenovo ThinkPad T420s R9-NP0D4 12/04

Cables Cable Type Shield Length (m) Ferrite Connection 1 Connection 2 Power PA 1.8m PA Tablet Computer Computer Ethernet No 5m No USB Ethernet Adapter Computer

USB PA 0.1m PA Tablet Computer USB Ethernet Adapter USB Yes 1.8m No Tablet Computer Unterminated Audio No 1.8m No Tablet Computer Unterminated PA = Cable is permanently attached to the device. Shielding and/or presence of ferrite may be unknown.WTD 12.5.23 

# CONFIGURATIONS

Configuration MCSO1607- 2

Software/Firmware Running during test Description Version WiFi Commands 1.0.8.18

EUT Description Manufacturer Model/Part Number Serial Number Tablet Computer Microsoft Corporation 1516 EV3

Peripherals in test setup boundary Description Manufacturer Model/Part Number Serial Number Power Adapter Unknown 1512 NA2171111 USB Ethernet Adapter Belkin F44047 D12-00135853

Remote Equipment Outside of Test Setup Boundary Description Manufacturer Model/Part Number Serial Number Computer Lenovo ThinkPad T420s R9-NP0D4 12/04

Cables Cable Type Shield Length (m) Ferrite Connection 1 Connection 2 Power PA 1.8m PA Tablet Computer Computer Ethernet No 5m No USB Ethernet Adapter Computer

USB PA 0.1m PA Tablet Computer USB Ethernet Adapter USB Yes 1.8m No Tablet Computer Unterminated Audio No 1.8m No Tablet Computer Unterminated PA = Cable is permanently attached to the device. Shielding and/or presence of ferrite may be unknown.WTD 12.5.23 

# MODIFICATIONS

Equipment Modifications

Item Date Test Modification Note Disposition of EUT

1 7/17/2012 SAR Evaluation Tested as delivered to Test Station. No EMI suppression devices were added or modified during this test. EUT remained at Northwest EMC following the test.

2 7/27/2012 SAR Evaluation Tested as delivered to Test Station. No EMI suppression devices were added or modified during this test. Scheduled testing was completed.WSTD.12.07.20 

# OUTPUT POWER

2.4 AND 5 GHz Bands

Per FCC KDB 248227, the conducted output power was measured at the “default test channels” and at the “required test channels” in each ba nd. Measurements were made while the EUT transmitted at the lowest, middle and t he highest data rates for each channel.

Per FCC KDB 248227, among the channels required for normal testing, SAR must be measured on the highest output channel (highlighted). When the SAR measured on the highest output channel is >0.8 W/kg, SAR evaluation for the other required test channels is necessary.

Output power measurements are on the following pages.WSTD.12.07.20 

# OUTPUT POWER

EUT: 1516 Work Order: MCSO1607 Serial Number: EV3 Date: 07/18/12 Customer: Microsoft Corporation Temperature: 24°C Attendees: Mike Boucher, Steve Stegner Relative Humidity: 45% Customer Project: None Bar. Pressure: 1010.5 mb Tested By:

Brandon Hobbs, Jennifer Herrett Job Site: EV02

Power: 120VAC/60Hz Configuration: NA


### TEST SPECIFICATIONS Specification: Method: FCC 2.1093:2011 FCC OET 65C:2001


### COMMENTS Conducted output power. 20 MHz Channel Bandwidths


### DEVIATIONS FROM TEST STANDARD
None


### RESULTS
Conducted Power (Average)

Channel Frequency Data Rate Modulation Antenna Port A Antenna Port B 

### (MHz) (Mbps) dBm W dBm W

1 BPSK 16.1 0.041 15.6 0.037

11 CCK 16.0 0.040 15.5 0.035

6 OFDM 15.8 0.038 15.2 0.033

36 OFDM 15.1 0.032 15.1 0.032

54 OFDM 15.0 0.031 15.2 0.033

7.2 (MCS0) OFDM 15.2 0.033 15.3 0.033

72.2 (MCS07) OFDM 15.7 0.037 15.2 0.033

14.4 (MCS08) OFDM 15.3 0.034 15.9 0.039 1 2412

144.4 (MCS15) OFDM 15.3 0.034 15.4 0.034

1 BPSK 15.7 0.037 16.2 0.042

11 CCK 15.6 0.037 15.2 0.033

6 OFDM 15.2 0.033 15.3 0.034

36 OFDM 15.1 0.033 15.8 0.038
54 OFDM 15.6 0.036 15.3 0.034
7.2 (MCS0) OFDM 15.2 0.033 15.4 0.035

72.2 (MCS07) OFDM 15.8 0.038 15.4 0.035

14.4 (MCS08) OFDM 15.3 0.034 15.0 0.032 6 2437

144.4 (MCS15) OFDM 15.4 0.035 15.5 0.036

1 BPSK 15.5 0.035 15.8 0.038

11 CCK 14.9 0.031 14.7 0.030

6 OFDM 15.1 0.032 15.4 0.034

36 OFDM 15.0 0.031 15.3 0.034

54 OFDM 14.3 0.027 15.4 0.034 11 2462

7.2 (MCS0) OFDM 14.5 0.028 15.5 0.035VERSION 

# OUTPUT POWER

Conducted Power (Average)
Channel Frequency Data Rate Modulation Antenna Port A Antenna Port B 

### (MHz) (Mbps) dBm W dBm W

72.2 (MCS07) OFDM 15.0 0.032 15.4 0.035

14.4 (MCS08) OFDM 15.1 0.033 15.0 0.032

144.4 (MCS15) OFDM 14.1 0.026 15.5 0.036VERSION 

# OUTPUT POWER

EUT: 1516 Work Order: MCSO1607 Serial Number: EV3 Date: 07/18/12 Customer: Microsoft Corporation Temperature: 24°C Attendees: Mike Boucher, Steve Stegner Relative Humidity: 45% Customer Project: None Bar. Pressure: 1010.5 mb Tested By:

Brandon Hobbs, Jennifer Herrett Job Site: EV02

Power: 120VAC/60Hz Configuration: NA


### TEST SPECIFICATIONS Specification: Method: FCC 2.1093:2011 FCC OET 65C:2001


### COMMENTS Conducted output power. 40 MHz Channel Bandwidths


### DEVIATIONS FROM TEST STANDARD
None


### RESULTS
Conducted Power (Average)
Channels Frequency Data Rate Modulation Antenna Port A Antenna Port B 

### (MHz) (Mbps) dBm W dBm W

7.2 (MCS0) OFDM 16.2 0.041 16.4 0.044 1/5 2422 14.4 (MCS08) OFDM 15.1 0.032 15.2 0.033

7.2 (MCS0) OFDM 16.2 0.041 16.5 0.045 4/8 2437 14.4 (MCS08) OFDM 15.1 0.032 15.3 0.034

7.2 (MCS0) OFDM 16.2 0.041 16.0 0.039 7/11 2452 14.4 (MCS08) OFDM 15.0 0.032 15.4 0.035VERSION 

# OUTPUT POWER

EUT: 1516 Work Order: MCSO1607 Serial Number: EV3 Date: 07/18/12 Customer: Microsoft Corporation Temperature: 24°C Attendees: Mike Boucher, Steve Stegner Relative Humidity: 45% Customer Project: None Bar. Pressure: 1010.5 mb Tested By:

Brandon Hobbs, Jennifer Herrett Job Site: EV02

Power: 120VAC/60Hz Configuration: NA


### TEST SPECIFICATIONS Specification: Method: FCC 2.1093:2011 FCC OET 65C:2001


### COMMENTS Conducted output power. 20 MHz Channel Bandwidths


### DEVIATIONS FROM TEST STANDARD
None


### RESULTS
Conducted Power (Average)

Channel Frequency Data Rate Modulation Antenna Port A Antenna Port B 

### (MHz) (Mbps) dBm W dBm W

6 OFDM 12.6 0.018 12.6 0.018

7.2 (MCS0) OFDM 11.9 0.015 11.4 0.014 36 5180

14.4 (MCS08) OFDM 11.5 0.014 11.4 0.014

6 OFDM 12.6 0.018 12.4 0.017

7.2 (MCS0) OFDM 11.9 0.015 12.3 0.017 40 5200

14.4 (MCS08) OFDM 11.5 0.014 11.8 0.015

6 OFDM 12.5 0.018 12.6 0.018

7.2 (MCS0) OFDM 12.3 0.017 12.1 0.016 44 5220

14.4 (MCS08) OFDM 11.9 0.015 11.8 0.015

6 OFDM 12.0 0.016 13.0 0.020

7.2 (MCS0) OFDM 11.8 0.015 12.4 0.017 48 5240

14.4 (MCS08) OFDM 11.5 0.014 12.1 0.016

6 OFDM 12.4 0.017 12.7 0.019

7.2 (MCS0) OFDM 11.8 0.015 12.2 0.016 52 5260

14.4 (MCS08) OFDM 11.4 0.014 11.8 0.015

6 OFDM 12.4 0.017 13.0 0.020

7.2 (MCS0) OFDM 11.7 0.015 12.4 0.017 56 5280

14.4 (MCS08) OFDM 11.3 0.014 12.0 0.016

6 OFDM 12.4 0.017 12.6 0.018

7.2 (MCS0) OFDM 11.7 0.015 12.1 0.016 60 5300

14.4 (MCS08) OFDM 11.3 0.013 11.7 0.015

6 OFDM 12.2 0.017 12.3 0.017 64 5320

7.2 (MCS0) OFDM 11.5 0.014 11.7 0.015VERSION 

# OUTPUT POWER

Conducted Power (Average)
Channel Frequency Data Rate Modulation Antenna Port A Antenna Port B 

### (MHz) (Mbps) dBm W dBm W

14.4 (MCS08) OFDM 11.1 0.013 11.3 0.013

6 OFDM 12.5 0.018 12.6 0.018

7.2 (MCS0) OFDM 11.9 0.015 12.0 0.016 100 5500

14.4 (MCS08) OFDM 11.5 0.014 11.6 0.014

6 OFDM 12.9 0.019 12.7 0.019

7.2 (MCS0) OFDM 12.2 0.017 12.2 0.017 104 5520

14.4 (MCS08) OFDM 11.8 0.015 11.8 0.015

6 OFDM 12.6 0.018 12.9 0.019

7.2 (MCS0) OFDM 11.9 0.016 12.3 0.017 108 5540

14.4 (MCS08) OFDM 11.5 0.014 12.0 0.016

6 OFDM 12.8 0.019 13.0 0.020

7.2 (MCS0) OFDM 12.1 0.016 12.5 0.018 112 5560

14.4 (MCS08) OFDM 11.7 0.015 12.1 0.016

6 OFDM 12.6 0.018 12.6 0.018

7.2 (MCS0) OFDM 12.4 0.017 12.5 0.018 116 5580

14.4 (MCS08) OFDM 12.0 0.016 11.6 0.015

6 OFDM 12.9 0.019 12.9 0.020

7.2 (MCS0) OFDM 12.2 0.017 12.4 0.017 132 5660

14.4 (MCS08) OFDM 11.7 0.015 11.9 0.016

6 OFDM 11.6 0.014 12.4 0.018

7.2 (MCS0) OFDM 11.9 0.015 11.9 0.015 136 5680

14.4 (MCS08) OFDM 11.4 0.014 11.5 0.014

6 OFDM 12.0 0.016 11.9 0.016

7.2 (MCS0) OFDM 11.8 0.015 11.9 0.016 140 5700

14.4 (MCS08) OFDM 11.3 0.014 11.5 0.014

6 OFDM 12.1 0.016 12.1 0.016

7.2 (MCS0) OFDM 11.4 0.014 12.5 0.018 149 5745

14.4 (MCS08) OFDM 10.8 0.012 11.6 0.015

6 OFDM 12.5 0.018 12.7 0.018

7.2 (MCS0) OFDM 11.8 0.015 12.1 0.016 153 5765

14.4 (MCS08) OFDM 11.3 0.014 11.7 0.015

6 OFDM 12.4 0.017 12.7 0.019

7.2 (MCS0) OFDM 11.7 0.015 12.1 0.016 157 5785

14.4 (MCS08) OFDM 11.3 0.013 11.6 0.014

6 OFDM 11.9 0.015 12.7 0.018

7.2 (MCS0) OFDM 11.7 0.015 12.1 0.016 161 5805

14.4 (MCS08) OFDM 11.2 0.013 11.6 0.015
165 5825 6 OFDM 12.2 0.017 12.6 0.018VERSION 

# OUTPUT POWER

Conducted Power (Average)
Channel Frequency Data Rate Modulation Antenna Port A Antenna Port B 

### (MHz) (Mbps) dBm W dBm W

7.2 (MCS0) OFDM 11.5 0.014 12.0 0.016

14.4 (MCS08) OFDM 11.1 0.013 11.6 0.014VERSION 

# OUTPUT POWER

EUT: 1516 Work Order: MCSO1607 Serial Number: EV3 Date: 07/18/12 Customer: Microsoft Corporation Temperature: 24°C Attendees: Mike Boucher, Steve Stegner Relative Humidity: 45% Customer Project: None Bar. Pressure: 1010.5 mb Tested By:

Brandon Hobbs, Jennifer Herrett Job Site: EV02

Power: 120VAC/60Hz Configuration: NA


### TEST SPECIFICATIONS Specification: Method: FCC 2.1093:2011 FCC OET 65C:2001


### COMMENTS Conducted output power. 40 MHz Channel Bandwidths


### DEVIATIONS FROM TEST STANDARD
None


### RESULTS
Conducted Power (Average)
Channels Frequency Data Rate Modulation Antenna Port A Antenna Port B 

### (MHz) (Mbps) dBm W dBm W

7.2 (MCS0) OFDM 12.2 0.016 12.4 0.017 36/40 5190 14.4 (MCS08) OFDM 12.1 0.016 11.8 0.015

7.2 (MCS0) OFDM 12.9 0.019 12.6 0.018 44/48 5230 14.4 (MCS08) OFDM 12.0 0.016 12.0 0.016

7.2 (MCS0) OFDM 12.8 0.019 12.5 0.018 52/56 5270 14.4 (MCS08) OFDM 11.9 0.015 12.0 0.016

7.2 (MCS0) OFDM 12.5 0.018 12.8 0.019 60/64 5310 14.4 (MCS08) OFDM 11.6 0.014 11.8 0.015

7.2 (MCS0) OFDM 12.7 0.018 12.8 0.019 100/104 5510 14.4 (MCS08) OFDM 11.7 0.015 11.7 0.015

7.2 (MCS0) OFDM 12.7 0.018 13.1 0.020 108/112 5550 14.4 (MCS08) OFDM 11.7 0.015 12.0 0.016

7.2 (MCS0) OFDM 12.9 0.019 13.0 0.020 132/136 5670 14.4 (MCS08) OFDM 11.9 0.015 12.0 0.016

7.2 (MCS0) OFDM 12.0 0.016 12.3 0.017 149/153 5755 14.4 (MCS08) OFDM 10.9 0.012 11.7 0.015

7.2 (MCS0) OFDM 12.4 0.017 12.3 0.017 157/161 5795 14.4 (MCS08) OFDM 11.4 0.014 11.7 0.015WSTE.12.07.12 

# TISSUE – EQUIVALENT LIQUID

Characterization of tissue-equivalent liquid dielectric properties

Per IEEE 1528: 2003, Section 5.2.2, the permittivity and co nductivity of the tissue ma terial should be measured at least within 24 hours of any full-compliance test. The measured values must be within +/- 5% of the target values. The temperature variation in the liquid during SAR measurements must be with in +/- 2 degrees C of that recorded when the dielectric properties were measured.

The dielectric parameters of the tissue-equivalent liquids were measured within 24 hours of the start of testing using the HP85070E dielectric probe kit. The dielectric measurements were made across the frequency range of the liquid. The attached data sheets show that the dielectric parameters of the liquid were within the required 5% tolerances.

Target values of dielectric parameters

Per FCC OET 65C, Appendix C:

“The head tissue dielectric parameters recommen ded by the IEEE SCC-34/SC-2 in P1528 have been incorporated in the following table. These head paramete rs are derived from planar layer models simulating the highest expected SAR for the dielectric properties an d tissue thickness variations in a human head. Other head and body tissue parameters that have not been specified in P1528 are derived from the tissue dielectric parameters computed from the 4-Cole-Cole equations and extrapolated according to the head parameters specified in P1528.”WSTE.12.07.12 

# TISSUE – EQUIVALENT LIQUID

Composition of Ingredients for Liquid Tissue Phantoms

Northwest EMC uses tissue-equivalent liquids prepared by SPEAG and confirmed by them to be within +/- 5% from the target values. Their recipes are based upon the following formulations as found in FCC OET 65C, Appendix C:

“The following tissue formulations are provided for re ference only as some of the parameters have not been thoroughly verified. The composition of ingredients may be modified accordingly to achieve the desired target tissue parameters required for routine SAR evaluation.”WSTE.12.07.12 

# TISSUE – EQUIVALENT LIQUID

Tissue : Body Work Order: MCSO1607 Serial Number: SAV Date: 07/16/2012 Customer: Microsoft Corporation Temperature: 23.8°C Customer Project: None Liquid Temperature: 21.8°C Tested By: Ethan Schoonover Relative Humidity: 47.1% Job Site: Cal Lab Bar. Pressure: 1012.2 mb


### TEST SPECIFICATIONS Specification: Method:
FCC 2.1093:2012 FCC OET 65C:2001



### RESULTS
Actual Values Target Values Deviation (%)

Frequency

(MHz) Relative Permittivity Conductivity Relative Permittivity Conductivity Relative Permittivity Conductivity

5800 46.5 6.279 48.2 6 3.53 -4.65

Frequency

(MHz) Relative Permittivity Conductivity

3400 50.5 3.236 3500 50.4 3.342 3600 50.3 3.449 3700 50.2 3.573 3900 49.9 3.782 4000 49.8 3.901 4100 49.6 4.013 4300 49.2 4.267 4400 49.1 4.405 4500 49 4.536 4600 48.8 4.67 4800 48.4 4.931 4850 48.3 4.994 4900 48.3 5.063 5000 48.1 5.197 5050 48 5.259 5100 47.8 5.322 5200 47.7 5.461 5250 47.6 5.525 5300 47.5 5.583 5350 47.4 5.659 5450 47.1 5.796 5500 47.1 5.86 5550 47 5.927 5650 46.8 6.069 5700 46.7 6.138 5750 46.6 6.195 5800 46.5 6.279 5850 46.4 6.351 5900 46.4 6.432WSTE.12.07.12 

# TISSUE – EQUIVALENT LIQUID

Tissue : Body Work Order: MCSO1607 Serial Number: EV3 Date: 07/25/2012 Customer: Microsoft Corporation Temperature: 23.2°C Customer Project: None Liquid Temperature: 22.7°C Tested By: Jennifer Herrett Relative Humidity: 45.8% Job Site: EV08 Bar. Pressure: 1023.9 mb


### TEST SPECIFICATIONS Specification: Method:
FCC 2.1093:2012 FCC OET 65C:2001



### RESULTS
Actual Values Target Values Deviation (%)

Frequency

(MHz) Relative Permittivity Conductivity Relative Permittivity Conductivity Relative Permittivity Conductivity

5800 46.3 6.201 48.2 6 3.94 -3.35

Frequency

(MHz) Relative Permittivity Conductivity

3400 50.1 3.249 3500 50.1 3.352 3600 50 3.456 3700 49.9 3.575 3900 49.5 3.777 4000 49.4 3.89 4100 49.2 4.002 4300 48.9 4.246 4400 48.8 4.377 4500 48.7 4.505 4600 48.5 4.637 4800 48.2 4.899 4850 48.1 4.961 4900 48.1 5.024 5000 47.9 5.151 5050 47.8 5.212 5100 47.5 5.275 5200 47.4 5.398 5250 47.3 5.462 5300 47.2 5.519 5350 47.1 5.589 5450 46.9 5.73 5500 46.9 5.789 5550 46.8 5.85 5650 46.6 5.994 5700 46.5 6.065 5750 46.4 6.12 5800 46.3 6.201 5850 46.2 6.256 5900 46.2 6.315WSTE.12.07.12 

# TISSUE – EQUIVALENT LIQUID

Tissue : Body Work Order: MCSO1607 Serial Number: EV3 Date: 07/25/2012 Customer: Microsoft Corporation Temperature: 24.3°C Customer Project: None Liquid Temperature: 22.6°C Tested By: Ethan Schoonover Relative Humidity: 41.7% Job Site: EV08 Bar. Pressure: 1015 mb


### TEST SPECIFICATIONS Specification: Method:
FCC 2.1093:2012 FCC OET 65C:2001



### RESULTS
Actual Values Target Values Deviation (%)

Frequency

(MHz) Relative Permittivity Conductivity Relative Permittivity Conductivity Relative Permittivity Conductivity

2450 50.1 1.943 52.7 1.95 4.93 0.36

Frequency

(MHz) Relative Permittivity Conductivity

1900 68.9 4.16 1925 55.3 0.955 1950 55.6 0.89 1975 55.4 0.922 2000 55.2 0.97 2025 55.1 1.023 2050 54.9 1.082 2100 54.4 1.2 2125 54.1 1.255 2150 53.8 1.32 2175 53.6 1.375 2200 53.2 1.427 2225 52.9 1.49 2250 52.6 1.546 2300 52 1.65 2325 51.7 1.7 2350 51.4 1.749 2375 51 1.806 2400 50.7 1.849 2425 50.4 1.901 2450 50.1 1.943 2500 49.5 2.029 2525 49.2 2.076 2550 48.9 2.126 2575 48.6 2.178 2600 48.2 2.221 2625 47.9 2.261 2675 47.3 2.331 2700 47 2.362WSSV.12.07.17 

# SAR SYSTEM VERIFICATION

REQUIREMENT Per IEEE 1528, Section 8.2.1, “System checks are performed prior to compliance tests and the results must always be within ± 10% of the target value corresponding to the test frequency, liquid, and the source used. The target values are 1 g or 10 g averaged SAR values m easured on systems having current system validation and calibration status, and using the system check setup as shown in Figure 14. These target values should be determined using a standard source.”



### TEST DESCRIPTION Within 24 hours of the start of testing, then ever y 72 hours thereafter, Northwest EMC used the system validation kit (calibrated reference dipole) to test whether the system was operating within its specifications. The validation was performed in the indicated bands by making SAR measurements of the reference dipole with the phantom filled with the tissue-equivalent liquid. First, a signal generator and power amplifier were used to produce a 100mW level as measured with a power mete r at the antenna terminals of the dipole. Then, the reference dipole was positioned below the bottom of the phantom and centered with its axis parallel to the longest side of the phantom. A low loss and low relative permittivity spacer was used to establish the correct distance between the center axis of the reference dipole and the liquid.

For the reference dipoles, the spacing distance s is given by:

s = 15mm, +/- 0.2mm for 300MHz ≤ f ≥ 1000 MHz: s = 10mm, +/- 0.2mm for 1000MHz ≤ f ≥ 6000MHz

The measured 1 g and 10 g spatial average SAR values were normalized to a 1W dipole input power for comparison to the calibration data. The results are summa rized in the attached table. The deviation is less than 10% in all cases, indicating that the syst em performance check was within tolerance.WSSV.12.07.17 

# SAR SYSTEM VERIFICATION

EUT: 1516 Work Order: MCSO1607 Customer: Microsoft Corporation Job Site: EV08 Attendees: None Customer Project: None


### TEST SPECIFICATIONS Specification: Method:

FCC 2.1093:2012 FCC 15.247:2012 FCC 15.407:2012 FCC OET 65C:2001 IEEE Std 1528:2003 FCC KDB 447498 D01 v04 FCC KDB 248227 D01 V01r02 FCC KDB 616217 D03 v01 FCC 865664 Health Safety Code 6:2009 RSS-102, Issue 4:2010


### COMMENTS
None


### DEVIATIONS FROM TEST STANDARD
None


### RESULTS

Measured Normalized to 1W Target (Normalized to 1W)

Get from Dipole Calibration Certificate % Difference Date Liquid part number and frequency Conducted Power into the Dipole

(dBm) Correction Factor 1g 10g 1g 10g 1g 10g 1g 10g

7/26/2012 MSL2450 2450MHz 20 10 5.13 2.37 51.3 23.7 50.4 23.7 1.79 0

7/23/2012 MSL2450 2450MHz 20 10 5.22 2.42 52.2 24.2 50.4 23.7 3.57 2.11WSSV.12.07.17 

# SAR SYSTEM VERIFICATION

Tested By: Jennifer Herrett Room Temperature (°C): 23.7°C Date: 7/26/2012 Liquid Temperature (°C): 22.7°C Serial Number: 855 Humidity (%RH): 46.9% Configuration: N/A Bar. Pressure (mb): 1015.6 mb Comments: None

HSL2450 System Check_7-26-12 DUT: Dipole 2450 MHz D2450V2; Type: D2450V2; Serial: D2450V2 - SN:855 Communication System: CW; Communication System Band: D2450 (2450.0 MHz); Frequency: 2450 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used: f = 2450 MHz; σ = 1.943 mho/m; ε r = 50.111; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

System Check/System Check/Area Scan (51x61x1): Measurement grid: dx=10mm, dy=10mm Maximum value of SAR (interpolated) = 5.37 mW/g

System Check/System Check/Zoom Scan (7x7x7) (7x7x7)/Cube 0: Measurement grid: dx=5mm, dy=5mm, dz=5mm Reference Value = 51.575 V/m; Power Drift = -0.02 dB Peak SAR (extrapolated) = 10.554 mW/g SAR(1 g) = 5.13 mW/g; SAR(10 g) = 2.37 mW/g Maximum value of SAR (measured) = 5.13 mW/g

System Check/System Check/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm Maximum value of Total (measured) = 9.77 mW/g

Approved ByWSSV.12.07.17 

# SAR SYSTEM VERIFICATION

HSL2450 System Check_7-26-12WSSV.12.07.17 

# SAR SYSTEM VERIFICATION

Tested By: Ethan Schoonover Room Temperature (°C): 24.3°C Date: 7/23/2012 Liquid Temperature (°C): 22.6°C Serial Number: ADL Humidity (%RH): 41.7% Configuration: N/A Bar. Pressure (mb): 1015 mb Comments: None

MSL2450 System Check 2450MHz_7-23-12 DUT: Dipole 2450 MHz D2450V2; Type: D2450V2; Serial: D2450V2 - SN:ADL Communication System: CW; Communication System Band: D2450 (2450.0 MHz); Frequency: 2450 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used: f = 2450 MHz; σ = 1.943 mho/m; ε r = 50.111; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

System Check/System Check/Area Scan (51x61x1): Measurement grid: dx=10mm, dy=10mm Maximum value of SAR (interpolated) = 5.50 mW/g

System Check/System Check/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm Maximum value of Total (measured) = 73.71 V/m

System Check/System Check/Zoom Scan (7x7x7) (7x7x7)/Cube 0: Measurement grid: dx=5mm, dy=5mm, dz=5mm Reference Value = 51.185 V/m; Power Drift = -0.02 dB Peak SAR (extrapolated) = 10.605 mW/g SAR(1 g) = 5.22 mW/g; SAR(10 g) = 2.42 mW/g Maximum value of SAR (measured) = 5.18 mW/g

Approved ByWSSV.12.07.17 

# SAR SYSTEM VERIFICATION

MSL2450 System Check 2450MHz_7-23-12WSSV.12.07.17 

# SAR SYSTEM VERIFICATION

REQUIREMENT Per IEEE 1528, Section 8.2.1, “System checks are performed prior to compliance tests and the results must always be within ± 10% of the target value corresponding to the test frequency, liquid, and the source used. The target values are 1 g or 10 g averaged SAR values m easured on systems having current system validation and calibration status, and using the system check setup as shown in Figure 14. These target values should be determined using a standard source.”



### TEST DESCRIPTION Within 24 hours of the start of testing, then ever y 72 hours thereafter, Northwest EMC used the system validation kit (calibrated reference dipole) to test whether the system was operating within its specifications. The validation was performed in the indicated bands by making SAR measurements of the reference dipole with the phantom filled with the tissue-equivalent liquid. First, a signal generator and power amplifier were used to produce a 100mW level as measured with a power mete r at the antenna terminals of the dipole. Then, the reference dipole was positioned below the bottom of the phantom and centered with its axis parallel to the longest side of the phantom. A low loss and low relative permittivity spacer was used to establish the correct distance between the center axis of the reference dipole and the liquid.

For the reference dipoles, the spacing distance s is given by:

s = 15mm, +/- 0.2mm for 300MHz ≤ f ≥ 1000 MHz: s = 10mm, +/- 0.2mm for 1000MHz ≤ f ≥ 6000MHz

The measured 1 g and 10 g spatial average SAR values were normalized to a 1W dipole input power for comparison to the calibration data. The results are summa rized in the attached table. The deviation is less than 10% in all cases, indicating that the syst em performance check was within tolerance.WSSV.12.07.17 

# SAR SYSTEM VERIFICATION

EUT: 1516 Work Order: MCSO1607 Customer: Microsoft Corporation Job Site: EV08 Attendees: None Customer Project: None


### TEST SPECIFICATIONS Specification: Method:

FCC 2.1093:2012 FCC 15.247:2012 FCC 15.407:2012 FCC OET 65C:2001 IEEE Std 1528:2003 FCC KDB 447498 D01 v04 FCC KDB 248227 D01 V01r02 FCC KDB 616217 D03 v01 FCC 865664 Health Safety Code 6:2009 RSS-102, Issue 4:2010


### COMMENTS
None


### DEVIATIONS FROM TEST STANDARD
None


### RESULTS

Measured Normalized to 1W Target (Normalized to 1W)

Get from Dipole Calibration Certificate % Difference Date Liquid part number and frequency Conducted Power into the Dipole

(dBm) Correction Factor 1g 10g 1g 10g 1g 10g 1g 10g

7/16/2012 MSL501 5200MHz 20 10 8.07 2.3 80.7 23 76. 5 21.6 5.49 6.48

7/16/2012 MSL501 5500MHz 20 10 7.82 2.2 78.2 22 83.3 23.4 -6.12 -5.98

7/16/2012 MSL501 5800MHz 18.45 14.29 5.21 1.47 74.45 21.01 78 21.9 -4.55 -4.06

7/19/2012 MSL501 5200MHz 20.71 8.49 9.58 2.76 81.33 23.43 76.5 21.6 6.31 8.47

7/19/2012 MSL501 5500MHz 20.33 9.27 9.27 2.61 85.93 24.19 83.3 23.4 3.16 3.38

7/19/2012 MSL501 5800MHz 19.28 11.8 6.11 1.74 72.1 20.53 78 21.9 -7.56 -6.26

7/23/2012 MSL501 5200MHz 20.41 9.1 8.31 2.38 75.62 21.66 76. 5 21.6 -1.15 0.28

7/23/2012 MSL501 5500MHz 20.41 9.1 9.56 2.75 87 25.02 83. 3 23.4 4.44 6.92

7/23/2012 MSL501 5800MHz 19.27 11.83 6.47 1.83 76.54 21.65 78 21.9 -1.87 -1.14

7/26/2012 MSL501 5800MHz 18.24 15 5.08 1.46 76.2 21.9 78 21.9 -2.31 0

7/26/2012 MSL501 5500MHz 19.36 11.59 7.59 2.14 87.97 24.8 83.3 23.4 5.61 5.98WSSV.12.07.17 

# SAR SYSTEM VERIFICATION

Tested By: Ethan Schoonover Room Temperature (°C): 22.2°C Date: 7/16/2012 Liquid Temperature (°C): 22.3°C Serial Number: ADM Humidity (%RH): 41.7% Configuration: N/A Bar. Pressure (mb): 1007 mb Comments: None

MSL501 System Check_5200MHz_7-16-12 DUT: Dipole D5GHzV2; Type: D5GHzV 2; Serial: D5GHzV2 - SN:1066 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5200 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used: f = 5200 MHz; σ = 5.461 mho/m; ε r = 47.665; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

• DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

System Check/System Check - Low Channel/Zoom Scan (7x9x7) (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 58.550 V/m; Power Drift = 0.23 dB Peak SAR (extrapolated) = 30.454 mW/g SAR(1 g) = 8.07 mW/g; SAR(10 g) = 2.3 mW/g Maximum value of SAR (measured) = 16.9 mW/g

System Check/System Check - Low Channel/Area Scan (51x61x1): Measurement grid: dx=10mm, dy=10mm Maximum value of SAR (interpolated) = 17.1 mW/g

System Check/System Check - Low Channel/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm Maximum value of Total (measured) = 60.12 V/m

Approved ByWSSV.12.07.17 

# SAR SYSTEM VERIFICATION

MSL501 System Check_5200MHz_7-16-12WSSV.12.07.17 

# SAR SYSTEM VERIFICATION

Tested By: Ethan Schoonover Room Temperature (°C): 22.2°C Date: 7/16/2012 Liquid Temperature (°C): 22.3°C Serial Number: ADM Humidity (%RH): 41.7% Configuration: N/A Bar. Pressure (mb): 1007 mb Comments: None

MSL501 System Check_5500MHz_7-16-12 DUT: Dipole D5GHzV2; Type: D5GHzV 2; Serial: D5GHzV2 - SN:1066 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5500 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used: f = 5500 MHz; σ = 5.86 mho/m; ε r = 47.139; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

• DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

System Check/System Check - Mid Channel/Area Scan (51x61x1): Measurement grid: dx=10mm, dy=10mm Maximum value of SAR (interpolated) = 17.5 mW/g

System Check/System Check - Mid Cha nnel/Zoom Scan (7x9x7) (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 57.599 V/m; Power Drift = 0.11 dB Peak SAR (extrapolated) = 30.793 mW/g SAR(1 g) = 7.82 mW/g; SAR(10 g) = 2.2 mW/g Maximum value of SAR (measured) = 16.6 mW/g

System Check/System Check - Mid Channel/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm Maximum value of Total (measured) = 58.32 V/m

Approved ByWSSV.12.07.17 

# SAR SYSTEM VERIFICATION

MSL501 System Check_5500MHz_7-16-12WSSV.12.07.17 

# SAR SYSTEM VERIFICATION

Tested By: Ethan Schoonover Room Temperature (°C): 22.2°C Date: 7/16/2012 Liquid Temperature (°C): 22.3°C Serial Number: ADM Humidity (%RH): 41.7% Configuration: N/A Bar. Pressure (mb): 1007 mb Comments: None

MSL501 System Check_5800MHz_7-16-12 DUT: Dipole D5GHzV2; Type: D5GHzV 2; Serial: D5GHzV2 - SN:1066 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5800 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used: f = 5800 MHz; σ = 6.279 mho/m; ε r = 46.517; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

• DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

System Check/System Check - Hi gh Channel/Area Scan (51x61x1): Measurement grid: dx=10mm, dy=10mm Maximum value of SAR (interpolated) = 11.1 mW/g

System Check/System Check - High Channel/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm Maximum value of Total (measured) = 37.92 V/m

System Check/System Check - High Chan nel/Zoom Scan (7x9x7) (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 37.253 V/m; Power Drift = 0.16 dB Peak SAR (extrapolated) = 21.177 mW/g SAR(1 g) = 5.21 mW/g; SAR(10 g) = 1.47 mW/g Maximum value of SAR (measured) = 11.2 mW/g

Approved ByWSSV.12.07.17 

# SAR SYSTEM VERIFICATION

MSL501 System Check_5800MHz_7-16-12WSSV.12.07.17 

# SAR SYSTEM VERIFICATION

Tested By: Jennifer Herrett Room Temperature (°C): 23.9°C Date: 7/19/2012 Liquid Temperature (°C): 23.9°C Serial Number: 1066 Humidity (%RH): 54.7% Configuration: N/A Bar. Pressure (mb): 1015 mb Comments: None

MSL501 System Checks_5200MHz_7-19-12 DUT: Dipole D5GHzV2; Type: D5GH zV2; Serial: D5GHzV2 - SN1066 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5200 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used: f = 5200 MHz; σ = 5.175 mho/m; ε r = 47.326; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

• DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

System Check/System Check - Low Channel/Zoom Scan (7x9x7) (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 67.120 V/m; Power Drift = 0.14 dB Peak SAR (extrapolated) = 35.286 mW/g SAR(1 g) = 9.58 mW/g; SAR(10 g) = 2.76 mW/g Maximum value of SAR (measured) = 19.8 mW/g

System Check/System Check - Low Channel/Area Scan (51x61x1): Measurement grid: dx=10mm, dy=10mm Maximum value of SAR (interpolated) = 20.3 mW/g

System Check/System Check - Low Channel/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm Maximum value of Total (measured) = 68.23 V/m

Approved ByWSSV.12.07.17 

# SAR SYSTEM VERIFICATION

MSL501 System Checks_5200MHz_7-19-12WSSV.12.07.17 

# SAR SYSTEM VERIFICATION

Tested By: Jennifer Herrett Room Temperature (°C): 23.9°C Date: 7/19/2012 Liquid Temperature (°C): 23.9°C Serial Number: 1066 Humidity (%RH): 54.7% Configuration: N/A Bar. Pressure (mb): 1015 mb Comments: None

MSL501 System Checks_5500MHz_7-19-12 DUT: Dipole D5GHzV2; Type: D5GH zV2; Serial: D5GHzV2 - SN1066 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5500 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used: f = 5500 MHz; σ = 5.549 mho/m; ε r = 46.752; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

• DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

System Check/System Check - Mid Channel/Area Scan (51x61x1): Measurement grid: dx=10mm, dy=10mm Maximum value of SAR (interpolated) = 20.3 mW/g

System Check/System Check - Mid Cha nnel/Zoom Scan (7x9x7) (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 65.236 V/m; Power Drift = 0.06 dB Peak SAR (extrapolated) = 35.587 mW/g SAR(1 g) = 9.27 mW/g; SAR(10 g) = 2.61 mW/g Maximum value of SAR (measured) = 19.4 mW/g

System Check/System Check - Mid Channel/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm Maximum value of Total (measured) = 65.71 V/m

Approved ByWSSV.12.07.17 

# SAR SYSTEM VERIFICATION

MSL501 System Checks_5500MHz_7-19-12WSSV.12.07.17 

# SAR SYSTEM VERIFICATION

Tested By: Jennifer Herrett Room Temperature (°C): 23.9°C Date: 7/19/2012 Liquid Temperature (°C): 23.9°C Serial Number: 1066 Humidity (%RH): 54.7% Configuration: N/A Bar. Pressure (mb): 1015 mb Comments: None

MSL501 System Checks_5800MHz_7-19-12 DUT: Dipole D5GHzV2; Type: D5GH zV2; Serial: D5GHzV2 - SN1066 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5800 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used: f = 5800 MHz; σ = 5.919 mho/m; ε r = 46.041; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

• DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

System Check/System Check - Hi gh Channel/Area Scan (51x61x1): Measurement grid: dx=10mm, dy=10mm Maximum value of SAR (interpolated) = 13.2 mW/g

System Check/System Check - High Channel/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm Maximum value of Total (measured) = 41.92 V/m

System Check/System Check - High Chan nel/Zoom Scan (7x9x7) (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 42.060 V/m; Power Drift = -0.03 dB Peak SAR (extrapolated) = 23.907 mW/g SAR(1 g) = 6.11 mW/g; SAR(10 g) = 1.74 mW/g Maximum value of SAR (measured) = 13.0 mW/g

Approved ByWSSV.12.07.17 

# SAR SYSTEM VERIFICATION

MSL501 System Checks_5800MHz_7-19-12WSSV.12.07.17 

# SAR SYSTEM VERIFICATION

Tested By: Ethan Schoonover Room Temperature (°C): 23.2°C Date: 7/23/2012 Liquid Temperature (°C): 21.5°C Serial Number: ADM Humidity (%RH): 41.7% Configuration: N/A Bar. Pressure (mb): 1015 mb Comments: None

MSL501 System Check_5200MHz_7-23-12 DUT: Dipole D5GHzV2; Type: D5GHzV 2; Serial: D5GHzV2 - SN:1066 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5200 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used: f = 5200 MHz; σ = 5.398 mho/m; ε r = 47.429; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

• DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

System Check/System Check - Low Channel/Zoom Scan (7x9x7) (7x7x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 60.329 V/m; Power Drift = 0.06 dB Peak SAR (extrapolated) = 30.630 mW/g SAR(1 g) = 8.31 mW/g; SAR(10 g) = 2.38 mW/g Maximum value of SAR (measured) = 17.4 mW/g

System Check/System Check - Low Channel/Area Scan (51x61x1): Measurement grid: dx=10mm, dy=10mm Maximum value of SAR (interpolated) = 17.6 mW/g

System Check/System Check - Low Channel/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm Maximum value of Total (measured) = 60.80 V/m

Approved ByWSSV.12.07.17 

# SAR SYSTEM VERIFICATION

MSL501 System Check_5200MHz_7-23-12WSSV.12.07.17 

# SAR SYSTEM VERIFICATION

Tested By: Ethan Schoonover Room Temperature (°C): 23.2°C Date: 7/23/2012 Liquid Temperature (°C): 21.5°C Serial Number: ADM Humidity (%RH): 41.7% Configuration: N/A Bar. Pressure (mb): 1015 mb Comments: None

MSL501 System Check_5500MHz_7-23-12 DUT: Dipole D5GHzV2; Type: D5GHzV 2; Serial: D5GHzV2 - SN:1066 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5500 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used: f = 5500 MHz; σ = 5.788 mho/m; ε r = 46.893; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

• DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

System Check/System Check - Mid Channel/Area Scan (51x61x1): Measurement grid: dx=10mm, dy=10mm Maximum value of SAR (interpolated) = 20.4 mW/g

System Check/System Check - Mid Cha nnel/Zoom Scan (7x9x7) (7x7x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 63.378 V/m; Power Drift = 0.13 dB Peak SAR (extrapolated) = 34.185 mW/g SAR(1 g) = 9.56 mW/g; SAR(10 g) = 2.75 mW/g Maximum value of SAR (measured) = 19.8 mW/g

System Check/System Check - Mid Channel/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm Maximum value of Total (measured) = 64.42 V/m

Approved ByWSSV.12.07.17 

# SAR SYSTEM VERIFICATION

MSL501 System Check_5500MHz_7-23-12WSSV.12.07.17 

# SAR SYSTEM VERIFICATION

Tested By: Ethan Schoonover Room Temperature (°C): 23.2°C Date: 7/23/2012 Liquid Temperature (°C): 21.5°C Serial Number: ADM Humidity (%RH): 41.7% Configuration: N/A Bar. Pressure (mb): 1015 mb Comments: None

MSL501 System Check_5800MHz_7-23-12 DUT: Dipole D5GHzV2; Type: D5GHzV 2; Serial: D5GHzV2 - SN:1066 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5800 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used: f = 5800 MHz; σ = 6.201 mho/m; ε r = 46.262; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

• DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

System Check/System Check - Hi gh Channel/Area Scan (51x61x1): Measurement grid: dx=10mm, dy=10mm Maximum value of SAR (interpolated) = 14.0 mW/g

System Check/System Check - High Channel/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm Maximum value of Total (measured) = 42.01 V/m

System Check/System Check - High Chan nel/Zoom Scan (7x9x7) (7x7x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 41.510 V/m; Power Drift = 0.10 dB Peak SAR (extrapolated) = 25.331 mW/g SAR(1 g) = 6.47 mW/g; SAR(10 g) = 1.83 mW/g Maximum value of SAR (measured) = 13.9 mW/g

Approved ByWSSV.12.07.17 

# SAR SYSTEM VERIFICATION

MSL501 System Check_5800MHz_7-23-12WSSV.12.07.17 

# SAR SYSTEM VERIFICATION

Tested By: Jennifer Herrett Room Temperature (°C): 23.4°C Date: 7/26/2012 Liquid Temperature (°C): 22.7°C Serial Number: 1066 Humidity (%RH): 44.9% Configuration: N/A Bar. Pressure (mb): 1015.6 mb Comments: None

MSL501 System Checks_5800MHz_7-26-12 DUT: Dipole D5GHzV2; Type: D5GHzV 2; Serial: D5GHzV2 - SN:1066 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5800 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used: f = 5800 MHz; σ = 6.201 mho/m; ε r = 46.262; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

• DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

System Check/System Check - Mid Channel/Area Scan (51x61x1): Measurement grid: dx=10mm, dy=10mm Maximum value of SAR (interpolated) = 10.8 mW/g

System Check/System Check - Mid Cha nnel/Zoom Scan (7x9x7) (8x8x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 46.187 V/m; Power Drift = 0.08 dB Peak SAR (extrapolated) = 19.687 mW/g SAR(1 g) = 5.08 mW/g; SAR(10 g) = 1.46 mW/g Maximum value of SAR (measured) = 10.8 mW/g

System Check/System Check - Mid Channel/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm Maximum value of Total (measured) = 13.5 mW/g

Approved ByWSSV.12.07.17 

# SAR SYSTEM VERIFICATION

MSL501 System Checks_5800MHz_7-26-12WSSV.12.07.17 

# SAR SYSTEM VERIFICATION

Tested By: Jennifer Herrett Room Temperature (°C): 23.4°C Date: 7/26/2012 Liquid Temperature (°C): 22.7°C Serial Number: 1066 Humidity (%RH): 44.9% Configuration: N/A Bar. Pressure (mb): 1015.6 mb Comments: None

MSL501 System Checks_5500MHz_7-26-12 DUT: Dipole D5GHzV2; Type: D5GHzV 2; Serial: D5GHzV2 - SN:1066 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5500 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used: f = 5500 MHz; σ = 5.788 mho/m; ε r = 46.893; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

• DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

System Check/System Check - Mid Channel/Area Scan (51x61x1): Measurement grid: dx=10mm, dy=10mm Maximum value of SAR (interpolated) = 16.6 mW/g

System Check/System Check - Mid Cha nnel/Zoom Scan (7x9x7) (8x8x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 57.543 V/m; Power Drift = 0.09 dB Peak SAR (extrapolated) = 29.132 mW/g SAR(1 g) = 7.59 mW/g; SAR(10 g) = 2.14 mW/g Maximum value of SAR (measured) = 16.1 mW/g

System Check/System Check - Mid Channel/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm Maximum value of Total (measured) = 19.6 mW/g

Approved ByWSSV.12.07.17 

# SAR SYSTEM VERIFICATION

MSL501 System Checks_5500MHz_7-26-12WTD 12.5.23 

# TEST RESULTS

Test Configurations

Test Locations

The FCC’s starting point for SAR guidance is KDB 44 7498 D01 Mobile Portable RF Exposure v04. Per Item 4(b), the bottom face (referred to as “back” in this report) and the top edge were tested. The antennas are located closest to the top edge.

Simultaneous Transmission

During testing, a KDB analysis was done to determine whether a SAR evaluation is required for simultaneous transmission. The condition of Item 4(b) of KDB 616217 was applied. Assuming a worst case SAR value of 1.19 W/kg from each antenna in MIMO mode, the equation from 4(b)(i) becomes:

5*[(1.19 + 1.19)/1.6] 1.5 = 9 cm

Since the antenna spacing of 14cm is greater than 9 cm, and there are no MPE exposure conditions to consider, simultaneous SAR is not required.

MIMO Evaluation

However, the FCC’s Guidance for SAR testing of 802.11 a/b/g device is found in KDB 248227. It states:

“SAR for MIMO is measured with all antennas transmitting simultaneously.

For many low-power devices, when the peak SAR loca tions are more than 5 cm apart, the 1-g SAR can usually be treated independently with little or no not iceable impact. Therefore spatial summing could be optional”

Although the highest conducted output power modes we re not MIMO, MIMO SAR evaluations were conducted in the 2.4 and 5 GHz bands to show that with a 14 cm antenna spacing, there were no overlapping SAR regions. The zoom scans of each hot spot were centered on the individual antennas. The maximum SAR measured for each MIMO mode was significantly lower than ot her modes reported in this SAR evaluation.

Summary

The following tables summarize the measured SAR values.

Per FCC KDB 248227, among the channels required for nor mal testing, SAR must be measured on the channel with the highest conducted output power. When the SAR measured on the highest output channel is >0.8 W/kg, SAR evaluation for the other requi red test channels is necessary.WSTD.12.07.20 

# SAR TEST DATA

EUT: 1516 Work Order: MCSO1607 Customer: Microsoft Corporation Job Site: EV08 Attendees: none Customer Project: None


### TEST SPECIFICATIONS Specification: Method:

FCC 2.1093:2012 FCC 15.247:2012 FCC 15.407:2012 FCC OET 65C:2001 IEEE Std 1528:2003 FCC KDB 447498 D01 v04 FCC KDB 248227 D01 V01r02 FCC KDB 616217 D03 v01 FCC 865664 Health Safety Code 6:2009 RSS-102, Issue 4:2010


### COMMENTS
None


### DEVIATIONS FROM TEST STANDARD
None


### RESULTS

Test Configuration Frequency Band Transmit Frequency

(MHz) Transmit Channel Data Rate (Mbps) Channel Bandwidth

(MHz) Antenna Port Output Power EUT Position SAR Drift During Test

(dB) Measured 1g SAR Level

(mW/g) Test#

Body 5.2 5230 44/48 7.2 (MCS0) 40 A 13 Top -0.07 0.413 9 Body 5.2 5230 44/48 7.2 (MCS0) 40 A 13 Back -0.04 1.39 10 Body 5.2 5190 36/40 7.2 (MCS0) 40 A 11 Back 0.03 1.2 10a Body 5.2 5180 36 6 20 A 13 Top -0.17 0.346 11 Body 5.2 5180 36 6 20 A 13 Back 0.04 1.31 12 Body 5.2 5240 48 6 20 A 12 Back -0.01 1.31 12a Body 5.2 5230 44/48 7.2 (MCS0) 40 B 13 Top -0.02 0.362 13 Body 5.2 5230 44/48 7.2 (MCS0) 40 B 13 Back 0.09 1.43 14 Body 5.2 5190 36/40 7.2 (MCS0) 40 B 11 Back 0 0.916 14a Body 5.2 5240 48 6 20 B 13 Top -0.22 0.396 15 Body 5.2 5240 48 6 20 B 13 Back 0.03 1.41 16 Body 5.2 5180 36 6 20 B 12 Back 0.08 1.09 16a Body 5.3 5270 52/56 7.2 (MCS0) 40 A 13 Top -0.16 0.474 17 Body 5.3 5270 52/56 7.2 (MCS0) 40 A 13 Back 0.02 1.34 18 Body 5.3 5310 60/64 7.2 (MCS0) 40 A 11 Back 0.05 1.1 18a Body 5.3 5260 52 6 20 A 13 Top -0.22 0.413 19 Body 5.3 5260 52 6 20 A 13 Back 0.01 1.31 20 Body 5.3 5320 64 6 20 A 12 Back 0.06 1.26 20a Body 5.3 5310 60/64 7.2 (MCS0) 40 B 13 Top 0.04 0.398 21 Body 5.3 5310 60/64 7.2 (MCS0) 40 B 13 Back 0 1.41 22 Body 5.3 5270 52/56 7.2 (MCS0) 40 B 11 Back 0.08 0.953 22a Body 5.3 5280 56 6 20 B 13 Top -0.08 0.373 23 Body 5.3 5280 56 6 20 B 13 Back -0.07 1.4 24 Body 5.3 5320 64 6 20 B 12 Back 0.18 0.998 24a Body 5.6 5510 100/104 7.2 (MCS0) 40 A 13 Top -0.17 0.43 25 Body 5.6 5510 100/104 7.2 (MCS0) 40 A 13 Back 0.06 1.17 26a Body 5.6 5550 108/112 7.2 (MCS0) 40 A 11 Back -0.01 1.46 26 Body 5.6 5670 132/136 7.2 (MCS0) 40 A 11 Back 0.16 0.999 26b Body 5.6 5520 104 6 20 A 13 Top -0.23 0.466 27 Body 5.6 5520 104 6 20 A 13 Back 0.11 1.46 28 Body 5.6 5680 136 6 20 A 12 Back -0.13 1.12 28b Body 5.6 5550 108/112 7.2 (MCS0) 40 B 13 Top -0.35 0.551 29 Body 5.6 5550 108/112 7.2 (MCS0) 40 B 11 Back 0.04 1.23 30b Body 5.6 5510 100/104 7.2 (MCS0) 40 B 11 Back -0.03 1.1 30c Body 5.6 5670 132/136 7.2 (MCS0) 40 B 11 Back -0.22 0.913 30d Body 5.6 5560 112 6 20 B 13 Top 0.02 0.51 31 Body 5.6 5560 112 6 20 B 12 Back -0.14 1.46 32a Body 5.6 5520 104 6 20 B 12 Back 0.13 1.28 32b Body 5.6 5680 136 6 20 B 12 Back -0.12 0.984 32c Body 5.8 5795 157/161 7.2 (MCS0) 40 A 13 Top -0.01 0.544 33 Body 5.8 5795 157/161 7.2 (MCS0) 40 A 13 Back -0.05 1.43 34 Body 5.8 5755 149/153 7.2 (MCS0) 40 A 11 Back 0.1 0.916 34a Body 5.8 5765 153 6 20 A 13 Top -0.06 0.519 35 Body 5.8 5765 153 6 20 A 13 Back -0.01 1.39 36 Body 5.8 5825 165 6 20 A 12 Back 0.09 1.14 36a Body 5.8 5795 157/161 7.2 (MCS0) 40 B 13 Top -0.08 0.499 37 Body 5.8 5795 157/161 7.2 (MCS0) 40 B 13 Back 0.08 1.5 38 Body 5.8 5755 149/153 7.2 (MCS0) 40 B 11 Back -0.08 0.983 38b Body 5.8 5785 157 6 20 B 13 Top -0.05 0.461 39 Body 5.8 5785 157 6 20 B 13 Back -0.02 1.42 40 Body 5.8 5745 149 6 20 B 12 Back -0.01 1.2 40b Body 5.8 5825 165 6 20 B 12 Back -0.11 1.14 40cWSTD.12.07.20 

# SAR TEST DATA

Tested By: Ethan Schoonover Room Temperature (°C): 23.5°C Date: 7/17/2012 Liquid Temperature (°C): 23.7°C Serial Number: EV3 Humidity (%RH): 46.6% Configuration: MCSO1607-1 Bar. Pressure (mb): 1010 mb

Test 9, 7-17-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5230 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5230 MHz; σ = 5.5 mho/m; ε r = 47.608; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Reference scan (41x101x1): Measurement grid: dx=30mm, dy=30mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.301 mW/g

Body/Body/Zoom Scan 2 (12x10x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 10.770 V/m; Power Drift = -0.07 dB Peak SAR (extrapolated) = 1.719 mW/g SAR(1 g) = 0.413 mW/g; SAR(10 g) = 0.145 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.765 mW/g

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.537 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 10.770 V/m; Power Drift = 0.01 dB Peak SAR (extrapolated) = 1.500 mW/g SAR(1 g) = 0.413 mW/g; SAR(10 g) = 0.158 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.845 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of Total (measured) = 0.218 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 9, 7-17-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Ethan Schoonover Room Temperature (°C): 23.3°C Date: 7/18/2012 Liquid Temperature (°C): 23.7°C Serial Number: EV3 Humidity (%RH): 48.3% Configuration: MCSO1607-1 Bar. Pressure (mb): 1016 mb

Test 10, 7-18-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5230 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5230 MHz; σ = 5.5 mho/m; ε r = 47.608; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Reference scan (71x101x1): Measurement grid: dx=30mm, dy=30mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 1.13 mW/g

Body/Body/Area scan (81x91x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 2.82 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 22.809 V/m; Power Drift = -0.04 dB Peak SAR (extrapolated) = 5.535 mW/g SAR(1 g) = 1.39 mW/g; SAR(10 g) = 0.424 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 2.88 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of Total (measured) = 0.760 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 10, 7-18-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 23°C Date: 7/20/2012 8:05:36 AM Liquid Temperature (°C): 22.4°C Serial Number: EV3 Humidity (%RH): 52.8% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1020.7 mb Comments: None

Test 10a, 7-20-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5190 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5190 MHz; σ = 5.448 mho/m; ε r = 47.683; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Reference scan (71x101x1): Measurement grid: dx=30mm, dy=30mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.503 mW/g

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 2.06 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 18.694 V/m; Power Drift = 0.03 dB Peak SAR (extrapolated) = 4.874 mW/g SAR(1 g) = 1.2 mW/g; SAR(10 g) = 0.366 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 2.58 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.670 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 10a, 7-20-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Ethan Schoonover Room Temperature (°C): 23.5°C Date: 7/17/2012 Liquid Temperature (°C): 23.7°C Serial Number: EV3 Humidity (%RH): 46.6% Configuration: MCSO1607-1 Bar. Pressure (mb): 1010mb

Test 11, 7-17-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5180 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5180 MHz; σ = 5.434 mho/m; ε r = 47.701; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Reference scan (21x101x1): Measurement grid: dx=30mm, dy=30mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.247 mW/g

Body/Body/Zoom Scan 2 (10x10x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 10.063 V/m; Power Drift = -0.17 dB Peak SAR (extrapolated) = 1.211 mW/g SAR(1 g) = 0.346 mW/g; SAR(10 g) = 0.131 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.666 mW/g

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.442 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 10.063 V/m; Power Drift = -0.06 dB Peak SAR (extrapolated) = 1.340 mW/g SAR(1 g) = 0.345 mW/g; SAR(10 g) = 0.131 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.705 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of Total (measured) = 0.194 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 11, 7-17-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Ethan Schoonover Room Temperature (°C): 23.3°C Date: 7/18/2012 Liquid Temperature (°C): 23.7°C Serial Number: EV3 Humidity (%RH): 48.3% Configuration: MCSO1607-1 Bar. Pressure (mb): 1016 mb

Test 12, 7-18-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5180 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5180 MHz; σ = 5.434 mho/m; ε r = 47.701; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (81x91x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 2.66 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 21.799 V/m; Power Drift = 0.04 dB Peak SAR (extrapolated) = 5.263 mW/g SAR(1 g) = 1.31 mW/g; SAR(10 g) = 0.396 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 2.69 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.722 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 12, 7-18-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 23°C Date: 7/20/2012 9:08:24 AM Liquid Temperature (°C): 22.4°C Serial Number: EV3 Humidity (%RH): 52.8% Configuration: MCSO1607 - 2 Bar. Pressure (mb): 1020.7 mb Comments: None

Test 12a, 7-20-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5240 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5240 MHz; σ = 5.513 mho/m; ε r = 47.589; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 2.34 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 19.695 V/m; Power Drift = -0.01 dB Peak SAR (extrapolated) = 5.506 mW/g SAR(1 g) = 1.31 mW/g; SAR(10 g) = 0.374 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 2.88 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.724 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 12a, 7-20-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Ethan Schoonover Room Temperature (°C): 22.6°C Date: 7/17/2012 Liquid Temperature (°C): 22.9°C Serial Number: EV3 Humidity (%RH): 47.3% Configuration: MCSO1607-1 Bar. Pressure (mb): 1010 mb

Test 13, 7-17-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5230 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5230 MHz; σ = 5.5 mho/m; ε r = 47.608; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Reference scan (21x101x1): Measurement grid: dx=30mm, dy=30mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.179 mW/g

Body/Body/Zoom Scan 2 (10x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 10.385 V/m; Power Drift = -0.13 dB Peak SAR (extrapolated) = 1.580 mW/g SAR(1 g) = 0.359 mW/g; SAR(10 g) = 0.117 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.716 mW/g

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.492 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 10.385 V/m; Power Drift = -0.02 dB Peak SAR (extrapolated) = 1.393 mW/g SAR(1 g) = 0.362 mW/g; SAR(10 g) = 0.124 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.770 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.177 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 13, 7-17-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Ethan Schoonover Room Temperature (°C): 21.3°C Date: 7/18/2012 Liquid Temperature (°C): 22.3°C Serial Number: EV3 Humidity (%RH): 47.0% Configuration: MCSO1607-1 Bar. Pressure (mb): 1016 mb

Test 14, 7-18-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5230 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5230 MHz; σ = 5.5 mho/m; ε r = 47.608; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Reference scan (71x101x1): Measurement grid: dx=30mm, dy=30mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.360 mW/g

Body/Body/Area scan (81x91x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 3.02 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 23.338 V/m; Power Drift = 0.09 dB Peak SAR (extrapolated) = 6.110 mW/g SAR(1 g) = 1.43 mW/g; SAR(10 g) = 0.415 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 3.10 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of Total (measured) = 0.780 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 14, 7-18-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 23°C Date: 7/20/2012 10:18:05 AM Liquid Temperature (°C): 22.4°C Serial Number: EV3 Humidity (%RH): 52.8% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1020.7 mb Comments: None

Test 14a, 7-20-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5190 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5190 MHz; σ = 5.448 mho/m; ε r = 47.683; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Reference scan (71x101x1): Measurement grid: dx=30mm, dy=30mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 1.55 mW/g

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 1.67 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 18.451 V/m; Power Drift = -0.00 dB Peak SAR (extrapolated) = 3.900 mW/g SAR(1 g) = 0.916 mW/g; SAR(10 g) = 0.215 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 1.97 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.422 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 14a, 7-20-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Ethan Schoonover Room Temperature (°C): 22.6°C Date: 7/17/2012 Liquid Temperature (°C): 22.9°C Serial Number: EV3 Humidity (%RH): 47.3% Configuration: MCSO1607-1 Bar. Pressure (mb): 1010 mb

Test 15, 7-17-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5240 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5240 MHz; σ = 5.513 mho/m; ε r = 47.589; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Reference scan (21x101x1): Measurement grid: dx=30mm, dy=30mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.205 mW/g

Body/Body/Zoom Scan 2 (11x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 11.148 V/m; Power Drift = -0.44 dB Peak SAR (extrapolated) = 1.505 mW/g SAR(1 g) = 0.389 mW/g; SAR(10 g) = 0.141 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.853 mW/g

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.755 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 11.148 V/m; Power Drift = -0.22 dB Peak SAR (extrapolated) = 1.621 mW/g SAR(1 g) = 0.396 mW/g; SAR(10 g) = 0.136 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.835 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of Total (measured) = 0.241 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 15, 7-17-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 23.8°C Date: 7/18/2012 Liquid Temperature (°C): 22.8°C Serial Number: EV3 Humidity (%RH): 48.5% Configuration: MCSO1607-1 Bar. Pressure (mb): 1016 mb

Test 16, 7-18-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5240 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5240 MHz; σ = 5.513 mho/m; ε r = 47.589; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Reference scan (71x101x1): Measurement grid: dx=30mm, dy=30mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.373 mW/g

Body/Body/Area scan (81x91x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 2.77 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 22.369 V/m; Power Drift = 0.03 dB Peak SAR (extrapolated) = 6.032 mW/g SAR(1 g) = 1.41 mW/g; SAR(10 g) = 0.419 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 2.82 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of Total (measured) = 0.735 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 16, 7-18-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 23°C Date: 7/20/2012 11:12:44 AM Liquid Temperature (°C): 22.4°C Serial Number: EV3 Humidity (%RH): 52.8% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1020.7 mb Comments: None

Test 16a, 7-20-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5180 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5180 MHz; σ = 5.434 mho/m; ε r = 47.701; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 1.88 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 19.702 V/m; Power Drift = 0.08 dB Peak SAR (extrapolated) = 4.712 mW/g SAR(1 g) = 1.09 mW/g; SAR(10 g) = 0.269 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 2.30 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.496 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 16a, 7-20-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Ethan Schoonover Room Temperature (°C): 24.6°C Date: 7/17/2012 Liquid Temperature (°C): 23.8°C Serial Number: EV3 Humidity (%RH): 49.3% Configuration: MCSO1607-1 Bar. Pressure (mb): 1010 mb

Test 17, 7-17-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5270 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5270 MHz; σ = 5.548 mho/m; ε r = 47.522; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Reference scan (21x101x1): Measurement grid: dx=30mm, dy=30mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.321 mW/g

Body/Body/Zoom Scan 2 (12x10x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 11.402 V/m; Power Drift = -0.16 dB Peak SAR (extrapolated) = 2.079 mW/g SAR(1 g) = 0.474 mW/g; SAR(10 g) = 0.165 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.870 mW/g

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.593 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 11.402 V/m; Power Drift = -0.07 dB Peak SAR (extrapolated) = 1.735 mW/g SAR(1 g) = 0.453 mW/g; SAR(10 g) = 0.161 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.957 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of Total (measured) = 0.235 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 17, 7-17-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 23.2°C Date: 7/18/2012 Liquid Temperature (°C): 22.8°C Serial Number: EV3 Humidity (%RH): 49.4% Configuration: MCSO1607-1 Bar. Pressure (mb): 1016 mb

Test 18, 7-18-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5260 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5260 MHz; σ = 5.537 mho/m; ε r = 47.546; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 2.48 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 22.211 V/m; Power Drift = 0.02 dB Peak SAR (extrapolated) = 5.348 mW/g SAR(1 g) = 1.34 mW/g; SAR(10 g) = 0.424 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 2.74 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.720 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 18, 7-18-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 23.3°C Date: 7/20/2012 2:19:53 PM Liquid Temperature (°C): 23.1°C Serial Number: EV3 Humidity (%RH): 49.8% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1020.7 mb Comments: None

Test 18a, 7-20-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5310 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5310 MHz; σ = 5.598 mho/m; ε r = 47.436; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 2.50 mW/g

Body/Body/Zoom Scan (8x8x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 23.233 V/m; Power Drift = 0.05 dB Peak SAR (extrapolated) = 4.837 mW/g SAR(1 g) = 1.1 mW/g; SAR(10 g) = 0.271 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 2.55 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.568 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 18a, 7-20-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Ethan Schoonover Room Temperature (°C): 24.6°C Date: 7/17/2012 Liquid Temperature (°C): 23.8°C Serial Number: EV3 Humidity (%RH): 49.3% Configuration: MCSO1607-1 Bar. Pressure (mb): 1010 mb

Test 19, 7-17-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5260 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5260 MHz; σ = 5.537 mho/m; ε r = 47.546; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Reference scan (21x101x1): Measurement grid: dx=30mm, dy=30mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.296 mW/g

Body/Body/Zoom Scan 2 (12x10x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 10.830 V/m; Power Drift = -0.22 dB Peak SAR (extrapolated) = 1.737 mW/g SAR(1 g) = 0.413 mW/g; SAR(10 g) = 0.144 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.755 mW/g

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.543 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 10.830 V/m; Power Drift = -0.09 dB Peak SAR (extrapolated) = 1.593 mW/g SAR(1 g) = 0.407 mW/g; SAR(10 g) = 0.145 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.857 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of Total (measured) = 0.216 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 19, 7-17-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 23.0°C Date: 7/18/2012 Liquid Temperature (°C): 22.2°C Serial Number: EV3 Humidity (%RH): 48.1% Configuration: MCSO1607-1 Bar. Pressure (mb): 1016 mb

Test 20, 7-18-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5260 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5260 MHz; σ = 5.537 mho/m; ε r = 47.546; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Reference scan (71x101x1): Measurement grid: dx=30mm, dy=30mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 1.07 mW/g

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 2.45 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 21.965 V/m; Power Drift = 0.01 dB Peak SAR (extrapolated) = 5.230 mW/g SAR(1 g) = 1.31 mW/g; SAR(10 g) = 0.412 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 2.69 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of Total (measured) = 0.712 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 20, 7-18-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 23.3°C Date: 7/20/2012 1:34:41 PM Liquid Temperature (°C): 23.1°C Serial Number: EV3 Humidity (%RH): 49.8% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1020.7 mb Comments: None

Test 20a, 7-20-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5320 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5320 MHz; σ = 5.613 mho/m; ε r = 47.421; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 2.88 mW/g

Body/Body/Zoom Scan (8x8x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 24.874 V/m; Power Drift = 0.06 dB Peak SAR (extrapolated) = 5.510 mW/g SAR(1 g) = 1.26 mW/g; SAR(10 g) = 0.310 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 2.93 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.645 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 20a, 7-20-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Ethan Schoonover Room Temperature (°C): 23.7°C Date: 7/17/2012 Liquid Temperature (°C): 23.1°C Serial Number: EV3 Humidity (%RH): 48.4% Configuration: MCSO1607-1 Bar. Pressure (mb): 1010 mb

Test 21, 7-17-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5310 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5310 MHz; σ = 5.598 mho/m; ε r = 47.436; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Reference scan (21x101x1): Measurement grid: dx=30mm, dy=30mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.216 mW/g

Body/Body/Zoom Scan 2 (10x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 10.623 V/m; Power Drift = 0.04 dB Peak SAR (extrapolated) = 1.540 mW/g SAR(1 g) = 0.398 mW/g; SAR(10 g) = 0.145 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.741 mW/g

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.506 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 10.623 V/m; Power Drift = 0.09 dB Peak SAR (extrapolated) = 1.522 mW/g SAR(1 g) = 0.380 mW/g; SAR(10 g) = 0.139 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.773 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of Total (measured) = 0.169 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 21, 7-17-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 23.8°C Date: 7/18/2012 Liquid Temperature (°C): 22.9°C Serial Number: EV3 Humidity (%RH): 48.3% Configuration: MCSO1607-1 Bar. Pressure (mb): 1016 mb

Test 22, 7-18-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5310 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5310 MHz; σ = 5.598 mho/m; ε r = 47.436; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 23.313 V/m; Power Drift = 0.00 dB Peak SAR (extrapolated) = 6.263 mW/g SAR(1 g) = 1.41 mW/g; SAR(10 g) = 0.434 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 2.80 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of Total (measured) = 0.705 mW/g

Body/Body/Area scan (81x91x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 2.82 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 22, 7-18-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 23°C Date: 7/20/2012 12:45:33 PM Liquid Temperature (°C): 22.7°C Serial Number: EV3 Humidity (%RH): 51.8% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1020.7 mb Comments: None

Test 22a, 7-20-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5270 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5270 MHz; σ = 5.548 mho/m; ε r = 47.522; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (81x91x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 1.80 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 18.053 V/m; Power Drift = 0.08 dB Peak SAR (extrapolated) = 4.188 mW/g SAR(1 g) = 0.953 mW/g; SAR(10 g) = 0.229 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 2.14 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.468 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 22a, 7-20-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 23.1°C Date: 7/18/2012 Liquid Temperature (°C): 23.0°C Serial Number: EV3 Humidity (%RH): 48.9% Configuration: MCSO1607-1 Bar. Pressure (mb): 1016 mb

Test 23, 7-18-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5280 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5280 MHz; σ = 5.56 mho/m; ε r = 47.499; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Zoom Scan (10x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 10.229 V/m; Power Drift = -0.08 dB Peak SAR (extrapolated) = 1.600 mW/g SAR(1 g) = 0.373 mW/g; SAR(10 g) = 0.144 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.797 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of Total (measured) = 0.171 mW/g

Body/Body/Reference scan (41x101x1): Measurement grid: dx=30mm, dy=30mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.284 mW/g

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.490 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 23, 7-18-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 23.3°C Date: 7/18/2012 Liquid Temperature (°C): 23.3°C Serial Number: EV3 Humidity (%RH): 49.1% Configuration: MCSO1607-1 Bar. Pressure (mb): 1016 mb

Test 24, 7-18-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5280 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5280 MHz; σ = 5.56 mho/m; ε r = 47.499; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 23.311 V/m; Power Drift = -0.07 dB Peak SAR (extrapolated) = 6.117 mW/g SAR(1 g) = 1.4 mW/g; SAR(10 g) = 0.427 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 2.79 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of Total (measured) = 0.703 mW/g

Body/Body/Reference scan (71x101x1): Measurement grid: dx=30mm, dy=30mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.420 mW/g

Body/Body/Area scan (81x91x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 2.82 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 24, 7-18-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 23°C Date: 7/20/2012 11:57:42 AM Liquid Temperature (°C): 22.7°C Serial Number: EV3 Humidity (%RH): 51.8% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1020.7 mb Comments: None

Test 24a, 7-20-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5320 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5320 MHz; σ = 5.613 mho/m; ε r = 47.421; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (81x91x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 1.86 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 18.263 V/m; Power Drift = 0.18 dB Peak SAR (extrapolated) = 4.563 mW/g SAR(1 g) = 0.998 mW/g; SAR(10 g) = 0.237 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 2.29 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.497 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 24a, 7-20-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 23°C Date: 7/19/2012 9:03:17 AM Liquid Temperature (°C): 22.9°C Serial Number: EV3 Humidity (%RH): 49.8% Configuration: MCSO1607-1 Bar. Pressure (mb): 1014.7 mb

Test 25, 7-19-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5510 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5510 MHz; σ = 5.56 mho/m; ε r = 46.728; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Reference scan (41x101x1): Measurement grid: dx=30mm, dy=30mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.253 mW/g

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.678 mW/g

Body/Body/Zoom Scan (10x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 11.867 V/m; Power Drift = -0.17 dB Peak SAR (extrapolated) = 1.680 mW/g SAR(1 g) = 0.430 mW/g; SAR(10 g) = 0.159 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.895 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.218 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 25, 7-19-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 23°C Date: 7/19/2012 7:42:58 AM Liquid Temperature (°C): 22.9°C Serial Number: EV3 Humidity (%RH): 49.8% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1014.7 mb

Test 26, 7-19-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5510 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5510 MHz; σ = 5.56 mho/m; ε r = 46.728; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 2.79 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 24.373 V/m; Power Drift = -0.01 dB Peak SAR (extrapolated) = 5.960 mW/g SAR(1 g) = 1.46 mW/g; SAR(10 g) = 0.451 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 3.00 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.715 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 26, 7-19-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 23.2°C Date: 7/23/2012 10:17:49 AM Liquid Temperature (°C): 22.7°C Serial Number: EV3 Humidity (%RH): 45.8% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1023.9 mb Comments: None

Test 26a, 7-23-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5550 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used: f = 5550 MHz; σ = 5.85 mho/m; ε r = 46.77; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm Maximum value of SAR (interpolated) = 2.71 mW/g

Body/Body/Zoom Scan (8x8x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 23.715 V/m; Power Drift = 0.06 dB Peak SAR (extrapolated) = 5.315 mW/g SAR(1 g) = 1.17 mW/g; SAR(10 g) = 0.282 mW/g Maximum value of SAR (measured) = 2.79 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm Maximum value of SAR (measured) = 0.587 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 26a, 7-23-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 21.8°C Date: 7/25/2012 9:07:38 AM Liquid Temperature (°C): 22.3°C Serial Number: EV3 Humidity (%RH): 45.7% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1015.7 mb Comments: None

Test 26b, 7-25-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5670 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5670 MHz; σ = 6.022 mho/m; ε r = 46.517; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 2.07 mW/g

Body/Body/Zoom Scan (8x8x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 19.248 V/m; Power Drift = 0.16 dB Peak SAR (extrapolated) = 4.333 mW/g SAR(1 g) = 0.999 mW/g; SAR(10 g) = 0.264 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 2.26 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.471 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 26b, 7-25-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 23°C Date: 7/19/2012 9:56:11 AM Liquid Temperature (°C): 22.9°C Serial Number: EV3 Humidity (%RH): 49.8% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1014.7 mb

Test 27, 7-19-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5520 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5520 MHz; σ = 5.571 mho/m; ε r = 46.703; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.742 mW/g

Body/Body/Zoom Scan (10x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 12.649 V/m; Power Drift = -0.23 dB Peak SAR (extrapolated) = 1.762 mW/g SAR(1 g) = 0.466 mW/g; SAR(10 g) = 0.165 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.975 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.254 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 27, 7-19-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 23°C Date: 7/19/2012 6:58:37 AM Liquid Temperature (°C): 22.9°C Serial Number: EV3 Humidity (%RH): 49.8% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1014.7 mb

Test 28, 7-19-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5520 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5520 MHz; σ = 5.571 mho/m; ε r = 46.703; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Reference scan (71x101x1): Measurement grid: dx=30mm, dy=30mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 1.01 mW/g

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 2.81 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 24.306 V/m; Power Drift = 0.11 dB Peak SAR (extrapolated) = 6.157 mW/g SAR(1 g) = 1.46 mW/g; SAR(10 g) = 0.453 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 2.98 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.721 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 28, 7-19-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 21.8°C Date: 7/25/2012 11:04:40 AM Liquid Temperature (°C): 22.3°C Serial Number: EV3 Humidity (%RH): 45.7% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1015.7 mb Comments: None

Test 28b, 7-25-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5670 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5670 MHz; σ = 6.022 mho/m; ε r = 46.517; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 2.14 mW/g

Body/Body/Zoom Scan (8x8x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 20.574 V/m; Power Drift = -0.13 dB Peak SAR (extrapolated) = 4.835 mW/g SAR(1 g) = 1.12 mW/g; SAR(10 g) = 0.317 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 2.46 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.490 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 28b, 7-25-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Ethan Schoonover Room Temperature (°C): 22.2 Date: 7/16/2012 Liquid Temperature (°C): 22 Serial Number: EV3 Humidity (%RH): 41.7 Configuration: MCSO1607-1 Bar. Pressure (mb): 1007

Test 29, 7-16-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5550 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used: f = 5550 MHz; σ = 5.927 mho/m; ε r = 47.041; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Reference scan (21x101x1): Measurement grid: dx=30mm, dy=30mm Maximum value of SAR (interpolated) = 0.414 mW/g

Body/Body/Zoom Scan 2 (10x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 12.582 V/m; Power Drift = -0.35 dB Peak SAR (extrapolated) = 2.524 mW/g SAR(1 g) = 0.551 mW/g; SAR(10 g) = 0.206 mW/g Maximum value of SAR (measured) = 1.04 mW/g

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm Maximum value of SAR (interpolated) = 0.740 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 12.582 V/m; Power Drift = -0.22 dB Peak SAR (extrapolated) = 2.138 mW/g SAR(1 g) = 0.505 mW/g; SAR(10 g) = 0.226 mW/g Maximum value of SAR (measured) = 1.13 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm Maximum value of SAR (measured) = 0.213 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 29, 7-16-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Ethan Schoonover Room Temperature (°C): 26.8°C Date: 7/19/2012 6:39:27 PM Liquid Temperature (°C): 23.6°C Serial Number: EV3 Humidity (%RH): 40.6% Configuration: MCSO1607-1 Bar. Pressure (mb): 1015 mb Comments:

Test 30b, 7-19-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5550 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used: f = 5550 MHz; σ = 5.927 mho/m; ε r = 47.041; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (81x91x1): Measurement grid: dx=10mm, dy=10mm Maximum value of SAR (interpolated) = 2.85 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 24.023 V/m; Power Drift = 0.04 dB Peak SAR (extrapolated) = 5.666 mW/g SAR(1 g) = 1.23 mW/g; SAR(10 g) = 0.354 mW/g Maximum value of SAR (measured) = 2.66 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm Maximum value of SAR (measured) = 0.619 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 30b, 7-19-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Ethan Schoonover Room Temperature (°C): 26.8°C Date: 7/19/2012 7:25:11 PM Liquid Temperature (°C): 23.6°C Serial Number: EV3 Humidity (%RH): 40.6% Configuration: MCSO1607-1 Bar. Pressure (mb): 1015 mb Comments:

Test 30c, 7-19-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5510 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5510 MHz; σ = 5.873 mho/m; ε r = 47.12; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (81x91x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 2.54 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 22.820 V/m; Power Drift = -0.03 dB Peak SAR (extrapolated) = 5.137 mW/g SAR(1 g) = 1.1 mW/g; SAR(10 g) = 0.315 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 2.33 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.570 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 30c, 7-19-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 21.8°C Date: 7/25/2012 12:03:27 PM Liquid Temperature (°C): 22.3°C Serial Number: EV3 Humidity (%RH): 45.7% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1015.7 mb Comments: None

Test 30d, 7-25-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5670 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5670 MHz; σ = 6.022 mho/m; ε r = 46.517; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 1.48 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 16.698 V/m; Power Drift = -0.22 dB Peak SAR (extrapolated) = 4.409 mW/g SAR(1 g) = 0.913 mW/g; SAR(10 g) = 0.240 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 2.01 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.398 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 30d, 7-25-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Ethan Schoonover Room Temperature (°C): Date: 7/19/2012 10:39:38 PM Liquid Temperature (°C): Serial Number: Humidity (%RH): Configuration: MCSO1607-1 Bar. Pressure (mb):

Test 31, 7-16-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5560 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5560 MHz; σ = 5.617 mho/m; ε r = 46.607; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Reference scan (21x101x1): Measurement grid: dx=30mm, dy=30mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.473 mW/g

Body/Body/Zoom Scan 2 (10x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 12.190 V/m; Power Drift = 0.02 dB Peak SAR (extrapolated) = 2.136 mW/g SAR(1 g) = 0.510 mW/g; SAR(10 g) = 0.198 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 1.07 mW/g

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.690 mW/g

Body/Body/Zoom Scan (11x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 12.190 V/m; Power Drift = 0.16 dB Peak SAR (extrapolated) = 2.043 mW/g SAR(1 g) = 0.483 mW/g; SAR(10 g) = 0.186 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 1.05 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.228 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 31, 7-16-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Ethan Schoonover Room Temperature (°C): 22.1°C Date: 7/19/2012 4:44:25 PM Liquid Temperature (°C): 22.3°C Serial Number: EV3 Humidity (%RH): 43.7% Configuration: MCSO1607-1 Bar. Pressure (mb): 1015 mb Comments:

Test 32a, 7-19-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5560 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5560 MHz; σ = 5.617 mho/m; ε r = 46.607; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Reference scan (71x101x1): Measurement grid: dx=30mm, dy=30mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 1.81 mW/g

Body/Body/Area scan (81x91x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 3.49 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 27.091 V/m; Power Drift = -0.14 dB Peak SAR (extrapolated) = 6.816 mW/g SAR(1 g) = 1.46 mW/g; SAR(10 g) = 0.394 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 3.18 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.694 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 32a, 7-19-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 23.2°C Date: 7/23/2012 12:42:46 PM Liquid Temperature (°C): 22.7°C Serial Number: EV3 Humidity (%RH): 45.8% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1023.9 mb Comments: None

Test 32b, 7-23-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5520 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5520 MHz; σ = 5.813 mho/m; ε r = 46.844; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Reference scan (71x101x1): Measurement grid: dx=30mm, dy=30mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 1.87 mW/g

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 2.13 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 19.192 V/m; Power Drift = 0.13 dB Peak SAR (extrapolated) = 5.990 mW/g SAR(1 g) = 1.28 mW/g; SAR(10 g) = 0.298 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 3.04 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.627 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 32b, 7-23-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 23.3°C Date: 7/25/2012 1:18:13 PM Liquid Temperature (°C): 22.3°C Serial Number: EV3 Humidity (%RH): 46.1% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1015.6 mb Comments: None

Test 32c, 7-25-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5680 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5680 MHz; σ = 6.037 mho/m; ε r = 46.496; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 1.62 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 16.803 V/m; Power Drift = -0.12 dB Peak SAR (extrapolated) = 4.828 mW/g SAR(1 g) = 0.984 mW/g; SAR(10 g) = 0.256 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 2.15 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.420 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 32c, 7-25-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 22.3 Date: 7/17/2012 Liquid Temperature (°C): 21.9 Serial Number: EV3 Humidity (%RH): 47.9 Configuration: MCSO1607-1 Bar. Pressure (mb): 1010

Test 33, 7-17-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5795 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5795 MHz; σ = 6.271 mho/m; ε r = 46.53; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Zoom Scan (8x8x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 12.199 V/m; Power Drift = -0.01 dB Peak SAR (extrapolated) = 2.116 mW/g SAR(1 g) = 0.544 mW/g; SAR(10 g) = 0.213 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 1.09 mW/g

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.756 mW/g

Body/Body/Zoom Scan 2 (10x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 12.199 V/m; Power Drift = -0.07 dB Peak SAR (extrapolated) = 2.165 mW/g SAR(1 g) = 0.516 mW/g; SAR(10 g) = 0.193 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.996 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.221 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 33, 7-17-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Ethan Schoonover Room Temperature (°C): 23.1 Date: 7/18/2012 Liquid Temperature (°C): 23.6 Serial Number: EV3 Humidity (%RH): 47.6 Configuration: MCSO1607-1 Bar. Pressure (mb): 1016

Test 34, 7-18-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5795 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5795 MHz; σ = 6.271 mho/m; ε r = 46.53; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (81x91x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 3.00 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 24.577 V/m; Power Drift = -0.05 dB Peak SAR (extrapolated) = 6.319 mW/g SAR(1 g) = 1.43 mW/g; SAR(10 g) = 0.450 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 3.00 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.654 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 34, 7-18-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 23.2°C Date: 7/23/2012 6:54:53 AM Liquid Temperature (°C): 22.7°C Serial Number: EV3 Humidity (%RH): 45.8% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1023.9 mb Comments: None

Test 34a, 7-23-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5755 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5755 MHz; σ = 6.203 mho/m; ε r = 46.635; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Reference scan (71x101x1): Measurement grid: dx=30mm, dy=30mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.601 mW/g

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 2.08 mW/g

Body/Body/Zoom Scan (8x8x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 20.444 V/m; Power Drift = 0.10 dB Peak SAR (extrapolated) = 4.391 mW/g SAR(1 g) = 0.916 mW/g; SAR(10 g) = 0.230 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 2.22 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.425 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 34a, 7-23-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 23.8 Date: 7/17/2012 Liquid Temperature (°C): 23.1 Serial Number: EV3 Humidity (%RH): 49.1 Configuration: MCSO1607-1 Bar. Pressure (mb): 1010

Test 35, 7-17-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5765 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5765 MHz; σ = 6.22 mho/m; ε r = 46.609; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 12.001 V/m; Power Drift = 0.01 dB Peak SAR (extrapolated) = 2.071 mW/g SAR(1 g) = 0.496 mW/g; SAR(10 g) = 0.181 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 1.05 mW/g

Body/Body/Reference scan (41x101x1): Measurement grid: dx=30mm, dy=30mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.290 mW/g

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.709 mW/g

Body/Body/Zoom Scan 2 (12x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 12.001 V/m; Power Drift = -0.06 dB Peak SAR (extrapolated) = 2.123 mW/g SAR(1 g) = 0.519 mW/g; SAR(10 g) = 0.190 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 1.03 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.236 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 35, 7-17-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Ethan Schoonover Room Temperature (°C): 23.1 Date: 7/18/2012 Liquid Temperature (°C): 23.6 Serial Number: EV3 Humidity (%RH): 47.6 Configuration: MCSO1607-1 Bar. Pressure (mb): 1016

Test 36, 7-18-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5765 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5765 MHz; σ = 6.22 mho/m; ε r = 46.609; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (81x91x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 2.95 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 24.318 V/m; Power Drift = -0.01 dB Peak SAR (extrapolated) = 6.229 mW/g SAR(1 g) = 1.39 mW/g; SAR(10 g) = 0.426 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 3.10 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.670 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 36, 7-18-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 23.2°C Date: 7/23/2012 8:52:05 AM Liquid Temperature (°C): 22.7°C Serial Number: EV3 Humidity (%RH): 45.8% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1023.9 mb Comments: None

Test 36a, 7-23-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5825 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5825 MHz; σ = 6.228 mho/m; ε r = 46.222; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 2.62 mW/g

Body/Body/Zoom Scan (8x8x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 22.733 V/m; Power Drift = 0.09 dB Peak SAR (extrapolated) = 5.733 mW/g SAR(1 g) = 1.14 mW/g; SAR(10 g) = 0.264 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 2.83 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.538 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 36a, 7-23-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 23.9 Date: 7/17/2012 Liquid Temperature (°C): 23.2 Serial Number: EV3 Humidity (%RH): 47.9 Configuration: MCSO1607-1 Bar. Pressure (mb): 1010

Test 37, 7-17-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5795 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5795 MHz; σ = 6.271 mho/m; ε r = 46.53; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Zoom Scan (8x8x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 11.399 V/m; Power Drift = -0.08 dB Peak SAR (extrapolated) = 2.024 mW/g SAR(1 g) = 0.499 mW/g; SAR(10 g) = 0.209 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 1.06 mW/g

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.674 mW/g

Body/Body/Zoom Scan 2 (10x10x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 11.399 V/m; Power Drift = -0.09 dB Peak SAR (extrapolated) = 1.969 mW/g SAR(1 g) = 0.480 mW/g; SAR(10 g) = 0.189 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 1.01 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.210 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 37, 7-17-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Ethan Schoonover Room Temperature (°C): 26.2 Date: 7/18/2012 Liquid Temperature (°C): 23.8 Serial Number: EV3 Humidity (%RH): 41.9 Configuration: MCSO1607-1 Bar. Pressure (mb): 1016

Test 38, 7-18-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5795 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5795 MHz; σ = 6.271 mho/m; ε r = 46.53; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (81x91x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 2.85 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 21.282 V/m; Power Drift = 0.08 dB Peak SAR (extrapolated) = 6.994 mW/g SAR(1 g) = 1.5 mW/g; SAR(10 g) = 0.435 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 3.46 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.741 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 38, 7-18-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Ethan Schoonover Room Temperature (°C): 23.3 Date: 7/19/2012 Liquid Temperature (°C): 22.9 Serial Number: EV3 Humidity (%RH): 39.1 Configuration: MCSO1607-1 Bar. Pressure (mb): 1015 Comments: none

Test 38b, 7-19-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5755 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5755 MHz; σ = 6.203 mho/m; ε r = 46.635; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (81x91x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 2.41 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 21.720 V/m; Power Drift = -0.08 dB Peak SAR (extrapolated) = 4.627 mW/g SAR(1 g) = 0.983 mW/g; SAR(10 g) = 0.278 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 2.26 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.491 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 38b, 7-19-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 23.4
Date: 7/17/2012 Liquid Temperature (°C): 22.7 Serial Number: EV3 Humidity (%RH): 47.3 Configuration: MCSO1607-1 Bar. Pressure (mb): 1010

Test 39, 7-17-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5785 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5785 MHz; σ = 6.254 mho/m; ε r = 46.556; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Zoom Scan (10x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 11.003 V/m; Power Drift = -0.15 dB Peak SAR (extrapolated) = 1.886 mW/g SAR(1 g) = 0.460 mW/g; SAR(10 g) = 0.180 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.989 mW/g

Body/Body/Reference scan (41x101x1): Measurement grid: dx=30mm, dy=30mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.459 mW/g

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.622 mW/g

Body/Body/Zoom Scan 2 (10x10x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 11.003 V/m; Power Drift = -0.05 dB Peak SAR (extrapolated) = 1.982 mW/g SAR(1 g) = 0.461 mW/g; SAR(10 g) = 0.194 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.983 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.186 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 39, 7-17-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Ethan Schoonover Room Temperature (°C): 26.2 Date: 7/18/2012 Liquid Temperature (°C): 23.8 Serial Number: EV3 Humidity (%RH): 41.9 Configuration: MCSO1607-1 Bar. Pressure (mb): 1016

Test 40, 7-18-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5785 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5785 MHz; σ = 6.254 mho/m; ε r = 46.556; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (81x91x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 2.77 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 21.171 V/m; Power Drift = -0.02 dB Peak SAR (extrapolated) = 6.671 mW/g SAR(1 g) = 1.42 mW/g; SAR(10 g) = 0.405 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 3.31 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.733 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 40, 7-18-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Ethan Schoonover Room Temperature (°C): 26.9°C Date: 7/19/2012 9:07:11 PM Liquid Temperature (°C): 23.8°C Serial Number: EV3 Humidity (%RH): 39.1% Configuration: MCSO1607-1 Bar. Pressure (mb): 1015 mb Comments:

Test 40b, 7-19-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5745 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5745 MHz; σ = 6.189 mho/m; ε r = 46.657; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (81x91x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 2.95 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 24.197 V/m; Power Drift = -0.01 dB Peak SAR (extrapolated) = 5.659 mW/g SAR(1 g) = 1.2 mW/g; SAR(10 g) = 0.330 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 2.79 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.607 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 40b, 7-19-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Ethan Schoonover Room Temperature (°C): 23.3 Date: 7/19/2012 Liquid Temperature (°C): 22.9 Serial Number: EV3 Humidity (%RH): 39.1 Configuration: MCSO1607-1 Bar. Pressure (mb): 1015 Comments: none

Test 40c, 7-19-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5825 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5825 MHz; σ = 5.952 mho/m; ε r = 45.982; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (81x91x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 2.80 mW/g

Body/Body/Zoom Scan (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 23.931 V/m; Power Drift = -0.11 dB Peak SAR (extrapolated) = 5.462 mW/g SAR(1 g) = 1.14 mW/g; SAR(10 g) = 0.325 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 2.65 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.559 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 40c, 7-19-12WSTD.12.07.20 

# SAR TEST DATA

Test 14, 7-18-12WSTD.12.07.20 

# SAR TEST DATA

Test 22, 7-18-12WSTD.12.07.20 

# SAR TEST DATA

Test 28a, 7-23-12WSTD.12.07.20 

# SAR TEST DATA

Test 38, 7-18-12WSTD.12.07.20 

# SAR TEST DATA

EUT: 1516 Work Order: MCSO1607 Customer: Microsoft Corporation Job Site: EV08 Attendees: none Customer Project: None


### TEST SPECIFICATIONS Specification: Method:

FCC 2.1093:2011 FCC 15.247:2011 FCC 15.407:2001 FCC OET 65C:2001 IEEE Std 1528:2003 FCC KDB 447498 D01 v04 FCC KDB 248227 D01 V01r02 FCC KDB 616217 D03 v01 FCC 865664 Health Safety Code 6:2009 RSS-102, Issue 4:2010


### COMMENTS Both antennas transmitting simultaneously in MCS8 MIMO. T hese tests show no overlapping SAR regions and lower SAR levels than other modes reported in this SAR evaluation


### DEVIATIONS FROM TEST STANDARD
None


### RESULTS

Test Configuration Frequency Band Transmit Frequency

(MHz) Transmit Channel Data Rate (Mbps) Channel Bandwidth

(MHz) Antenna Port Output Power EUT Position SAR Drift During Test

(dB) Measured 1g SAR Level

(mW/g) Test#

Body 5.6 5580 116 MCS8 20 A&B 12 Back 0.09 1.06 41 Body 5.6 5550 108/112 MCS8 40 A&B 11 Back 0.21 0.767 42WSTD.12.07.20 

# SAR TEST DATA

Tested By: Ethan Schoonover Room Temperature (°C): 21.6°C Date: 7/25/2012 3:25:58 PM Liquid Temperature (°C): 22.1°C Serial Number: EV3 Humidity (%RH): 44.6% Configuration: MCSO1607-1 Bar. Pressure (mb): 1011 mb Comments:

Test 41, 7-25-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5580 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5580 MHz; σ = 5.892 mho/m; ε r = 46.694; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Zoom Scan (8x8x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 19.327 V/m; Power Drift = 0.09 dB Peak SAR (extrapolated) = 4.482 mW/g SAR(1 g) = 1.06 mW/g; SAR(10 g) = 0.298 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 2.24 mW/g

Body/Body/Area scan 2 (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 1.47 mW/g

Body/Body/Reference scan (71x101x1): Measurement grid: dx=30mm, dy=30mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.535 mW/g

Body/Body/Zoom Scan 3 (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 16.280 V/m; Power Drift = 0.18 dB Peak SAR (extrapolated) = 4.003 mW/g SAR(1 g) = 0.903 mW/g; SAR(10 g) = 0.256 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 1.91 mW/g

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 1.98 mW/gWSTD.12.07.20 

# SAR TEST DATA

Body/Body/Z Scan 2 (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.408 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.500 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 41, 7-25-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Ethan Schoonover Room Temperature (°C): 21.6°C Date: 7/25/2012 4:58:23 PM Liquid Temperature (°C): 22.1°C Serial Number: EV3 Humidity (%RH): 44.6% Configuration: MCSO1607-1 Bar. Pressure (mb): 1011 mb Comments:

Test 42, 7-25-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D5GHz (5000.0 - 6000.0 MHz); Frequency: 5580 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 5580 MHz; σ = 5.892 mho/m; ε r = 46.694; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Zoom Scan (8x8x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 16.747 V/m; Power Drift = 0.21 dB Peak SAR (extrapolated) = 2.856 mW/g SAR(1 g) = 0.767 mW/g; SAR(10 g) = 0.210 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 1.70 mW/g

Body/Body/Area scan 2 (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 1.24 mW/g

Body/Body/Reference scan (71x101x1): Measurement grid: dx=30mm, dy=30mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.410 mW/g

Body/Body/Zoom Scan 3 (9x9x9)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=2.5mm Reference Value = 15.219 V/m; Power Drift = -0.20 dB Peak SAR (extrapolated) = 3.432 mW/g SAR(1 g) = 0.766 mW/g; SAR(10 g) = 0.198 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 1.69 mW/g

Body/Body/Area scan (81x81x1): Measurement grid: dx=10mm, dy=10mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 1.53 mW/g

Body/Body/Z Scan 2 (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.377 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mmWSTD.12.07.20 

# SAR TEST DATA

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.407 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 42, 7-25-12WSTD.12.07.20 

# SAR TEST DATA

Test 41, 7-25-12WSTD.12.07.20 

# SAR TEST DATA

EUT: 1516 Work Order: MCSO1607 Customer: Microsoft Corporation Job Site: EV08 Attendees: None Customer Project: None


### TEST SPECIFICATIONS Specification: Method:

FCC 2.1093:2011 FCC 15.247:2011 FCC 15.407:2001 FCC OET 65C:2001 IEEE Std 1528:2003 FCC KDB 447498 D01 v04 FCC KDB 248227 D01 V01r02 FCC KDB 616217 D03 v01 FCC 865664 Health Safety Code 6:2009 RSS-102, Issue 4:2010


### COMMENTS
None


### DEVIATIONS FROM TEST STANDARD
None


### RESULTS

Test Configuration Frequency Band Transmit Frequency

(MHz) Transmit Channel Data Rate (Mbps) "Channel Bandwidth Antenna Port EUT Output Power (dBm) EUT Position SAR Drift During Test

(dB) Measured 1g SAR Level

(mW/g) Test#

Body 2.4 2437 4/8 7.2 (MCS0) 40 A 16 Top -0.05 0.376 1 Body 2.4 2437 4/8 7.2 (MCS0) 40 A 16 Back -0.09 1.35 2 Body 2.4 2422 1/5 7.2 (MCS0) 40 A 16 Back -0.13 1.32 2a Body 2.4 2452 7/11 7.2 (MCS0) 40 A 16 Back -0.13 1.37 2b Body 2.4 2412 1 1 20 A 16 Top -0.12 0.411 3 Body 2.4 2412 1 1 20 A 16 Back -0.1 1.44 4 Body 2.4 2437 6 1 20 A 16 Back -0.12 1.38 4a Body 2.4 2462 11 1 20 A 16 Back 0.09 1.36 4b Body 2.4 2437 4/8 7.2 (MCS0) 40 B 16 Top -0.12 0.285 5 Body 2.4 2437 4/8 7.2 (MCS0) 40 B 16 Back -0.07 0.879 6 Body 2.4 2422 1/5 7.2 (MCS0) 40 B 16 Back -0.06 0.905 6a Body 2.4 2452 7/11 7.2 (MCS0) 40 B 16 Back -0.03 0.949 6b Body 2.4 2437 6 1 20 B 16 Top -0.12 0.28 7 Body 2.4 2437 6 1 20 B 16 Back -0.04 0.893 8WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 21.5°C Date: 7/24/2012 11:51:54 AM Liquid Temperature (°C): 21.1°C Serial Number: EV3 Humidity (%RH): 47.5% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1016.9 mb Comments: None

Test 1, 7-24-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D2450 (2450.0 MHz); Frequency: 2437 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 , Medium parameters used (interpolated): f = 2437 MHz; σ = 1.921 mho/m; ε r = 50.286; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (71x81x1): Measurement grid: dx=15mm, dy=15mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.485 mW/g

Body/Body/Zoom Scan (7x8x7)/Cube 0: Measurement grid: dx=5mm, dy=5mm, dz=5mm Reference Value = 15.513 V/m; Power Drift = -0.05 dB Peak SAR (extrapolated) = 1.030 mW/g SAR(1 g) = 0.376 mW/g; SAR(10 g) = 0.156 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.543 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.208 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 1, 7-24-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 22.3°C Date: 7/24/2012 7:27:57 AM Liquid Temperature (°C): 22.1°C Serial Number: EV3 Humidity (%RH): 48.9% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1016.7 mb Comments: None

Test 2, 7-24-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D2450 (2450.0 MHz); Frequency: 2437 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 , Medium parameters used (interpolated): f = 2437 MHz; σ = 1.921 mho/m; ε r = 50.286; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Reference scan (71x101x1): Measurement grid: dx=30mm, dy=30mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.438 mW/g

Body/Body/Area scan (71x71x1): Measurement grid: dx=15mm, dy=15mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 1.49 mW/g

Body/Body/Zoom Scan (7x7x7)/Cube 0: Measurement grid: dx=5mm, dy=5mm, dz=5mm Reference Value = 26.703 V/m; Power Drift = -0.09 dB Peak SAR (extrapolated) = 3.331 mW/g SAR(1 g) = 1.35 mW/g; SAR(10 g) = 0.529 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 1.83 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.866 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 2, 7-24-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 22.3°C Date: 7/24/2012 1:26:34 PM Liquid Temperature (°C): 22.1°C Serial Number: EV3 Humidity (%RH): 48.9% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1016.7 mb Comments: None

Test 2a, 7-24-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D2450 (2450.0 MHz); Frequency: 2422 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 , Medium parameters used (interpolated): f = 2422 MHz; σ = 1.895 mho/m; ε r = 50.481; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (71x81x1): Measurement grid: dx=15mm, dy=15mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 1.61 mW/g

Body/Body/Zoom Scan (7x7x7)/Cube 0: Measurement grid: dx=5mm, dy=5mm, dz=5mm Reference Value = 28.505 V/m; Power Drift = -0.13 dB Peak SAR (extrapolated) = 3.255 mW/g SAR(1 g) = 1.32 mW/g; SAR(10 g) = 0.516 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 1.77 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.856 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 2a, 7-24-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Ethan Schoonover Room Temperature (°C): 23.8°C Date: 7/24/2012 Liquid Temperature (°C): 22.8°C Serial Number: EV3 Humidity (%RH): 38.7% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1016 mb Comments: Output power = 16dBm

Test 2b, 7-24-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D2450 (2450.0 MHz); Frequency: 2452 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 , Medium parameters used (interpolated): f = 2452 MHz; σ = 1.946 mho/m; ε r = 50.089; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (71x81x1): Measurement grid: dx=15mm, dy=15mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 1.68 mW/g

Body/Body/Zoom Scan (7x7x7)/Cube 0: Measurement grid: dx=5mm, dy=5mm, dz=5mm Reference Value = 28.833 V/m; Power Drift = -0.13 dB Peak SAR (extrapolated) = 3.390 mW/g SAR(1 g) = 1.37 mW/g; SAR(10 g) = 0.537 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 1.85 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.876 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 2b, 7-24-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 22.3°C Date: 7/24/2012 9:58:34 AM Liquid Temperature (°C): 22.1°C Serial Number: EV3 Humidity (%RH): 48.9% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1016.7 mb Comments: None

Test 3, 7-24-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D2450 (2450.0 MHz); Frequency: 2412 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 , Medium parameters used (interpolated): f = 2412 MHz; σ = 1.874 mho/m; ε r = 50.591; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Reference scan (41x101x1): Measurement grid: dx=30mm, dy=30mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.208 mW/g

Body/Body/Area scan (71x81x1): Measurement grid: dx=15mm, dy=15mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.537 mW/g

Body/Body/Zoom Scan (7x8x7)/Cube 0: Measurement grid: dx=5mm, dy=5mm, dz=5mm Reference Value = 16.680 V/m; Power Drift = -0.12 dB Peak SAR (extrapolated) = 1.112 mW/g SAR(1 g) = 0.411 mW/g; SAR(10 g) = 0.168 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.591 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.228 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 3, 7-24-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 22.3°C Date: 7/24/2012 9:04:03 AM Liquid Temperature (°C): 22.1°C Serial Number: EV3 Humidity (%RH): 48.9% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1016.7 mb Comments: None

Test 4, 7-24-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D2450 (2450.0 MHz); Frequency: 2412 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 , Medium parameters used (interpolated): f = 2412 MHz; σ = 1.874 mho/m; ε r = 50.591; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (71x71x1): Measurement grid: dx=15mm, dy=15mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 1.60 mW/g

Body/Body/Zoom Scan (7x7x7)/Cube 0: Measurement grid: dx=5mm, dy=5mm, dz=5mm Reference Value = 27.840 V/m; Power Drift = -0.10 dB Peak SAR (extrapolated) = 3.535 mW/g SAR(1 g) = 1.44 mW/g; SAR(10 g) = 0.561 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 1.90 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.922 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 4, 7-24-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Ethan Schoonover Room Temperature (°C): 23.8°C Date: 7/24/2012 Liquid Temperature (°C): 22.8°C Serial Number: EV3 Humidity (%RH): 38.7% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1016 mb Comments: Output power = 16dBm

Test 4a, 7-24-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D2450 (2450.0 MHz); Frequency: 2437 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 , Medium parameters used (interpolated): f = 2437 MHz; σ = 1.921 mho/m; ε r = 50.286; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (71x81x1): Measurement grid: dx=15mm, dy=15mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 1.70 mW/g

Body/Body/Zoom Scan (7x7x7)/Cube 0: Measurement grid: dx=5mm, dy=5mm, dz=5mm Reference Value = 29.219 V/m; Power Drift = -0.12 dB Peak SAR (extrapolated) = 3.387 mW/g SAR(1 g) = 1.38 mW/g; SAR(10 g) = 0.538 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 1.87 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.892 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 4a, 7-24-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Ethan Schoonover Room Temperature (°C): 23.8°C Date: 7/24/2012 Liquid Temperature (°C): 22.8°C Serial Number: EV3 Humidity (%RH): 38.7% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1016 mb Comments: Output power = 16dBm

Test 4b, 7-24-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D2450 (2450.0 MHz); Frequency: 2462 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 , Medium parameters used (interpolated): f = 2462 MHz; σ = 1.963 mho/m; ε r = 49.976; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (71x81x1): Measurement grid: dx=15mm, dy=15mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 1.66 mW/g

Body/Body/Zoom Scan (7x7x7)/Cube 0: Measurement grid: dx=5mm, dy=5mm, dz=5mm Reference Value = 28.462 V/m; Power Drift = -0.09 dB Peak SAR (extrapolated) = 3.370 mW/g SAR(1 g) = 1.36 mW/g; SAR(10 g) = 0.531 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 1.90 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.872 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 4b, 7-24-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 22.3°C Date: 7/24/2012 11:08:34 AM Liquid Temperature (°C): 22.1°C Serial Number: EV3 Humidity (%RH): 48.9% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1016.7 mb Comments: None

Test 5, 7-24-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D2450 (2450.0 MHz); Frequency: 2437 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 , Medium parameters used (interpolated): f = 2437 MHz; σ = 1.921 mho/m; ε r = 50.286; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (71x71x1): Measurement grid: dx=15mm, dy=15mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.393 mW/g

Body/Body/Zoom Scan (7x7x7)/Cube 0: Measurement grid: dx=5mm, dy=5mm, dz=5mm Reference Value = 14.013 V/m; Power Drift = -0.12 dB Peak SAR (extrapolated) = 0.744 mW/g SAR(1 g) = 0.285 mW/g; SAR(10 g) = 0.123 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.398 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.198 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 5, 7-24-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 22.3°C Date: 7/24/2012 8:08:09 AM Liquid Temperature (°C): 22.1°C Serial Number: EV3 Humidity (%RH): 48.9% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1016.7 mb Comments: None

Test 6, 7-24-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D2450 (2450.0 MHz); Frequency: 2437 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 , Medium parameters used (interpolated): f = 2437 MHz; σ = 1.921 mho/m; ε r = 50.286; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Reference scan (71x101x1): Measurement grid: dx=30mm, dy=30mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.952 mW/g

Body/Body/Area scan (71x71x1): Measurement grid: dx=15mm, dy=15mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 1.24 mW/g

Body/Body/Zoom Scan (7x7x7)/Cube 0: Measurement grid: dx=5mm, dy=5mm, dz=5mm Reference Value = 24.403 V/m; Power Drift = -0.07 dB Peak SAR (extrapolated) = 2.195 mW/g SAR(1 g) = 0.879 mW/g; SAR(10 g) = 0.401 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 1.15 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.511 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 6, 7-24-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Ethan Schoonover Room Temperature (°C): 24.5°C Date: 7/24/2012 Liquid Temperature (°C): 23.0°C Serial Number: EV3 Humidity (%RH): 40.3% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1016 mb Comments: Output power = 16dBm

Test 6a, 7-24-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D2450 (2450.0 MHz); Frequency: 2422 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 , Medium parameters used (interpolated): f = 2422 MHz; σ = 1.895 mho/m; ε r = 50.481; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (71x71x1): Measurement grid: dx=15mm, dy=15mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 1.30 mW/g

Body/Body/Zoom Scan (7x7x7)/Cube 0: Measurement grid: dx=5mm, dy=5mm, dz=5mm Reference Value = 24.847 V/m; Power Drift = -0.06 dB Peak SAR (extrapolated) = 2.276 mW/g SAR(1 g) = 0.905 mW/g; SAR(10 g) = 0.405 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 1.20 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.520 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 6a, 7-24-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Ethan Schoonover Room Temperature (°C): 24.5°C Date: 7/24/2012 Liquid Temperature (°C): 23.0°C Serial Number: EV3 Humidity (%RH): 40.3% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1016 mb Comments: Output power = 16dBm

Test 6b, 7-24-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D2450 (2450.0 MHz); Frequency: 2452 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 , Medium parameters used (interpolated): f = 2452 MHz; σ = 1.946 mho/m; ε r = 50.089; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Area scan (71x71x1): Measurement grid: dx=15mm, dy=15mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 1.41 mW/g

Body/Body/Zoom Scan (7x7x7)/Cube 0: Measurement grid: dx=5mm, dy=5mm, dz=5mm Reference Value = 25.275 V/m; Power Drift = -0.03 dB Peak SAR (extrapolated) = 2.425 mW/g SAR(1 g) = 0.949 mW/g; SAR(10 g) = 0.428 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 1.26 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.516 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 6b, 7-24-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 22.3°C Date: 7/24/2012 10:40:35 AM Liquid Temperature (°C): 22.1°C Serial Number: EV3 Humidity (%RH): 48.9% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1016.7 mb Comments: None

Test 7, 7-24-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D2450 (2450.0 MHz); Frequency: 2437 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 , Medium parameters used (interpolated): f = 2437 MHz; σ = 1.921 mho/m; ε r = 50.286; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Reference scan (41x101x1): Measurement grid: dx=30mm, dy=30mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.247 mW/g

Body/Body/Area scan (71x71x1): Measurement grid: dx=15mm, dy=15mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.386 mW/g

Body/Body/Zoom Scan (7x7x7)/Cube 0: Measurement grid: dx=5mm, dy=5mm, dz=5mm Reference Value = 13.948 V/m; Power Drift = -0.12 dB Peak SAR (extrapolated) = 0.710 mW/g SAR(1 g) = 0.280 mW/g; SAR(10 g) = 0.120 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.392 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.193 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 7, 7-24-12WSTD.12.07.20 

# SAR TEST DATA

Tested By: Ethan Schoonover Room Temperature (°C): 24.5°C Date: 7/24/2012 Liquid Temperature (°C): 23.0°C Serial Number: EV3 Humidity (%RH): 40.3% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1016 mb Comments: Output power = 16dBm

Test 8, 7-24-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D2450 (2450.0 MHz); Frequency: 2412 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 , Medium parameters used (interpolated): f = 2412 MHz; σ = 1.874 mho/m; ε r = 50.591; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of Total (measured) = 0.500 mWg

Body/Body/Reference scan (71x101x1): Measurement grid: dx=30mm, dy=30mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.993 mW/g

Body/Body/Area scan (71x71x1): Measurement grid: dx=15mm, dy=15mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 1.29 mW/g

Body/Body/Zoom Scan (7x7x7)/Cube 0: Measurement grid: dx=5mm, dy=5mm, dz=5mm Reference Value = 24.853 V/m; Power Drift = -0.04 dB Peak SAR (extrapolated) = 2.251 mW/g SAR(1 g) = 0.893 mW/g; SAR(10 g) = 0.403 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 1.18 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 8, 7-24-12WSTD.12.07.20 

# SAR TEST DATA

Test 4, 7-24-12WSTD.12.07.20 

# SAR TEST DATA

EUT: 1516 Work Order: MCSO1607 Customer: Microsoft Corporation Job Site: EV08 Attendees: None Customer Project: None


### TEST SPECIFICATIONS Specification: Method:

FCC 2.1093:2011 FCC 15.247:2011 FCC 15.407:2001 FCC OET 65C:2001 IEEE Std 1528:2003 FCC KDB 447498 D01 v04 FCC KDB 248227 D01 V01r02 FCC KDB 616217 D03 v01 FCC 865664 Health Safety Code 6:2009 RSS-102, Issue 4:2010


### COMMENTS Both antennas transmitting simultaneously in MCS8 MIMO. T hese tests show no overlapping SAR regions and lower SAR levels than other modes reported in this SAR evaluation.


### DEVIATIONS FROM TEST STANDARD
None


### RESULTS

Test Configuration Frequency Band Transmit Frequency

(MHz) Transmit Channel Data Rate (Mbps) Channel Bandwidth

(MHz) Antenna Port EUT Output Power (dBM) EUT Position SAR Drift During Test

(dB) Measured 1g SAR Level

(mW/g) Test#

Body 2.4GHz 2412 1 MCS8 20 A & B 16 Back -0.13 1.07 43 Body 2.4GHz 2452 7/11 MCS8 40 A & B 16 Back -0.12 1.19 44WSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 23.4°C Date: 7/27/2012 8:17:21 AM Liquid Temperature (°C): 22.1°C Serial Number: EV3 Humidity (%RH): 45.6% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1015.5 mb Comments: None

Test 43, 7-27-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D2450 (2450.0 MHz); Frequency: 2412 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 2412 MHz; σ = 1.874 mho/m; ε r = 50.591; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Reference scan (71x101x1): Measurement grid: dx=30mm, dy=30mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.616 mW/g

Body/Body/Area scan (71x71x1): Measurement grid: dx=15mm, dy=15mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.987 mW/g

Body/Body/Zoom Scan (7x7x7)/Cube 0: Measurement grid: dx=5mm, dy=5mm, dz=5mm Reference Value = 20.700 V/m; Power Drift = -0.13 dB Peak SAR (extrapolated) = 2.662 mW/g SAR(1 g) = 1.07 mW/g; SAR(10 g) = 0.422 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 1.45 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.701 mW/g

Body/Body/Area scan 2 (71x71x1): Measurement grid: dx=15mm, dy=15mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.964 mW/g

Body/Body/Zoom Scan 2 (7x7x7)/Cube 0: Measurement grid: dx=5mm, dy=5mm, dz=5mm Reference Value = 18.709 V/m; Power Drift = 0.17 dB Peak SAR (extrapolated) = 1.729 mW/g SAR(1 g) = 0.711 mW/g; SAR(10 g) = 0.320 mW/g

Body/Body/Z Scan 2 (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mmWSTD.12.07.20 

# SAR TEST DATA

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.420 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 43, 7-27-12

Test 43, 7-27-12 Reference ScanWSTD.12.07.20 

# SAR TEST DATA

Tested By: Jennifer Herrett Room Temperature (°C): 22.3°C Date: 7/27/2012 10:47:48 AM Liquid Temperature (°C): 22.1°C Serial Number: EV3 Humidity (%RH): 45.6% Configuration: MCSO1607 - 1 Bar. Pressure (mb): 1015.5 mb Comments: None

Test 44, 7-27-12 DUT: Tablet Computer; Type: 1516; Serial: EV3 Communication System: CW; Communication System Band: D2450 (2450.0 MHz); Frequency: 2452 MHz;Communication System PAR: 0 dB; PMF: 1 Medium parameters used (interpolated): f = 2452 MHz; σ = 1.946 mho/m; ε r = 50.089; ρ = 1000 kg/m 3 , Medium parameters used: σ = 0 mho/m, ε r = 1; ρ = 1000 kg/m 3 Phantom section: Flat Section Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007) DASY Configuration:

 DASY52 52.8.1(838); SEMCAD X 14.6.5(6469)

Body/Body/Reference scan (71x101x1): Measurement grid: dx=30mm, dy=30mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 0.720 mW/g

Body/Body/Area scan (71x71x1): Measurement grid: dx=15mm, dy=15mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 1.12 mW/g

Body/Body/Zoom Scan (7x7x7)/Cube 0: Measurement grid: dx=5mm, dy=5mm, dz=5mm Reference Value = 21.857 V/m; Power Drift = -0.12 dB Peak SAR (extrapolated) = 2.951 mW/g SAR(1 g) = 1.19 mW/g; SAR(10 g) = 0.468 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 1.62 mW/g

Body/Body/Z Scan (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.762 mW/g

Body/Body/Area scan 2 (71x71x1): Measurement grid: dx=15mm, dy=15mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (interpolated) = 1.15 mW/g

Body/Body/Zoom Scan 2 (7x7x7)/Cube 0: Measurement grid: dx=5mm, dy=5mm, dz=5mm Reference Value = 20.435 V/m; Power Drift = 0.13 dB Peak SAR (extrapolated) = 2.000 mW/g SAR(1 g) = 0.802 mW/g; SAR(10 g) = 0.367 mW/g

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 1.06 mW/gWSTD.12.07.20 

# SAR TEST DATA

Body/Body/Z Scan 2 (1x1x21): Measurement grid: dx=20mm, dy=20mm, dz=5mm

Info: Interpolated medium parameters used for SAR evaluation. Maximum value of SAR (measured) = 0.456 mW/g

Approved ByWSTD.12.07.20 

# SAR TEST DATA

Test 44, 7-27-12

Test 44, 7-27-12 Reference ScanWSTD.12.07.20 

# SAR TEST DATA

Test 44, 7-27-12WSTD.12.07.20 

# SAR TEST DATA

Test 44, 7-27-12WSMS 2012.08.15 

# SYSTEM AND TEST SITE DESCRIPTION

SAR MEASUREMENT SYSTEM

Schmid & Partner Engineering AG, DASY52

Northwest EMC selected the leader in SAR evaluation system s to provide the measurement tools for this evaluation. SPEAG’s DASY52 is the fastest and most accurate scanner on the market. It is fully compatible with all world-wide standards for transmitters operating at the ear or within 20cm of the body. It provides full compatibility with IEC 62209-1, IE C 62209-2, IEEE 1528 as well as national adaptations such as FCC OET-65c and Korean Std. MIC #2000-93

The DASY52 system for performing compliance tests consists of the following items:

 A standard high precision 6-axis robot (Staubli TX=RX fam ily) with controller, teach pendant and software. An arm extension for accommodating the data acquisition electronics (DAE).

 An isotropic field probe optimized and calibrated for the targeted measurement.

 A data acquisition electronics (DAE) whic h performs the signal amplification, signal multiplexing, AD-conversion, offset measurements, mechanical surface detection, col lision detection, etc. The unit is battery powered with standard or rechargeable batteries. The sign al is optically transmitted to the EOC.

 The Electro-optical converter (EOC) performs the conversion from optical to electrical signals for the digital communication to the DAE. To use optical surface detecti on, a special version of t he EOC is required. The EOC signal is transmitted to the measurement server.

 The function of the measurement server is to perform the time critical tasks such as signal filtering, control of the robot operation and fast movement interrupts.

 The Light Beam used is for probe alignment. This improv es the (absolute) accuracy of the probe positioning.

 A computer running WinXP and the DASY5 software.

 Remote control and teach pendant as well as additional circ uitry for robot safety such as warning lamps, etc.

 The SAM twin phantom, oval flat phantom, device holder, tissue simulating liquids, and validation dipole kits.WSMS 2012.08.15 

# SYSTEM AND TEST SITE DESCRIPTION

TEST SITE

Northwest EMC, Lab EV08

The SAR measurement system is located in a semi-anechoic c hamber. This provides an ambient free environment that also eliminates reflections.

The chamber is 12 ft wide by 16 ft long x 8 ft high. A dedicat ed HVAC unit provides +/- 1 degree C temperature control.WSEQ.12.07.12 

# TEST EQUIPMENT

TEST EQUIPMENT Description Manufacturer Model ID Last Cal. Interval Antenna, Dipole 2450MHz SAR SPEAG D2450V 2 ADL 12/09/2011 12 mo Humidity Temperature Meter Omegaette HH311 DTX 03/29/2011 24 mo Humidity Temperature Meter Omegaette HH311 DTY 03/29/2011 24 mo Dielectric Probe Kit Agilent 85070E IPP 09/08/2010 24 mo Network Analyzer Hewlett Packard N5230A NAD 06/19/2012 12 mo Robot Arm Staeubli TX60LSPEAG SAA NCR 0 mo Phantom, 2mm Oval ELI4 (Body) SPEAG QD OVA 001 BB SAC NCR 0 mo Light Beam Unit SPEAG SE UKS 030 AA SAD NCR 0 mo SAR Probe SPEAG EX3DV4 SAG 11/17/2011 12 mo DAE SPEAG SD 000 D04 EJ SAH 11/08/2011 12 mo Robot Controller Staeubli CS8C SAI NCR 0 mo Robot Chasis and power Supply Staeubli N/A SAJ NCR 0 mo DASY5 Measurement Server Staeubli DAYS5 SAK NCR 0 mo Body Solution SPEAG MSL 2450 SAM Within 24 hours of a measurement Device Holder SPEAG N/A SAW NCR 0 mo Power Sensor Agilent E9300H SQO 06/06/2011 24 mo Power Meter Agilent N1913A SQR 06/06/2011 24 mo MXG Analog Signal Generator Agilent N5181A TIG NCR 0 mo Amplifier Mini Circuits ZVE-3W-83+ TTA NCR 0 mo Antenna, Dipole 5.1-5.8GHz SAR SPEAG D5GH zV2 ADM 12/14/2011 12 mo Body Solution SPEAG MSL 501 SAV Within 24 hours of a measurment SAR Probe SPEAG EX3DV4 R074 11/17/2011 12 mo Power Sensor Gigatronics 80701A SPL 07/08/2011 24 mo Power Meter Gigatronics 8650271 SPM 01/09/2012 24 moWSMBL 2012.08.15 

# MEASUREMENT UNCERTAINTY

MEASUREMENT UNCERTAINTY BUDGETS PER IEEE 1528:2003

300-3000 MHz Range

Uncertainty Component Tolerance (+/- %) Probability Distribution Divisor c i (1g) c i (10g) u i (1g) (+/-%) u i (10g) (+/-%) v i
Measurement System

Probe calibration (k=1) 5.5 normal 1 1 1 5.5 5.5 ∞

Axial isotropy 4.7 rectangular 1.732 0.707 0.707 1.9 1.9 ∞

Hemispherical isotropy 9.6 rectangular 1.732 0.707 0.707 3.9 3.9 ∞

Boundary effect 1.0 rectangular 1.732 1 1 0.6 0.6 ∞

Linearity 4.7 rectangular 1.732 1 1 2.7 2.7 ∞

System detection limits 1.0 rectangular 1.732 1 1 0.6 0.6 ∞

Readout electronics 0.3 normal 1 1 1 0.3 0.3 ∞

Response time 0.8 rectangular 1.732 1 1 0.5 0.5 ∞

Integration time 2.6 rectangular 1.732 1 1 1.5 1.5 ∞

RF ambient conditions - noise 1.7 rectangular 1.732 1 1 1.0 1.0 ∞

RF Ambient Reflections 0.0 rectangular 1.732 1 1 0.0 0.0 ∞

Probe positioner mechanical tolerance 0.4 rectangular 1.732 1 1 0.2 0.2 ∞ Probe positioner with respect to phantom shell 2.9 rectangular 1.732 1 1 1.7 1.7 ∞ Extrapolation, interpolation, and integration algorithms for max. SAR evaluation 1.0 rectangular 1.732 1 1 0.6 0.6 ∞

Test Sample Related

Device Positioning 2.9 normal 1 1 1 2.9 2.9 145

Device Holder 3.6 normal 1 1 1 3.6 3.6 5

Power Drift 5.0 rectangular 1.732 1 1 2.9 2.9 ∞

Phantom and tissue parameters

Phantom Uncertainty - shell thickness tolerances 4.0 rectangular 1.732 1 1 2.3 2.3 ∞ Liquid conductivity - deviation from target values 5.0 rectangular 1.732 0.64 0.43 1.8 1.2 ∞ Liquid conductivity - measurement uncertainty 6.5 normal 1 0.64 0.43 4.2 2.8 ∞ Liquid permittivity - deviation from target values 5.0 rectangular 1.732 0.6 0.49 1.7 1.4 ∞

Liquid permittivity - measurement uncertainty 3.2 normal 1 0.6 0.49 1.9 1.6 ∞

11.2 10.6 387

22.5 21.2 Combined Standard Uncertainty

Expanded Measurement Uncertainty (95% Confidence/ RSS

normal (k=2)WSMBH 2012.08.15 

# MEASUREMENT UNCERTAINTY

MEASUREMENT UNCERTAINTY BUDGETS PER IEEE 1528:2003

3000-6000 MHz Range

Uncertainty Component Tolerance (+/- %) Probability Distribution Divisor c i (1g) c i (10g) u i (1g) (+/-%) u i (10g) (+/-%) v i
Measurement System

Probe calibration (k=1) 6.55 n o r m a l 1 1 16

. 6 6

. 6 ∞

Axial isotropy 4.7 rectangular 1.732 0.707 0.707 1.9 1.9 ∞

Hemispherical isotropy 9.6 rectangular 1.732 0.707 0.707 3.9 3.9 ∞

Boundary effect 2.0 rectangular 1.732 1 1 1.2 1.2 ∞

Linearity 4.7 rectangular 1.732 1 1 2.7 2.7 ∞

System detection limits 1.0 rectangular 1.732 1 1 0.6 0.6 ∞

Readout electronics 0.3 n o r m a l 1 1 10

. 3 0

. 3 ∞

Response time 0.8 rectangular 1.732 1 1 0.5 0.5 ∞

Integration time 2.6 rectangular 1.732 1 1 1.5 1.5 ∞

RF ambient conditions - noise 1.7 rectangular 1.732 1 1 1.0 1.0 ∞

RF Ambient Reflections 0.0 rectangular 1.732 1 1 0.0 0.0 ∞

Probe positioner mechanical tolerance 0.8 rectangular 1.732 1 1 0.5 0.5 ∞ Probe positioner with respect to phantom shell 9.9 rectangular 1.732 1 1 5.7 5.7 ∞ Extrapolation, interpolation, and integration algorithms for max. SAR evaluation 4.0 rectangular 1.732 1 1 2.3 2.3 ∞

Test Sample Related

Device Positioning 2.9 normal 1 1 1 2.9 2.9 145

Device Holder 3.6 normal 1 1 1 3.6 3.6 5

Power Drift 5.0 rectangular 1.732 1 1 2.9 2.9 ∞

Phantom and tissue parameters

Phantom Uncertainty - shell thickness tolerances 4.0 rectangular 1.732 1 1 2.3 2.3 ∞ Liquid conductivity - deviation from target values 5.0 rectangular 1.732 0.64 0.43 1.8 1.2 ∞ Liquid conductivity - measurement uncertainty 6.5 normal 1 0.64 0.43 4.2 2.8 ∞ Liquid permittivity - deviation from target values 5.0 rectangular 1.732 0.6 0.49 1.7 1.4 ∞

Liquid permittivity - measurement uncertainty 3.2 normal 1 0.6 0.49 1.9 1.6 ∞

13.2 12.7 330

26.5 25.4 Combined Standard Uncertainty RSS

Expanded Measurement Uncertainty (95% Confidence/ normal (k=2)REV 2012.08.02 

# PROBE CALIBRATION

PROBE CALIBRATION
Please see attached calibration data

### Calibration Laboratory of Schweizerischer Kalibrierdienst 

### 5 Schmid & Partner Service suisse d'etalonnage 

### C Servizio svizzero di taratura 

### Engineering AG 

### 5 Zeughausstrasse 43, 8004 Zurich, Switzerland Swiss Calibration Service

Accredited by the Swiss Accreditation Service (SAS) Accreditation No.: SCS 108

The Swiss Accreditation Service is one of the signatories to the EA
Multilateral Agreement for the recognition of calibration certificates

Client 

### Northwest EMC Certificate No: 

### EX3-3746_Nov11



## ICALIBRATION CERTIFICATE

Object EX3DV4

- SN:3746

Calibration procedure(s) QA CAL-01.v8, QA CAL-14.v3

, QA CAL-23.v4

, QA CAL-2S.v4 Calibration procedure for dosimetric E-field probes

I

Calibration date : November 17,2011 

### J

This calibration certificate documents the traceability to national standards, which realize the physical units of measurements (SI)

. The measurements and the uncertainties with confidence probability are given on the following pages and are part of the certificate.

All calibrations have been conducted in the closed laboratory facility: environment temperature (22 ± 3tC and humidity < 70%.

Calibration Equipment used (M&TE critical for calibration)

Primary Standards 10 Cal Date (Certificate No.) Scheduled Calibration

Power meter E4419B GB41293874 31-Mar-11 (No

. 217-01372) Apr-12

PowersensorE4412A MY41498087 31-Mar-11 (No. 217-01372) Apr-12

Reference 3 dB Attenuator SN : S5054 (3c) 29-Mar-11 (No. 217-01369) Apr-12

Reference 20 dB Attenuator SN : S5086 (20b) 29-Mar-11 (No

. 217-01367) Apr-12

Reference 30 dB Attenuator SN: S5129 (30b) 29-Mar-11 (No. 217-01370) Apr-12
Reference Probe ES3DV2 SN : 3013 29-Dec-10 (No

. ES3

- 3013 Dec10) Dec-11

DAE4 SN: 654 3-May-11 (No

. DAE4-654_May11) May-12

Secondary Standards 10 Check Date (in house) Scheduled Check

RF generator HP 8648C US3642U01700 4-Aug-99 (in house check Apr -11 ) In house check : Apr-13

Network Analyzer HP 8753E US37390585 18-0ct-01 (in house check Oct-11) In house check: Oct-12

Name Function
Calibrated by: Katja Pokovic Technical Manager

Approved by : Niels Kuster Quality Manager 

# /

Issued : November 17

, 2011

This calibration certificate shall not be reproduced except in full without written approval of the laboratory.

Certificate No: EX3-3746

_ Nov11 Page 1 of 11 

# I

### Calibration Laboratory of Schweizerischer Kalibrierdienst 

### S Schmid & Partner Service suisse d'etalonnage 

### C Servizio svizzero di taratura 

### Engineering AG 

### S Zeughausstrasse 43, 8004 Zurich, Switzerland Swiss Calibration Service

Accredited by the Swiss Accreditation Service (SAS) Accreditation No.: SCS 108

The Swiss Accreditation Service is one of the signatories to the EA
Multilateral Agreement for the recognition of calibration certificates 

### Glossary:
TSL tissue simulating liquid NORMx,y,z sensitivity in free space ConvF sensitivity in TSL / NORMx,y,z DCP diode compression point CF crest factor (1/duty_cycle) of the RF signal A,B,C modulation dependent linearization parameters Polarization <p <p rotation around probe axis
Polarization S S rotation around an axis that is in the plane normal to probe axis (at measurement center), i.e., S = 0 is normal to probe axis



### Calibration is Performed According to the Following Standards: a) IEEE Std 1528-2003, "IEEE Recommended Practice for Determining the Peak Spatial-Averaged Specific Absorption Rate (SAR) in the Human Head from Wireless Communications Devices: Measurement Techniques", December 2003 b) IEC 62209-1, "Procedure to measure the Specific Absorption Rate (SAR) for hand-held devices used in close proximity to the ear (frequency range of 300 MHz to 3 GHz)", February 2005



### Methods Applied and Interpretation of Parameters:

• NORMx,Y,z: Assessed for E-field polarization -S = 0 (f**~** 900 MHz in TEM-cell; f> 1800 MHz : R22 waveguide). NORMx,y,z are only intermediate values

, i.e., the uncertainties of NORMx,y,z does not affect the E 2 -field uncertainty inside TSL (see below ConvF).

• NORM(f)x,Y,z = NORMx

, Y

, z * frequency_response (see Frequency Response Chart). This linearization is implemented in DASY4 software versions later than 4

. 2. The uncertainty of the frequency response is included in the stated uncertainty of ConvF.

• OCPx,Y

, z: DCP are numerical linearization parameters assessed based on the data of power sweep with CW signal (no uncertainty required). DCP does not depend on frequency nor media.

• PAR: PAR is the Peak to Average Ratio that is not calibrated but determined based on the signal characteristics

• Ax

, Y,z ; Bx,Y

, z; Cx

, y,z , VRx,Y,z: A, B, C are numerical linearization parameters assessed based on the data of power sweep for specific modulation signal. The parameters do not depend on frequency nor media

. VR is the maximum calibration range expressed in RMS voltage across the diode

.

• ConvF and Boundary Effect Parameters: Assessed in flat phantom using E-field (or Temperature Transfer Standard for f**~** 800 MHz) and inside waveguide using analytical field distributions based on power measurements for f > 800 MHz. The same setups are used for assessment of the parameters applied for boundary compensation (alpha, depth) of which typical uncertainty values are given. These parameters are used in DASY 4 software to improve probe accuracy close to the boundary. The sensitivity in TSL corresponds to NORMx

, y,z * ConvF whereby the uncertainty corresponds to that given for ConvF. A frequency dependent ConvF is used in DASY version 4.4 and higher which allows extending the validity from ± 50 MHz to ± 100 MHz.

• Spherical isotropy**(3~** deviation from isotropy): in a field of low gradients realized using a flat phantom exposed by a patch antenna.

• Sensor Offset: The sensor offset corresponds to the offset of virtual measurement center from the probe tip (on probe axis). No tolerance required

.

Certificate No : EX3-3746_Nov11 Page 2 of 11EX3DV4

- SN : 3746 November 17

, 2011



# Probe 

# EX3DV4

SN:3746

Manufactured: March 26, 2010 Calibrated: November 17,2011

Calibrated for DASY/EASY Systems
(Note : non-compatible with DASY2 system!)

Certificate No: EX3-3746_Nov11 Page 3 of 11EX3DV4- SN:3746 November 17, 2011



# DASY/EASY

- Parameters of Probe: EX3DV4

- SN:3746

Basic Calibration Parameters
Sensor X Sensor Y Sensor Z Unc (k=2)

Norm (JlV/(V/m)2)A 0.49 0.47 0.50 

### ± 10.1 % DCP (mV)t:l 97.5 98

. 1 98.0



### Modulation Calibration Parameters

UID Communication System Name PAR A
dB B
dB C
dB VR mV Unct: (k=2)

10000 CW 0.00 X 0

. 00 0.00 1

. 00 112.2 ±3.0 %

Y 0

. 00 0

. 00 1

. 00 116.0

Z 0.00 0.00 1.00 114.8
10061 IEEE 802.11 b WiFi 2.4 GHz (DSSS, 11 Mbps) 3.60 X 6.31 77

. 5 22.0 114.7 ±2.2 %

y 4.08 73

. 8 21

. 3 111

. 0

Z 6.71 80.1 23.4 115.3
10069 IEEE 802.11 a/h WiFi 5 GHz (OFDM

, 54 Mbps) 12

. 20 X 14.34 73.4 26

. 1 129

. 1 ±4.6 %

y 12.54 71.3 25.5 113.0

Z 14

. 02 73.2 26.1 128.5
10077 IEEE 802.11 g WiFi 2.4 GHz (DSSS/OFDM, 54 Mbps) 13.12 X 13.85 73.4 27.0 110

. 1 ±5

. 2 %

y 13.00 73.9 28.0 139.2

Z 13.46 73.0 26.9 108.5



### The reported uncertainty of measurement is stated as the standard uncertainty of measurement multiplied by the coverage factor k=2, which for a normal distribution corresponds to a coverage probability of approximately 95%.

A The uncertainties of NormX

, Y,Z do not affect the E 2 -field uncertainty inside TSL (see Pages 5 and 6).

B Numerical linearization parameter: uncertainty not required

. E Uncertainty is determined using the max. deviation from linear response applying rectangular distribution and is expressed for the square of the field value.

Certificate No: EX3-3746_Nov11 Page 4 of 11EX3DV4- SN:3746 November 17,2011



# DASY/EASY

- Parameters of Probe: EX3DV4

- SN:3746

Calibration Parameter Determined in Head Tissue Simulating Media

f (MHzl c Relative Permittivity F Conductivity (S/ml F ConvF X ConvF Y ConvF Z Alpha Depth (mm) Unct. (k=2)

2450 39.2 1.80 7.14 7.14 7.14 0.46 0.91 ± 12

. 0 %

5200 36.0 4

. 66 4.90 4.90 4.90 0.35 1.80 ± 13.1 %

5300 35.9 4.76 4.67 4.67 4.67 0.35 1.80 ± 13.1 %

5500 35.6 4.96 4.56 4.56 4.56 0.40 1.80 ± 13.1 %

5800 35.3 5.27 4

. 23 4.23 4.23 0.50 1.80 ± 13.1 

### %

C Frequency validity of ± 100 MHz only applies for DASY v4.4 and higher (see Page 2), else it is restricted to ± 50 MHz

. The uncertainty is the RSS of the ConvF uncertainty at calibration frequency and the uncertainty for the indicated frequency band

.
F At frequencies below 3 GHz, the validity of tissue parameters ( £ and a) can be relaxed to ± 10% if liquid compensation formula is applied to measured SAR values

. At frequencies above 3 GHz, the validity of tissue parameters (£ and a) is restricted to ± 5%

. The uncertainty is the RSS of the ConvF uncertainty for indicated target tissue parameters

.

Certificate No: EX3-3746_Nov11 Page 5 of 11EX3DV4- SN:3746 November 17, 2011



# DASY/EASY

- Parameters of Probe: EX3DV4

- SN:3746

Calibration Parameter Determined in Body Tissue Simulating Media

f (MHz) c Relative Permittivity F Conductivity (S/m) F ConvF X ConvF Y ConvF Z**AI~ha** Depth (mm) Unct. (k=2)

2450 52.7 1.95 6.77 6.77 6.77 0.80 0.50 ± 12.0 %

5200 49.0 5.30 4.10 4.10 4.10 0.52 1.90 ± 13.1 %

5300 48.9 5.42 3.92 3.92 3.92 0.52 1.90 ± 13.1 %

5500 48.6 5.65 3.72 3.72 3.72 0.55 1.90 ± 13.1 %

5800 48.2 6.00 3.85 3.85 3.85 0.60 1.90 ± 13.1 %

C Frequency validity of ± 100 MHz only applies for DASY v4.4 and higher (see Page 2), else it is restricted to ± 50 MHz. The uncertainty is the RSS of the ConvF uncertainty at calibration frequency and the uncertainty for the indicated frequency band.

F At frequencies below 3 GHz, the validity of tissue parameters (£ and a) can be relaxed to ± 10% if liquid compensation formula is applied to measured SAR values. At frequencies above 3 GHz, the validity of tissue parameters (£ and a) is restricted to ± 5%. The uncertainty is the RSS of the ConvF uncertainty for indicated target tissue parameters.

Certificate No : EX3-3746_Nov11 Page 6 of 11EX3DV4- SN:3746 November 17,2011 

# Frequency Response of E-Field (TEM-Cell:ifi110 EXX, Waveguide: R22)**1.5~-- --** **----~------~------------------------------ --** **-------,**

1.4

_____ 1.3

"0 Q)**.~** ro 1.2
E 

# o S 1.1

Q) 

## &. 1.0

(j)

### ~ 0 9 

## G .

c

### ~ 0.8 ()**~** I..L 0

. 7

0.6 . 

### .
, ,

----- . . - ------------.--'-

----------------**-----:-------- -.-.-.----** **---;------------------ ----~--** **----- ---** **.--.-----.-~--.---- -----** 

.. . .... ..... ..... . . . ...**~** .......... . . . ..........**~** . ........ ....... ...... :

...... .. ..... . .........**~** .... . .... . . .. ..... . ......... .. .. ... ........... :

..... .

, 

### .

.... .......... ... .. . ...**~** .. .. . ... . . . ......... . . . .............. . ... . ... !

.... . . ... .......... . .. .**~** ......... . .. ... . . .....**~** . ... .. . ................**~** ..... .



# .**~** 

# :

.**~** 

# .**------ -----** **------: ----** **-- ---------------:·_----** **-- ---** **-- --** **------.; ----------.-.--------** **~--------- --** **------- _--!-----------------------"!-----** 

- • : : : : : 

## i

. : 

# . : :

.... .... .. . ....... "

...**~** .... . ...... . ...... .. . . . ..... ..... . . .. . .. ··· ·: i ··
. ..... .... "

.. .. . . ... ... .. '."...

. . .. , .... . ... . . ... ... 

# ··i·

.. ···

······
.. · ·
.. ····
.. ·· · ··,·········
.. · ·· · · ··
.. ···c

.. · ·
.. -···· ·
.. ... . ........ . .................

### ~ .... . ..... ..... ... ....

### ~- ....... .... ...... ... .. ;

..... .

, • I • • ,

- - -- - - - . - - - - - - - - - - - -- - -!-

- - - - - - - - - -- - - - - -- - - - -- -!-

- - - - - - - . - . - - . - - - - - - - -**~** -. -- - - - - - - - - - - - - - - - - - - -!-

- - - -- •

- - - - - • - •

- - - - - - - - -: - - - - • - - - - - - - - - - - -- - • - - -**~. ~**  - -**~** •

, , , : : :
: : :

. , , 

### : 

### : :

- . - - . -
.**~** - - -. - - . - - - - - . -**-~** - - -- - - -- -**~** - - - - - - -**~** - - --**-~** - - - - - -- - _. - - - - - - -- - - -
. ! -. - - - - - - - - - - - -- . -- - - - - -**~** - - - -- - - - - - - - - - - - - - - - - -**~** --
. - - - - . - - - - - . - - -- . - - - -**~** -**~** - -. ­

, " ,.

• • • I I :

. :

. , , ,

, ,

---- -- - - - - --- -- -- ----**--~** - .. --- --. -- --------- ---:------- - -- - -- -- -- -- - - --:-- .- . -- -- . . ---- -- - --**---~--** -- - -

-- ---- -- .---- --**-~.** --- - -- - - ------- -- --**---~-** -- --­**0.5+- ~~~** **~~ -r~~** **~~ ~~** **~~~~ ~-r~** **~~ ~~~**  

## o 500 1000 1500 2000 2500 3000 

### f [MHz] 

### I i j TEM R22

Uncertainty of Frequency Response of E-field: ± 6.3% (k=2)

### ~

Certificate No: EX3-3746_Nov11 Page 7 of 11EX3DV4- SN:3746 November 17, 2011



# Receiving Pattern (<1», 

# 3 

# = 0°

f=600 MHz,TEM f=1800 MHz,R22

90**-~**

1 35 4 5 1 3 5

, , 45

e- - I " '0



### i



## . '**~, ~** 



# . _ 

# . • I

. - -'-- 1

,

180 

# . ..

00 . ', 

# 1.



# . . I 1 0 "'

, 1 "

,**~ ')**  04 p-B 0 8 1 8 

## 9**~** 0 2 04 08 

### ,I**~6** • I

· '


## . 

# . "


# . ..

_ 

# . 

# , . . . - 

# . • r**~** I I ,;

t

..



# . 

# . . ,


# . -#­**~** 225 '3 15 225 ' 315

270

., - 270

­ 

# • TI



### - _1 

# •**.I** jil

Tot X Y Z Tot X Y Z



### o

. 5



### co :£ 

# e 00 

### W

-05 , "

- - - -- - - - -- -- --- -- - - - - - - - -- - -- ---**~** - - - - - - - - -- - --- - -- - - - - -- - -- - - --- - - - - - - - -- -**~** - - -- --- - - - - -- - - --- - -. :.

-- - - . - -- - - - - -- - - - - - -**~** -- - - - --- ----- - - -- - - -**~** - - ---- - - -- - ­

: : : I :

_ . - ..",,­ ' v::.:"\):i

-**¥i-~ ..$$~~** **: ""** ***'r"~-~~"""'~ -fl** **="~~** ' ""'-**~' tJ:.::** **'\.- "O-'~** 

- - -

- - - - - - --- - - - - - - - -

- - - - - - -

- - - - -**~** ---- - - - - - -- - - - - --- - - -**~** - - -- - - - - - - - -

- - -- - - - - ;

-- - - - - - - - - - - ---- - - - - .:.

- - - - - - - - - - - - - - - - - -- -**~** - - - - - - - - - -- - - - - - - - - -**~** - - - - - - - - - - -­
, ,.
, .

-150 -100 -50 50 100 150
[!J [!]**~** 

### W 100 M Hz 600 tv1Hz 1800 MHz 2500 t-/IHz

Uncertainty of Axial Isotropy Assessment: 

### ± 0,5% (k=2)

Certificate No: EX3-3746_Nov11 Page 8 of 11EX3DV4- SN : 3746 November 17,2011



# Dynamic Range f(SAR head ) 

### (TEM cell, 

### f = 900 MHz)

10- 2 10- 1 10° 10 1 10 2 SAR [mW/cm3]**~ ~**  

### [!] X compensated X not compensated Y compensated 

# rn 

# • rn
Y not compensated Z compensated Z not com pensated**2'---"~~7 ~~,,------"~~~~--~--~""'~'~--"~--""~~----~O-~~** **.**

- "

. ..::



### ... . ... 1-

... . . .. . ........**, .'.'.** **' .**  . . "

.. , ..**,.~** ..**,.,~** .. '1":--[**.~**

.... 

## g

LU - 0­

. . ...

r­ -1

- " " "

. : ;: :;; , , . , : . . ,. . .. " , . . , . , .. . . 

# .•**~lK '** **~~ ~~** **, :4:** **:!** 

# ..**:~** .1':

## :~**.' ;(** **:i. :~,** **; :!** **¥**



# . .. . .... · · ·1' + :

. : ,. .. . ,**_2,r---~~~ ~~** **~·+_~~~·~:~ ~:** **· :·** **r-~~~ ·** **· :·** **~~~~ ::** **~:--~~~.~.~** I I I I 10- 3 10- 2 10- 1 10 0 10 1 10 2 SAR [mW/crn3] 

# w**~** 

# w
X compensated X not compensated Y com pensated [ • I

. j 

# - . [ Y not compensated Z compensated Z not compensated

Uncertainty of Linearity Assessment: 

### ± 0.6% (k=2)

Certificate No: EX3-3746_Nov11 Page 9 of 11EX3DV4- SN : 3746 November 17, 2011


# Conversion Factor Assessment

f = 2450 MHz

, WGLS R22 (H_convF)

40

35

30 •



# .

.

15

10

5**~** 

## . 0 I

0 10 1 S 20 2 5 30 35 40 0 5 1 0 1 5 io 2 5 30 35 40

z [m m] z [mIni**~ ~I** **~** _1

a rlaiyttc -a i mea s u r ed n a lyt lCa i rre as ll e d 

# Deviation from Isotropy in Liquid

Error 

### (cp, S), f = 

### 900 MHz

c o**~ ~**  

### o f 

## = 2450 MHz

, WGLS R22 (M_convF)

40

3 5

30 

# ,



# .

. 25 f 

### ­**~** , ; 

# .

0> 

# . ! 20

er: « (j) i 5

! ! , ! ! ! ! ! ! 0

, I I

1

. 0
0.8
0.6
0.4
0

. 2

0

. 0**k~~~ ~~~!!!!** **~~** -0.2



### o

-1.0 -0

. 8 -0.6 -0.4 -0

. 2 0.0 0

. 2 0.4 0

. 6 0

. 8 1

. 0 Uncertainty of Spherical Isotropy Assessment: 

### ± 2.6% (k=2)

Certificate No : EX3-3746_Nov11 Page 10 of 11EX3DV4- SN:3746 November 17

, 2011



# DASY/EASY

- Parameters of Probe: EX3DV4

- SN:3746

Other Probe Parameters

Sensor Arrangement Triangular

Connector Angle ( 0 ) Not applicable

Mechanical Surface Detection Mode enabled
Optical Surface Detection Mode disabled

Probe Overall Length 337 mm

Probe Body Diameter 10 mm
Tip Length 9mm
Tip Diameter 2.5mm

Probe Tip to Sensor X Calibration Point 1 mm

Probe Tip to Sensor Y Calibration Point 1 mm

Probe Tip to Sensor Z Calibration Point 1 mm

Recommended Measurement Distance from Surface 2mm

Certificate No : EX3-3746_Nov11 Page 11 of 11REV 2012.08.02 

# DIPOLE CALIBRATION

DIPOLE CALIBRATION
Please see attached calibration data

# I 

# I\lJL

Calibration Laboratory of Schweizerischer Kalibrierdienst 

# s Schmid 

### & Partner 

### C Service suisse d'etalonnage
Servizio svizzero di taratura 

### Engineering AG Zeughausstrasse 43,8004 Zurich, Switzerland 

# s Swiss Calibration Service

Accredited by the Swiss Accreditation Service (SAS) Accreditation No.: 

### SCS 108

The Swiss Accreditation Service is one of the signatories to the EA
Multilateral Agreement for the recognition of calibration certificates

Client 

### Northwest EMC Certificate No: 

### D2450V2-855_Dec11



# ' •

. CALIBRATION CERTIFICATE

Object 

### D2450V2

- SN: 855 

## I

Calibration procedure(s) 

### QA CAL-OS.v8 � Calibration procedure for dipole validation kits above 700 MHz �


# I

Calibration date: 

### December 09, 2011**----------~~--------------------------------~**

This calibration certificate documents the traceability to national standards

, which realize the physical units of measurements (SI). �
The measurements and the uncertainties with confiden c e probability are given on the following pages and are part of the certificate. �

All calibrations have been conducted in the closed laboratory facility: environment temperature (22 ± 3) O C and humidity < 70 %

. �

Calibration Equipment used (M& TE critical for calibration) �

Primary Standards

Power meter EPM·442A
Power sensor HP 8481A
Reference 20 dB Attenuator Type-N mismatch combination
Reference Probe ES3DV3

DAE4

Secondary Standards

Power sensor HP 8481A
RF generator R&S SMT-06
Network Analyzer HP 8753E

Calibrated by:

Approved by : ID # Cal Date (Certificate No.) Scheduled Calibration
GB37480704 05·0ct·ll (No. 217·01451) Oct-12
US37292783 05-0ct-ll (No. 217·01451) Oct-12
SN : 5086 (20g) 29-Ma r -l1 (No. 217-01368) Apr-12
SN: 5047.2/06327 29-Mar-ll (No

. 217-01371) Apr-12
SN : 3205 29-Apr-ll (No. ES3-3205_Aprll) Apr-12
SN: 601 04

- Jul-l1 (No

. DAE4-601_Jull1) JUI-12

ID # Check Date (in house) Scheduled Check

MY41092317 18-0ct-02 (in house check Oct-ll ) In house check: Oct-13

100005 04-Aug-99 (in house check Oct-l1) In house check : Oct-13
US37390585 S4206 18-0ct-Ol (in house check Oct-ll) In house check : Oct-12

Name Function Signature

Dimce lliev Laboratory Technician

Katja Pokovic Technical Manager

Issued: December 9, 2011

This calibration certificate shall not be reproduced except in full without written approval of the laboratory.

Certificate No: D2450V2-855_Dec11 Page 1 of 8

### Calibration Laboratory of 

### S Schweizerischer Kalibrierdienst 

### Schmid 

### & Partner Service suisse d'etalonnage 

### C Servizio svizzero di taratura 

### Engineering AG Zeughausstrasse 43, 8004 Zurich, Switzerland 

### S Swiss Calibration Service

Accredited by the Swiss Accreditation Service (SAS) Accreditation No

. : 

### SCS 108

The Swiss Accreditation Service is one of the signatories to the EA
Multilateral Agreement for the recognition of calibration certificates 

### Glossary: TSL tissue simulating liquid ConvF sensitivity in TSL 

## 1 NORM x,y,z NIA 

### not applicable or not measured



### Calibration is Performed According to the Following Standards: a) IEEE Std 1528-2003, "IEEE Recommended Practice for Determining the Peak SpatialAveraged Specific Absorption Rate (SAR) in the Human Head from Wireless Communications Devices: Measurement Techniques", December 2003 b) IEC 62209-1, "Procedure to measure the Specific Absorption Rate (SAR) for hand-held devices used in close proximity to the ear (frequency range of 300 MHz to 3 GHz)", February 2005 c) Federal Communications Commission Office of Engineering & 

### Technology (FCC OET), "Evaluating Compliance with FCC Guidelines for Human Exposure to Radiofrequency Electromagnetic Fields; Additional Information for Evaluating Compliance of Mobile and Portable Devices with FCC Limits for Human Exposure to Radiofrequency Emissions", Supplement C (Edition 01-01) to Bulletin 65



### Additional Documentation: d) DASY 4/5 

### System Handbook



### Methods Applied and Interpretation of Parameters:

• Measurement Conditions: 

### Further details are available from the Validation Report at the end of the certificate. All figures stated in the certificate are valid at the frequency indicated.

• Antenna Parameters with TSL: 

### The dipole is mounted with the spacer to position its feed pOint exactly below the center marking of the flat phantom section, with the arms oriented parallel to the body axis.

• Feed Point Impedance and Return Loss: 

### These parameters are measured with the dipole positioned under the liquid filled phantom. The impedance stated is transformed from the measurement at the SMA connector to the feed point. The Return Loss ensures low reflected power. No uncertainty required.

• Electrical Delay: 

### One-way delay between the SMA connector and the antenna feed pOint. No uncertainty required.

• SAR measured: 

### SAR measured at the stated antenna input power.

• SAR normalized: 

### SAR as measured, normalized to an input power of 1 W at the antenna connector.

• SAR for nominal TSL parameters: 

### The measured TSL parameters are used to calculate the nominal SAR result.

Certificate No: D2450V2-855_Dec11 Page 2 of 8

### Measurement Conditions
t

. f DASY system con Iguratlon, as ar as no gIven on page

DASV Version DASY5 V52.8.0

Extrapolation

Phantom Advanced Extrapolation

Modular Flat Phantom

Distance Dipole Center

- TSL

Zoom Scan Resolution

Frequency 10 mm
dx, dy, dz 

## = 5 mm

2450 MHz ± 1 MHz with Spacer



### Head TSL parameters
Th f II' t d I I f e 0 oWIng parame ers an ca cu a Ions were app 

### r Ie 

### d Temperature Permittivity Conductivity

Nominal Head TSL parameters 22

. 0 ° C 39.2 1.80 mho/m

Measured Head TSL parameters (22

. 0 ± 0.2) °C 39.5 ± 6 % 1.87 mho/m ± 6 %

Head TSL temperature change during test < 0.5 ° C

---­ --- ­



### SAR result with Head TSL

SAR averaged over 1 cm 3 (1 g) of Head TSL Condition

SAR measured 250 mW input power 13.7mW/g

SAR for nominal Head TSL parameters normalized to 1 W 53.9 mW /g ± 17.0 % (k=2)

SAR averaged over 10 cm 3 (10 g) of Head TSL condition

SAR measured 250 mW input power 6.38 mW I g

SAR for nominal Head TSL parameters normalized to 1 W 25.3 mW /g ± 16.5 % (k=2)



### Body TSL parameters
Th f II d I I

. e 0 oWIng parameters an ca cu atlons were app 

### r Ie 

### d Temperature Permittivity Conductivity

Nominal Body TSL parameters 22.0°C 52.7 1.95 mho/m

Measured Body TSL parameters (22

. 0 ± 0

. 2) ° C 50

. 7±6% 2

. 04 mho/m ± 6 %
Body TSL temperature change during test < 0.5 ° C



### ---­ ---­



### SAR result with Body TSL

SAR averaged over 1 cm 3 (1 g) of Body TSL Condition

SAR measured 250 mW input power 13.0 mW I g

SAR for nominal Body TSL parameters normalized to 1 W 50.4 mW / g ± 17.0 % (k=2)

SAR averaged over 10 cm 3 (10 g) of Body TSL condition

SAR measured 250 mW input power 6.02 mW I g
SAR for nominal Body TSL parameters normalized to 1 W 23.7 mW / g ± 16.5 % (k=2)

Certificate No : D2450V2-855_Dec11 Page 3 of 8

### Appendix

Antenna Parameters with Head TSL

Impedance, transformed to feed point 52.9 

### n + 4.5 

### jn

Return Loss

- 25.7 dB



### Antenna Parameters with Body TSL

Impedance, transformed to feed pOint 50.4 

### n + 5

. 3 

### jn

Return Loss

- 25.5 dB



### General Antenna Parameters and Design

Electrical Delay (one direction) 1.157 ns

After long term use with 100W radiated power, only a slight warming of the dipole near the feedpoint can be measured.

The dipole is made of standard semirigid coaxial cable. The center conductor of the feeding line is directly connected to the second arm of the dipole. The antenna is therefore short-circuited for DC-signals

. On some of the dipoles, small end caps

are added to the dipole arms in order to improve matching when loaded according to the position as explained in the "Measurement Conditions" paragraph

. The SAR data are not affected by this change. The overall dipole length is still according to the Standard. No excessive force must be applied to the dipole arms, because they might bend or the soldered connections near the feedpoint may be damaged.



### Additional EUT Data

Manufactured by SPEAG

Manufactured on November 10, 2009

Certificate No: D2450V2-855_Dec11 Page 4 of 8

### DASY5 Validation Report for Head TSL

Date: 09

. 12.20 II



### Test Laboratory: SPEAG, Zurich, Switzerland



### DUT: Dipole 2450 MHz; Type: D2450V2; Serial: D2450V2

- SN: 855

Communication System: CW; Frequency: 2450 MHz Medium parameters used: f = 2450 MHz; cr = 1.87 mho/m; tr 

### = 39.5; P = 1000 kg/m 3



### Phantom section: Flat Section
Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007)

DASY52 Configuration :

• Probe: ES3DV3

- SN3205; ConvF(4.45, 4.45, 4.45); Calibrated: 29

. 04 .2 011

• Sensor-Surface: 3mm (Mechanical Surface Detection)

• Electronics: DAE4 Sn60 I; 

### Calibrated: 04.07.20 II



### • Phantom: Flat Phantom 5.0 (front); Type: QDOOOP50AA; Serial : 1001

• DASY52 52.8.0(692 ); SEMCAD X 14.6.4(4989)



## Dipole Calibration for Head TissueIPin=250 mW, d=10mmlZoom Scan (7x7x7)/Cube 0:
Measurement grid: dx=5mm, dy=5mm, dz=5mm Reference Value = 100.7 Vim; Power Drift = 0.07 dB Peak SAR (extrapolated) = 28.3310 

### SAR(1 g) 

# = 13.7 mW/g; SAR(lO g) 

# = 6.38 mW/g Maximum value of SAR (measured) = 17

. 684 mW/g

dB o

- 4.80

- 9

. 60

- 14

. 40

-19.20

- 24.00 

# o dB = 17.680mW/g = 24.95 dB mW/g

Certificate No : D2450V2-855_Dec11 Page 5 of 8

### -- Impedance Measurement Plot for Head TSL

'3 Dec 21311 13: :32: 30 [ill]**~311** 1 U Fe- 1: 52.912**~'l** 4

.. 5527**~). 2'~5** **.75** pH

. J 2**450.0~H)** 000 t'lHz



# , t

De 1

I
Cor-'

H V8 16

1 1 

# r.

-

t 

### I 

### -"I

CH2 LOG ;) dB! REF -20 dB 1:-25.595**d~** 2 4 50. 000 130 13 MHz



# -

Cor

Mil,:! 16

- 00

- 0 0**~ r~** 



### --sT ;RT2'25"0. 00'000 0 r1

- - s**IP .,**  Eo 5

- . H z ---'----I-----I-----'-**----** - .L. O

Certificate No: D2450V2-855_Dec11 Page 6 of 8

### DASY5 Validation Report for Body TSL

Date : 08.12.20 II



### Test Laboratory: SPEAG, Zurich, Switzerland



### DUT: Dipole 2450 MHz; Type: D2450V2; Serial: D2450V2

- SN: 855

Communication System: CW; Frequency: 2450 

### MHz Medium parameters used : f 

### = 2450 

### MHz; (J 

### = 2.04 mho/m ; Cr 

### = 50.7; 

### P = 1000 kg/m 3



### Phantom section: Flat Section
Measurement Standard: DASYS (IEEEIIEC/ANSI C63

. 19-2007)

DASYS2 Configuration:

• Probe: ES3DV3

- SN320S; ConvF(4.26, 4.26, 4.26); Calibrated: 29.04.2011

• Sensor-Surface: 3mm (Mechanical Surface Detection)

• Electronics: DAE4 Sn601; Cahbrated : 04.07.2011

• Phantom: Flat Phantom 5

. 0 

### (back); Type: QDOOOPSOAA; Serial: 1002

• DASYS2 52.8.0(692); 

### SEMCAD X 14

. 6.4(4989)



## Dipole Calibration for Body TissuelPin=250 mW, d=10mm/Zoom Scan (7x7x7)/Cube 0:
Measurement grid: dx=Smm, dy=Smm, dz=Smm Reference Value 

# = 95.074 

### Vim; Power Drift 

# = -0

. 0092 dB Peak SAR (extrapolated) 

# = 27

. 0840 

### SAR(l g) 

# = 13 mW/g; SAR(lO g) 

# = 6.02 mW/g Maximum value of SAR (measured) = 17

. 188 mW/g

dB 

## o

-5.00

-10.00

-15.00

-20.00

-25.00


# o dB 

# = 17. 190mW/g 

# = 24.71 dB mW/g

Certificate No : D2450V2-855

_ Dec11 Page 7 of 8

### Impedance Measurement Plot for Body TSL

8 De c 2011 11:1219:43

i U F:3 2450.220 01210**~1Hz** 

# (**--.-,~~**

'

. I



# ,

HId

CH2

Avg 15

HId ':-11**.** LDG 5 dB " REF

-**~'0** dB

- ,:=c r­**~** 2 '- 25 455 d B '> 4 502'> 0000 MH z



# _

..... 

# " T

- -

-



## - 

# --­**~**



# -- 

# --­

-----­ 

## I**~** 

## -

----- "'-... 

### -------r-/'i 

## I 

# -**---"-~~',- l**  -­ f--­**I--** .... 

### '3 -.­ /­
2 

## I



# - 

# L 

# ' '''**~\ ;7** **~'--** 

# - -­ 

# - 

# -i L

'-­ --­ 

# i



# - ST A RT 2 250

. 13013 01313 NHz S T OP 2 80 0 .121130 0 00 MHz CHi 11at- -k er

- s

:3 : 55

. bE.1) "

27

. 3 44**~':!** 2

. b0000 13Hz

C H2 Nad :**er ~** 
3:- 11.51 5 dB 2 .5000 121 13Hz

Certificate No : D2450V2-855

_ Dec11 Page 8 of 8

### Calibration Laboratory of Schmid & Partner Engineering AG Zeughausstrasse 43, 8004 Zurich, Switzerland

Accredited by the Swiss Accreditation Service (SAS)

The Swiss Accreditation Service is one of the signatories to the EA

Multilateral Agreement for the recognition of calibration certificates

Client 

### Northwest EMC 

# s Schweizerischer Kalibrierdienst 

### C Service suisse d'etalonnage
Servizio svizzero di taratura 

# s Swiss Calibration Service

Accreditation No

. : 

### SCS 108

Certificate No: 

### D5GHzV2-1066_Dec11



# CALIBRATION CERTIFICATE

Object 

### D5GHzV2

- SN: 1066

Calibration procedure(s) 

### QA CAL-22_v1 Calibration procedure for dipole validation kits between 3-6 GHz


# I

Calibration date : 

### December 14, 2011 

## I

This calibration certificate documents the traceability to national standards. which realize the physi c al units of measurements (SI).
The measurements and the uncertainties with confidence probability are given on the following pages and are part of the certificate.

All calibrations have been conducted in the closed laboratory facility : environment temperature (22 ± 3) O C and humidity < 70 %.

Calibration Equipment used (M&TE critical for calibration)

Primary Standards

Power meter EPM-442A
Power s ensor HP 8481A
Reference 20 dB Attenuator Type-N mismatch combination
Reference Probe EX3DV4

DAE4

Secondary Standards

Power sensor HP 8481A

RF generator R&S SMT-06

Network Analyzer HP 8753E

Calibrated by:

Approved by: ID # Cal Date (Certificate No.) Scheduled Calibration

GB37480704 05-0ct-11 (No. 217-01451) Oct-12
US37292783 05-0ct-11 (No. 217-01451) Oct-12
SN : 5086 (20g) 29-Mar-11 (No. 217-01368) Apr-12
SN : 5047.2/06327 29-Mar-11 (No

. 217-01371) Apr-12
SN: 3503 04-Mar-11 (No

. EX3-3503_Mar11) Mar-12
SN: 601 04-Jul-11 (No

. DAE4-601_JuI11) Jul-12

ID # Check Date (in house) Scheduled Check

MY41092317 18-0ct-02 (in house check Oct-11) In house check : Oct-13

100005 04-Aug-99 (in house check Oct-11) In house check : Oct-13
US37390585 S4206 18-0ct-01 (in house check Oct-11) In house check: Oct-12

Name Function Signature

Jeton Kastrati Laboratory Technician 

# f**~ L--**  

# ________ _

Katja Pokovic Technical Manager

# V~~~

Issued : December 14, 2011

This calibration certificate shall not be reproduced except in full without written approval of the laboratory.

Certificate No: D5GHzV2-1066_Dec11 Page 1 of 12

### Calibration Laboratory of 

## S Schweizerischer Kalibrierdienst 

### Schmid 

### & Partner Service suisse d'etalonnage 

### C Servizio svizzero di taratura 

### Engineering AG Zeughausstrasse 43, 8004 Zurich, Switzerland 

## S Swiss Calibration Service

Accredited by the Swiss Accreditation Service (SAS) Accreditation No.: 

### SCS 108

The Swiss Accreditation Service is one of the signatories to the EA
Multilateral Agreement for the recognition of calibration certificates 

### Glossary: TSL tissue simulating liquid ConvF sensitivity in TSL 

### 1 NORM x , y,z N/A not applicable or not measured



### Calibration is Performed According to the Following Standards: a) IEEE Std 1528-2003, "IEEE Recommended Practice for Determining the Peak SpatialAveraged Specific Absorption Rate (SAR) in the Human Head from Wireless Communications Devices: Measurement Techniques", December 2003 b) IEC 62209-1, "Procedure to measure the Specific Absorption Rate (SAR) for hand-held devices used in close proximity to the ear (frequency range of 300 MHz to 3 GHz)", February 2005 c) Federal Communications Commission Office of Engineering & 

### Technology (FCC OET),

" Evaluating Compliance with FCC Guidelines for Human Exposure to Radiofrequency Electromagnetic Fields; Additional Information for Evaluating Compliance of Mobile and Portable Devices with FCC Limits for Human Exposure to Radiofrequency Emissions", Supplement C (Edition 01-01) to Bulletin 65



### Additional Documentation: d) DASY 4/5 System Handbook



### Methods Applied and Interpretation of Parameters:

• Measurement Conditions: 

### Further details are available from the Validation Report at the end of the certificate

. All figures stated in the certificate are valid at the frequency indicated

.

• Antenna Parameters with TSL: 

### The dipole is mounted with the spacer to position its feed point exactly below the center marking of the flat phantom section, with the arms oriented parallel to the body axis.

• Feed Point Impedance and Return Loss: 

### These parameters are measured with the dipole positioned under the liquid filled phantom. The impedance stated is transformed from the measurement at the SMA connector to the feed point. The Return Loss ensures low reflected power. No uncertainty required.

• Electrical Delay: 

### One-way delay between the SMA connector and the antenna feed point. No uncertainty required.

• SAR measured: 

### SAR measured at the stated antenna input power.

• SAR normalized: 

### SAR as measured, normalized to an input power of 1 W at the antenna connector.

• SAR for nominal TSL parameters: 

### The measured TSL parameters are used to calculate the nominal SAR result.

Certificate No: D5GHzV2-1066

_ Dec11 Page 2 of 12

### Measurement Conditions
t

. Iven on page
DASY Version system con Iguratlon, as D A SY f ar as no**~** DASY5 V52

. 8

. 0

Advanced Extrapolation Extrapolation

Modular Flat Phantom V5

. 0 Phantom

10 mm with Spacer Distance Dipole Center

- TSL

Graded Ratio = 1.4 (Z direction)

5200 MHz ± 1 MHz Frequency Zoom Scan Resolution dx, dy = 4.0 mm, dz = 1.4 mm

5500 MHz ± 1 MHz
5800 MHz ± 1 MHz



### Head TSL parameters at 5200 MHz
Th f· d I I

. I" d e ollowlng parameters an ca cu atlons were app Ie Temperature Permittivity Conductivity

Nominal Head TSL parameters 22.0 ° C 36

. 0 4.66 mho/m

Measured Head TSL parameters (22.0 ± 0

. 2) ° C 36

. 1 ± 6 % 4

. 65 mho/m ± 6 %

Head TSL temperature change during test < 0

. 5 ° C



### ---­ 

### ---­

SAR result with Head TSL at 5200 MHz

SAR averaged over 1 cm 3 (1 g) of Head TSL Condition

SAR measured 100 mW input power 8.13 mW I g

SAR for nominal Head TSL parameters normalized to 1 W 81.3 mW /g ± 17.0 % (k=2)

SAR averaged over 10 cm 3 (10 g) of Head TSL condition

SAR measured 100 mW input power 2

. 32 mW I g

SAR for nominal Head TSL parameters normalized to 1W 23.2 mW /g ± 16.5 % (k=2)



### Head TSL parameters at 5500 MHz
The following parameters and calculations were applied Temperature Permittivity Conductivity

Nominal Head TSL parameters 22.0 ° C 35.6 4.96 mho/m

Measured Head TSL parameters (22

. 0 ± 0

. 2) °C 35.6 ±6 % 4.96 mho/m ± 6 %

Head TSL temperature change during test < 0

. 5 ° C



### ---­ -

... 

### _­



### SAR result with Head TSL at 5500 MHz

SAR averaged over 1 cm 3 (1 g) of Head TSL Condition

SAR measured 100 mW input power 8.53 mW I g

SAR for nominal Head TSL parameters normalized to 1 W 85.3 mW / g ± 17.0 % (k=2)

SAR averaged over 10 cm 3 (10 g) of Head TSL condition

SAR measured 100 mW input power 2.41 mW I g

SAR for nominal Head TSL parameters normalized to 1 W 24.1 mW / g ± 16.5 % (k=2)

Certificate No : D5GHzV2-1066_Dec11 Page 3 of 12

### Head TSL parameters at 5800 MHz
The following parameters and calculations were applied Temperature Permittivity Conductivity

Nominal Head TSL parameters 22.0°C 35

. 3 5

. 27 mho/m

Measured Head TSL parameters (22.0 ± 0

. 2) ° C 35.1±6% 5.27 mho/m ± 6 %

Head TSL temperature change during test < 0

. 5 ° C

---­ ---­



### SAR result with Head TSL at 5800 MHz

SAR averaged over 1 cm 3 (1 g) of Head TSL Condition

SAR measured 100 mW input power 7

. 86 mW I 9

SAR for nominal Head TSL parameters normalized to 1 W 78.5 mW I g ± 17.0 % (k=2)

SAR averaged over 10 cm 3 (10 g) of Head TSL condition

SAR measured 100 mW input power 2.22 mW I 9

SAR for nominal Head TSL parameters normalized to 1 W 22.2 mW I g ± 16.5 % (k=2)



### Body TSL parameters at 5200 MHz
Thfll · t dllt" I"d e 0**oWln~arame** ers an ca cu a Ions were app Ie Temperature Permittivity Conductivity

Nominal Body TSL parameters 22

. 0 ° C 49.0 5

. 30 mho/m

Measured Body TSL parameters (22.0 ± 0.2) ° C 49

. 6 ± 6 % 5.44 mho/m ± 6 %
Body TSL temperature change during test < 0.5 ° C



### ---­ ---­



### SAR result with Body TSL at 5200 MHz

SAR averaged over 1 cm 3 (1 g) of Body TSL Condition

SAR measured 100 mW input power 7.51mW/g

SAR for nominal Body TSL parameters normalized to 1 W 75.3 mW I g ± 18.1 % (k=2)

SAR averaged over 10 cm 3 (10 g) of Body TSL condition

SAR measured 100 mW input power 2.09 mW I 9

SAR for nominal Body TSL parameters normalized to 1 W 21.0 mW I g ± 17.6 % (k=2)

Certificate No: D5GHzV2-1066_Dec11 Page 4 of 12

### Body TSL parameters at 5500 MHz
The following parameters and calculations were applied Temperature Permittivity Conductivity

Nominal Body TSL parameters 22

. 0 ° C 48.6 5.65 mho/m

Measured Body TSL parameters (22.0 ± 0

. 2) ° C 49.0 ± 6 % 5.86 mho/m ± 6 %
Body TSL temperature change during test < 0.5 ° C

.. --­ ---­



### SAR result with Body TSL at 5500 MHz

SAR averaged over 1 cm 3 (1 g) of Body TSL Condition

SAR measured 100 mW input power 8

. 04 mW 1 9

SAR for nominal Body TSL parameters normalized to 1 W BO.7 mW / g ± 1B.1 % (k=2)

SAR averaged over 10 cm 3 (10 g) of Body TSL condition

SAR measured 100 mW input power 2.22 mW 

### 1 9

SAR for nominal Body TSL parameters normalized to 1 W 22.3 mW / g ± 17.6 % (k=2)



### Body TSL parameters at 5800 IVIHz
The following parameters and calculations were applied

. Temperature Permittivity Conductivity

Nominal Body TSL parameters 22.0 ° C 48.2 6

. 00 mho/m

Measured Body TSL parameters (22

. 0 ± 0.2) ° C 48.4 ± 6 % 6

. 28 mho/m ± 6 %
Body TSL temperature change during test < 0.5 ° C

---­ ---­



### SAR result with Body TSL at 5800 MHz

SAR averaged over 1 cm 3 (1 g) of Body TSL Condition

SAR measured 100 mW input power 7.54 mW 1 9

SAR for nominal Body TSL parameters normalized to 1 W 75.6 mW / g ± 1B.1 % (k=2)

SAR averaged over 10 cm 3 (10 g) of Body TSL condition

SAR measured 100 mW input power 2.07 mW 

### Ig

SAR for nominal Body TSL parameters normalized to 1 W 20.B mW / g ± 17.6 % (k=2)

Certificate No: D5GHzV2-1066_Dec11 Page 5 of 12

### Appendix

Antenna Parameters with Head TSL at 5200 MHz

Impedance, transformed to feed point 51.8 Q - 5

. 1 iQ

Return Loss

- 25.6 dB



### Antenna Parameters with Head TSL at 5500 MHz

Impedance, transformed to feed point 53.2 Q

- 2.3 iQ

Return Loss

- 28.4 dB



### Antenna Parameters with Head TSL at 5800 MHz

Impedance, transformed to feed point 55

. 5 Q - 1.0 iQ

Return Loss

- 25.5 dB



### Antenna Parameters with Body TSL at 5200 MHz

Impedance, transformed to feed point 51

. 2 Q - 4.7 iQ

Return Loss

- 26.4 dB



### Antenna Parameters with Body TSL at 5500 MHz

Impedance, transformed to feed point 53.5 Q

- 0

. 2 iQ

Return Loss

- 29.4 dB



### Antenna Parameters with Body TSL at 5800 MHz

Impedance, transformed to feed point 56.4 Q + 1 .6 iQ

Return Loss

- 24.1 dB



### General Antenna Parameters and Design

Electrical Delay (one direction) 1.197 ns

After long term use with 100W radiated power, only a slight warming of the dipole near the feedpoint can be measured. The dipole is made of standard semirigid coaxial cable. The center conductor of the feeding line is directly connected to the second arm of the dipole. The antenna is therefore short-circuited for DC-signals. On some of the dipoles, small end caps

are added to the dipole arms in order to improve matching when loaded according to the position as explained in the "Measurement Conditions" paragraph

. The SAR data are not affected by this change. The overall dipole length is still according to the Standard. No excessive force must be applied to the dipole arms, because they might bend or the soldered connections near the feedpoint may be damaged

.



### Additional EUT Data

Manufactured by SPEAG

Manufactured on November 27, 2006

Certificate No : D5GHzV2-1066_Dec11 Page 6 of 12

### DASY5 Validation Report for Head TSL

Date: 14.12.2011

Test Laboratory: SPEAG, Zurich, Switzerland



### DUT: Dipole 5GHz; Type: D5GHzV2; Serial: D5GHzV2

- SN: 1066

Communication System: CW; Frequency: 5200 MHz, Frequency: 5500 MHz, Frequency: 5800 MHz Medium parameters used: f = 5200 MHz; () = 4.65 mho/m; lOr 

### = 36.1; p = 1000 kg/m 3

, 

### Medium parameters used: f = 5500 MHz; () = 4.96 mho/m; lOr 

### = 35.6; p = 1000 kg/m 3

, 

### Medium parameters used: f = 5800 MHz; () = 5.27 mho/m; lOr 

### = 35.1; P = 1000 kg/m 3



### Phantom section: Flat Section
Measurement Standard: DASY5 

### (IEEE/IEC/ANSI C63.19-2007)

DASY52 Configuration:

• Probe: EX3DV4

- SN3503; ConvF(5.41, 5.41, 5.41), ConvF(4.91, 4.91, 4.91), ConvF(4.81, 4.81, 4.81); Calibrated: 04.03.2011

• Sensor-Surface: l.4mm (Mechanical Surface Detection)

• Electronics: DAE4 Sn601; Calibrated: 04.07.2011

• Phantom: Flat Phantom 5.0 (front); Type: QDOOOP50AA; Serial: 1001

• DASY52 52.8.0(692); SEMCAD X 14.6A(4989)



## Dipole Calibration for Head TissueIPin=100mW, dist=10mm, f=5200 MHz/Zoom Scan, dist=1.4mm (8x8x7)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz= lAmm 

### Reference Value = 64.855 

### Vim; Power Drift = 0.10 dB
Peak SAR (extrapolated) = 30.2380 

### SAR(1 g) = 

### 8.13 mW/g; 

### SAR(1O g) = 

### 2.32 mW/g 

### Maximum value of SAR (measured) = 18A18 

### m W Ig



## Dipole Calibration for Head TissueIPin=100mW, dist=10mm, f=5500 MHz/Zoom Scan, dist=l.4mm (8x8x7)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=IAmm Reference Value = 64.965 

### Vim; Power Drift = 0.04 dB Peak SAR (extrapolated) = 33.8680 

### SAR(1 g) = 

### 8.53 mW/g; 

### SAR(1O g) = 

### 2.41 mW/g 

### Maximum value of SAR (measured) = 19.692 mW/g



## Dipole Calibration for Head TissueIPin=100mW, dist=10mm, f=5800 MHz/Zoom Scan, dist=1.4mm (8x8x7)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=IAmm 

### Reference Value = 61.095 

### Vim; Power Drift = 0.03 dB Peak SAR (extrapolated) = 33.1420 

### SAR(1 g) 

# = 7.86 mW/g; 

### SAR(10 g) 

# = 2.22 mW/g 

### Maximum value of SAR (measured) = 18.642 mW/g

Certificate No: DSGHzV2-1066_Dec11 Page 7 of 12

### dB 

# o

-10.00

-20.00

-30.00

-40.00

-50.00


## o dB 

# = 18.640mW/g 

# = 25.41 

### dB mW/g

Certificate No : DSGHzV2

- 1066_Dec11 Page 8 of 12

### Impedance Measurement Plot for Head TSL

14 Dec 2011 09:59:02

1 U FS 3: 55.518 I; -'3'34

. 14 n "'>1**27.~,[12** pF 580[1.[100 00[1 NHz



# *

De 1 1: 51.:::01 Ii -5.0525**~** 5.20000 13Hz

2: 53.2l.31**~~**

-2.2 ' 3 88**{~** 5.5000(1 13Hz

AV9**1~,**

Hid 

### --lCH2 Sll LOG 5 dBI REF -20 dB 3:-25.493 dB 5 800 000 0010**~1Hz** I
I -

t-----t

-­ r-­

I .



# -

- Cor

AV8 15

Hid

- ST ART 5 000.13130 (HW t1Hz STOP 5 00(1.(1(1(1 000 t1Hz CH2**~1arkers**
1:-25.51:.2 dB 5.213000 6Hz

2:-28.355 dB 5.50000 6Hz

Certificate No: D5GHzV2·1066_Dec11 Page 9 of 12

### DASY5 Validation Report for Body TSL

Date: 13.12

. 2011

Test Laboratory: SPEAG, Zurich, Switzerland

DUT: Dipole SGHz; Type: DSGHzV2; Serial: DSGHzV2

- SN: 1066

Communication System: CW; Frequency: 5200 MHz, Frequency: 5500 MHz, Frequency: 5800 MHz Medium parameters used: f = 5200 MHz; () = 5.44 mho/m; lOr 

### = 49.6; 

### P = 1000 kg/m 3

, Medium parameters used: f = 5500 MHz; () = 5

. 86 mho/m; lOr 

### = 49; p = 1000 kg/m 3

, Medium parameters used: f = 5800 MHz; ()

= 6.28 mho/m ; lOr 

### = 48.4; p = 1000 kg/m 3



### Phantom section: Flat Section
Measurement Standard: DASY5 (IEEE/IEC/ANSI C63.19-2007)

DASY52 Configuration :

• Probe: EX3DV4

- SN3503; ConvF(4.91, 4.91, 4

. 91), ConvF(4.43, 4.43

, 4.43), ConvF(4.38, 4.38,
4

. 38); Calibrated: 04.03.2011

• Sensor-Surface: l.4mm (Mechanical Surface Detection)

• Electronics: DAE4 Sn60 I; 

### Calibrated: 04

. 07.20 II



### • Phantom: Flat Phantom 5.0 (back); Type: QDOOOP50AA; Serial: 1002

• DASY52 52.8.0(692); SEMCAD X 14.6.4(4989)



## Dipole Calibration for Body TissueIPin=100mW, dist=10mm, f=5200 MHz/Zoom Scan, dist=l.4mm (8x8x7)/Cube 0: Measurement grid: dx=4mm, dy=4mm

, dz= l.4mm Reference Value = 58.272 Vim; Power Drift = -0.0057 dB Peak SAR (extrapolated) = 29.4900 SAR(1 g) 

# = 7.51 mW/g; SAR(1O g) 

# = 2.09 mW/g Maximum value of SAR (measured) = 17.296 mW/g



## Dipole Calibration for Body TissueIPin=100mW, dist=10mm, f=5500 MHz/Zoom Scan, dist=1.4mm (8x8x7)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz=1.4mm Reference Value = 58

. 543 Vim; Power Drift =
- 0.01 dB Peak SAR (extrapolated) = 34.4970 SAR(1 g) 

# = 8.04 mW/g; SAR(1O g) 

# = 2.22 mW/g Maximum value of SAR (measured) = 19

. 193 mW/g



## Dipole Calibration for Body TissueIPin=100mW, dist=10mm, f=5800 MHz/Zoom Scan, dist=1.4mm (8x8x7)/Cube 0: Measurement grid: dx=4mm, dy=4mm, dz= 1.4mm Reference Value = 54.820 Vim; Power Drift = -0.0098 dB Peak SAR (extrapolated) = 35

. 3730 SAR(1 g) 

# = 7.54 mW/g; SAR(10 g) 

# = 2.07 mW/g Maximum value of SAR (measured) = 18.371 mW/g

Certificate No: D5GHzV2-1066_Dec11 Page 10 of 12dB 

## o

-10.00

-20.00

-30.00

-40.00

-50.00 

# o dB = 

### 18.370mW/g = 

### 25

. 28 

### dB mW/g

Certificate No: D5GHzV2-1066_Dec11 Page 11 of 12

### Impedance Measurement Plot for Body TSL

13 Dec 2011 10:52:37

1 U FS 1: 51.it::3 " -4.7188 \, t

.. 48E

, : 3 pF 5 200.(1013 1300 r'lHz

\.­ --. .; 

# * /
De I I 

## I' /

# ~I-- -,---I 

# ***~** (

'

. ,;.

# ~

MV g j I 

## L .,-,/ 15 /. I

, .. / 

## ",**I--:>~**



# <.. HId 

### -1---'"'

CH2

Cor

A\i8 16

HId

::::TuP 5**.eeu** \:lell NH z Cl1**-.J .**  L O G 5 dB I REF -20 dB 1'-"6 371 dB 5 ?00 130 0 0 00 MH z • L • 

# - - -

-



# -**.** 

## "\ So

. 

# \

-

.'t 

### 'v 

## r- /



# _ \ /r /'-..-..



# -

! \



# --- S TAF.:T 5 131313.000 13th) r'lHz 

# \/v ._ .. _.

;--**~**

/"'.. 

# I**~** 

# V= -O-- !**~** "V**~** '" ,;,

- 

### - - - 

### - CH 1 r·l

. ,w· k e r";

2: 53.516**~** -ltA.06**1lI~ ,**  5.50130£1 13Hz

3 : 5':

. .428 " ' 1.5425**(~** 5.80(11313 13Hz

CH2**~lark** t?t " ;

2:-29.363 dB 5.5001313 GHz

3:-24.1135 dB
5.81313100 GHz

Certificate No: D5GHzV2-1066_Dec11 Page 12 of 12