Shopping Cart Rapid Order

Username: 

Password: 

New User? Forgot Password?

SCIENTIFIC SAFETY HEALTHCARE SCIENCE EDUCATION

Scientific Home Browse by: Product - Supplier - Application Help Order Catalogs MSDS Part of Thermo Fisher Scientific

 Search

C USTOM S EARCHES

Submit

Antibodies

Certificate Of Analysis

MSDS Search

Substructure

:

 Shopping Cart View Cart

No items in cart 

For security reasons you will 

be automatically logged out 

after 30 minutes of inactivity.

Back to Browse Results

 

Thermo Scientific* TVA1000B Toxic Vapor Analyzer 

19-171-032 

The only over-the-shoulder portable vapor analyzer that offers both PID (Photo Ionization Detection) and FID 

(Flame Ionization Detection) in a single, easy-to-use instrument 

Analyzer monitors and logs organic and inorganic vapors simultaneously; suitable for use in a broad range of 

applications including fugitive emissions monitoring, emergency response, hazardous waste site evaluation, 

industrial hygiene, underground storage tank investigations, and natural gas leak detection.

Click in the table below 

to view accessories.

Description

n Provides fast, accurate readings of organic and inorganic vapors in accordance with EPA Method 21 specifications for monitoring 

fugitive emissions 

n Portable and lightweight 

n Multiple response factors and curves 

n Multi-point calibration 

n On-board datalogging 

n 8-hour battery life

n Bright and easy-to-read displays with large keypad for simple operation with or without gloves 

n Menu-driven operations; datalogging activated by a thumb-switch on handheld probe 

n Standard probe shows measurement values with units displayed on %, ppm, or ppb and bar with concentration level 

n Data can be downloaded to Windows*-based program using supplied software

n Minimum Detectable Limit: 100ppb benzene (PID); 300ppb hexane (FID) (laboratory conditions) 

n Fuel: None required (PID); 99.995% hydrogen (FID) 

n Portable Operation Time: 8 hours (with reference operating conditions)

Specifications & Ordering Information:

Standard accessories include battery pack and charger; shoulder strap; audible alarms; PC interface cable; ThermoConnect* software; water trap probe and 

filter; hydrogen cylinder and refill adapter; instructions and response book.Enhanced probe with larger display and menu -driven access to analyzer functions 

available separately through your Fisher Safety Customer Service Representative. Calibration kits sold separately. 

Certifications: FM (Class 1, Div. 1, Groups A, B, C and D Hazardous Location, Temp. Class T4)

Compliance: CE

Specification for TVA 1000B Toxic Vapor Analyzer

Measurement 

Ranges

Dynamic Range (PID): 0 to 2,000ppm 

Dynamic Range (FID): 0 to 50,000ppm

Accuracy

FID=±25% of reading or ±2.5ppm, whichever is greater from 1.0 to 10000ppm; PID=±25% of reading or ±2.5ppm, 

whichever is greater from 0.5 to 500ppm

Repeatability ±1%

Flowrate 1000 cc (nominal)

Display Bar Graph and 4-digit LCD

Response Time

<3.5 seconds for 90% of final value, using 100ppm of isobutylene; <3.5 seconds for 90% of final value, using 10000ppm of 

methane

Alarms High, Low, and STEL (short term exposure limit)

Input Output Analog Output: 0 to 2VDC

Data Memory 900 to 18,000 points, mode specific

Temperature 

Range

0° to +50°C (32° to 122°F)

Dimensions 13.5 x 10.3 x 3.2 in. (343 x 262 x 81mm)

Weight Approx. 12 lb. (5.4kg) 

Description Mfr. No. Cat. No. Quantity Price

FID Only Monitor

Items Details Images

Fisher Scientific - Thermo Scientific* TVA1000B Toxic Vapor Analyzer

http://www.fishersci.com/ecomm/servlet/fsproductdetail?catalogId=29104&productId=6471343... Page 1 of 2
Without Probe or Case 81010

19-171-031

Each for $9,378.11

With Standard Probe and Case 81020

19-171-032

Each for $10,630.06

Calibration Kit for TVA-1000B FID Monitor CR009UY 19-171-035 Each for $1,123.68

FID/PID Monitor

Without Probe or Case 81030

19-171-033

Each for $12,801.83

With Standard Probe and Case 81040

19-171-034

Each for $14,012.97

Calibration Kit for TVA-1000B FID/PID Monitor CR012UH 19-171-036 Each for $1,405.95

Add Items to Cart

Note: Probe required to operate analyzer or optional Bluetooth connectivity device. 

 

Back to Browse Results

Trademarks | Terms and Conditions | Legal Notices | Privacy Policy | Thermo Fisher Scientific Inc. 

2011 Thermo Fisher Scientific Inc. - All rights reserved. 

Fisher Scientific - Thermo Scientific* TVA1000B Toxic Vapor Analyzer

http://www.fishersci.com/ecomm/servlet/fsproductdetail?catalogId=29104&productId=6471343... Page 2 of 2