DKK-TOA Environnement SA Horiba SIR SA TeledyneMonitor Labs

Model GFC-311E1 Serinus 30 1 & Opsis AB OPS30 1 Model EC9830 CO12M 1 Model APMA-370 1 Model SIR S-50061 Model TML 30 1 Model 300E1 Model 300EU 1 Model 300EM Model 48i 1 Model 48i-TLE1 Range(s) 0-5,

0-10, 0-20, 0-50, 0-100 ppm 0-200 ppm 0-200 ppm Ranges: 0-10/ 25 / 50 / 100 / 200 ppm or user selectable ranges Standard:

0-10/20/50/100 ppm; 0-5/10/20/50 ppm
Optional (measurable): 4 ranges selectable from 0-100 ppm, within 10 times range ratio 0-1, 50, 200 ppm 0-1 ppm

0-1000 ppm full sale, user selectable 0-1 ppm

0-1000 ppm

user selectable 0-100 ppb

0-100 ppm

user selectable 0-5 ppm

0-5000 ppm

user selectable 0-1, 2, 5, 10, 20,

50, 100, 200, 500, 1000, 2000, 5000 and 10000 ppm
Custom Ranges: 0-1 to 10000 ppm 0-1, 5, 10, 20,

50, 100, 200, 500, 1000ppm
Custom Ranges: 0-1 to 1000ppm

Noise Footnote 3 < 20 ppb with Kalman filter active 25 ppb with Kalman filter active Footnote 3 Footnote 3 0.02 ppm Zero Noise: <0.02 ppm Zero Noise:

< 0.02 ppm Zero Noise:

< 0.01 ppm RMS Zero Noise:

< 0.1 ppm Zero Noise:

0.02ppm RMS (60 second averaging time) Zero Noise:

0.02ppm RMS (30 second averaging time)

Lower detectable limit 0.4 ppm < 40 ppb with Kalman filter active 50 ppb with Kalman filter active 50 ppb Footnote 3 0.04 ppm 0.04 ppm 0.04 ppm 0.02 ppm 0.2 ppm 0.04 ppm 0.04ppm (30 second averaging time) Interferents Footnote 2 Footnote 2

- Footnote 2 Footnote 2 Footnote 2 Footnote 2 Footnote 2 Footnote 2

- Footnote 2 Footnote 2 Zero drift Within 0.1ppm/24

hours Footnote 3

- < 0.1 ppm / 24 h; < 0.1 ppm / 7 days Footnote 3 0.01 ppm/24 h <0.1 ppm/ 24 h
<0.2ppm/ 7 days < 0.1 ppm/

24 h
0.2 ppm/ 7 days < 10 ppb/

24 h
20 ppb/ 7 days < 0.5 ppm/

24 h
< 1 ppm/ 7 days <0.1ppm/

24 hour < 0.1 ppm/

24 h

Span drift Within ±2%FS/24 hours Footnote 3

- < 1% / 24 h: < 1% / 7 days Footnote 3 0.5% F.S. <0.5% of reading/ 24 h
1% of reading/ 7 days < 0.5% of

reading/ 24 h

1% of reading/ 7 days < 0.5% of

reading/ 24 h

1% of reading/ 7 days < 0.5% of

reading/ 24 h

1% of reading/ 7 days ±1% full scale/

24 h ±1% full scale/

24 h

Lag time Footnote 3 Footnote 3

- Footnote 3 Footnote 3 15 sec 10 sec 10 sec 10 sec 10 sec Footnote 3 Footnote 3 ECOTECH Performance Parameter 

# Modern CO Instrument Performance Data

Thermo 

## Office of Research and Development

September 2010 Non-Dispersive IR Photometry

Teledyne APIDKK-TOA Environnement SA Horiba SIR SA TeledyneMonitor Labs

Model GFC-311E1 Serinus 30 1 & Opsis AB OPS30 1 Model EC9830 CO12M 1 Model APMA-370 1 Model SIR S-50061 Model TML 30 1 Model 300E1 Model 300EU 1 Model 300EM Model 48i 1 Model 48i-TLE1 ECOTECH Performance Parameter 

# Modern CO Instrument Performance Data

Thermo 

## Office of Research and Development

September 2010 Non-Dispersive IR Photometry

Teledyne API

Rise time (response) Footnote 3 Footnote 3

- Automatic & programmable Min 30 sec Footnote 3 60 sec (95% FS) <60 seconds to

95% <60 seconds

to 95% <30 seconds

to 95% <60 seconds

to 95% 60 seconds

(30 second average time) 60 seconds

(30 second average time) Fall time Footnote 3 Footnote 3

- Footnote 3 Footnote 3 60 sec (95% FS) <60 seconds to

95% <60 seconds

to 95% <30 seconds

to 95% <60 seconds

to 95% Footnote 3 Footnote 3

Precision Footnote 3 Footnote 3

- Footnote 3 Footnote 3 ± 0.1ppm 0.5% of reading 0.5% of reading 0.5% of

reading 1% of reading ± 0.1ppm Footnote 3

Information Source (unless footnoted) DKK-TOA web site ECOTECH web site ECOTECH web site Environnement

SA web site Horiba web site SIR SA web site Teledyne web site Teledyne-api

web site Teledyne-api

web site Teledyne-api

web site Theremo web

site Theremo web

site

3Test informaiton included in FEM application 1 EPA-designated FEM 2 Interferences tested according to 40 CFR Part 53