(918) 307-8865 or (888) 461-8778 www.airhygiene.com Corporate Headquarters 5634 S. 122nd E. Ave. Ste. F Tulsa, Oklahoma 74146 Las Vegas Office 5925 E. Lake Mead Blvd. Las Vegas, Nevada 89156 Philadelphia Office 8900 State Road Philadelphia, Pennsylvania 19136 Houston Office 1920 Treble Drive Humble, Texas 77338 

# STACK EMISSIONS STUDY FROM THE CATERPILLAR G3520C UNITS 1-6 PREPARED FOR ENERGY DEVELOPMENTS COVEL GARDENS LANDFILL GAS POWER STATION REGULATED ENTITY #: RN104156062 CUSTOMER REFERENCE #: CN601527534 SAN ANTONIO, TEXAS January 4, 2007

## STACK EMISSIONS STUDY FROM THE CATERPILLAR G3520C UNITS 1 - 6 PREPARED FOR ENERGY DEVELOPMENTS COVEL GARDENS LANDFILL GAS POWER STATION REGULATED ENTITY #: RN104156062 CUSTOMER REFERENCE #: CN601527534 SAN ANTONIO, TEXAS JANUARY 4, 2007

Prepared and Reviewed by:

Patrick K. McGovern, Jr., Test Manager



### --------------------------------------------- ------------------------------------------- John M. Plummer, Project Manager Quinn A. Bierman, President

### TABLE OF CONTENTS

1.0 INTRODUCTION ______________________________________________________ 1 1.1 TEST PURPOSE AND OBJECTIVES __________________________________ 1 1.2 SUMMARY OF TEST PROGRAM _____________________________________ 1 1.2.1 Participating Organizations__________________________________________ 1 1.2.2 Industry _________________________________________________________ 1 1.2.3 Plant Location ____________________________________________________ 1 1.2.4 Equipment Tested _________________________________________________ 1 1.2.5 Emission Points___________________________________________________ 1 1.2.6 Pollutants Measured _______________________________________________ 1 1.2.7 Date of Emission Test______________________________________________ 1 1.3 KEY PERSONNEL __________________________________________________ 2 2.0 SUMMARY OF TEST RESULTS ________________________________________ 2 2.1 EMISSION TEST RESULTS __________________________________________ 2 3.0 SOURCE OPERATION _________________________________________________ 9 3.1 PROCESS DESCRIPTION ____________________________________________ 9 4.0 SAMPLING AND ANALYTICAL PROCEDURES __________________________ 9 4.1 TEST METHODS ____________________________________________________ 9 4.2 INSTRUMENT CONFIGURATION AND OPERATIONS __________________ 9

APPENDICES Appendix A Results and Calculations Appendix B Quality Assurance Activities Appendix C Calibration Gas Certifications

### ed-07-sanantonio-tx-ecom#1.doc Page 1 

### Stack Emissions Study from the Caterpillar G3520C, Units 1 - 6 prepared for Energy Developments

San Antonio, Texas January 4, 2007 1.0 INTRODUCTION Air Hygiene International, Inc. (Air Hygiene) h as completed the stack sampling survey for nitrogen oxides (NOx), carbon monoxide (CO), and oxygen (O 2 

### ) from the exhausts of the Caterpillar G3520C, Units 1 - 6 at Energy Developments’ Covel Gard ens Landfill Gas Power Station near San Antonio, Texas. This report details the background in formation, test results, process description, sampling/analysis methodology, and quality control/ assurance activities of the stack sampling survey conducted on January 4, 2007. 

### 1.1 TEST PURPOSE AND OBJECTIVES The purpose of the test was to conduct the quarterly emission test to document emission levels of selected pollutants at normal operation levels of the designated engines. The information will be used to confirm compliance with the air quality perm it requirements. The specific objective was to determine the emission concentration of NOx, CO, and O 2 

### from the exhausts of Energy Developments’ Caterpillar G3520C, Units 1 - 6 dur ing normal operating conditions. 

### 1.2 SUMMARY OF TEST PROGRAM The following list details pertinent inform ation related to this specific project: 1.2.1 Participating Organizations: 3⁄4 Energy Developments 3⁄4 Air Hygiene 1.2.2 Industry: 3⁄4 Electric Utility 1.2.3 Plant Location: 3⁄4 Covel Gardens Landfill Gas Power Station near San Antonio, Texas 1.2.4 Equipment Tested: 3⁄4 Caterpillar G3520C, Units 1 - 6, Emission Point Numbers E1 - E6 1.2.5 Emission Points: 3⁄4 Exhausts of the Caterpillar G3520C, Units 1 - 6 1.2.6 Pollutants Measured: 3⁄4 NOx 3⁄4 CO 3⁄4 O 2 

### 1.2.7 Date of Emission Test: 3⁄4 January 4, 2007

### ed-07-sanantonio-tx-ecom#1.doc Page 2 

### 1.3 KEY PERSONNEL Energy Developments: Liza Mireles 713-300-3323 Air Hygiene: Patrick McGovern 918-307-8865



### 2.0 SUMMARY OF TEST RESULTS 2.1 EMISSION TEST RESULTS Results from the sampling conducted at Covel Gardens Landfill Gas Power Station from the Caterpillar G3520C, Units 1 - 6 exha usts on January 4, 2007 are summarized in the following tables.

### ed-07-sanantonio-tx-ecom#1.doc Page 3 

### TABLE 2.1 UNIT 1 EMISSIONS REPORT Company: Date: Engine Tested: Location: 

### Stack Gas Flow Rate: Method 2

Test # Brake Horsepower O 2 Conc. (%) Fuel Heating Value (Btu/SCF) [HHV] F Factor-Dry Oxy. (DSCFex/MMBtu) Fuel Flow (SCF/hr) Stack Flow (SCF/hr) 1 2,172.0 10.9 474.54 9,444.22 N/A 250,000.00 2 2,172.0 11.4 474.54 9,444.22 N/A 250,000.00 3 2,172.0 11.2 474.54 9,444.22 N/A 250,000.00 Average 2,172.0 11.2 474. 54 9,444.22 N/A 250,000.00 

### NOx Mass Emission Rate

Test # Brake Horsepower NOx Conc. (ppmvd) MW E (g/hp*hr) E (lb/hr) E (ton/yr) 1 2,172.0 96.9 46.01 0.60 2.89 12.67 2 2,172.0 91.8 46.01 0.57 2.74 12.00 3 2,172.0 72.5 46.01 0.45 2.16 9.48 Average 2,172.0 87.0 46.01 0.54 2.60 11.38 

### CO Mass Emission Rate

Test # Brake Horsepower CO Conc. (ppmvd) MW E (g/hp*hr) E (lb/hr) E (ton/yr) 1 2,172.0 652.5 28.00 2.48 11.86 51.93 2 2,172.0 576.1 28.00 2.19 10.47 45.85 3 2,172.0 624.3 28.00 2.37 11.34 49.69 Average 2,172.0 617.6 28.00 2.34 11.22 49.16 

### January 4, 2007 Energy Development Caterpillar G3520 C, Unit #1 Covel Gardens Power Station

### ed-07-sanantonio-tx-ecom#1.doc Page 4 

### TABLE 2.2 UNIT 2 EMISSIONS REPORT Company: Date: Engine Tested: Location: 

### Stack Gas Flow Rate: Method 2

Test # Brake Horsepower O 2 Conc. (%) Fuel Heating Value (Btu/SCF) [HHV] F Factor-Dry Oxy. (DSCFex/MMBtu) Fuel Flow (SCF/hr) Stack Flow (SCF/hr) 1 2,172.0 9.8 474.54 9,637.46 N/A 250,000.00 2 2,172.0 9.9 474.54 9,637.46 N/A 250,000.00 3 2,172.0 9.4 474.54 9,637.46 N/A 250,000.00 Average 2,172.0 9.7 474.54 9,637.46 N/A 250,000.00


### NOx Mass Emission Rate

Test # Brake Horsepower NOx Conc. (ppmvd) MW E (g/hp*hr) E (lb/hr) E (ton/yr) 1 2,172.0 96.9 46.01 0.60 2.89 12.67 2 2,172.0 91.1 46.01 0.57 2.72 11.92 3 2,172.0 62.3 46.01 0.39 1.86 8.15 Average 2,172.0 83.4 46.01 0.52 2.49 10.91 

### CO Mass Emission Rate

Test # Brake Horsepower CO Conc. (ppmvd) MW E (g/hp*hr) E (lb/hr) E (ton/yr) 1 2,172.0 662.4 28.00 2.51 12.04 52.72 2 2,172.0 660.5 28.00 2.51 12.00 52.57 3 2,172.0 738.8 28.00 2.80 13.42 58.80 Average 2,172.0 687.2 28.00 2.61 12.49 54.69 

### January 4, 2007 Energy Development Caterpillar G3520 C, Unit #2 Covel Gardens Power Station

### ed-07-sanantonio-tx-ecom#1.doc Page 5 

### TABLE 2.3 UNIT 3 EMISSIONS REPORT Company: Date: Engine Tested: Location: 

### Stack Gas Flow Rate: Method 2

Test # Brake Horsepower O 2 Conc. (%) Fuel Heating Value (Btu/SCF) [HHV] F Factor-Dry Oxy. (DSCFex/MMBtu) Fuel Flow (SCF/hr) Stack Flow (SCF/hr) 1 2,172.0 9.5 474.54 9,637.46 N/A 250,000.00 2 2,172.0 9.7 474.54 9,637.46 N/A 250,000.00 3 2,172.0 9.6 474.54 9,637.46 N/A 250,000.00 Average 2,172.0 9.6 474.54 9,637.46 N/A 250,000.00 

### NOx Mass Emission Rate

Test # Brake Horsepower NOx Conc. (ppmvd) MW E (g/hp*hr) E (lb/hr) E (ton/yr) 1 2,172.0 96.7 46.01 0.60 2.89 12.64 2 2,172.0 75.4 46.01 0.47 2.25 9.86 3 2,172.0 82.8 46.01 0.52 2.47 10.82 Average 2,172.0 84.9 46.01 0.53 2.54 11.11 

### CO Mass Emission Rate

Test # Brake Horsepower CO Conc. (ppmvd) MW E (g/hp*hr) E (lb/hr) E (ton/yr) 1 2,172.0 584.6 28.00 2.22 10.62 46.53 2 2,172.0 584.8 28.00 2.22 10.63 46.54 3 2,172.0 625.0 28.00 2.37 11.36 49.74 Average 2,172.0 598.1 28.00 2.27 10.87 47.60 

### January 4, 2007 Energy Development Caterpillar G3520 C, Unit #3 Covel Gardens Power Station

### ed-07-sanantonio-tx-ecom#1.doc Page 6 

### TABLE 2.4 UNIT 4 EMISSIONS REPORT Company: Date: Engine Tested: Location: 

### Stack Gas Flow Rate: Method 2

Test # Brake Horsepower O 2 Conc. (%) Fuel Heating Value (Btu/SCF) [HHV] F Factor-Dry Oxy. (DSCFex/MMBtu) Fuel Flow (SCF/hr) Stack Flow (SCF/hr) 1 2,172.0 9.4 474.54 9,637.46 N/A 250,000.00 2 2,172.0 9.4 474.54 9,637.46 N/A 250,000.00 3 2,172.0 9.5 474.54 9,637.46 N/A 250,000.00 Average 2,172.0 9.4 474.54 9,637.46 N/A 250,000.00 

### NOx Mass Emission Rate

Test # Brake Horsepower NOx Conc. (ppmvd) MW E (g/hp*hr) E (lb/hr) E (ton/yr) 1 2,172.0 70.1 46.01 0.44 2.09 9.17 2 2,172.0 70.0 46.01 0.44 2.09 9.15 3 2,172.0 69.3 46.01 0.43 2.07 9.06 Average 2,172.0 69.8 46.01 0.44 2.08 9.13 

### CO Mass Emission Rate

Test # Brake Horsepower CO Conc. (ppmvd) MW E (g/hp*hr) E (lb/hr) E (ton/yr) 1 2,172.0 732.5 28.00 2.78 13.31 58.30 2 2,172.0 714.0 28.00 2.71 12.97 56.83 3 2,172.0 733.0 28.00 2.78 13.32 58.34 Average 2,172.0 726.5 28.00 2.76 13.20 57.82 

### January 4, 2007 Energy Development Caterpillar G3520 C, Unit #4 Covel Gardens Power Station

### ed-07-sanantonio-tx-ecom#1.doc Page 7 

### TABLE 2.5 UNIT 5 EMISSIONS REPORT Company: Date: Engine Tested: Location: 

### Stack Gas Flow Rate: Method 2

Test # Brake Horsepower O 2 Conc. (%) Fuel Heating Value (Btu/SCF) [HHV] F Factor-Dry Oxy. (DSCFex/MMBtu) Fuel Flow (SCF/hr) Stack Flow (SCF/hr) 1 2,172.0 9.3 474.54 9,637.46 N/A 250,000.00 2 2,172.0 9.4 474.54 9,637.46 N/A 250,000.00 3 2,172.0 9.1 474.54 9,637.46 N/A 250,000.00 Average 2,172.0 9.3 474.54 9,637.46 N/A 250,000.00 

### NOx Mass Emission Rate

Test # Brake Horsepower NOx Conc. (ppmvd) MW E (g/hp*hr) E (lb/hr) E (ton/yr) 1 2,172.0 88.6 46.01 0.55 2.65 11.59 2 2,172.0 83.1 46.01 0.52 2.48 10.87 3 2,172.0 101.6 46.01 0.63 3.03 13.29 Average 2,172.0 91.1 46.01 0.57 2.72 11.92 

### CO Mass Emission Rate

Test # Brake Horsepower CO Conc. (ppmvd) MW E (g/hp*hr) E (lb/hr) E (ton/yr) 1 2,172.0 573.5 28.00 2.18 10.42 45.64 2 2,172.0 546.9 28.00 2.08 9.94 43.53 3 2,172.0 574.0 28.00 2.18 10.43 45.68 Average 2,172.0 564.8 28.00 2.14 10.26 44.95 

### January 4, 2007 Energy Development Caterpillar G3520 C, Unit #5 Covel Gardens Power Station

### ed-07-sanantonio-tx-ecom#1.doc Page 8 

### TABLE 2.6 UNIT 6 EMISSIONS REPORT Company: Date: Engine Tested: Location: 

### Stack Gas Flow Rate: Method 2

Test # Brake Horsepower O 2 Conc. (%) Fuel Heating Value (Btu/SCF) [HHV] F Factor-Dry Oxy. (DSCFex/MMBtu) Fuel Flow (SCF/hr) Stack Flow (SCF/hr) 1 2,172.0 9.4 474.54 9,637.46 N/A 250,000.00 2 2,172.0 9.4 474.54 9,637.46 N/A 250,000.00 3 2,172.0 9.2 474.54 9,637.46 N/A 250,000.00 Average 2,172.0 9.3 474.54 9,637.46 N/A 250,000.00 

### NOx Mass Emission Rate

Test # Brake Horsepower NOx Conc. (ppmvd) MW E (g/hp*hr) E (l b/hr) E (ton/yr) 1 2,172.0 75.3 46.01 0.47 2.25 9.84 2 2,172.0 75.1 46.01 0.47 2.24 9.82 3 2,172.0 73.6 46.01 0.46 2.20 9.63 Average 2,172.0 74.7 46.01 0.47 2.23 9.77 

### CO Mass Emission Rate

Test # Brake Horsepower CO Conc. (ppmvd) MW E (g/hp*hr) E (l b/hr) E (ton/yr) 1 2,172.0 718.8 28.00 2.73 13.06 57.20 2 2,172.0 711.0 28.00 2.70 12.92 56.59 3 2,172.0 689.3 28.00 2.62 12.52 54.86 Average 2,172.0 706.3 28.00 2.68 12.83 56.22 

### January 4, 2007 Energy Development Caterpillar G3520 C, Unit #6 Covel Gardens Power Station

### ed-07-sanantonio-tx-ecom#1.doc Page 9 

### 3.0 SOURCE OPERATION 3.1 PROCESS DESCRIPTION The units tested were Caterpillar G3520C, Units 1 - 6. The engines are rated at 2,172 horsepower. The engines emissions are vented to the atmos phere through an exhaust system extending from the engine. The sample ports were located on the exha ust stack s . All exhaust samples were continuously drawn from the exhaust stack s at the sample ports.



### 4.0 SAMPLING AND ANALYTICAL PROCEDURES 4.1 TEST METHODS The emission tests on the Caterpillar G3520C, Units 1 - 6 at the Covel Gardens Landfill Gas Power Station was performed following EPA methods described by the Code of Federal Regulations (CFR), Title 40, Part 60, Appendix A and Title 40, Part 63, Appendix A, Conditional Test Method (CTM)- 034. Table 4.1 below outlines the specific methods performed on January 4, 2007. TABLE 4.1 SUMMARY OF SAMPLING METHODS

Pollutant or Parameter Sampling Method Analysis Method Stack Flow Rate EPA Method 2 S-type Pitot Tube O 2 

### , NOx, CO EPA CTM-034 Electrochemical Cell



### 4.2 INSTRUMENT CONFIGURATION AND OPERATIONS FOR GAS ANALYSIS The sampling and analysis procedures used during th ese tests conform in principle with the methods outlined in the Code of Federal Regulations (CFR), Title 40, Part 60, Appendix A, Methods 2, and CFR, Title 40, Part 63, Appendix A, Conditional Test Method (CTM)-034. Figure 4.1 depicts the sample system used for the NOx, CO, and O 2 

### tests. A stainless steel probe was inserted into the sample port of the stack to extr act gas measurements from the emission stream. The gas sample was continuously pulled through the probe and transported via heat-traced Teflon® tubing to the ECOM-AC portable gas analyzer.
Gaseous calibration standards were provided in aluminum cylinders w ith the concentrations certified by the vendor. EPA Protocol No. 1 was used to determine the cylinder concentrations where applicable (i.e. NOx calibration gases).

### ed-07-sanantonio-tx-ecom#1.doc Page 10 Table 4.2 provides a description of the analyzer used for the instrument portion of the tests. All data from the continuous monitoring instrument was reco rded using the Advanced DAS ECOM Software. Data records can be found in Appendix A of this report. Three test runs of approximately 10 minutes each were conducted on the engines in the “as found” operating condition of the Caterpillar G 3520C, Units 1 - 6 for NOx, CO, and O 2 

### . Exhaust flow rates were measured on Unit 2 at high load during the May 9, 2006 initial compliance demonstration. The Unit 2 exhaust flow rate determined in accordance with EPA Method 2 is representative of all units and is used to calculate emission rates

. The stack gas analysis for O 2 

### , NOx, and CO concentrations was performed in accordance with the procedures set forth in EPA CTM-034. The analyzer uses an electrochemical cell. TABLE 4.2 ANALYTICAL INSTRUMENTATION

Parameter Model & Manufacturer Max. Ranges Sensitivity Detection Principle

O 2 

### , NOx, and CO ECOM - AC 21%, 4000 ppm, 40,000

ppm 0.1 ppm or % Electrochemical CellFigure 4.1PORTABLE ANALYZERE:\SHARED\DRAWINGS\ECOM SCHEMATIC.PPT 04/04/06 DRAWING:06-001 TKG COPYRIGHT©2006 AIR HYGIENE INTERNATIONAL, INC.5634 S. 122ndEast Ave, Suite FTulsa, Oklahoma 74146www.airhygiene.com(888) 461-8778

STACK flowsampleconfiguration SAMPLE LINEGASCONDITIONERCALIBRATION/DRIFTGAS BOTTLES calibration/ drift lineO2ANALYZERCOANALYZERNOxANALYZERTHCANALYZERSO2ANALYZER (optional)HYPERLOGGERDATARECORDERcalibrationconfigurationPROBEPROBE ELECTROCHEMICAL CELL

# APPENDIX A Results and Calculations

### App. A 

### EMISSION CALCULATIONS

Mass Emission Rate Calculations


## G

MW Q CRA hr lb E S 

## × × = 6 

## 10 ) / ( CRA = corrected raw average Qs = stack gas flow rate (Method 19) MW = molecular weight (lb/lb-mol) G = ideal gas conversion factor



### bhp E hr hp g E hr lb 

### 6

. 453 ) / ( / 

### ×
= ⋅ bhp = brake horsepower



## 2000 ) / ( / year hr lb 

### hr E yr ton E ×
= hr year 

### = hours per year

### 123456 Air Fuel Ratio 27 26.8 26.1 26.4 25 26.8 Engine Speed in RPM 1200 1200 1200 1200 1200 1200 Load in K W 1600 1600 1600 1600 1600 1600 Exhaust Temperature 753 749 743 744 749 746 UNIT # ENGINE OPERATING DATA 1-4-07

# Unit 1Facility

Phone Serial: # 91210CVNXH Gas type O 2 % CO ppm NO ppm NO 2 pp m Concent. 20.9 886 138 47.7

Date: 1/4/2007 Cell Temp Flow Rat e Three O 2 % CO ppm NO ppm NO 2 pp m Phase s 10.0 678.0 8.0 122.3 90.6 2.5 10.2 665.8 6.8 116.5 90.6 2.5 10.4 672.5 6.0 130.8 90.6 2.5 10.6 660.8 6.0 124.5 90.6 2.5 

## Phase 10.7 667.3 6.0 103.3 90.5 2.5 10.8 654.0 7.0 89.0 10.9 649.0 7.0 88.0 10.9 646.0 7.0 87.0 10.9 646.0 7.0 86.0 90.4 2.5 10.9 654.0 7.0 95.0 11.0 657.0 7.0 90.0 11.0 654.0 7.0 90.0 11.0 659.0 7.0 88.0 90.4 2.5
10.9 652.4 7.0 89.1
1.1% 1.0% 0.0% 6.6% Yes or No Yes or No Yes or No Yes or No 11.0 656.0 7.0 90.8 90.4 2.5 11.1 632.3 7.3 80.7 90.6 2.5 11.3 347.3 9.0 30.8 90.6 2.5 11.7 21.3 8.3 24.0 90.5 2.5

Stop Time: Covel Gardens Power Station Address 8611 Covel Road San Antonio, Texas 78252 Emission Point: E1 (Unit 1)

210-623-1314
Calibration Gas Verification Information ECOM-AC Analyzer make & model:
Calibration Gas Info. (manufacturer,expiration,etc) 

## MEASUREMENT CYCLE ( circle measurement task below ) Source Test (run1) Analyzer Response



## RAMP- t 1 t 2 

## UP t 3 t 4 t 5 t 5:15 

## TEST t 5:30 t 5:45 

## DATA t 6:00 t 6:15 

## Phase t 6:30 t 6:45 t 7:00 Mean Avera g e Concentration ( sum of t5:15 throu g h t7:00/8 ) Maximum Deviation (no single reading exceeds ± 2% of mean average) Acce p table "Test Data Phase" t 7

## REt 8 t 9 t 10 t 11 

## Phase t 12 t 13



### Yes or No 

### Yes or No Sam p le Flow Rate are within s p ecifications ( ±10% or as verified ) Start time: 13:56:51

14:06:21 t 14 t 15
Cell Tem p erature ( ±10F for each run, Not to exceed 20F for test da y) and 

## FRESH

### Facility

Phone Serial: # 91210CVN Gas type O 2 

### % CO ppm NO ppm NO 2 

### pp m Concent. 20.9 886 138 47.7

Date: 1/4/2007 

### Cell Temp Flow Rate
Three O 2 

### % CO ppm NO ppm NO 2 

### pp m Phases 13.2 148.5 8.5 70.5 89.8 2.5 12.9 523.0 9.3 79.5 89.8 2.5 12.7 572.5 9.3 79.8 90.0 2.5 12.5 579.5 9.3 78.0 90.2 2.5 

## Phase 12.3 565.0 9.8 70.0 90.4 2.5 12.3 573.0 9.0 69.0 11.6 584.0 15.0 71.0 11.5 579.0 16.0 71.0 11.4 570.0 16.0 78.0 90.6 2.5 11.3 579.0 16.0 84.0 11.2 572.0 16.0 81.0 11.2 575.0 15.0 81.0 11.2 577.0 14.0 82.0 90.4 2.5
11.5 576.1 14.6 77.1
7.3% 1.4% 38.5% 10.5% Yes or No Yes or No Yes or No Yes or No 

### 11.2 575.8 15.3 82.0 90.4 2.5 11.2 18.0 13.7 53.7 89.5 2.5 12.1 13.8 12.5 23.5 89.2 2.5 12.8 12.0 10.5 20.8 89.1 2.5

Stop Time: 

### Yes or No 

### Yes or No Sample Flow Rate are within specifications (±10% or as verified) 

### Start time: 14:08:18

14:18:03 t 14 t 15 Cell Temperature (±10F for each run, Not to exceed 20F for test day) and 

## FRESH t 10 t 11 

## Phase t 12 t 13 exceeds ± 2% of mean average) 

### Acce p table "Test Data Phase" t 7

## REt 8 t 9 t 7:00 Mean Avera g e Concentration ( sum of t5:15 throu g h t7:00/8 ) Maximum Deviation (no single reading 

## DATA t 6:00 t 6:15 

## Phase t 6:30 t 6:45 t 5:15 

## TEST t 5:30 t 5:45 

## UP t 3 t 4 t 5 Analyzer Response



## RAMP- t 1 t 2 

## Source Test (run 2) Calibration Gas Info. 

### (manufacturer,expiration,etc) 

## MEASUREMENT CYCLE ( circle measurement task below ) 210-623-1314
Calibration Gas Verification Information ECOM-AC Analyzer make & model: Covel Gardens Power Station Address 8611 Covel Road San Antonio, Texas 78252 Emission Point: E1 (Unit 1)

### Facility

Phone Serial: # 91210CVN Gas type O 2 

### % CO ppm NO ppm NO 2 

### pp m Concent. 20.9 886 138 47.7

Date: 1/4/2007 

### Cell Temp Flow Rate
Three O 2 

### % CO ppm NO ppm NO 2 

### pp m Phases 18.0 10.0 19.5 40.3 88.8 2.5 15.0 319.5 15.8 65.8 88.9 2.5 12.8 624.0 15.0 58.3 89.1 2.5 12.2 628.2 11.3 55.5 89.3 2.5 

## Phase 11.7 625.7 10.0 60.8 89.4 2.5 11.5 624.8 9.0 68.0 11.4 625.0 9.0 69.0 11.5 626.0 8.0 66.0 11.4 623.0 8.0 64.0 89.4 2.5 11.1 621.0 9.0 54.0 10.8 626.0 12.0 55.0 10.6 625.0 16.0 57.0 10.3 625.0 21.0 56.0 89.4 2.5
11.1 624.5 11.5 61.1
7.0% 0.6% 82.6% 12.9% Yes or No Yes or No Yes or No Yes or No 

### 10.7 624.3 14.5 55.5 89.4 2.5 10.4 282.3 18.7 31.0 89.4 2.5 10.5 30.0 12.5 19.8 89.1 2.5 11.1 7.3 8.5 18.0 88.8 2.5

Stop Time: Covel Gardens Power Station Address 8611 Covel Road San Antonio, Texas 78252 Emission Point: E1 (Unit 1)

210-623-1314
Calibration Gas Verification Information ECOM-AC Analyzer make & model:
Calibration Gas Info. 

### (manufacturer,expiration,etc) 

## MEASUREMENT CYCLE ( circle measurement task below ) Source Test (run 3) Analyzer Response



## RAMP- t 1 t 2 

## UP t 3 t 4 t 5 t 5:15 

## TEST t 5:30 t 5:45 

## DATA t 6:00 t 6:15 

## Phase t 6:30 t 6:45 t 7:00 Mean Avera g e Concentration ( sum of t5:15 throu g h t7:00/8 ) Maximum Deviation (no single reading exceeds ± 2% of mean average) 

### Acce p table "Test Data Phase" t 7

## REt 8 t 9 t 10 t 11 

## Phase t 12 t 13



### Yes or No 

### Yes or No Sample Flow Rate are within specifications (±10% or as verified) 

### Start time: 14:18:59

14:28:44 t 14 t 15 Cell Temperature (±10F for each run, Not to exceed 20F for test day) and 

## FRESH

# Unit 2Facility

Phone Serial: # 91210CVNXH Gas type O 2 % CO ppm NO ppm NO 2 pp m Concent. 20.9 886 138 47.7

Date: 1/4/2007 Cell Temp Flow Rat e Three O 2 % CO ppm NO ppm NO 2 pp m Phase s 9.8 719.3 11.0 90.5 133.7 2.5 9.8 719.3 9.3 90.0 132.9 2.5 9.8 712.0 10.8 93.5 132.4 2.5 14.9 664.3 8.0 61.0 109.5 2.5 

## Phase 9.9 363.0 8.5 96.5 118.5 2.5 9.9 635.0 9.0 93.0 9.8 665.0 9.0 93.0 9.8 683.0 9.0 93.0 9.9 691.0 10.0 92.0 118.1 2.5 9.8 691.0 10.0 88.0 9.8 688.0 11.0 85.0 9.9 687.0 11.0 81.0 9.8 687.0 12.0 84.0 118.0 2.5
9.8 678.4 10.1 88.6
0.6% 6.4% 18.5% 8.6% Yes or No Yes or No Yes or No Yes or No 9.8 688.3 11.0 84.5 118.0 2.5 12.0 660.0 13.7 72.7 118.8 2.5 20.1 125.6 7.0 28.5 118.8 2.5 20.9 8.0 4.3 6.3 118.6 2.5

Stop Time: 

### Yes or No 

### Yes or No Sam p le Flow Rate are within s p ecifications ( ±10% or as verified ) Start time: 11:53:42

12:03:12 t 14 t 15
Cell Tem p erature ( ±10F for each run, Not to exceed 20F for test da y) and 

## FRESH t 10 t 11 

## Phase t 12 t 13 exceeds ± 2% of mean average) Acce p table "Test Data Phase" t 7

## REt 8 t 9 t 7:00 Mean Avera g e Concentration ( sum of t5:15 throu g h t7:00/8 ) Maximum Deviation (no single reading 

## DATA t 6:00 t 6:15 

## Phase t 6:30 t 6:45 t 5:15 

## TEST t 5:30 t 5:45 

## UP t 3 t 4 t 5 Analyzer Response



## RAMP- t 1 t 2 Calibration Gas Info. (manufacturer,expiration,etc) 

## MEASUREMENT CYCLE ( circle measurement task below ) Source Test (run1) 210-623-1314
Calibration Gas Verification Information ECOM-AC Analyzer make & model: Covel Gardens Power Station Address 8611 Covel Road San Antonio, Texas 78252 Emission Point: E2 (Unit 2)

### Facility

Phone Serial: # 91210CVN Gas type O 2 

### % CO ppm NO ppm NO 2 

### pp m Concent. 20.9 886 138 47.7

Date: 10/24/2006 

### Cell Temp Flow Rate
Three O 2 

### % CO ppm NO ppm NO 2 

### pp m Phases 9.9 525.5 16.5 63.5 109.4 2.5 9.9 594.5 18.5 58.8 109.8 2.5 9.9 617.3 21.0 55.3 110.0 2.5 9.9 633.8 25.0 55.3 109.9 2.5 

## Phase 9.9 645.8 19.0 67.3 109.7 2.5 9.9 651.0 16.0 72.0 10.0 653.0 17.0 71.0 9.9 657.0 17.0 72.0 9.9 661.0 18.0 75.0 109.7 2.5 9.9 667.0 19.0 76.0 9.9 672.0 19.0 76.0 9.9 673.0 20.0 74.0 11.0 674.0 21.0 65.0 109.3 2.5
10.1 663.5 18.4 72.6
9.5% 1.9% 14.3% 10.5% Yes or No Yes or No Yes or No Yes or No 

### 10.2 671.5 19.8 72.8 109.3 2.5 19.6 665.7 11.7 30.3 108.1 2.5 20.6 234.3 4.5 19.8 108.0 2.5 20.9 9.7 4.0 6.9 107.8 2.5

Stop Time: Covel Gardens Power Station Address 8611 Covel Road San Antonio, Texas 78252 Emission Point: E2 (Unit 2)

210-623-1314
Calibration Gas Verification Information ECOM-AC Analyzer make & model:



## Source Test (run 2) Calibration Gas Info. 

### (manufacturer,expiration,etc) 

## MEASUREMENT CYCLE ( circle measurement task below )

Analyzer Response



## RAMP- t 1 t 2 

## UP t 3 t 4 t 5 t 5:15 

## TEST t 5:30 t 5:45 

## DATA t 6:00 t 6:15 

## Phase t 6:30 t 6:45 t 7:00 Mean Avera g e Concentration ( sum of t5:15 throu g h t7:00/8 ) Maximum Deviation (no single reading exceeds ± 2% of mean average) 

### Acce p table "Test Data Phase" t 7

## REt 8 t 9 t 10 t 11 

## Phase t 12 t 13



### Yes or No 

### Yes or No Sample Flow Rate are within specifications (±10% or as verified) 

### Start time: 12:05:15

12:15:00 t 14 t 15 Cell Temperature (±10F for each run, Not to exceed 20F for test day) and 

## FRESH

### Facility

Phone Serial: # 91210CVN Gas type O 2 

### % CO ppm NO ppm NO 2 

### pp m Concent. 20.9 886 138 47.7

Date: 10/24/2006 

### Cell Temp Flow Rate
Three O 2 

### % CO ppm NO ppm NO 2 

### pp m Phases 10.3 601.3 38.0 6.3 91.0 2.5 10.3 650.5 36.3 11.5 91.8 2.5 10.2 685.3 34.3 15.0 95.1 2.5 10.1 718.8 31.3 21.8 95.6 2.5 

## Phase 10.1 733.3 27.5 33.0 94.7 2.5 10.1 734.0 27.0 34.0 10.1 735.0 27.0 34.0 10.0 735.0 27.0 35.0 10.1 736.0 27.0 35.0 93.8 2.5 10.1 736.0 27.0 35.0 10.0 734.0 27.0 35.0 10.0 734.0 28.0 37.0 10.0 734.0 28.0 36.0 93.4 2.5
10.1 734.8 27.3 35.1
0.5% 0.2% 2.8% 5.3% Yes or No Yes or No Yes or No Yes or No 

### 10.0 734.5 27.5 35.8 93.4 2.5 17.3 720.3 13.3 19.7 93.1 2.5 20.8 308.8 3.0 8.0 92.9 2.5 20.8 8.4 2.0 6.8 92.5 2.5

Stop Time: 

### Yes or No 

### Yes or No Sample Flow Rate are within specifications (±10% or as verified) 

### Start time: 13:01:52

13:11:37 t 14 t 15 Cell Temperature (±10F for each run, Not to exceed 20F for test day) and 

## FRESH t 10 t 11 

## Phase t 12 t 13 exceeds ± 2% of mean average) 

### Acce p table "Test Data Phase" t 7

## REt 8 t 9 t 7:00 Mean Avera g e Concentration ( sum of t5:15 throu g h t7:00/8 ) Maximum Deviation (no single reading 

## DATA t 6:00 t 6:15 

## Phase t 6:30 t 6:45 t 5:15 

## TEST t 5:30 t 5:45 

## UP t 3 t 4 t 5 Analyzer Response



## RAMP- t 1 t 2 Calibration Gas Info. 

### (manufacturer,expiration,etc) 

## MEASUREMENT CYCLE ( circle measurement task below ) Source Test (run 3) 210-623-1314
Calibration Gas Verification Information ECOM-AC Analyzer make & model: Covel Gardens Power Station Address 8611 Covel Road San Antonio, Texas 78252 Emission Point: E2 (Unit 2)

# Unit 3Facility

Phone Serial: # 91210CVNXH Gas type O 2 % CO ppm NO ppm NO 2 pp m Concent. 20.9 886 138 47.7

Date: 1/4/2007 Cell Temp Flow Rat e Three O 2 % CO ppm NO ppm NO 2 pp m Phase s 9.3 566.8 35.5 60.3 93.5 2.5 9.2 579.0 22.8 88.3 93.3 2.5 9.3 585.0 24.5 89.3 93.1 2.5 9.3 589.8 27.0 82.3 93.1 2.5 

## Phase 9.5 592.3 18.0 81.3 93.5 2.5 9.5 588.0 17.0 78.0 9.5 584.0 18.0 77.0 9.5 582.0 18.0 78.0 9.4 582.0 17.0 81.0 93.7 2.5 9.5 583.0 16.0 84.0 9.6 583.0 15.0 82.0 9.6 584.0 15.0 78.0 9.9 580.0 22.0 70.0 93.7 2.5
9.6 583.3 17.3 78.5
3.5% 0.8% 27.5% 10.8% Yes or No Yes or No Yes or No Yes or No 9.7 582.5 17.0 78.5 93.7 2.5 14.4 573.0 21.0 50.3 93.5 2.5 20.4 541.0 6.3 24.0 93.3 2.5 20.9 16.3 4.3 8.5 93.2 2.5

Stop Time: Covel Gardens Power Station Address 8611 Covel Road San Antonio, Texas 78252 Emission Point: E3 (Unit 3)

210-623-1314
Calibration Gas Verification Information ECOM-AC Analyzer make & model:
Calibration Gas Info. (manufacturer,expiration,etc) 

## MEASUREMENT CYCLE ( circle measurement task below ) Source Test (run1) Analyzer Response



## RAMP- t 1 t 2 

## UP t 3 t 4 t 5 t 5:15 

## TEST t 5:30 t 5:45 

## DATA t 6:00 t 6:15 

## Phase t 6:30 t 6:45 t 7:00 Mean Avera g e Concentration ( sum of t5:15 throu g h t7:00/8 ) Maximum Deviation (no single reading exceeds ± 2% of mean average) Acce p table "Test Data Phase" t 7

## REt 8 t 9 t 10 t 11 

## Phase t 12 t 13



### Yes or No 

### Yes or No Sam p le Flow Rate are within s p ecifications ( ±10% or as verified ) Start time: 11:13:16

11:22:46 t 14 t 15
Cell Tem p erature ( ±10F for each run, Not to exceed 20F for test da y) and 

## FRESH

### Facility

Phone Serial: # 91210CVN Gas type O 2 

### % CO ppm NO ppm NO 2 

### pp m Concent. 20.9 886 138 47.7

Date: 10/24/2006 

### Cell Temp Flow Rate
Three O 2 

### % CO ppm NO ppm NO 2 

### pp m Phases 9.8 28.5 12.5 61.3 93.2 2.5 9.6 333.3 18.3 55.5 93.3 2.5 9.6 510.0 17.5 54.5 93.3 2.5 9.6 548.0 14.3 63.8 93.3 2.5 

## Phase 9.7 568.0 16.3 59.3 93.3 2.5 9.7 573.0 17.0 56.0 9.6 576.0 15.0 58.0 9.7 579.0 13.0 61.0 9.7 582.0 13.0 63.0 93.3 2.5 9.6 586.0 13.0 63.0 9.6 590.0 14.0 64.0 9.6 594.0 14.0 63.0 9.7 598.0 14.0 62.0 93.2 2.5
9.7 584.8 14.1 61.3
0.5% 2.3% 20.4% 8.6% Yes or No Yes or No Yes or No Yes or No 

### 9.6 592.0 13.8 63.0 93.2 2.5 9.6 602.0 11.3 68.7 93.2 2.5 15.1 619.5 8.0 54.8 93.2 2.5 20.9 12.3 3.8 7.4 93.2 2.5

Stop Time: 

### Yes or No 

### Yes or No Sample Flow Rate are within specifications (±10% or as verified) 

### Start time: 11:23:41

11:33:26 t 14 t 15 Cell Temperature (±10F for each run, Not to exceed 20F for test day) and 

## FRESH t 10 t 11 

## Phase t 12 t 13 exceeds ± 2% of mean average) 

### Acce p table "Test Data Phase" t 7

## REt 8 t 9 t 7:00 Mean Avera g e Concentration ( sum of t5:15 throu g h t7:00/8 ) Maximum Deviation (no single reading 

## DATA t 6:00 t 6:15 

## Phase t 6:30 t 6:45 t 5:15 

## TEST t 5:30 t 5:45 

## UP t 3 t 4 t 5 Analyzer Response



## RAMP- t 1 t 2 

## Source Test (run 2) Calibration Gas Info. 

### (manufacturer,expiration,etc) 

## MEASUREMENT CYCLE ( circle measurement task below ) 210-623-1314
Calibration Gas Verification Information ECOM-AC Analyzer make & model: Covel Gardens Power Station Address 8611 Covel Road San Antonio, Texas 78252 Emission Point: E3 (Unit 3)

### Facility

Phone Serial: # 91210CVN Gas type O 2 

### % CO ppm NO ppm NO 2 

### pp m Concent. 20.9 886 138 47.7

Date: 10/24/2006 

### Cell Temp Flow Rate
Three O 2 

### % CO ppm NO ppm NO 2 

### pp m Phases 10.9 47.0 5.3 56.5 93.0 2.5 9.6 462.0 11.0 68.8 93.3 2.5 9.6 579.0 13.5 63.8 93.3 2.5 9.6 601.5 16.0 59.8 93.0 2.5 

## Phase 9.6 609.5 13.0 63.5 92.7 2.5 9.6 615.0 10.0 73.0 9.5 620.0 10.0 75.0 9.6 626.0 10.0 75.0 9.6 630.0 10.0 74.0 92.3 2.5 9.6 631.0 11.0 71.0 9.6 632.0 11.0 71.0 9.6 635.0 12.0 70.0 10.4 638.0 12.0 66.0 91.9 2.5
9.7 628.4 10.8 71.9
7.4% 2.1% 11.6% 8.2% Yes or No Yes or No Yes or No Yes or No 

### 9.8 634.0 11.5 69.5 91.9 2.5 19.9 624.3 5.7 27.7 91.4 2.5 20.6 189.5 3.3 16.8 90.9 2.5 20.9 14.8 3.0 6.1 90.8 2.5

Stop Time: Covel Gardens Power Station Address 8611 Covel Road San Antonio, Texas 78252 Emission Point: E3 (Unit 3)

210-623-1314
Calibration Gas Verification Information ECOM-AC Analyzer make & model:
Calibration Gas Info. 

### (manufacturer,expiration,etc) 

## MEASUREMENT CYCLE ( circle measurement task below ) Source Test (run 3) Analyzer Response



## RAMP- t 1 t 2 

## UP t 3 t 4 t 5 t 5:15 

## TEST t 5:30 t 5:45 

## DATA t 6:00 t 6:15 

## Phase t 6:30 t 6:45 t 7:00 Mean Avera g e Concentration ( sum of t5:15 throu g h t7:00/8 ) Maximum Deviation (no single reading exceeds ± 2% of mean average) 

### Acce p table "Test Data Phase" t 7

## REt 8 t 9 t 10 t 11 

## Phase t 12 t 13



### Yes or No 

### Yes or No Sample Flow Rate are within specifications (±10% or as verified) 

### Start time: 11:34:59

11:22:46 t 14 t 15 Cell Temperature (±10F for each run, Not to exceed 20F for test day) and 

## FRESH

# Unit 4Facility

Phone Serial: # 91210CVNXH Gas type O 2 % CO ppm NO ppm NO 2 pp m Concent. 20.9 886 138 47.7

Date: 1/4/2007 Cell Temp Flow Rat e Three O 2 % CO ppm NO ppm NO 2 pp m Phase s 9.5 738.8 3.8 70.5 81.7 2.5 9.4 745.5 3.5 76.3 81.9 2.5 9.4 738.0 3.0 72.8 82.3 2.5 9.4 737.0 2.8 69.8 82.7 2.5 

## Phase 9.4 728.0 2.3 69.3 83.4 2.5 9.3 726.0 2.0 69.0 9.4 728.0 2.0 64.0 9.5 729.0 2.0 64.0 9.5 736.0 2.0 65.0 83.9 2.5 9.3 741.0 2.0 72.0 9.3 739.0 1.0 71.0 9.2 735.0 1.0 73.0 19.9 561.0 1.0 30.0 84.4 2.5
10.7 711.9 1.6 63.5
86.4% 21.2% 38.5% 52.8% Yes or No Yes or No Yes or No Yes or No 11.9 694.0 1.3 61.5 84.4 2.5 21.0 72.7 0.3 6.7 84.5 2.5 21.0 14.8 0.0 4.0 84.1 2.5 21.0 9.5 0.0 2.8 83.6 2.5

Stop Time: Covel Gardens Power Station Address 8611 Covel Road San Antonio, Texas 78252 Emission Point: E4 (Unit 4)

210-623-1314
Calibration Gas Verification Information ECOM-AC Analyzer make & model:
Calibration Gas Info. (manufacturer,expiration,etc) 

## MEASUREMENT CYCLE ( circle measurement task below ) Source Test (run1) Analyzer Response



## RAMP- t 1 t 2 

## UP t 3 t 4 t 5 t 5:15 

## TEST t 5:30 t 5:45 

## DATA t 6:00 t 6:15 

## Phase t 6:30 t 6:45 t 7:00 Mean Avera g e Concentration ( sum of t5:15 throu g h t7:00/8 ) Maximum Deviation (no single reading exceeds ± 2% of mean average) Acce p table "Test Data Phase" t 7

## REt 8 t 9 t 10 t 11 

## Phase t 12 t 13



### Yes or No 

### Yes or No Sam p le Flow Rate are within s p ecifications ( ±10% or as verified ) Start time: 17:10:06

17:19:36 t 14 t 15
Cell Tem p erature ( ±10F for each run, Not to exceed 20F for test da y) and 

## FRESH

### Facility

Phone Serial: # 91210CVN Gas type O 2 

### % CO ppm NO ppm NO 2 

### pp m Concent. 20.9 886 138 47.7

Date: 10/24/2006 

### Cell Temp Flow Rate
Three O 2 

### % CO ppm NO ppm NO 2 

### pp m Phases 10.0 414.8 0.5 54.8 83.4 2.5 9.4 622.5 1.0 65.8 84.3 2.5 9.4 664.0 1.0 66.8 84.8 2.5 9.4 694.8 1.0 70.5 85.3 2.5 

## Phase 9.4 711.8 1.0 71.5 85.7 2.5 9.3 716.0 1.0 69.0 9.3 712.0 1.0 68.0 9.4 709.0 1.0 67.0 9.3 710.0 1.0 72.0 86.0 2.5 9.3 716.0 1.0 69.0 9.5 714.0 1.0 64.0 9.4 718.0 1.0 71.0 9.3 717.0 2.0 71.0 86.1 2.5
9.4 714.0 1.1 68.9
1.6% 0.7% 77.8% 7.1% Yes or No Yes or No Yes or No Yes or No 

### 9.4 716.3 1.3 68.8 86.1 2.5 13.2 589.0 1.7 50.7 86.1 2.5 21.0 39.3 0.3 4.0 85.9 2.5 21.0 11.8 0.0 2.3 85.4 2.5

Stop Time: 

### Yes or No 

### Yes or No Sample Flow Rate are within specifications (±10% or as verified) 

### Start time: 17:20:13

17:29:58 t 14 t 15 Cell Temperature (±10F for each run, Not to exceed 20F for test day) and 

## FRESH t 10 t 11 

## Phase t 12 t 13 exceeds ± 2% of mean average) 

### Acce p table "Test Data Phase" t 7

## REt 8 t 9 t 7:00 Mean Avera g e Concentration ( sum of t5:15 throu g h t7:00/8 ) Maximum Deviation (no single reading 

## DATA t 6:00 t 6:15 

## Phase t 6:30 t 6:45 t 5:15 

## TEST t 5:30 t 5:45 

## UP t 3 t 4 t 5 Analyzer Response



## RAMP- t 1 t 2 

## Source Test (run 2) Calibration Gas Info. 

### (manufacturer,expiration,etc) 

## MEASUREMENT CYCLE ( circle measurement task below ) 210-623-1314
Calibration Gas Verification Information ECOM-AC Analyzer make & model: Covel Gardens Power Station Address 8611 Covel Road San Antonio, Texas 78252 Emission Point: E4 (Unit 4)

### Facility

Phone Serial: # 91210CVN Gas type O 2 

### % CO ppm NO ppm NO 2 

### pp m Concent. 20.9 886 138 47.7

Date: 10/24/2006 

### Cell Temp Flow Rate
Three O 2 

### % CO ppm NO ppm NO 2 

### pp m Phases 9.4 572.0 0.0 66.3 85.1 2.5 9.3 651.0 0.8 71.3 85.7 2.5 9.4 688.5 1.0 72.0 86.1 2.5 9.4 718.3 1.0 70.0 86.5 2.5 

## Phase 9.4 732.3 1.8 69.8 86.7 2.5 9.4 734.0 1.0 69.0 9.5 732.0 1.0 67.0 9.4 732.0 1.0 67.0 9.4 731.0 1.0 69.0 86.8 2.5 9.6 731.0 2.0 67.0 9.5 735.0 2.0 66.0 9.4 734.0 2.0 70.0 14.3 731.0 2.0 60.0 87.0 2.5
10.1 732.5 1.5 66.9
42.1% 0.3% 33.3% 10.3% Yes or No Yes or No Yes or No Yes or No 

### 10.7 732.8 2.0 65.8 87.0 2.5 19.9 132.3 0.7 11.0 87.1 2.5 20.9 18.5 0.0 3.0 87.2 2.5 21.0 9.5 0.0 2.0 87.2 2.5

Stop Time: Covel Gardens Power Station Address 8611 Covel Road San Antonio, Texas 78252 Emission Point: E4 (Unit 4)

210-623-1314
Calibration Gas Verification Information ECOM-AC Analyzer make & model:
Calibration Gas Info. 

### (manufacturer,expiration,etc) 

## MEASUREMENT CYCLE ( circle measurement task below ) Source Test (run 3) Analyzer Response



## RAMP- t 1 t 2 

## UP t 3 t 4 t 5 t 5:15 

## TEST t 5:30 t 5:45 

## DATA t 6:00 t 6:15 

## Phase t 6:30 t 6:45 t 7:00 Mean Avera g e Concentration ( sum of t5:15 throu g h t7:00/8 ) Maximum Deviation (no single reading exceeds ± 2% of mean average) 

### Acce p table "Test Data Phase" t 7

## REt 8 t 9 t 10 t 11 

## Phase t 12 t 13



### Yes or No 

### Yes or No Sample Flow Rate are within specifications (±10% or as verified) 

### Start time: 17:30:30

17:40:15 t 14 t 15 Cell Temperature (±10F for each run, Not to exceed 20F for test day) and 

## FRESH

# Unit 5Facility

Phone Serial: # 91210CVNXH Gas type O 2 % CO ppm NO ppm NO 2 pp m Concent. 20.9 886 138 47.7

Date: 1/4/2007 Cell Temp Flow Rat e Three O 2 % CO ppm NO ppm NO 2 pp m Phase s 9.3 570.3 9.0 73.3 88.4 2.5 9.4 565.0 6.5 75.0 88.3 2.5 9.3 568.8 6.8 78.0 88.3 2.5 9.3 570.3 8.3 79.8 88.3 2.5 

## Phase 9.3 572.8 9.5 77.8 88.3 2.5 9.3 573.0 8.0 82.0 9.3 574.0 7.0 81.0 9.2 572.0 6.0 83.0 9.2 571.0 6.0 85.0 88.5 2.5 9.3 574.0 6.0 83.0 9.3 575.0 7.0 80.0 9.2 575.0 7.0 81.0 9.3 574.0 7.0 82.0 88.5 2.5
9.3 573.5 6.8 82.1
0.7% 0.4% 18.5% 3.5% Yes or No Yes or No Yes or No Yes or No 9.3 574.5 6.8 81.5 88.5 2.5 15.9 511.3 4.7 52.0 88.5 2.5 20.7 41.5 2.0 13.8 88.5 2.5 20.7 8.0 3.0 8.8 88.4 2.5

Stop Time: 

### Yes or No 

### Yes or No Sam p le Flow Rate are within s p ecifications ( ±10% or as verified ) Start time:

9:27:51

09:37:36 t 14 t 15
Cell Tem p erature ( ±10F for each run, Not to exceed 20F for test da y) and 

## FRESH t 10 t 11 

## Phase t 12 t 13 exceeds ± 2% of mean average) Acce p table "Test Data Phase" t 7

## REt 8 t 9 t 7:00 Mean Avera g e Concentration ( sum of t5:15 throu g h t7:00/8 ) Maximum Deviation (no single reading 

## DATA t 6:00 t 6:15 

## Phase t 6:30 t 6:45 t 5:15 

## TEST t 5:30 t 5:45 

## UP t 3 t 4 t 5 Analyzer Response



## RAMP- t 1 t 2 Calibration Gas Info. (manufacturer,expiration,etc) 

## MEASUREMENT CYCLE ( circle measurement task below ) Source Test (run1) 210-623-1314
Calibration Gas Verification Information ECOM-AC Analyzer make & model: Covel Gardens Power Station Address 8611 Covel Road San Antonio, Texas 78252 Emission Point: E5 (Unit 5)

### Facility

Phone Serial: # 91210CVN Gas type O 2 

### % CO ppm NO ppm NO 2 

### pp m Concent. 20.9 886 138 47.7

Date: 10/24/2006 

### Cell Temp Flow Rate
Three O 2 

### % CO ppm NO ppm NO 2 

### pp m Phases 12.7 35.8 3.8 61.5 89.3 2.5 9.6 444.8 8.8 69.0 89.3 2.5 9.5 504.5 10.3 66.0 89.3 2.5 9.5 519.8 12.3 66.0 89.7 2.5 

## Phase 9.6 533.5 10.0 70.3 89.7 2.5 9.5 540.0 8.0 72.0 9.4 540.0 8.0 74.0 9.4 543.0 8.0 75.0 9.4 547.0 8.0 75.0 89.6 2.5 9.4 548.0 9.0 76.0 9.5 551.0 9.0 75.0 9.4 553.0 9.0 74.0 9.5 553.0 9.0 76.0 89.6 2.5
9.4 546.9 8.5 74.6
0.7% 1.3% 5.9% 3.5% Yes or No Yes or No Yes or No Yes or No 

### 9.5 551.3 9.0 75.3 89.6 2.5 16.0 506.7 6.3 50.0 89.5 2.5 20.6 52.0 3.3 15.0 89.5 2.5 20.6 6.5 2.3 10.3 89.5 2.5

Stop Time: Covel Gardens Power Station Address 8611 Covel Road San Antonio, Texas 78252 Emission Point: E5 (Unit 5)

210-623-1314
Calibration Gas Verification Information ECOM-AC Analyzer make & model:



## Source Test (run 2) Calibration Gas Info. 

### (manufacturer,expiration,etc) 

## MEASUREMENT CYCLE ( circle measurement task below )

Analyzer Response



## RAMP- t 1 t 2 

## UP t 3 t 4 t 5 t 5:15 

## TEST t 5:30 t 5:45 

## DATA t 6:00 t 6:15 

## Phase t 6:30 t 6:45 t 7:00 Mean Avera g e Concentration ( sum of t5:15 throu g h t7:00/8 ) Maximum Deviation (no single reading exceeds ± 2% of mean average) 

### Acce p table "Test Data Phase" t 7

## REt 8 t 9 t 10 t 11 

## Phase t 12 t 13



### Yes or No 

### Yes or No Sample Flow Rate are within specifications (±10% or as verified) 

### Start time: 09:38:26

09:48:11 t 14 t 15 Cell Temperature (±10F for each run, Not to exceed 20F for test day) and 

## FRESH

### Facility

Phone Serial: # 91210CVN Gas type O 2 

### % CO ppm NO ppm NO 2 

### pp m Concent. 20.9 886 138 47.7

Date: 10/24/2006 

### Cell Temp Flow Rate
Three O 2 

### % CO ppm NO ppm NO 2 

### pp m Phases 9.4 457.3 11.8 71.8 92.4 2.5 9.3 524.5 14.0 72.0 92.8 2.5 9.4 541.0 15.3 72.3 90.6 2.5 9.3 551.3 11.0 85.5 91.0 2.5 

## Phase 9.1 556.3 14.8 86.0 91.0 2.5 9.1 564.0 21.0 78.0 9.0 569.0 19.0 86.0 9.0 572.0 18.0 79.0 9.0 577.0 17.0 81.0 90.7 2.5 9.5 583.0 17.0 79.0 9.1 586.0 16.0 82.0 9.1 576.0 18.0 91.0 9.1 565.0 20.0 91.0 90.3 2.5
9.1 574.0 18.3 83.4
4.3% 2.1% 15.1% 9.1% Yes or No Yes or No Yes or No Yes or No 

### 9.2 577.5 17.8 85.8 90.3 2.5 18.7 558.3 12.7 48.3 91.1 2.5 20.6 232.3 5.0 23.3 91.2 2.5 20.6 11.8 3.5 10.8 91.2 2.5

Stop Time: 

### Yes or No 

### Yes or No Sample Flow Rate are within specifications (±10% or as verified) 

### Start time: 09:49:00

09:58:45 t 14 t 15 Cell Temperature (±10F for each run, Not to exceed 20F for test day) and 

## FRESH t 10 t 11 

## Phase t 12 t 13 exceeds ± 2% of mean average) 

### Acce p table "Test Data Phase" t 7

## REt 8 t 9 t 7:00 Mean Avera g e Concentration ( sum of t5:15 throu g h t7:00/8 ) Maximum Deviation (no single reading 

## DATA t 6:00 t 6:15 

## Phase t 6:30 t 6:45 t 5:15 

## TEST t 5:30 t 5:45 

## UP t 3 t 4 t 5 Analyzer Response



## RAMP- t 1 t 2 Calibration Gas Info. 

### (manufacturer,expiration,etc) 

## MEASUREMENT CYCLE ( circle measurement task below ) Source Test (run 3) 210-623-1314
Calibration Gas Verification Information ECOM-AC Analyzer make & model: Covel Gardens Power Station Address 8611 Covel Road San Antonio, Texas 78252 Emission Point: E5 (Unit 5)

# Unit 6Facility

Phone Serial: # 91210CVNXH Gas type O 2 % CO ppm NO ppm NO 2 pp m Concent. 20.9 886 138 47.7

Date: 1/4/2007 Cell Temp Flow Rat e Three O 2 % CO ppm NO ppm NO 2 pp m Phase s 9.4 692.8 14.0 62.3 85.5 2.5 9.4 700.5 13.5 64.8 85.7 2.5 9.4 714.5 13.8 64.8 85.9 2.5 9.3 713.3 13.0 67.0 86.1 2.5 

## Phase 9.4 708.3 12.0 63.0 86.3 2.5 9.4 728.0 12.0 63.0 9.4 724.0 12.0 63.0 9.4 723.0 12.0 63.0 9.3 717.0 12.0 64.0 86.4 2.5 9.3 715.0 12.0 63.0 9.3 713.0 11.0 65.0 9.2 711.0 11.0 66.0 9.2 707.0 11.0 66.0 86.4 2.5
9.3 717.3 11.6 64.1
1.2% 1.5% 5.4% 2.9% Yes or No Yes or No Yes or No Yes or No 9.3 711.5 11.3 65.0 86.4 2.5 9.3 708.0 11.0 65.7 86.3 2.5 9.3 710.8 10.0 65.8 86.3 2.5 12.1 618.0 8.0 53.5 86.3 2.5

Stop Time: Covel Gardens Power Station Address 8611 Covel Road San Antonio, Texas 78252 Emission Point: E6 (Unit 6)

210-623-1314
Calibration Gas Verification Information ECOM-AC Analyzer make & model:
Calibration Gas Info. (manufacturer,expiration,etc) 

## MEASUREMENT CYCLE ( circle measurement task below ) Source Test (run1) Analyzer Response



## RAMP- t 1 t 2 

## UP t 3 t 4 t 5 t 5:15 

## TEST t 5:30 t 5:45 

## DATA t 6:00 t 6:15 

## Phase t 6:30 t 6:45 t 7:00 Mean Avera g e Concentration ( sum of t5:15 throu g h t7:00/8 ) Maximum Deviation (no single reading exceeds ± 2% of mean average) Acce p table "Test Data Phase" t 7

## REt 8 t 9 t 10 t 11 

## Phase t 12 t 13



### Yes or No 

### Yes or No Sam p le Flow Rate are within s p ecifications ( ±10% or as verified ) Start time: 15:53:27

16:02:57 t 14 t 15
Cell Tem p erature ( ±10F for each run, Not to exceed 20F for test da y) and 

## FRESH

### Facility

Phone Serial: # 91210CVN Gas type O 2 

### % CO ppm NO ppm NO 2 

### pp m Concent. 20.9 886 138 47.7

Date: 1/4/2007 

### Cell Temp Flow Rate
Three O 2 

### % CO ppm NO ppm NO 2 

### pp m Phases 9.8 409.0 8.5 55.3 86.1 2.5 9.3 619.0 9.8 70.3 86.3 2.5 9.3 655.3 8.8 68.3 86.3 2.5 9.3 683.0 8.5 67.8 86.2 2.5 

## Phase 9.3 695.0 8.5 68.8 86.4 2.5 9.4 703.0 8.0 67.0 9.5 702.0 8.0 66.0 9.4 704.0 8.0 69.0 9.4 709.0 8.0 67.0 86.5 2.5 9.5 721.0 8.0 64.0 9.4 727.0 8.0 66.0 9.3 716.0 8.0 68.0 9.2 706.0 8.0 70.0 86.4 2.5
9.4 711.0 8.0 67.1
2.0% 2.3% 0.0% 4.7% Yes or No Yes or No Yes or No Yes or No 

### 9.4 717.5 8.0 67.0 86.4 2.5 16.5 471.7 3.7 36.3 86.4 2.5 20.8 27.5 0.3 4.0 86.1 2.5 20.8 11.0 0.0 2.5 85.6 2.5

Stop Time: 

### Yes or No 

### Yes or No Sample Flow Rate are within specifications (±10% or as verified) 

### Start time: 16:03:41

16:13:27 t 14 t 15 Cell Temperature (±10F for each run, Not to exceed 20F for test day) and 

## FRESH t 10 t 11 

## Phase t 12 t 13 exceeds ± 2% of mean average) 

### Acce p table "Test Data Phase" t 7

## REt 8 t 9 t 7:00 Mean Avera g e Concentration ( sum of t5:15 throu g h t7:00/8 ) Maximum Deviation (no single reading 

## DATA t 6:00 t 6:15 

## Phase t 6:30 t 6:45 t 5:15 

## TEST t 5:30 t 5:45 

## UP t 3 t 4 t 5 Analyzer Response



## RAMP- t 1 t 2 

## Source Test (run 2) Calibration Gas Info. 

### (manufacturer,expiration,etc) 

## MEASUREMENT CYCLE ( circle measurement task below ) 210-623-1314
Calibration Gas Verification Information ECOM-AC Analyzer make & model: Covel Gardens Power Station Address 8611 Covel Road San Antonio, Texas 78252 Emission Point: E6 (Unit 6)

### Facility

Phone Serial: # 91210CVN Gas type O 2 

### % CO ppm NO ppm NO 2 

### pp m Concent. 20.9 886 138 47.7

Date: 1/4/2007 

### Cell Temp Flow Rate
Three O 2 

### % CO ppm NO ppm NO 2 

### pp m Phases 9.3 524.8 5.0 63.3 84.9 2.5 9.2 605.3 5.8 67.5 85.1 2.5 9.2 637.5 5.3 66.5 85.3 2.5 9.3 657.8 5.8 66.0 85.5 2.5 

## Phase 9.3 679.3 5.5 66.8 85.6 2.5 9.2 681.0 5.0 70.0 9.2 683.0 5.0 68.0 9.3 689.0 5.0 66.0 9.2 694.0 5.0 70.0 86.0 2.5 9.2 695.0 5.0 70.0 9.2 694.0 5.0 67.0 9.2 692.0 5.0 69.0 9.2 688.0 5.0 71.0 86.1 2.5
9.2 689.5 5.0 68.9
0.9% 1.2% 0.0% 4.2% Yes or No Yes or No Yes or No Yes or No 

### 9.2 692.3 5.0 69.3 86.1 2.5 20.7 162.7 0.7 8.3 86.0 2.5 20.8 16.0 0.0 2.5 85.7 2.5 20.8 9.0 0.0 2.0 85.2 2.5

Stop Time: Covel Gardens Power Station Address 8611 Covel Road San Antonio, Texas 78252 Emission Point: E6 (Unit 6)

210-623-1314
Calibration Gas Verification Information ECOM-AC Analyzer make & model:
Calibration Gas Info. 

### (manufacturer,expiration,etc) 

## MEASUREMENT CYCLE ( circle measurement task below ) Source Test (run 3) Analyzer Response



## RAMP- t 1 t 2 

## UP t 3 t 4 t 5 t 5:15 

## TEST t 5:30 t 5:45 

## DATA t 6:00 t 6:15 

## Phase t 6:30 t 6:45 t 7:00 Mean Avera g e Concentration ( sum of t5:15 throu g h t7:00/8 ) Maximum Deviation (no single reading exceeds ± 2% of mean average) 

### Acce p table "Test Data Phase" t 7

## REt 8 t 9 t 10 t 11 

## Phase t 12 t 13



### Yes or No 

### Yes or No Sample Flow Rate are within specifications (±10% or as verified) 

### Start time: 16:13:59

16:23:44 t 14 t 15 Cell Temperature (±10F for each run, Not to exceed 20F for test day) and 

## FRESH

# APPENDIX B Quality Assurance Activities

### App. B QUALITY ASSURANCE PROGRAM SUMMARY Air Hygiene ensures the quality and validity of its emission measurement and reporting procedures through a rigorous quality assurance (QA) program. The program is developed and admini stered by an internal QA team and encompasses five major areas: 1. QA reviews of reports, laboratory work, and field-testing 2. Equipment calibration and maintenance 3. Chain-of-custody 4. Training 5. Knowledge of current test methods Each of these areas is discussed individually below. 

### QA Reviews Air Hygiene’s review procedure includes review of each source test report, along with laboratory and fieldwork, by the QA Team. The most important review is the one that takes place be fore a test program begins. The QA Team works closely with technical division personnel to prepare and re view test protocols. Test protocol re view includes selection of appropriate test procedures, evaluation of interferences or other restrictions th at might preclude use of standard test procedures, and evaluati on and/or development of alternate procedures. 

### Equipment Calibration and Maintenance The equipment used to conduct the emission m easurements is maintained according to the manufacturer’s instructions to ensure proper operation. In addition to the maintenance program, ca librations are carried out on each measurement device according to the schedule outlined by the Environmental Protection Agency. Quality control checks are also conducted in the field for each test program. 

### Chain-of-Custody Air Hygiene maintains full chain-of-custody documentation on all samples and data sheets. In addition to normal documentation of changes between field sample custodi ans, laboratory personnel, and field test personnel, Air Hygiene documents every individual who handles any test component in the field (e.g., probe wash, impinger loading and recovery, filter loading and recovery, etc.). Samples are stored in a locked area to wh ich only Air Hygiene personnel have access. Field data sheets are secured at Air Hygiene’s offices upon return from the field. 

### Training Personnel’s training is essential to ensure quality testing. Air Hygiene has formal and informal training programs, which include: 1. Attendance at EPA-sponsored training courses 2. Enrollment in EPA correspondence courses 3 A requirement for all technicians to read and understand Air Hygiene’s QA manual 4. In-house training and QA meetings on a regular basis 5. Maintenance of training records 

### Knowledge of Current Test Methods With the constant updating of standard test methods and the wide variety of emerging test procedures, it is essential that any qualified source tester keep abreast of ne w developments. Air Hygiene subscribes to services, which provide updates on EPA reference methods, rules, and regulations. Additionally, source test personnel regularly attend and present papers at testing and emission-related seminars and conferences . Air Hygiene personnel maintain membersh ip in the Air and Waste Management Association and the American I ndustrial Hygiene Association.Facility

Phone

Gas type O 2 % CO ppm NO ppm NO 2 ppm Concent. 886

Date: 1/3/2007 Cell Temp Flow Rate

Three Start time: O 2 % CO ppm NO ppm NO 2 ppm Phases 18.6 657.7 1.0 0.0 77 2.5 7.6 789.3 1.0 0.0 77 2.5 0.0 854.5 1.0 0.0 77 2.5 0.0 886.0 1.0 0.0 77 2.5 

### Phase 0.0 887.0 1.0 0.0 77 2.5 0.0 886.0 1.0 0.0 0.0 886.0 1.0 0.0 0.0 886.0 1.0 0.0 0.0 886.0 1.0 0.0 77 2.5 0.0 887.0 1.0 0.0 0.0 886.0 1.0 0.0 0.0 886.0 1.0 0.0 0.0 886.0 1.0 0.0 77 2.5

0.0 886.1 1.0 0.0

0.0% 0.1% 0.0% 0.0% Yes or No Yes or No Yes or No Yes or No 18.9 762.5 1.0 0.0 77 2.5 21.0 149.6 1.0 0.0 77 2.5 21.0 21.0 2.0 0.0 77 2.5 21.0 6.0 1.0 0.0 77 2.5

Sto p Time:
Sample Flow Rate are within specifications (±10% or as verified) Yes or No Yes or No t 14 t 15
Cell Temperature (±10F for each run, Not to exceed 20F for test day) and 14:44:46 

### FRESH
Phase t 7 t 8 t 9 t 10 t 11 t 12 t 13 Mean Average Concentration (sum of t5:15 through t7:00/8) Maximum Deviation (no single reading exceeds ± 2% of mean average) 

### Phase t 5:30 t 5:45 t 6:00 t 6:15

Acceptable "Test Data Phase"



### REt 4 t 5 t 5:15

t 6:30 t 6:45 t 7:00 

### TEST
DATA UPRAMPREPEATABILITY (CO) 

### MEASUREMENT CYCLE (circle measurement task below)

t 2 t 3 Address Emission Point: C0#1
San Antonio, Texas 78252 210-623-1314 Covel Gardens Power Station 8611 Covel Road

Analyzer make & model:ECOM-AC Serial # 91210CVNXH

Analyzer Response

t 1 14:35:15 Calibration Gas Verification Information Calibration Gas Info. (manufacturer,expiration,etc)Facility

Phone

Gas type O 2 % CO ppm NO ppm NO 2 ppm Concent. 886

Date: 1/3/2007 Cell Temp Flow Rate

Three Start time: O 2 % CO ppm NO ppm NO 2 ppm Phases 18.8 734.1 1.0 0.0 76 2.5 2.1 809.4 1.0 0.0 76 2.5 0.0 886.0 1.0 0.0 76 2.5 0.0 886.0 1.0 0.0 76 2.5 

### Phase 0.0 886.0 1.0 0.0 76 2.5 0.0 886.0 1.0 0.0 0.0 886.0 1.0 0.0 0.0 886.0 1.0 0.0 0.0 886.0 1.0 0.0 76 2.5 0.0 886.0 1.0 0.0 0.0 886.0 1.0 0.0 0.0 886.0 1.0 0.0 0.0 886.0 1.0 0.0 76 2.5

0.0 886.0 1.0 0.0

0.0% 0.0% 0.0% 0.0% Yes or No Yes or No Yes or No Yes or No 18.8 723.6 1.0 0.0 76 2.5 21.0 187.4 1.0 0.0 76 2.5 21.0 18.6 1.0 0.0 76 2.5 21.0 7.0 1.0 0.0 76 2.5

Sto p Time: Analyzer Response

t 1 t 2 t 3 14:47:15 Address Emission Point: CO#2
San Antonio, Texas 78252 210-623-1314 Covel Gardens Power Station 8611 Covel Road



### MEASUREMENT CYCLE (circle measurement task below) Analyzer make & model:ECOM-AC Serial # 91210CVNXH Calibration Gas Verification Information Calibration Gas Info. (manufacturer,expiration,etc)
REPEATABILITY (CO)



### UPRAMPAcceptable "Test Data Phase" t 5:30 t 5:45 t 6:00 t 6:15

Mean Average Concentration (sum of t5:15 through t7:00/8)



### REt 4 t 5 t 5:15

t 6:30 t 6:45 t 7:00 

### TEST
DATA
Phase

Maximum Deviation (no single reading exceeds ± 2% of mean average)



### FRESH
Phase t 7 t 8 t 9 t 10 t 11 t 12 t 13

Sample Flow Rate are within specifications (±10% or as verified) Yes or No Yes or No t 14 t 15
Cell Temperature (±10F for each run, Not to exceed 20F for test day) and 14:56:46Facility

Phone

Gas type O 2 % CO ppm NO ppm NO 2 ppm Concent. 886

Date: 1/3/2007 Cell Temp Flow Rate

Three Start time: O 2 % CO ppm NO ppm NO 2 ppm Phases 8.6 574.3 2.0 0.0 76 2.5 2.1 763.5 2.0 0.0 76 2.5 1.1 844.7 2.0 0.0 76 2.5 0.0 879.5 2.0 0.0 76 2.5 

### Phase 0.0 886.0 1.0 0.0 76 2.5 0.0 886.0 1.0 0.0 0.0 886.0 1.0 0.0 0.0 886.0 1.0 0.0 0.0 886.0 1.0 0.0 75 2.5 0.0 886.0 1.0 0.0 0.0 886.0 1.0 0.0 0.0 886.0 1.0 0.0 0.0 886.0 1.0 0.0 75 2.5

0.0 886.0 1.0 0.0

0.0% 0.0% 0.0% 0.0% Yes or No Yes or No Yes or No Yes or No 18.9 753.0 1.0 0.0 75 2.5 21.0 209.0 1.0 0.0 75 2.5 21.0 21.0 1.0 0.0 75 2.5 21.0 7.0 1.0 0.0 75 2.5

Sto p Time: t 13

Sample Flow Rate are within specifications (±10% or as verified) Yes or No Yes or No t 14 t 15
Cell Temperature (±10F for each run, Not to exceed 20F for test day) and 15:08:46 Maximum Deviation (no single reading exceeds ± 2% of mean average)



### FRESH
Phase t 7 t 8 t 9 t 10 t 11 t 12

### REt 4 t 5 t 5:15

t 6:30 t 6:45 t 7:00 

### TEST
DATA
Phase REPEATABILITY (CO)



### UPRAMPAcceptable "Test Data Phase" t 5:30 t 5:45 t 6:00 t 6:15

Mean Average Concentration (sum of t5:15 through t7:00/8) 

### MEASUREMENT CYCLE (circle measurement task below) Analyzer make & model:ECOM-AC Serial # 91210CVNXH Calibration Gas Verification Information Calibration Gas Info. (manufacturer,expiration,etc) Address Emission Point: CO#3
San Antonio, Texas 78252 210-623-1314 Covel Gardens Power Station 8611 Covel Road

Analyzer Response

t 1 t 2 t 3 14:59:15Facility

Phone

Gas type O 2 % CO ppm NO ppm NO 2 ppm Concent. 886

Date: 1/3/2007 Cell Temp Flow Rate

Three Start time: O 2 % CO ppm NO ppm NO 2 ppm Phases 11.5 569.3 2.0 0.0 76 2.5 0.3 886.0 2.0 0.0 76 2.5 0.0 886.0 2.0 0.0 76 2.5 0.0 886.0 2.0 0.0 76 2.5 

### Phase 0.0 886.0 1.0 0.0 76 2.5 0.0 886.0 1.0 0.0 0.0 886.0 1.0 0.0 0.0 886.0 1.0 0.0 0.0 886.0 1.0 0.0 75 2.5 0.0 886.0 1.0 0.0 0.0 886.0 1.0 0.0 0.0 886.0 1.0 0.0 0.0 886.0 1.0 0.0 75 2.5

0.0 886.0 1.0 0.0

0.0% 0.0% 0.0% 0.0% Yes or No Yes or No Yes or No Yes or No 18.9 265.0 1.0 0.0 75 2.5 20.8 52.2 1.0 0.0 75 2.5 21.0 21.0 1.0 0.0 75 2.5 21.0 7.0 1.0 0.0 75 2.5

Sto p Time: Analyzer Response

t 1 t 2 t 3 15:11:15 Address Emission Point: CO#4
San Antonio, Texas 78252 210-623-1314 Covel Gardens Power Station 8611 Covel Road



### MEASUREMENT CYCLE (circle measurement task below) Analyzer make & model:ECOM-AC Serial # 91210CVNXH Calibration Gas Verification Information Calibration Gas Info. (manufacturer,expiration,etc)
REPEATABILITY (CO)



### UPRAMPAcceptable "Test Data Phase" t 5:30 t 5:45 t 6:00 t 6:15

Mean Average Concentration (sum of t5:15 through t7:00/8)



### REt 4 t 5 t 5:15

t 6:30 t 6:45 t 7:00 

### TEST
DATA
Phase

Maximum Deviation (no single reading exceeds ± 2% of mean average)



### FRESH
Phase t 7 t 8 t 9 t 10 t 11 t 12 t 13

Sample Flow Rate are within specifications (±10% or as verified) Yes or No Yes or No t 14 t 15
Cell Temperature (±10F for each run, Not to exceed 20F for test day) and 15:20:46Facility

Phone

Gas type O 2 % CO ppm NO ppm NO 2 ppm Concent. 138

Date: 1/3/2007 Cell Temp Flow Rate

Three Start time: O 2 % CO ppm NO ppm NO 2 ppm Phases 0.4 0.0 138.0 0.0 76 2.5 0.2 0.0 138.0 0.0 76 2.5 0.3 0.0 138.0 0.0 76 2.5 0.0 0.0 138.0 0.0 76 2.5 

### Phase 0.0 0.0 138.0 0.0 76 2.5 0.0 0.0 138.0 0.0 0.0 0.0 138.0 0.0 0.0 0.0 138.0 0.0 0.0 0.0 138.0 0.0 75 2.5 0.0 0.0 138.0 0.0 0.0 0.0 138.0 0.0 0.0 0.0 138.0 0.0 0.0 0.0 138.0 0.0 75 2.5

0.0 0.0 138.0 0.0

0.0% 0.0% 0.0% 0.0% Yes or No Yes or No Yes or No Yes or No 19.3 0.0 67.4 0.0 75 2.5 20.8 0.0 34.0 0.0 75 2.5 21.0 0.0 15.0 0.0 75 2.5 21.0 0.0 6.0 0.0 75 2.5

Sto p Time: t 13

Sample Flow Rate are within specifications (±10% or as verified) Yes or No Yes or No t 14 t 15
Cell Temperature (±10F for each run, Not to exceed 20F for test day) and 15:34:46 Maximum Deviation (no single reading exceeds ± 2% of mean average)



### FRESH
Phase t 7 t 8 t 9 t 10 t 11 t 12

### REt 4 t 5 t 5:15

t 6:30 t 6:45 t 7:00 

### TEST
DATA
Phase REPEATABILITY (NO)



### UPRAMPAcceptable "Test Data Phase" t 5:30 t 5:45 t 6:00 t 6:15

Mean Average Concentration (sum of t5:15 through t7:00/8) 

### MEASUREMENT CYCLE (circle measurement task below) Analyzer make & model:ECOM-AC Serial # 91210CVNXH Calibration Gas Verification Information Calibration Gas Info. (manufacturer,expiration,etc) Address Emission Point: NO#1
San Antonio, Texas 78252 210-623-1314 Covel Gardens Power Station 8611 Covel Road

Analyzer Response

t 1 t 2 t 3 15:25:15Facility

Phone

Gas type O 2 % CO ppm NO ppm NO 2 ppm Concent. 138

Date: 1/3/2007 Cell Temp Flow Rate

Three Start time: O 2 % CO ppm NO ppm NO 2 ppm Phases 0.8 0.0 122.3 0.0 76 2.5 0.7 0.0 129.7 0.0 76 2.5 0.0 0.0 138.0 0.0 76 2.5 0.0 0.0 138.0 0.0 76 2.5 

### Phase 0.0 0.0 138.0 0.0 76 2.5 0.0 0.0 138.0 0.0 0.0 0.0 138.0 0.0 0.0 0.0 138.0 0.0 0.0 0.0 138.0 0.0 75 2.5 0.0 0.0 138.0 0.0 0.0 0.0 138.0 0.0 0.0 0.0 138.0 0.0 0.0 0.0 138.0 0.0 75 2.5

0.0 0.0 138.0 0.0

0.0% 0.0% 0.0% 0.0% Yes or No Yes or No Yes or No Yes or No 2.4 0.0 127.0 0.0 75 2.5 20.8 0.0 17.0 0.0 75 2.5 21.0 0.0 11.0 0.0 75 2.5 21.0 0.0 7.0 0.0 75 2.5

Sto p Time: Analyzer Response

t 1 t 2 t 3 15:36:15 Address Emission Point: NO#2
San Antonio, Texas 78252 210-623-1314 Covel Gardens Power Station 8611 Covel Road



### MEASUREMENT CYCLE (circle measurement task below) Analyzer make & model:ECOM-AC Serial # 91210CVNXH Calibration Gas Verification Information Calibration Gas Info. (manufacturer,expiration,etc)
REPEATABILITY (NO)



### UPRAMPAcceptable "Test Data Phase" t 5:30 t 5:45 t 6:00 t 6:15

Mean Average Concentration (sum of t5:15 through t7:00/8)



### REt 4 t 5 t 5:15

t 6:30 t 6:45 t 7:00 

### TEST
DATA
Phase

Maximum Deviation (no single reading exceeds ± 2% of mean average)



### FRESH
Phase t 7 t 8 t 9 t 10 t 11 t 12 t 13

Sample Flow Rate are within specifications (±10% or as verified) Yes or No Yes or No t 14 t 15
Cell Temperature (±10F for each run, Not to exceed 20F for test day) and 15:45:46Facility

Phone

Gas type O 2 % CO ppm NO ppm NO 2 ppm Concent. 138

Date: 1/3/2007 Cell Temp Flow Rate

Three Start time: O 2 % CO ppm NO ppm NO 2 ppm Phases 3.7 0.0 84.2 0.0 76 2.5 1.6 0.0 126.3 0.0 76 2.5 0.0 0.0 138.0 0.0 76 2.5 0.0 0.0 138.0 0.0 76 2.5 

### Phase 0.0 0.0 138.0 0.0 76 2.5 0.0 0.0 138.0 0.0 0.0 0.0 138.0 0.0 0.0 0.0 138.0 0.0 0.0 0.0 138.0 0.0 75 2.5 0.0 0.0 138.0 0.0 0.0 0.0 138.0 0.0 0.0 0.0 138.0 0.0 0.0 0.0 138.0 0.0 75 2.5

0.0 0.0 138.0 0.0

0.0% 0.0% 0.0% 0.0% Yes or No Yes or No Yes or No Yes or No 1.7 0.0 124.0 0.0 75 2.5 20.8 0.0 17.0 0.0 75 2.5 21.0 0.0 11.0 0.0 75 2.5 21.0 0.0 6.0 0.0 75 2.5

Sto p Time: t 13

Sample Flow Rate are within specifications (±10% or as verified) Yes or No Yes or No t 14 t 15
Cell Temperature (±10F for each run, Not to exceed 20F for test day) and 15:56:46 Maximum Deviation (no single reading exceeds ± 2% of mean average)



### FRESH
Phase t 7 t 8 t 9 t 10 t 11 t 12

### REt 4 t 5 t 5:15

t 6:30 t 6:45 t 7:00 

### TEST
DATA
Phase REPEATABILITY (NO)



### UPRAMPAcceptable "Test Data Phase" t 5:30 t 5:45 t 6:00 t 6:15

Mean Average Concentration (sum of t5:15 through t7:00/8) 

### MEASUREMENT CYCLE (circle measurement task below) Analyzer make & model:ECOM-AC Serial # 91210CVNXH Calibration Gas Verification Information Calibration Gas Info. (manufacturer,expiration,etc) Address Emission Point: NO#3
San Antonio, Texas 78252 210-623-1314 Covel Gardens Power Station 8611 Covel Road

Analyzer Response

t 1 t 2 t 3 15:47:15Facility

Phone

Gas type O 2 % CO ppm NO ppm NO 2 ppm Concent. 138

Date: 1/3/2007 Cell Temp Flow Rate

Three Start time: O 2 % CO ppm NO ppm NO 2 ppm Phases 3.9 0.0 78.3 0.0 76 2.5 0.8 0.0 137.1 0.0 76 2.5 0.6 0.0 137.9 0.0 76 2.5 0.0 0.0 138.0 0.0 76 2.5 

### Phase 0.0 0.0 138.0 0.0 76 2.5 0.0 0.0 138.0 0.0 0.0 0.0 138.0 0.0 0.0 0.0 138.0 0.0 0.0 0.0 138.0 0.0 75 2.5 0.0 0.0 138.0 0.0 0.0 0.0 138.0 0.0 0.0 0.0 138.0 0.0 0.0 0.0 138.0 0.0 75 2.5

0.0 0.0 138.0 0.0

0.0% 0.0% 0.0% 0.0% Yes or No Yes or No Yes or No Yes or No 0.0 0.0 121.3 0.0 75 2.5 20.8 0.0 18.0 0.0 75 2.5 21.0 0.0 11.0 0.0 75 2.5 21.0 0.0 7.0 0.0 75 2.5

Sto p Time: Analyzer Response

t 1 t 2 t 3 15:58:15 Address Emission Point: NO#4
San Antonio, Texas 78252 210-623-1314 Covel Gardens Power Station 8611 Covel Road



### MEASUREMENT CYCLE (circle measurement task below) Analyzer make & model:ECOM-AC Serial # 91210CVNXH Calibration Gas Verification Information Calibration Gas Info. (manufacturer,expiration,etc)
REPEATABILITY (NO)



### UPRAMPAcceptable "Test Data Phase" t 5:30 t 5:45 t 6:00 t 6:15

Mean Average Concentration (sum of t5:15 through t7:00/8)



### REt 4 t 5 t 5:15

t 6:30 t 6:45 t 7:00 

### TEST
DATA
Phase

Maximum Deviation (no single reading exceeds ± 2% of mean average)



### FRESH
Phase t 7 t 8 t 9 t 10 t 11 t 12 t 13

Sample Flow Rate are within specifications (±10% or as verified) Yes or No Yes or No t 14 t 15
Cell Temperature (±10F for each run, Not to exceed 20F for test day) and 16:07:46Facility

Phone

Gas type O 2 % CO ppm NO ppm NO 2 ppm Concent. 47.7

Date: 1/3/2007 Cell Temp Flow Rate

Three Start time: O 2 % CO ppm NO ppm NO 2 ppm Phases 3.9 0.0 3.0 32.6 76 2.5 1.6 0.0 2.0 45.8 76 2.5 0.0 0.0 1.0 48.0 76 2.5 0.0 0.0 1.0 48.0 76 2.5 

### Phase 0.0 0.0 1.0 48.0 76 2.5 0.0 0.0 1.0 48.0 0.0 0.0 1.0 48.0 0.0 0.0 1.0 48.0 0.0 0.0 1.0 48.0 75 2.5 0.0 0.0 1.0 48.0 0.0 0.0 1.0 48.0 0.0 0.0 1.0 48.0 0.0 0.0 1.0 48.0 75 2.5

0.0 0.0 1.0 48.0

0.0% 0.0% 0.0% 0.0% Yes or No Yes or No Yes or No Yes or No 0.0 0.0 1.0 3.0 75 2.5 20.8 0.0 1.0 3.0 75 2.5 21.0 0.0 1.0 3.0 75 2.5 21.0 0.0 1.0 3.0 75 2.5

Sto p Time: t 13

Sample Flow Rate are within specifications (±10% or as verified) Yes or No Yes or No t 14 t 15
Cell Temperature (±10F for each run, Not to exceed 20F for test day) and 16:19:46 Maximum Deviation (no single reading exceeds ± 2% of mean average)



### FRESH
Phase t 7 t 8 t 9 t 10 t 11 t 12

### REt 4 t 5 t 5:15

t 6:30 t 6:45 t 7:00 

### TEST
DATA
Phase REPEATABILITY (NO 2 )



### UPRAMPAcceptable "Test Data Phase" t 5:30 t 5:45 t 6:00 t 6:15

Mean Average Concentration (sum of t5:15 through t7:00/8) 

### MEASUREMENT CYCLE (circle measurement task below) Analyzer make & model:ECOM-AC Serial # 91210CVNXH Calibration Gas Verification Information Calibration Gas Info. (manufacturer,expiration,etc) Address Emission Point: NO 2 #1
San Antonio, Texas 78252 210-623-1314 Covel Gardens Power Station 8611 Covel Road

Analyzer Response

t 1 t 2 t 3 16:10:15Facility

Phone

Gas type O 2 % CO ppm NO ppm NO 2 ppm Concent. 47.7

Date: 1/3/2007 Cell Temp Flow Rate

Three Start time: O 2 % CO ppm NO ppm NO 2 ppm Phases 3.9 0.0 3.0 31.3 76 2.5 1.6 0.0 2.0 44.3 76 2.5 0.0 0.0 1.0 48.0 76 2.5 0.0 0.0 1.0 48.0 76 2.5 

### Phase 0.0 0.0 1.0 48.0 76 2.5 0.0 0.0 1.0 48.0 0.0 0.0 1.0 48.0 0.0 0.0 1.0 48.0 0.0 0.0 1.0 48.0 75 2.5 0.0 0.0 1.0 48.0 0.0 0.0 1.0 48.0 0.0 0.0 1.0 48.0 0.0 0.0 1.0 48.0 75 2.5

0.0 0.0 1.0 48.0

0.0% 0.0% 0.0% 0.0% Yes or No Yes or No Yes or No Yes or No 0.0 0.0 1.0 3.0 75 2.5 20.8 0.0 1.0 3.0 75 2.5 21.0 0.0 1.0 3.0 75 2.5 21.0 0.0 1.0 3.0 75 2.5

Sto p Time: Analyzer Response

t 1 t 2 t 3 16:20:15 Address Emission Point: NO 2 #2
San Antonio, Texas 78252 210-623-1314 Covel Gardens Power Station 8611 Covel Road



### MEASUREMENT CYCLE (circle measurement task below) Analyzer make & model:ECOM-AC Serial # 91210CVNXH Calibration Gas Verification Information Calibration Gas Info. (manufacturer,expiration,etc)
REPEATABILITY (NO 2 )



### UPRAMPAcceptable "Test Data Phase" t 5:30 t 5:45 t 6:00 t 6:15

Mean Average Concentration (sum of t5:15 through t7:00/8)



### REt 4 t 5 t 5:15

t 6:30 t 6:45 t 7:00 

### TEST
DATA
Phase

Maximum Deviation (no single reading exceeds ± 2% of mean average)



### FRESH
Phase t 7 t 8 t 9 t 10 t 11 t 12 t 13

Sample Flow Rate are within specifications (±10% or as verified) Yes or No Yes or No t 14 t 15
Cell Temperature (±10F for each run, Not to exceed 20F for test day) and 16:29:46Facility

Phone

Gas type O 2 % CO ppm NO ppm NO 2 ppm Concent. 47.7

Date: 1/3/2007 Cell Temp Flow Rate

Three Start time: O 2 % CO ppm NO ppm NO 2 ppm Phases 3.9 0.0 3.0 42.1 76 2.5 1.2 0.0 2.0 45.4 76 2.5 0.0 0.0 2.0 48.0 76 2.5 0.0 0.0 2.0 48.0 76 2.5 

### Phase 0.0 0.0 2.0 48.0 76 2.5 0.0 0.0 2.0 48.0 0.0 0.0 2.0 48.0 0.0 0.0 2.0 48.0 0.0 0.0 2.0 48.0 75 2.5 0.0 0.0 2.0 48.0 0.0 0.0 2.0 48.0 0.0 0.0 2.0 48.0 0.0 0.0 2.0 48.0 75 2.5

0.0 0.0 2.0 48.0

0.0% 0.0% 0.0% 0.0% Yes or No Yes or No Yes or No Yes or No 17.3 0.0 1.0 18.4 75 2.5 20.8 0.0 1.0 2.0 75 2.5 20.9 0.0 1.0 2.0 75 2.5 20.9 0.0 1.0 2.0 75 2.5

Sto p Time: t 13

Sample Flow Rate are within specifications (±10% or as verified) Yes or No Yes or No t 14 t 15
Cell Temperature (±10F for each run, Not to exceed 20F for test day) and 16:41:46 Maximum Deviation (no single reading exceeds ± 2% of mean average)



### FRESH
Phase t 7 t 8 t 9 t 10 t 11 t 12

### REt 4 t 5 t 5:15

t 6:30 t 6:45 t 7:00 

### TEST
DATA
Phase REPEATABILITY (NO 2 )



### UPRAMPAcceptable "Test Data Phase" t 5:30 t 5:45 t 6:00 t 6:15

Mean Average Concentration (sum of t5:15 through t7:00/8) 

### MEASUREMENT CYCLE (circle measurement task below) Analyzer make & model:ECOM-AC Serial # 91210CVNXH Calibration Gas Verification Information Calibration Gas Info. (manufacturer,expiration,etc) Address Emission Point: NO 2 #3
San Antonio, Texas 78252 210-623-1314 Covel Gardens Power Station 8611 Covel Road

Analyzer Response

t 1 t 2 t 3 16:32:15Facility

Phone

Gas type O 2 % CO ppm NO ppm NO 2 ppm Concent. 47.7

Date: 1/3/2007 Cell Temp Flow Rate

Three Start time: O 2 % CO ppm NO ppm NO 2 ppm Phases 0.4 0.0 3.0 47.0 76 2.5 0.2 0.0 2.0 47.0 76 2.5 0.0 0.0 2.0 47.0 76 2.5 0.0 0.0 2.0 47.0 76 2.5 

### Phase 0.0 0.0 2.0 47.0 76 2.5 0.0 0.0 2.0 47.0 0.0 0.0 2.0 47.0 0.0 0.0 2.0 47.0 0.0 0.0 2.0 47.0 75 2.5 0.0 0.0 2.0 47.0 0.0 0.0 2.0 47.0 0.0 0.0 2.0 47.0 0.0 0.0 2.0 47.0 75 2.5

0.0 0.0 2.0 47.0

0.0% 0.0% 0.0% 0.0% Yes or No Yes or No Yes or No Yes or No 0.0 0.0 2.0 3.0 75 2.5 20.8 0.0 1.0 2.0 75 2.5 20.9 0.0 1.0 2.0 75 2.5 20.9 0.0 1.0 1.0 75 2.5

Sto p Time: Analyzer Response

t 1 t 2 t 3 16:44:15 Address Emission Point: NO 2 #4
San Antonio, Texas 78252 210-623-1314 Covel Gardens Power Station 8611 Covel Road



### MEASUREMENT CYCLE (circle measurement task below) Analyzer make & model:ECOM-AC Serial # 91210CVNXH Calibration Gas Verification Information Calibration Gas Info. (manufacturer,expiration,etc)
REPEATABILITY (NO 2 )



### UPRAMPAcceptable "Test Data Phase" t 5:30 t 5:45 t 6:00 t 6:15

Mean Average Concentration (sum of t5:15 through t7:00/8)



### REt 4 t 5 t 5:15

t 6:30 t 6:45 t 7:00 

### TEST
DATA
Phase

Maximum Deviation (no single reading exceeds ± 2% of mean average)



### FRESH
Phase t 7 t 8 t 9 t 10 t 11 t 12 t 13

Sample Flow Rate are within specifications (±10% or as verified) Yes or No Yes or No t 14 t 15
Cell Temperature (±10F for each run, Not to exceed 20F for test day) and 16:53:46Facility

Phone Serial: # 91210CVNXH Gas type O 2 % CO ppm NO ppm NO 2 pp m Concent. 138

Date: 1/4/2007 Cell Temp Flow Rat e Three O 2 % CO ppm NO ppm NO 2 pp m Phase s 0.6 119.8 146.0 2.0 84.5 2.5 0.5 119.0 146.0 2.0 84.4 2.5 0.5 115.3 144.5 2.0 84.6 2.5 0.5 111.8 138.0 2.0 84.8 2.5 

## Phase 0.5 0.0 138.0 2.0 85.1 2.5 0.5 0.0 138.0 2.0 0.5 0.0 138.0 2.0 0.5 0.0 138.0 2.0 0.5 0.0 138.0 2.0 85.2 2.5 0.5 0.0 138.0 2.0 0.5 0.0 138.0 2.0 0.5 0.0 138.0 2.0 0.5 0.0 139.0 2.0 85.1 2.5
0.5 0.0 138.1 2.0
0.0% 0.0% 0.6% 0.0% Yes or No Yes or No Yes or No Yes or No 0.5 0.0 138.3 2.0 85.1 2.5 5.7 0.0 106.0 2.0 85.0 2.5 20.4 0.0 6.0 2.0 85.0 2.5 20.6 0.0 3.0 2.0 85.1 2.5 20.9 0.0 7.1 2.0

Stop Time: Covel Gardens Power Station Address 8611 Covel Road San Antonio, Texas 78252 Emission Point: PRE-CAL

210-623-1314
Calibration Gas Verification Information ECOM-AC Analyzer make & model:
Calibration Gas Info. (manufacturer,expiration,etc) 

## MEASUREMENT CYCLE ( circle measurement task below ) Pre-Cal Test (NO) Analyzer Response



## RAMP- t 1 t 2 

## UP t 3 t 4 t 5 t 5:15 

## TEST t 5:30 t 5:45 

## DATA t 6:00 t 6:15 

## Phase t 6:30 t 6:45 t 7:00 Mean Avera g e Concentration ( sum of t5:15 throu g h t7:00/8 ) Maximum Deviation (no single reading exceeds ± 2% of mean average) Acce p table "Test Data Phase" t 7

## REt 8 t 9 t 10 t 11 

## Phase t 12 t 13



### Yes or No 

### Yes or No Sam p le Flow Rate are within s p ecifications ( ±10% or as verified ) Start time: 07:49:12

07:58:42 t 14 t 15
Cell Tem p erature ( ±10F for each run, Not to exceed 20F for test da y) and 

## FRESH

### Facility

Phone Serial: # 91210CVN Gas type O 2 

### % CO ppm NO ppm NO 2 

### pp m Concent. 886

Date: 1/4/2007 

### Cell Temp Flow Rate
Three O 2 

### % CO ppm NO ppm NO 2 

### pp m Phases 0.1 877.3 0.0 0.0 84.8 2.5 0.1 885.8 0.0 0.0 85.1 2.5 0.2 885.5 7.5 0.5 85.4 2.5 0.2 885.8 5.8 1.0 85.6 2.5 

## Phase 0.1 886.0 3.5 1.0 85.7 2.5 0.1 886.0 2.0 1.0 0.1 886.0 2.0 1.0 0.1 886.0 2.0 1.0 0.1 886.0 2.0 1.0 85.8 2.5 0.1 886.0 4.0 1.0 0.6 886.0 4.0 1.0 0.2 886.0 4.0 1.0 0.2 886.0 5.0 1.0 86.0 2.5
0.2 886.0 3.1 1.0
220.0% 0.0% 60.0% 0.0% Yes or No Yes or No Yes or No Yes or No 

### 0.3 886.0 4.3 1.0 86.0 2.5 20.6 31.7 4.3 1.0 86.1 2.5 21.0 4.8 1.8 1.0 86.2 2.5 20.4 3.0 1.0 1.0 86.4 2.5

Stop Time: 

### Yes or No 

### Yes or No Sample Flow Rate are within specifications (±10% or as verified) 

### Start time:

8:10:11

8:19:56 t 14 t 15 Cell Temperature (±10F for each run, Not to exceed 20F for test day) and 

## FRESH t 10 t 11 

## Phase t 12 t 13 exceeds ± 2% of mean average) 

### Acce p table "Test Data Phase" t 7

## REt 8 t 9 t 7:00 Mean Avera g e Concentration ( sum of t5:15 throu g h t7:00/8 ) Maximum Deviation (no single reading 

## DATA t 6:00 t 6:15 

## Phase t 6:30 t 6:45 t 5:15 

## TEST t 5:30 t 5:45 

## UP t 3 t 4 t 5 Analyzer Response



## RAMP- t 1 t 2 

## Pre-Cal (CO) Calibration Gas Info. 

### (manufacturer,expiration,etc) 

## MEASUREMENT CYCLE ( circle measurement task below ) 210-623-1314
Calibration Gas Verification Information ECOM-AC Analyzer make & model: Covel Gardens Power Station Address 8611 Covel Road San Antonio, Texas 78252 Emission Point: PRE-CAL

### Facility

Phone Serial: # 91210CVN Gas type O 2 

### % CO ppm NO ppm NO 2 

### pp m Concent. 47.7

Date: 1/4/2007 

### Cell Temp Flow Rate
Three O 2 

### % CO ppm NO ppm NO 2 

### pp m Phases 20.8 4.0 0.0 0.0 84.9 2.5 20.8 4.0 0.0 0.0 85.2 2.5 7.0 4.0 9.5 45.8 85.4 2.5 0.2 4.0 4.8 114.8 85.6 2.5 

## Phase 0.1 4.0 3.0 49.8 85.8 2.5 0.1 4.0 2.0 48.0 0.1 4.0 2.0 48.0 0.1 4.0 2.0 48.0 0.1 4.0 4.0 48.0 85.9 2.5 0.6 4.0 2.0 48.0 0.2 4.0 2.0 48.0 0.2 4.0 2.0 48.0 0.1 4.0 2.0 48.0 86.0 2.5
0.2 4.0 2.3 48.0
220.0% 0.0% 0.0% 0.0% Yes or No Yes or No Yes or No Yes or No 

### 0.3 4.0 2.0 48.0 86.0 2.5 21.0 4.0 3.3 1.7 86.2 2.5 21.0 3.8 1.5 1.0 86.2 2.5 20.3 3.6 1.2 1.0 86.4 2.5

Stop Time: Covel Gardens Power Station Address 8611 Covel Road San Antonio, Texas 78252 Emission Point: PRE-CAL

210-623-1314
Calibration Gas Verification Information ECOM-AC Analyzer make & model:
Calibration Gas Info. 

### (manufacturer,expiration,etc) 

## MEASUREMENT CYCLE ( circle measurement task below ) Pre-Cal (NO2) Analyzer Response



## RAMP- t 1 t 2 

## UP t 3 t 4 t 5 t 5:15 

## TEST t 5:30 t 5:45 

## DATA t 6:00 t 6:15 

## Phase t 6:30 t 6:45 t 7:00 Mean Avera g e Concentration ( sum of t5:15 throu g h t7:00/8 ) Maximum Deviation (no single reading exceeds ± 2% of mean average) 

### Acce p table "Test Data Phase" t 7

## REt 8 t 9 t 10 t 11 

## Phase t 12 t 13



### Yes or No 

### Yes or No Sample Flow Rate are within specifications (±10% or as verified) 

### Start time: 07:59:10

08:08:56 t 14 t 15 Cell Temperature (±10F for each run, Not to exceed 20F for test day) and 

## FRESHFacility

Phone Serial: # 91210CVNXH Gas type O 2 % CO ppm NO ppm NO 2 pp m Concent. 138

Date: 1/4/2007 Cell Temp Flow Rat e Three O 2 % CO ppm NO ppm NO 2 pp m Phase s 0.0 4.0 117.0 0.0 87 2.5 0.0 3.0 138.0 0.0 86.9 2.5 0.0 2.0 138.0 0.0 86.8 2.5 0.0 2.0 139.0 0.0 86.7 2.5 

## Phase 0.0 2.0 139.0 0.0 86.6 2.5 0.0 1.0 138.0 0.0 0.0 1.0 138.0 0.0 0.0 1.0 138.0 0.0 0.0 1.0 138.0 0.0 86.5 2.5 0.0 1.0 138.0 0.0 0.0 1.0 138.0 0.0 0.0 1.0 138.0 0.0 0.0 1.0 138.0 0.0 86.4 2.5
0.0 1.0 138.0 0.0
0.0% 0.0% 0.0% 0.0% Yes or No Yes or No Yes or No Yes or No 3.6 1.0 103.0 0.0 86.4 2.5 13.7 1.0 67.0 0.0 86.4 2.5 20.9 1.0 11.0 0.0 86.4 2.5 21.0 1.0 1.3 0.0 86.4 2.5

Stop Time: Covel Gardens Power Station Address 8611 Covel Road San Antonio, Texas 78252 Emission Point: E1 (Unit 1)

210-623-1314
Calibration Gas Verification Information ECOM-AC Analyzer make & model:
Calibration Gas Info. (manufacturer,expiration,etc) 

## MEASUREMENT CYCLE ( circle measurement task below ) Post-Cal Test (NO) Analyzer Response



## RAMP- t 1 t 2 

## UP t 3 t 4 t 5 t 5:15 

## TEST t 5:30 t 5:45 

## DATA t 6:00 t 6:15 

## Phase t 6:30 t 6:45 t 7:00 Mean Avera g e Concentration ( sum of t5:15 throu g h t7:00/8 ) Maximum Deviation (no single reading exceeds ± 2% of mean average) Acce p table "Test Data Phase" t 7

## REt 8 t 9 t 10 t 11 

## Phase t 12 t 13



### Yes or No 

### Yes or No Sam p le Flow Rate are within s p ecifications ( ±10% or as verified ) Start time: 17:42:47

17:52:17 t 14 t 15
Cell Tem p erature ( ±10F for each run, Not to exceed 20F for test da y) and 

## FRESH

### Facility

Phone Serial: # 91210CVN Gas type O 2 

### % CO ppm NO ppm NO 2 

### pp m Concent. 886

Date: 1/4/2007 

### Cell Temp Flow Rate
Three O 2 

### % CO ppm NO ppm NO 2 

### pp m Phases 4.6 445.7 1.8 1.0 86.2 2.5 0.1 902.3 1.0 1.0 86.2 2.5 0.1 900.8 1.0 1.0 86.2 2.5 0.1 889.3 1.0 1.0 86.2 2.5 

## Phase 0.1 890.5 1.0 1.0 86.2 2.5 0.1 886.0 1.0 1.0 0.1 885.0 1.0 1.0 0.1 884.0 1.0 1.0 0.1 883.0 1.0 1.0 86.8 2.5 0.1 881.0 1.0 1.0 0.1 880.0 1.0 1.0 0.0 886.0 1.0 1.0 0.0 884.0 1.0 1.0 86.7 2.5
0.1 883.6 1.0 1.0
100.0% 0.4% 0.0% 0.0% Yes or No Yes or No Yes or No Yes or No 

### 0.1 882.8 1.0 1.0 86.7 2.5 13.5 547.7 1.0 1.0 86.7 2.5 20.8 33.0 1.3 1.0 86.7 2.5 20.8 13.0 1.3 1.0 86.7 2.5

Stop Time: 

### Yes or No 

### Yes or No Sample Flow Rate are within specifications (±10% or as verified) 

### Start time:

18:04:11

18:13:56 t 14 t 15 Cell Temperature (±10F for each run, Not to exceed 20F for test day) and 

## FRESH t 10 t 11 

## Phase t 12 t 13 exceeds ± 2% of mean average) 

### Acce p table "Test Data Phase" t 7

## REt 8 t 9 t 7:00 Mean Avera g e Concentration ( sum of t5:15 throu g h t7:00/8 ) Maximum Deviation (no single reading 

## DATA t 6:00 t 6:15 

## Phase t 6:30 t 6:45 t 5:15 

## TEST t 5:30 t 5:45 

## UP t 3 t 4 t 5 Analyzer Response



## RAMP- t 1 t 2 

## Post-Cal Test (CO) Calibration Gas Info. 

### (manufacturer,expiration,etc) 

## MEASUREMENT CYCLE ( circle measurement task below ) 210-623-1314
Calibration Gas Verification Information ECOM-AC Analyzer make & model: Covel Gardens Power Station Address 8611 Covel Road San Antonio, Texas 78252 Emission Point: E1 (Unit 1)

### Facility

Phone Serial: # 91210CVN Gas type O 2 

### % CO ppm NO ppm NO 2 

### pp m Concent. 47.7

Date: 1/4/2007 

### Cell Temp Flow Rate
Three O 2 

### % CO ppm NO ppm NO 2 

### pp m Phases 21.0 4.0 0.0 0.0 87.2 2.5 1.7 4.0 1.8 17.0 87.2 2.5 0.0 4.0 3.5 35.0 87.2 2.5 0.0 4.0 2.0 40.3 87.2 2.5 

## Phase 0.0 4.0 2.0 48.0 87.2 2.5 0.0 3.0 2.0 49.0 0.0 3.0 2.0 49.0 0.0 3.0 2.0 49.0 0.0 3.0 2.0 50.0 87.2 2.5 0.0 3.0 2.0 50.0 0.0 3.0 2.0 50.0 0.0 3.0 2.0 50.0 0.0 3.0 2.0 51.0 87.2 2.5
0.0 3.0 2.0 49.8
0.0% 0.0% 0.0% 2.5% Yes or No Yes or No Yes or No Yes or No 

### 0.0 3.0 2.0 50.3 87.2 2.5 20.6 3.0 1.3 7.3 87.2 2.5 21.0 3.0 0.0 0.5 87.2 2.5 21.0 3.0 0.0 0.0 87.2 2.5

Stop Time: Covel Gardens Power Station Address 8611 Covel Road San Antonio, Texas 78252 Emission Point: E1 (Unit 1)

210-623-1314
Calibration Gas Verification Information ECOM-AC Analyzer make & model:
Calibration Gas Info. 

### (manufacturer,expiration,etc) 

## MEASUREMENT CYCLE ( circle measurement task below ) Post-Cal (NO2) Analyzer Response



## RAMP- t 1 t 2 

## UP t 3 t 4 t 5 t 5:15 

## TEST t 5:30 t 5:45 

## DATA t 6:00 t 6:15 

## Phase t 6:30 t 6:45 t 7:00 Mean Avera g e Concentration ( sum of t5:15 throu g h t7:00/8 ) Maximum Deviation (no single reading exceeds ± 2% of mean average) 

### Acce p table "Test Data Phase" t 7

## REt 8 t 9 t 10 t 11 

## Phase t 12 t 13



### Yes or No 

### Yes or No Sample Flow Rate are within specifications (±10% or as verified) 

### Start time: 17:53:06

18:02:36 t 14 t 15 Cell Temperature (±10F for each run, Not to exceed 20F for test day) and 

## FRESH

# APPENDIX C Calibration Gas CertificationsCC113410 (886).max