13339 Hagers Ferry Road

Huntersville, NC 28078-7929

McGuire Nuclear Complex - MG03A2

Phone: 980-875-5245 Fax: 980-875-4349

Order Summary Report

Analytical Laboratory

Order Number: J12080348

Project Name: Somerville Waste Water

Lab Contact: Jason C Perkins

Date: 9/6/2012

Customer Address: 253 Plant Allen Road

Customer Name(s): BILL KENNEDY, RON LAWS, ALLEN STOWE, ROBBIN JOLLY, DON SCRUG

Belmont, NC 28012

Phone: 980-875-5348

Report Authorized By:

(Signature)

Program Comments:

Please contact the Program Manager (Jason C Perkins) with any questions regarding this report.

Data Flags & Calculations:

Any analytical tests or individual analytes within a test flagged with a Qualifier indicate a deviation from the method quality 

system or quality control requirement. The qualifier description is found at the end of the Certificate of Analysis (sample results) 

under the qualifiers heading. All results are reported on a dry weight basis unless otherwise noted. Subcontracted data 

included on the Duke Certificate of Analysis is to be used as information only. Certified vendor results can be found in the 

subcontracted lab final report. Duke Energy Analytical Laboratory subcontracts analyses to other vendor laboratories that have 

been qualified by Duke Energy to perform these analyses except where noted.

Data Package:

This data package includes analytical results that are applicable only to the samples described in this narrative. An estimation of 

the uncertainty of measurement for the results in the report is available upon request. This report shall not be reproduced, except 

in full, without the written consent of the Analytical Laboratory. Please contact the Analytical laboratory with any questions. The 

order of individual sections within this report is as follows:

Job Summary Report, Sample Identification, Technical Validation of Data Package, Analytical Laboratory Certificate of Analysis, 

Analytical Laboratory QC Reports, Sub-contracted Laboratory Results, Customer Specific Data Sheets, Reports & 

Documentation, Customer Database Entries, Test Case Narratives, Chain of Custody (COC)

Certification:

The Analytical Laboratory holds the following State Certifications : North Carolina (DENR) Certificate #248, South Carolina 

(DHEC) Laboratory ID # 99005. Contact the Analytical Laboratory for definitive information about the certification status of 

specific methods. 

Page 1 of 39
Sample ID's & Descriptions:

Sample ID Plant/Station

Collection 

Date and Time Collected By Sample Description

2012018172 ALLEN 17-Aug-12 8:01 AM Chris Williams FGD Purge Eff 

2012018173 ALLEN 17-Aug-12 8:34 AM Chris Williams EQ Tank Eff 

2012018174 ALLEN 17-Aug-12 8:37 AM Chris Williams BioReactor 1 Inf 

2012018175 ALLEN 17-Aug-12 8:37 AM Chris Williams BioReactor 1 Inf BLANK 

2012018176 ALLEN 17-Aug-12 8:51 AM Chris Williams BioReactor 2 Inf 

2012018177 ALLEN 17-Aug-12 8:51 AM Chris Williams BioReactor 2 Inf BLANK 

2012018178 ALLEN 17-Aug-12 8:45 AM Chris Williams BioReactor 2 Eff 

2012018179 ALLEN 17-Aug-12 8:45 AM Chris Williams BioReactor 2 Eff BLANK 

2012018180 ALLEN 17-Aug-12 9:01 AM Chris Williams Filter Blk 

9 Total Samples

Page 2 of 39
COC and .pdf report are in agreement with sample totals 

and analyses (compliance programs and procedures).

All Results are less than the laboratory reporting limits.

All laboratory QA/QC requirements are acceptable.

Yes No

Technical Validation Review

Checklist:

Yes No

Yes No

Report Sections Included:

Job Summary Report Sub-contracted Laboratory Results

Sample Identification Customer Specific Data Sheets, Reports, & Documentation

Technical Validation of Data Package Customer Database Entries

Analytical Laboratory Certificate of Analysis

Analytical Laboratory QC Report

Chain of Custody

Reviewed By: DataBase Administrator Date: 9/6/2012

Electronic Data Deliverable (EDD) Sent Separately

Page 3 of 39
Certificate of Laboratory Analysis

This report shall not be reproduced, except in full.

Order # J12080348

2012018172

Collection Date: 17-Aug-12 8:01 AM

Site: FGD Purge Eff 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

DF Analyst

ALKALINITY - (Analysis Performed by Prism Labs)

Vendor Parameter Complete Vendor Method V_PRISM

INORGANIC IONS BY IC

Bromide 11 mg/L 5 50 EPA 300.0 8/22/2012 12:09:00 PM JAHERMA

Chloride 2300 mg/L 50 500 EPA 300.0 8/22/2012 12:09:00 PM JAHERMA

Sulfate 1900 mg/L 50 500 EPA 300.0 8/22/2012 12:09:00 PM JAHERMA

MERCURY (COLD VAPOR) IN WATER

Mercury (Hg) 54.3 ug/L 2.5 50 EPA 245.1 8/23/2012 12:59:25 PM AGIBBS

DISSOLVED METALS BY ICP

Manganese (Mn) < 0.05 mg/L 0.05 10 EPA 200.7 8/28/2012 2:49:00 PM MHH7131

TOTAL RECOVERABLE METALS BY ICP

Boron (B) 191 mg/L 0.5 10 EPA 200.7 8/28/2012 11:48:00 AM MHH7131

Calcium (Ca) 1830 mg/L 0.1 10 EPA 200.7 8/28/2012 11:48:00 AM MHH7131

Iron (Fe) 98.1 mg/L 0.1 10 EPA 200.7 8/28/2012 11:48:00 AM MHH7131

Lithium (Li) 0.146 mg/L 0.05 10 EPA 200.7 8/28/2012 11:48:00 AM MHH7131

Magnesium (Mg) 732 mg/L 0.05 10 EPA 200.7 8/28/2012 11:48:00 AM MHH7131

Manganese (Mn) 2.80 mg/L 0.05 10 EPA 200.7 8/28/2012 11:48:00 AM MHH7131

Potassium (K) 37.1 mg/L 1 10 EPA 200.7 8/28/2012 11:48:00 AM MHH7131

Sodium (Na) 34.3 mg/L 0.5 10 EPA 200.7 8/28/2012 11:48:00 AM MHH7131

DISSOLVED METALS BY ICP-MS

Selenium (Se) 898 ug/L 10 10 EPA 200.8 8/31/2012 1:27:00 PM KRICHAR

TOTAL RECOVERABLE METALS BY ICP-MS

Arsenic (As) 140 ug/L 10 10 EPA 200.8 8/29/2012 12:58:00 PM KRICHAR

Cadmium (Cd) < 10 ug/L 10 10 EPA 200.8 8/29/2012 12:58:00 PM KRICHAR

Chromium (Cr) 200 ug/L 10 10 EPA 200.8 8/29/2012 12:58:00 PM KRICHAR

Copper (Cu) 132 ug/L 10 10 EPA 200.8 8/29/2012 12:58:00 PM KRICHAR

Nickel (Ni) 203 ug/L 10 10 EPA 200.8 8/29/2012 12:58:00 PM KRICHAR

Selenium (Se) 951 ug/L 10 10 EPA 200.8 8/29/2012 12:58:00 PM KRICHAR

Silver (Ag) < 10 ug/L 10 10 EPA 200.8 8/29/2012 12:58:00 PM KRICHAR

Zinc (Zn) 242 ug/L 10 10 EPA 200.8 8/29/2012 12:58:00 PM KRICHAR

SELENIUM SPECIATION - (Analysis Performed by Applied Speciation and Consulting, LLC)

Vendor Parameter Complete Vendor Method V_AS&C

TOTAL DISSOLVED SOLIDS

TDS 11000 mg/L 200 1 SM2540C 8/22/2012 4:00:00 PM TJA7067

TOTAL SUSPENDED SOLIDS

TSS 3000 mg/L 250 1 SM2540D 8/23/2012 10:08:00 AM TJA7067

Page 4 of 39
Certificate of Laboratory Analysis

This report shall not be reproduced, except in full.

Order # J12080348

2012018172

Collection Date: 17-Aug-12 8:01 AM

Site: FGD Purge Eff 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

DF Analyst

2012018173

Collection Date: 17-Aug-12 8:34 AM

Site: EQ Tank Eff 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

DF Analyst

MERCURY (COLD VAPOR) IN WATER

Mercury (Hg) 40.2 ug/L 2.5 50 EPA 245.1 8/23/2012 1:01:48 PM AGIBBS

DISSOLVED METALS BY ICP

Manganese (Mn) < 0.05 mg/L 0.05 10 EPA 200.7 8/28/2012 2:53:00 PM MHH7131

TOTAL RECOVERABLE METALS BY ICP

Boron (B) 191 mg/L 0.5 10 EPA 200.7 8/28/2012 11:52:00 AM MHH7131

Calcium (Ca) 1640 mg/L 0.1 10 EPA 200.7 8/28/2012 11:52:00 AM MHH7131

Iron (Fe) 74.3 mg/L 0.1 10 EPA 200.7 8/28/2012 11:52:00 AM MHH7131

Lithium (Li) 0.132 mg/L 0.05 10 EPA 200.7 8/28/2012 11:52:00 AM MHH7131

Magnesium (Mg) 689 mg/L 0.05 10 EPA 200.7 8/28/2012 11:52:00 AM MHH7131

Manganese (Mn) 2.15 mg/L 0.05 10 EPA 200.7 8/28/2012 11:52:00 AM MHH7131

Potassium (K) 32.9 mg/L 1 10 EPA 200.7 8/28/2012 11:52:00 AM MHH7131

Sodium (Na) 41.8 mg/L 0.5 10 EPA 200.7 8/28/2012 11:52:00 AM MHH7131

DISSOLVED METALS BY ICP-MS

Selenium (Se) 755 ug/L 10 10 EPA 200.8 8/31/2012 1:30:00 PM KRICHAR

TOTAL RECOVERABLE METALS BY ICP-MS

Arsenic (As) 111 ug/L 10 10 EPA 200.8 8/29/2012 1:01:00 PM KRICHAR

Cadmium (Cd) < 10 ug/L 10 10 EPA 200.8 8/29/2012 1:01:00 PM KRICHAR

Chromium (Cr) 164 ug/L 10 10 EPA 200.8 8/29/2012 1:01:00 PM KRICHAR

Copper (Cu) 107 ug/L 10 10 EPA 200.8 8/29/2012 1:01:00 PM KRICHAR

Nickel (Ni) 175 ug/L 10 10 EPA 200.8 8/29/2012 1:01:00 PM KRICHAR

Selenium (Se) 864 ug/L 10 10 EPA 200.8 8/29/2012 1:01:00 PM KRICHAR

Silver (Ag) < 10 ug/L 10 10 EPA 200.8 8/29/2012 1:01:00 PM KRICHAR

Zinc (Zn) 199 ug/L 10 10 EPA 200.8 8/29/2012 1:01:00 PM KRICHAR

2012018174

Collection Date: 17-Aug-12 8:37 AM

Site: BioReactor 1 Inf 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

DF Analyst

ALKALINITY - (Analysis Performed by Prism Labs)

Vendor Parameter Complete Vendor Method V_PRISM

Page 5 of 39
Certificate of Laboratory Analysis

This report shall not be reproduced, except in full.

Order # J12080348

2012018174

Collection Date: 17-Aug-12 8:37 AM

Site: BioReactor 1 Inf 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

DF Analyst

MERCURY 1631 - (Analysis Performed by Brooks Rand Labs LLC)

Vendor Parameter Complete Vendor Method V_BRAND

MERCURY 1631 - DISSOLVED - (Analysis Performed by Brooks Rand Labs LLC)

Vendor Parameter Complete Vendor Method V_BRAND

DISSOLVED METALS BY ICP

Manganese (Mn) < 0.05 mg/L 0.05 10 EPA 200.7 8/28/2012 2:57:00 PM MHH7131

TOTAL RECOVERABLE METALS BY ICP

Boron (B) 183 mg/L 0.5 10 EPA 200.7 8/28/2012 11:56:00 AM MHH7131

Calcium (Ca) 1520 mg/L 0.1 10 EPA 200.7 8/28/2012 11:56:00 AM MHH7131

Iron (Fe) < 0.1 mg/L 0.1 10 EPA 200.7 8/28/2012 11:56:00 AM MHH7131

Lithium (Li) 0.062 mg/L 0.05 10 EPA 200.7 8/28/2012 11:56:00 AM MHH7131

Magnesium (Mg) 535 mg/L 0.05 10 EPA 200.7 8/28/2012 11:56:00 AM MHH7131

Manganese (Mn) < 0.05 mg/L 0.05 10 EPA 200.7 8/28/2012 11:56:00 AM MHH7131

Potassium (K) 17.5 mg/L 1 10 EPA 200.7 8/28/2012 11:56:00 AM MHH7131

Sodium (Na) 73.2 mg/L 0.5 10 EPA 200.7 8/28/2012 11:56:00 AM MHH7131

DISSOLVED METALS BY ICP-MS

Selenium (Se) 691 ug/L 10 10 EPA 200.8 8/31/2012 1:34:00 PM KRICHAR

TOTAL RECOVERABLE METALS BY ICP-MS

Arsenic (As) < 10 ug/L 10 10 EPA 200.8 8/29/2012 1:04:00 PM KRICHAR

Cadmium (Cd) < 10 ug/L 10 10 EPA 200.8 8/29/2012 1:04:00 PM KRICHAR

Chromium (Cr) 13.6 ug/L 10 10 EPA 200.8 8/29/2012 1:04:00 PM KRICHAR

Copper (Cu) < 10 ug/L 10 10 EPA 200.8 8/29/2012 1:04:00 PM KRICHAR

Nickel (Ni) < 10 ug/L 10 10 EPA 200.8 8/29/2012 1:04:00 PM KRICHAR

Selenium (Se) 741 ug/L 10 10 EPA 200.8 8/29/2012 1:04:00 PM KRICHAR

Silver (Ag) < 10 ug/L 10 10 EPA 200.8 8/29/2012 1:04:00 PM KRICHAR

Zinc (Zn) < 10 ug/L 10 10 EPA 200.8 8/29/2012 1:04:00 PM KRICHAR

SELENIUM SPECIATION - (Analysis Performed by Applied Speciation and Consulting, LLC)

Vendor Parameter Complete Vendor Method V_AS&C

2012018175

Collection Date: 17-Aug-12 8:37 AM

Site: BioReactor 1 Inf BLANK 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

DF Analyst

MERCURY 1631 - (Analysis Performed by Brooks Rand Labs LLC)

Vendor Parameter Complete Vendor Method V_BRAND

Page 6 of 39
Certificate of Laboratory Analysis

This report shall not be reproduced, except in full.

Order # J12080348

2012018175

Collection Date: 17-Aug-12 8:37 AM

Site: BioReactor 1 Inf BLANK 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

DF Analyst

MERCURY 1631 - DISSOLVED - (Analysis Performed by Brooks Rand Labs LLC)

Vendor Parameter Complete Vendor Method V_BRAND

2012018176

Collection Date: 17-Aug-12 8:51 AM

Site: BioReactor 2 Inf 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

DF Analyst

ALKALINITY - (Analysis Performed by Prism Labs)

Vendor Parameter Complete Vendor Method V_PRISM

MERCURY 1631 - (Analysis Performed by Brooks Rand Labs LLC)

Vendor Parameter Complete Vendor Method V_BRAND

MERCURY 1631 - DISSOLVED - (Analysis Performed by Brooks Rand Labs LLC)

Vendor Parameter Complete Vendor Method V_BRAND

DISSOLVED METALS BY ICP

Manganese (Mn) < 0.05 mg/L 0.05 10 EPA 200.7 8/28/2012 3:01:00 PM MHH7131

TOTAL RECOVERABLE METALS BY ICP

Boron (B) 193 mg/L 0.5 10 EPA 200.7 8/28/2012 11:59:00 AM MHH7131

Calcium (Ca) 1590 mg/L 0.1 10 EPA 200.7 8/28/2012 11:59:00 AM MHH7131

Iron (Fe) < 0.1 mg/L 0.1 10 EPA 200.7 8/28/2012 11:59:00 AM MHH7131

Lithium (Li) 0.065 mg/L 0.05 10 EPA 200.7 8/28/2012 11:59:00 AM MHH7131

Magnesium (Mg) 552 mg/L 0.05 10 EPA 200.7 8/28/2012 11:59:00 AM MHH7131

Manganese (Mn) < 0.05 mg/L 0.05 10 EPA 200.7 8/28/2012 11:59:00 AM MHH7131

Potassium (K) 31.1 mg/L 1 10 EPA 200.7 8/28/2012 11:59:00 AM MHH7131

Sodium (Na) 74.1 mg/L 0.5 10 EPA 200.7 8/28/2012 11:59:00 AM MHH7131

DISSOLVED METALS BY ICP-MS

Selenium (Se) 77.3 ug/L 10 10 EPA 200.8 8/31/2012 1:38:00 PM KRICHAR

TOTAL RECOVERABLE METALS BY ICP-MS

Arsenic (As) < 10 ug/L 10 10 EPA 200.8 8/29/2012 1:07:00 PM KRICHAR

Cadmium (Cd) < 10 ug/L 10 10 EPA 200.8 8/29/2012 1:07:00 PM KRICHAR

Chromium (Cr) < 10 ug/L 10 10 EPA 200.8 8/29/2012 1:07:00 PM KRICHAR

Copper (Cu) < 10 ug/L 10 10 EPA 200.8 8/29/2012 1:07:00 PM KRICHAR

Nickel (Ni) < 10 ug/L 10 10 EPA 200.8 8/29/2012 1:07:00 PM KRICHAR

Selenium (Se) 118 ug/L 10 10 EPA 200.8 8/29/2012 1:07:00 PM KRICHAR

Silver (Ag) < 10 ug/L 10 10 EPA 200.8 8/29/2012 1:07:00 PM KRICHAR

Zinc (Zn) < 10 ug/L 10 10 EPA 200.8 8/29/2012 1:07:00 PM KRICHAR

Page 7 of 39
Certificate of Laboratory Analysis

This report shall not be reproduced, except in full.

Order # J12080348

2012018176

Collection Date: 17-Aug-12 8:51 AM

Site: BioReactor 2 Inf 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

DF Analyst

SELENIUM SPECIATION - (Analysis Performed by Applied Speciation and Consulting, LLC)

Vendor Parameter Complete Vendor Method V_AS&C

2012018177

Collection Date: 17-Aug-12 8:51 AM

Site: BioReactor 2 Inf BLANK 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

DF Analyst

MERCURY 1631 - (Analysis Performed by Brooks Rand Labs LLC)

Vendor Parameter Complete Vendor Method V_BRAND

MERCURY 1631 - DISSOLVED - (Analysis Performed by Brooks Rand Labs LLC)

Vendor Parameter Complete Vendor Method V_BRAND

2012018178

Collection Date: 17-Aug-12 8:45 AM

Site: BioReactor 2 Eff 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

DF Analyst

ALKALINITY - (Analysis Performed by Prism Labs)

Vendor Parameter Complete Vendor Method V_PRISM

INORGANIC IONS BY IC

Bromide 10 mg/L 5 50 EPA 300.0 8/22/2012 12:27:00 PM JAHERMA

Chloride 2700 mg/L 50 500 EPA 300.0 8/22/2012 12:27:00 PM JAHERMA

Sulfate 2000 mg/L 50 500 EPA 300.0 8/22/2012 12:27:00 PM JAHERMA

MERCURY 1631 - (Analysis Performed by Brooks Rand Labs LLC)

Vendor Parameter Complete Vendor Method V_BRAND

MERCURY 1631 - DISSOLVED - (Analysis Performed by Brooks Rand Labs LLC)

Vendor Parameter Complete Vendor Method V_BRAND

DISSOLVED METALS BY ICP

Manganese (Mn) < 0.05 mg/L 0.05 10 EPA 200.7 8/28/2012 3:05:00 PM MHH7131

Page 8 of 39
Certificate of Laboratory Analysis

This report shall not be reproduced, except in full.

Order # J12080348

2012018178

Collection Date: 17-Aug-12 8:45 AM

Site: BioReactor 2 Eff 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

DF Analyst

TOTAL RECOVERABLE METALS BY ICP

Boron (B) 205 mg/L 0.5 10 EPA 200.7 8/28/2012 12:03:00 PM MHH7131

Calcium (Ca) 1480 mg/L 0.1 10 EPA 200.7 8/28/2012 12:03:00 PM MHH7131

Iron (Fe) 0.483 mg/L 0.1 10 EPA 200.7 8/28/2012 12:03:00 PM MHH7131

Lithium (Li) 0.065 mg/L 0.05 10 EPA 200.7 8/28/2012 12:03:00 PM MHH7131

Magnesium (Mg) 555 mg/L 0.05 10 EPA 200.7 8/28/2012 12:03:00 PM MHH7131

Manganese (Mn) < 0.05 mg/L 0.05 10 EPA 200.7 8/28/2012 12:03:00 PM MHH7131

Potassium (K) 28.6 mg/L 1 10 EPA 200.7 8/28/2012 12:03:00 PM MHH7131

Sodium (Na) 68.9 mg/L 0.5 10 EPA 200.7 8/28/2012 12:03:00 PM MHH7131

DISSOLVED METALS BY ICP-MS

Selenium (Se) < 5 ug/L 5 5 EPA 200.8 8/31/2012 1:42:00 PM KRICHAR

TOTAL RECOVERABLE METALS BY ICP-MS

Arsenic (As) < 5 ug/L 5 5 EPA 200.8 8/29/2012 1:10:00 PM KRICHAR

Cadmium (Cd) < 5 ug/L 5 5 EPA 200.8 8/29/2012 1:10:00 PM KRICHAR

Chromium (Cr) < 5 ug/L 5 5 EPA 200.8 8/29/2012 1:10:00 PM KRICHAR

Copper (Cu) < 5 ug/L 5 5 EPA 200.8 8/29/2012 1:10:00 PM KRICHAR

Nickel (Ni) < 5 ug/L 5 5 EPA 200.8 8/29/2012 1:10:00 PM KRICHAR

Selenium (Se) 5.16 ug/L 5 5 EPA 200.8 8/29/2012 1:10:00 PM KRICHAR

Silver (Ag) < 5 ug/L 5 5 EPA 200.8 8/29/2012 1:10:00 PM KRICHAR

Zinc (Zn) < 5 ug/L 5 5 EPA 200.8 8/29/2012 1:10:00 PM KRICHAR

SELENIUM SPECIATION - (Analysis Performed by Applied Speciation and Consulting, LLC)

Vendor Parameter Complete Vendor Method V_AS&C

2012018179

Collection Date: 17-Aug-12 8:45 AM

Site: BioReactor 2 Eff BLANK 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

DF Analyst

MERCURY 1631 - (Analysis Performed by Brooks Rand Labs LLC)

Vendor Parameter Complete Vendor Method V_BRAND

MERCURY 1631 - DISSOLVED - (Analysis Performed by Brooks Rand Labs LLC)

Vendor Parameter Complete Vendor Method V_BRAND

Page 9 of 39
Certificate of Laboratory Analysis

This report shall not be reproduced, except in full.

Order # J12080348

2012018180

Collection Date: 17-Aug-12 9:01 AM

Site: Filter Blk 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

DF Analyst

DISSOLVED METALS BY ICP

Manganese (Mn) < 0.005 mg/L 0.005 1 EPA 200.7 8/28/2012 3:09:00 PM MHH7131

DISSOLVED METALS BY ICP-MS

Selenium (Se) < 1 ug/L 1 1 EPA 200.8 8/31/2012 12:36:00 PM KRICHAR

Page 10 of 39
 

September 6, 2012 

Jay Perkins 

Duke Energy Analytical Laboratory 

Mail Code MGO3A2 (Building 7405) 

13339 Hagers Ferry Rd. 

Huntersville, NC 28078 

(704) 875-5245 

Project: Allen Shay/MillerCreek (LIMS #J12080348) 

Dear Mr. Perkins, 

Attached is the report associated with four (4) aqueous samples submitted for selenium 

speciation analysis on August 21, 2012. The samples were received in a sealed cooler at 0.7°C 

on August 22, 2012. Selenium speciation analysis was performed via ion chromatography 

inductively coupled plasma collision reaction cell mass spectrometry (IC-ICP-CRC-MS). Any 

issues associated with the analysis are addressed in the following report. 

If you have any questions, please feel free to contact me at your convenience. 

 Sincerely, 

 

 Russell Gerads 

 Vice President 

 Applied Speciation and Consulting, LLC 

 

Page 11 of 39
Applied Speciation and Consulting, LLC 

Report prepared for: 

Jay Perkins 

Duke Energy Analytical Laboratory 

Mail Code MGO3A2 (Building 7405) 

13339 Hagers Ferry Rd. 

Huntersville, NC 28078 

Project: Allen Shay/MillerCreek (LIMS #J12080348) 

September 6, 2012 

1. Sample Reception 

Four (4) aqueous samples in 125mL HDPE bottles (provided by Applied Speciation and 

Consulting) were submitted for selenium speciation analysis on August 21, 2012. The 

samples were received on August 22, 2012 in a sealed container at 0.7°C. 

The samples were received in a laminar flow clean hood, void of trace metals contamination 

and ultra-violet radiation, and were designated discrete sample identifiers. An aliquot of 

each sample was filtered (0.45μm) and each filtrate was stored in a secure, monitored 

cryofreezer (maintained at a temperature of -80oC) until selenium speciation analysis could 

be performed via ion chromatography inductively coupled plasma collision reaction cell 

mass spectrometry (IC-ICP-CRC-MS). 

2. Sample Preparation 

All sample preparation is performed in laminar flow clean hoods known to be free from trace 

metals contamination. All applied water for dilutions and sample preservatives are 

monitored for contamination to account for any biases associated with the sample results. 

Selenium Speciation Analysis by IC-ICP-CRC-MS Prior to analysis, an aliquot of each 

sample was filtered with a syringe filter (0.45μm) and injected directly into a sealed 

autosampler vial. No further sample preparation was performed as any chemical alteration of 

a sample may shift the equilibrium of the system, resulting in changes in speciation ratios. 

3. Sample Analysis 

All sample analysis is preceded by a minimum of a five-point calibration curve spanning the 

entire concentration range of interest. Calibration curves are performed at the beginning of 

each analytical day. All calibration curves, associated with each species of interest, are 

Page 12 of 39
standardized by linear regression resulting in a response factor. All sample results are 

instrument blank corrected to account for any operational biases associated with the 

analytical platform. 

Prior to sample analysis, all calibration curves are verified using second source standards 

which are identified as initial calibration verification standards (ICV). 

Ongoing instrument performance is identified by the analysis of continuing calibration 

verification standards (CCV) and continuing calibration blanks (CCB) at a minimal interval 

of every ten analytical runs. 

Selenium Speciation Analysis by IC-ICP-CRC-MS Each sample for selenium speciation 

analysis was analyzed by ion chromatography inductively coupled plasma collision reaction 

cell mass spectrometry (IC-ICP-CRC-MS) on August 29, 2012. An aliquot of each sample is

injected onto an anion exchange column and mobilized by a basic (pH > 7) gradient. The 

eluting selenium species are then introduced into a radio frequency (RF) plasma where 

energy-transfer processes cause desolvation, atomization, and ionization. The ions are 

extracted from the plasma through a differentially-pumped vacuum interface and travel 

through a pressurized chamber (CRC) containing a reaction gas which preferentially reacts 

with interfering ions of the same target mass to charge ratios (m/z). A solid-state detector 

detects ions transmitted through the mass analyzer and the resulting current is processed by a 

data handling system. 

Retention times for each eluting species are compared to known standards for species 

identification. 

4. Analytical Issues 

The overall analyses went well and no significant analytical issues were encountered. All 

quality control parameters associated with the samples were within acceptance limits.

The estimated method detection limits (eMDLs) for selenite, selenate, and selenocyanate are 

generated from replicate analyses of the lowest standard in the calibration curve. Not all 

selenium species are present in preparation blanks; therefore, eMDL calculations based on 

preparation blanks are artificially biased low. 

The eMDL for methylseleninic acid and selenomethionine is calculated from the average 

eMDL of selenite, selenate, and selenocyanate. The calibration does not contain 

methylseleninic acid or selenomethionine due to impurities in these standards which would 

bias the results for other selenium species. 

Page 13 of 39
If you have any questions or concerns regarding this report, please feel free to contact me. 

Sincerely, 

Russell Gerads 

Vice President 

Applied Speciation and Consulting, LLC 

Page 14 of 39
Sample Results

Sample ID Se(IV) Se(VI) SeCN MeSe(IV) SeMe

Unknown Se 

Species (n)

FGD Purge Eff ND (<6.8) 564 ND (<2.1) ND (<3.4) ND (<3.4) 0.0 (0)

BioReactor 1 Inf ND (<1.7) 657 ND (<0.51) ND (<0.85) ND (<0.85) 0.0 (0)

BioReactor 2 Inf 61.3 12.7 ND (<0.51) 2.33 ND (<0.85) 1.12 (1)

BioReactor 2 Eff ND (<1.7) ND (<0.34) ND (<0.51) ND (<0.85) ND (<0.85) 0.0 (0)

All results reflect the applied dilution and are reported in μg/L

ND = Not detected at the applied dilution

SeCN = Selenocyanate

MeSe(IV) = Methylseleninic acid

SeMe = Selenomethionine

Unknown Se Species = Total concentration of all unknown Se species observed by IC-ICP-MS

n = number of unknown Se species observed

Report Generated by: Russell Gerads

Applied Speciation and Consulting, LLC

Selenium Speciation Results for Duke Energy

Project Name: Allen Shay/MillerCreek

Contact: Jay Perkins

Date: September 6, 2012

LIMS #J12080348

Page 15 of 39
Quality Control Summary - Preparation Blank Summary

Analyte (μg/L) PBW1 PBW2 PBW3 PBW4 Mean StdDev eMDL* eMDL 250x eMDL 1000x

Se(IV) 0.000 0.000 0.000 0.000 0.000 0.000 0.0068 1.7 6.8

Se(VI) 0.016 0.000 0.000 0.000 0.004 0.008 0.0014 0.34 1.4

SeCN 0.000 0.000 0.000 0.000 0.000 0.000 0.0021 0.51 2.1

MeSe(IV) 0.000 0.000 0.000 0.000 0.000 0.000 0.0034 0.85 3.4

SeMe 0.000 0.000 0.000 0.000 0.000 0.000 0.0034 0.85 3.4

eMDL = Estimated Method Detection Limit

*Please see narrative regarding eMDL calculations

Quality Control Summary - Certified Reference Materials

Analyte (μg/L) CRM True Value Result Recovery

Se(IV) LCS 9.57 9.47 99.0

Se(VI) LCS 9.48 9.33 98.4

SeCN LCS 8.92 8.63 96.7

MeSe(IV) LCS 6.47 6.14 95.0

SeMe LCS 9.32 8.81 94.5

Applied Speciation and Consulting, LLC

Selenium Speciation Results for Duke Energy

Project Name: Allen Shay/MillerCreek

Contact: Jay Perkins

LIMS #J12080348

Date: September 6, 2012

Report Generated by: Russell Gerads

Page 16 of 39
Quality Control Summary - Matrix Duplicates

Analyte (μg/L) Sample ID Rep 1 Rep 2 Mean RPD

Se(IV) Batch QC 33.6 34.3 33.9 2.1

Se(VI) Batch QC 23.02 22.64 22.83 1.6

SeCN Batch QC 0.71 ND (<0.51) NC NC

MeSe(IV) Batch QC 1.45 1.58 1.52 8.4

SeMe Batch QC ND (<0.85) ND (<0.85) NC NC

ND = Not detected at the applied dilution

NC = Value was not calculated due to one or more concentrations below the eMDL

Quality Control Summary - Matrix Spike/ Matrix Spike Duplicate

Analyte (μg/L) Sample ID Spike Conc MS Result Recovery Spike Conc MSD Result Recovery RPD

Se(IV) Batch QC 1390 1412 99.2 1390 1413 99.2 0.0

Se(VI) Batch QC 1261 1262 98.3 1261 1263 98.3 0.0

SeCN Batch QC 1144 1108 96.8 1144 1107 96.8 0.1

Applied Speciation and Consulting, LLC

Selenium Speciation Results for Duke Energy

Project Name: Allen Shay/MillerCreek

Contact: Jay Perkins

LIMS #J12080348

Date: September 6, 2012

Report Generated by: Russell Gerads

Page 17 of 39
Page 18 of 39
 

3958 6th Ave NW • Seattle, WA 98107 • T: 206-632-6206 • F: 206-632-6017 • www.brooksrand.com • brl@brooksrand.com 

August 31, 2012 

Duke Energy 

ATTN: Jay Perkins 

Scientific Support-Laboratory

13339 Hagers Ferry Road 

Huntersville NC 28078 

jcperkins@duke-energy.com 

labcustomer@duke-energy.com 

RE: Project DUK-HV1201 Client Project: J12080348 

Dear Mr. Perkins, 

On August 22, 2012, Brooks Rand Labs (BRL) received three (3) wastewater samples and three 

(3) corresponding field blanks. An aliquot was removed from each sample bottle and filtered into 

a separate container designed for dissolved mercury (Hg) analysis. The sample volume from 

the original container was logged-in for total Hg analysis. All samples were received, prepared, 

analyzed, and stored according to BRL SOPs and EPA methodology. 

Data used for regulatory purposes has a 24 hour filtration holding time requirement. Nonregulatory purposed data has a 48 hour filtration holding time. The samples were received 

outside of the 48 hour filtration requirement and the sample results were qualified H. 

The results were blank-corrected as described in the calculations section of the relevant SOP(s) 

and may have been evaluated using reporting limits that have been adjusted to account for 

sample aliquot size. Please refer to the Sample Results page for sample-specific MDLs, MRLs, 

and other details. 

The instrument blank (IBL) IBL1 had an abnormal peak. The IBL was re-analyzed as IBL5 which 

produced a normal peak. IBL5 was reported in place of IBL1. 

The sample BioReactor 2 Eff Hg Blk (1234011-11) had an elevated total Hg concentration of 

0.67 ng/L. The sample was re-analyzed and confirmed the initial result. The associated sample 

BioReactor 2 Eff (1234011-09) has a concentration greater than 10x the concentration of the 

blank. No significant contamination was suspected. All data was reported without qualification, 

aside from concentration qualifiers, and all associated quality control sample results met the 

acceptance criteria. 

BRL, an accredited laboratory, certifies the reported results of all analyses for which BRL is 

NELAP accredited meet all NELAP requirements. For more details, see the Report Information

page of the report. Please feel free to contact me if you have any questions regarding this 

report. 

Sincerely, 

 

Lydia Greaves 

Project Manager 

lydia@brooksrand.com

Page 19 of 39
Project ID: DUK-HV1201

PM: Tiffany Stilwater

Client PM: Jay Perkins

 Client PO: 141391

Definition of Data Qualifiers

(Effective 9/23/09)

Laboratory Accreditation

BRL is accredited by the National Environmental Laboratory Accreditation Program (NELAP) through the State of Florida 

Department of Health, Bureau of Laboratories (E87982) and is certified to perform many environmental analyses. BRL is 

also certified by many other states to perform environmental analyses. For a current list of our accreditations /certifications, 

please visit our website at <http://www.brooksrand.com/default.asp?contentID=586>. Results reported relate only to the 

samples listed in the report.

Report Information

BLK

BRL

BS

CAL

CCV

CRM

D

DUP

ICV

MSD

ND

NR

PS

REC

RPD

RSD

SCV

SOP

method blank 

Brooks Rand Labs

laboratory fortified blank

calibration standard

continuing calibration verification

certified reference material

dissolved fraction

duplicate

initial calibration verification

matrix spike duplicate

non-detect

non-reportable

post preparation spike

percent recovery

relative percent difference

relative standard deviation

secondary calibration verification

standard operating procedure

MDL

MRL

MS

method detection limit

method reporting limit

matrix spike

SRM

T

COC

standard reference material

total recoverable fraction

chain of custody record 

Common Abbreviations

These qualifiers are based on those previously utilized by Brooks Rand Labs, those found in the EPA SOW ILM03.0, 

Exhibit B, Section III, pg. B-18, and the USEPA Contract Laboratory Program National Functional Guidelines for Inorganic 

Superfund Data Review; USEPA; January 2010. These supersede all previous qualifiers ever employed by BRL.

B Detected by the instrument, the result is > the MDL but ≤ the MRL. Result is reported and considered an estimate.

E An estimated value due to the presence of interferences. A full explanation is presented in the narrative.

H Holding time and/or preservation requirements not met. Result is estimated.

J Estimated value. A full explanation is presented in the narrative.

J-M Duplicate precision (RPD) for associated QC sample was not within acceptance criteria. Result is estimated.

J-N Spike recovery for associated QC sample was not within acceptance criteria. Result is estimated.

M Duplicate precision (RPD) was not within acceptance criteria. Result is estimated.

N Spike recovery was not within acceptance criteria. Result is estimated.

R Rejected, unusable value. A full explanation is presented in the narrative.

U Result is ≤ the MDL or client requested reporting limit (CRRL). Result reported as the MDL or CRRL.

X Result is not BLK-corrected and is within 10x the absolute value of the highest detectable BLK in the batch. 

Result is estimated.

Field Quality Control Samples

Please be notified that certain EPA methods require the collection of field quality control samples of an appropriate type 

and frequency; failure to do so is considered a deviation from some methods and for compliance purposes should only be 

done with the approval of regulatory authorities. Please see the specific EPA methods for details regarding required field 

quality control samples.

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.com

Page 20 of 39
Project ID: DUK-HV1201

PM: Tiffany Stilwater

Client PM: Jay Perkins

 Client PO: 141391

Sample Information

Sample Lab ID Report Matrix Type Sampled Received

BioReactor 1 Inf 1234011-01 Influent Sample 08/17/2012 08/22/2012

BioReactor 1 Inf 1234011-02 Influent Sample 08/17/2012 08/22/2012

BioReactor 1 Inf Hg Blk 1234011-03 DIW Field Blank 08/17/2012 08/22/2012

BioReactor 1 Inf Hg Blk 1234011-04 DIW Field Blank 08/17/2012 08/22/2012

BioReactor 2 Inf 1234011-05 Influent Sample 08/17/2012 08/22/2012

BioReactor 2 Inf 1234011-06 Influent Sample 08/17/2012 08/22/2012

BioReactor 2 Inf Hg Blk 1234011-07 DIW Field Blank 08/17/2012 08/22/2012

BioReactor 2 Inf Hg Blk 1234011-08 DIW Field Blank 08/17/2012 08/22/2012

BioReactor 2 Eff 1234011-09 Effluent Sample 08/17/2012 08/22/2012

BioReactor 2 Eff 1234011-10 Effluent Sample 08/17/2012 08/22/2012

BioReactor 2 Eff Hg Blk 1234011-11 DIW Field Blank 08/17/2012 08/22/2012

BioReactor 2 Eff Hg Blk 1234011-12 DIW Field Blank 08/17/2012 08/22/2012

Batch Summary

Analyte Lab Matrix Method Prepared Analyzed Batch Sequence

Hg Water EPA 1631 08/24/2012 08/28/2012 B121527 1200668

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.com

Page 21 of 39
Project ID: DUK-HV1201

PM: Tiffany Stilwater

Client PM: Jay Perkins

 Client PO: 141391

Sample Results

Sample Analyte Report Matrix Basis Result Qualifier MDL MRL Unit Batch Sequence

BioReactor 1 Inf

1234011-01 Hg Influent T 14000 50.5 135 ng/L B121527 1200668

1234011-02 Hg Influent D 12200 H 30.3 80.8 ng/L B121527 1200668

BioReactor 1 Inf Hg Blk

1234011-03 Hg DIW T 0.15 U 0.15 0.41 ng/L B121527 1200668

1234011-04 Hg DIW D 0.15 H, U 0.15 0.40 ng/L B121527 1200668

BioReactor 2 Eff

1234011-09 Hg Effluent T 63.2 1.52 4.04 ng/L B121527 1200668

1234011-10 Hg Effluent D 15.4 H 0.31 0.82 ng/L B121527 1200668

BioReactor 2 Eff Hg Blk

1234011-11 Hg DIW T 0.67 0.15 0.41 ng/L B121527 1200668

1234011-12 Hg DIW D 0.15 H, U 0.15 0.41 ng/L B121527 1200668

BioReactor 2 Inf

1234011-05 Hg Influent T 1100 15.3 40.8 ng/L B121527 1200668

1234011-06 Hg Influent D 54.2 H 0.30 0.81 ng/L B121527 1200668

BioReactor 2 Inf Hg Blk

1234011-07 Hg DIW T 0.15 U 0.15 0.40 ng/L B121527 1200668

1234011-08 Hg DIW D 0.15 H, U 0.15 0.41 ng/L B121527 1200668

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.com

Page 22 of 39
Project ID: DUK-HV1201

PM: Tiffany Stilwater

Client PM: Jay Perkins

 Client PO: 141391

Accuracy & Precision Summary

Batch: B121527

Analyte Native Spike Result Units REC & Limits RPD & Limits

Method: EPA 1631

Lab Matrix: Water

Sample

B121527-SRM1 Certified Reference Material (1232065, NIST 1641d 1000x dilution)

Hg 15.68 16.69 ng/L 106% 85-115

B121527-MS2 Matrix Spike (1234011-05)

Hg 1103 5510 6736 ng/L 102% 71-125

B121527-MSD2 Matrix Spike Duplicate (1234011-05)

Hg 1103 5510 7098 ng/L 109% 71-125 245%

B121527-MS3 Matrix Spike (1234034-03)

Hg 644.0 3030 3845 ng/L 106% 71-125

B121527-MSD3 Matrix Spike Duplicate (1234034-03)

Hg 644.0 3030 3773 ng/L 103% 71-125 242%

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.com

Page 23 of 39
Project ID: DUK-HV1201

PM: Tiffany Stilwater

Client PM: Jay Perkins

 Client PO: 141391

Method Blanks & Reporting Limits

Batch: B121527

Method: EPA 1631

Matrix: Water

Analyte: Hg

Sample Result Units

B121527-BLK1 0.14 ng/L

B121527-BLK2 0.10 ng/L

B121527-BLK3 0.08 ng/L

B121527-BLK4 0.12 ng/L

Average: 0.11 Standard Deviation: 0.03 MDL: 0.15

Limit: 0.50 Limit: 0.10 MRL: 0.40

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.com

Page 24 of 39
Project ID: DUK-HV1201

PM: Tiffany Stilwater

Client PM: Jay Perkins

 Client PO: 141391

Instrument Calibration

Sequence: 1200668

Instrument: THG-06(MerxT) Method: EPA 1631

Date: 08/28/2012

Total Mercury and Mercury Speciation by CVAFS

Analyte: Hg

Lab ID True Value Result Units REC & Limits

1200668-IBL2 5.89 pg of Hg

1200668-IBL3 5.89 pg of Hg

1200668-IBL4 9.85 pg of Hg

1200668-CAL1 25.00 21.87 pg of Hg 87%

1200668-CAL2 100.0 105.0 pg of Hg 105%

1200668-CAL3 500.0 515.2 pg of Hg 103%

1200668-CAL4 2500 2614 pg of Hg 105%

1200668-CAL5 10000 10230 pg of Hg 102%

1200668-ICV1 1568 1669 pg of Hg 106% 85-115

1200668-CCV1 500.0 508.8 pg of Hg 102% 77-123

1200668-CCB1 9.43 pg of Hg

1200668-IBL5 6.15 pg of Hg

1200668-ICB1 6.87 pg of Hg

1200668-CCB2 44.4 pg of Hg

1200668-CCB3 11.2 pg of Hg

1200668-CCB4 8.20 pg of Hg

1200668-CCV2 500.0 513.5 pg of Hg 103% 77-123

1200668-CCV3 500.0 410.0 pg of Hg 82% 77-123

1200668-CCV4 500.0 548.0 pg of Hg 110% 77-123

1200668-CCV5 500.0 514.9 pg of Hg 103% 77-123

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.com

Page 25 of 39
Project ID: DUK-HV1201

PM: Tiffany Stilwater

Client PM: Jay Perkins

 Client PO: 141391

Sample Containers

Lab ID: 1234011-01 Report Matrix: Influent

Sample: BioReactor 1 Inf Sample Type: Sample Received: 08/22/2012

Collected: 08/17/2012

Des Container Size Lot Preservation P-Lot pH Ship. Cont.

A 71666330

10

Bottle FLPE Hg-T 500 mL none n/a Cooler

Lab ID: 1234011-02 Report Matrix: Influent

Sample: BioReactor 1 Inf Sample Type: Sample Received: 08/22/2012

Collected: 08/17/2012

Des Container Size Lot Preservation P-Lot pH Ship. Cont.

A 71659890

20

Bottle FLPE Hg-T 250 mL none n/a Cooler

Comments: Split from THg Container

Lab ID: 1234011-03 Report Matrix: DIW

Sample: BioReactor 1 Inf Hg Blk Sample Type: Field Blank Received: 08/22/2012

Collected: 08/17/2012

Des Container Size Lot Preservation P-Lot pH Ship. Cont.

A 71666330

10

Bottle FLPE Hg-T 500 mL none n/a Cooler

Lab ID: 1234011-04 Report Matrix: DIW

Sample: BioReactor 1 Inf Hg Blk Sample Type: Field Blank Received: 08/22/2012

Collected: 08/17/2012

Des Container Size Lot Preservation P-Lot pH Ship. Cont.

A 71659890

20

Bottle FLPE Hg-T 250 mL none n/a Cooler

Comments: Split from THg Container

Lab ID: 1234011-05 Report Matrix: Influent

Sample: BioReactor 2 Inf Sample Type: Sample Received: 08/22/2012

Collected: 08/17/2012

Des Container Size Lot Preservation P-Lot pH Ship. Cont.

A 71666330

10

Bottle FLPE Hg-T 500 mL none n/a Cooler

Lab ID: 1234011-06 Report Matrix: Influent

Sample: BioReactor 2 Inf Sample Type: Sample Received: 08/22/2012

Collected: 08/17/2012

Des Container Size Lot Preservation P-Lot pH Ship. Cont.

A 71659890

20

Bottle FLPE Hg-T 250 mL none n/a Cooler

Comments: Split from THg Container

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.com

Page 26 of 39
Project ID: DUK-HV1201

PM: Tiffany Stilwater

Client PM: Jay Perkins

 Client PO: 141391

Sample Containers

Lab ID: 1234011-07 Report Matrix: DIW

Sample: BioReactor 2 Inf Hg Blk Sample Type: Field Blank Received: 08/22/2012

Collected: 08/17/2012

Des Container Size Lot Preservation P-Lot pH Ship. Cont.

A 71666330

10

Bottle FLPE Hg-T 500 mL none n/a Cooler

Lab ID: 1234011-08 Report Matrix: DIW

Sample: BioReactor 2 Inf Hg Blk Sample Type: Field Blank Received: 08/22/2012

Collected: 08/17/2012

Des Container Size Lot Preservation P-Lot pH Ship. Cont.

A 71659890

20

Bottle FLPE Hg-T 250 mL none n/a Cooler

Comments: Split from THg Container

Lab ID: 1234011-09 Report Matrix: Effluent

Sample: BioReactor 2 Eff Sample Type: Sample Received: 08/22/2012

Collected: 08/17/2012

Des Container Size Lot Preservation P-Lot pH Ship. Cont.

A 71666330

10

Bottle FLPE Hg-T 500 mL none n/a Cooler

Lab ID: 1234011-10 Report Matrix: Effluent

Sample: BioReactor 2 Eff Sample Type: Sample Received: 08/22/2012

Collected: 08/17/2012

Des Container Size Lot Preservation P-Lot pH Ship. Cont.

A 71659890

20

Bottle FLPE Hg-T 250 mL none n/a Cooler

Comments: Split from THg Container

Lab ID: 1234011-11 Report Matrix: DIW

Sample: BioReactor 2 Eff Hg Blk Sample Type: Field Blank Received: 08/22/2012

Collected: 08/17/2012

Des Container Size Lot Preservation P-Lot pH Ship. Cont.

A 71666330

10

Bottle FLPE Hg-T 500 mL none n/a Cooler

Lab ID: 1234011-12 Report Matrix: DIW

Sample: BioReactor 2 Eff Hg Blk Sample Type: Field Blank Received: 08/22/2012

Collected: 08/17/2012

Des Container Size Lot Preservation P-Lot pH Ship. Cont.

A 71659890

20

Bottle FLPE Hg-T 250 mL none n/a Cooler

Comments: Split from THg Container

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.com

Page 27 of 39
Project ID: DUK-HV1201

PM: Tiffany Stilwater

Client PM: Jay Perkins

 Client PO: 141391

Shipping Containers

Cooler

Tracking No: 5353 0519 3373 via FedEx

Temperature: -0.6 °C

Coolant Type: Ice

Description: Cooler

Damaged in transit? No

Returned to client? No

Custody seals present? No

Custody seals intact? No

COC present? Yes

Received: August 22, 2012 8:50

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.com

Page 28 of 39
Page 29 of 39
This data package contains the analytical results for the project identified above and includes a Case Narrative, Sample 

Results and Chain of Custody. Unless otherwise noted, all samples were received in acceptable condition and processed 

according to the referenced methods. 

Data qualifiers are flagged individually on each sample. A key reference for the data qualifiers appears at the end of this case 

narrative.

NC Certification No. 402

SC Certification No. 99012

NC Drinking Water Cert No. 37735

Case Narrative

08/27/2012

Duke Energy Corporation (04)

Jay Perkins

13339 Hagers Ferry Road

Huntersville, NC 28078

Project: Allen Shay/Miller Creek

Project No.: J12080348

Lab Submittal Date: 08/21/2012

Prism Work Order: 2080547

VA Certification No. 1287

Please call if you have any questions relating to this analytical report.

Respectfully,

PRISM LABORATORIES, INC.

VP Laboratory Services Reviewed By

Data Qualifiers Key Reference:

HT Sample received and analyzed outside of the hold time.

MDL Method Detection Limit

BRL Below Reporting Limit

RPD Relative Percent Difference

*

Results reported to the reporting limit. All other results are reported to the MDL with values between MDL and 

reporting limit indicated with a J.

This report should not be reproduced, except in its entirety, without the written consent of Prism Laboratories, Inc.

449 Springbrook Road - P.O. Box 240543 - Charlotte, NC 28224-0543

Phone: 704/529-6364 - Toll Free Number: 1-800/529-6364 - Fax: 704/525-0409 Page 1 of 9

Page 30 of 39
Sample Receipt Summary

08/27/2012

Prism Work Order: 2080547

Client Sample ID Lab Sample ID Matrix Date Sampled Date Received

2012018172/FGD Purge Eff 2080547-01 Water 08/17/12 08/21/12

2012018174/BioReactor 1 Inf 2080547-02 Water 08/17/12 08/21/12

2012018176/BioReactor 2 Inf 2080547-03 Water 08/17/12 08/21/12

2012018178/BioReactor 2 Eff 2080547-04 Water 08/17/12 08/21/12

Samples received in good condition at 2.9 degrees C unless otherwise noted.

This report should not be reproduced, except in its entirety, without the written consent of Prism Laboratories, Inc.

449 Springbrook Road - P.O. Box 240543 - Charlotte, NC 28224-0543

Phone: 704/529-6364 - Toll Free Number: 1-800/529-6364 - Fax: 704/525-0409 Page 2 of 9

Page 31 of 39
Laboratory Report

08/27/2012

Parameter Result Batch

ID

Report

Limit

Analysis

Date/Time

Dilution Method

Factor

Units MDL Analyst

13339 Hagers Ferry Road Project No.: 

Duke Energy Corporation (04) Project: Allen Shay/Miller Creek

Huntersville, NC 28078

Attn: Jay Perkins

Sample Matrix: Water

Client Sample ID: 

Prism Sample ID: 2080547-01

Prism Work Order: 2080547

Time Collected: 08/17/12 08:01

Time Submitted: 08/21/12 17:03

2012018172/FGD Purge Eff

J12080348

General Chemistry Parameters

pH 7.0 HT pH Units 1 *SM4500-H B 8/23/12 11:00 JAB P2H0516

Total Alkalinity 52 mg/L 5.0 0.66 1 *SM2320 B 8/23/12 11:00 JAB P2H0512

Carbonate Alkalinity BRL mg/L 5.0 0.66 1 *SM2320 B 8/23/12 11:00 JAB P2H0513

Bicarbonate Alkalinity 52 mg/L 5.0 0.66 1 *SM2320 B 8/23/12 11:00 JAB P2H0514

This report should not be reproduced, except in its entirety, without the written consent of Prism Laboratories, Inc.

449 Springbrook Road - P.O. Box 240543 - Charlotte, NC 28224-0543

Phone: 704/529-6364 - Toll Free Number: 1-800/529-6364 - Fax: 704/525-0409 Page 3 of 9

Page 32 of 39
Laboratory Report

08/27/2012

Parameter Result Batch

ID

Report

Limit

Analysis

Date/Time

Dilution Method

Factor

Units MDL Analyst

13339 Hagers Ferry Road Project No.: 

Duke Energy Corporation (04) Project: Allen Shay/Miller Creek

Huntersville, NC 28078

Attn: Jay Perkins

Sample Matrix: Water

Client Sample ID: 

Prism Sample ID: 2080547-02

Prism Work Order: 2080547

Time Collected: 08/17/12 08:37

Time Submitted: 08/21/12 17:03

2012018174/BioReactor 1 Inf

J12080348

General Chemistry Parameters

pH 7.4 HT pH Units 1 *SM4500-H B 8/23/12 11:00 JAB P2H0516

Total Alkalinity 54 mg/L 5.0 0.66 1 *SM2320 B 8/23/12 11:00 JAB P2H0512

Carbonate Alkalinity BRL mg/L 5.0 0.66 1 *SM2320 B 8/23/12 11:00 JAB P2H0513

Bicarbonate Alkalinity 54 mg/L 5.0 0.66 1 *SM2320 B 8/23/12 11:00 JAB P2H0514

This report should not be reproduced, except in its entirety, without the written consent of Prism Laboratories, Inc.

449 Springbrook Road - P.O. Box 240543 - Charlotte, NC 28224-0543

Phone: 704/529-6364 - Toll Free Number: 1-800/529-6364 - Fax: 704/525-0409 Page 4 of 9

Page 33 of 39
Laboratory Report

08/27/2012

Parameter Result Batch

ID

Report

Limit

Analysis

Date/Time

Dilution Method

Factor

Units MDL Analyst

13339 Hagers Ferry Road Project No.: 

Duke Energy Corporation (04) Project: Allen Shay/Miller Creek

Huntersville, NC 28078

Attn: Jay Perkins

Sample Matrix: Water

Client Sample ID: 

Prism Sample ID: 2080547-03

Prism Work Order: 2080547

Time Collected: 08/17/12 08:51

Time Submitted: 08/21/12 17:03

2012018176/BioReactor 2 Inf

J12080348

General Chemistry Parameters

pH 7.3 HT pH Units 1 *SM4500-H B 8/23/12 11:00 JAB P2H0516

Total Alkalinity 290 mg/L 5.0 0.66 1 *SM2320 B 8/23/12 11:00 JAB P2H0512

Carbonate Alkalinity BRL mg/L 5.0 0.66 1 *SM2320 B 8/23/12 11:00 JAB P2H0513

Bicarbonate Alkalinity 290 mg/L 5.0 0.66 1 *SM2320 B 8/23/12 11:00 JAB P2H0514

This report should not be reproduced, except in its entirety, without the written consent of Prism Laboratories, Inc.

449 Springbrook Road - P.O. Box 240543 - Charlotte, NC 28224-0543

Phone: 704/529-6364 - Toll Free Number: 1-800/529-6364 - Fax: 704/525-0409 Page 5 of 9

Page 34 of 39
Laboratory Report

08/27/2012

Parameter Result Batch

ID

Report

Limit

Analysis

Date/Time

Dilution Method

Factor

Units MDL Analyst

13339 Hagers Ferry Road Project No.: 

Duke Energy Corporation (04) Project: Allen Shay/Miller Creek

Huntersville, NC 28078

Attn: Jay Perkins

Sample Matrix: Water

Client Sample ID: 

Prism Sample ID: 2080547-04

Prism Work Order: 2080547

Time Collected: 08/17/12 08:45

Time Submitted: 08/21/12 17:03

2012018178/BioReactor 2 Eff

J12080348

General Chemistry Parameters

pH 7.3 HT pH Units 1 *SM4500-H B 8/23/12 11:00 JAB P2H0516

Total Alkalinity 190 mg/L 5.0 0.66 1 *SM2320 B 8/23/12 11:00 JAB P2H0512

Carbonate Alkalinity BRL mg/L 5.0 0.66 1 *SM2320 B 8/23/12 11:00 JAB P2H0513

Bicarbonate Alkalinity 190 mg/L 5.0 0.66 1 *SM2320 B 8/23/12 11:00 JAB P2H0514

This report should not be reproduced, except in its entirety, without the written consent of Prism Laboratories, Inc.

449 Springbrook Road - P.O. Box 240543 - Charlotte, NC 28224-0543

Phone: 704/529-6364 - Toll Free Number: 1-800/529-6364 - Fax: 704/525-0409 Page 6 of 9

Page 35 of 39
8/27/12

Level II QC Report

13339 Hagers Ferry Road Project No: J12080348

Duke Energy Corporation (04) Project: Allen Shay/Miller Creek

Huntersville, NC 28078

Attn: Jay Perkins

Prism Work Order: 2080547

Time Submitted: 8/21/2012 5:03:00PM

Result Limit

Reporting

Units Level

Spike

Result

Source

%REC

%REC

Limits RPD

RPD

 Analyte Limit Notes 

General Chemistry Parameters - Quality Control

Batch P2H0512 - NO PREP

Blank (P2H0512-BLK1) Prepared & Analyzed: 08/23/12 

Total Alkalinity BRL 5.0 mg/L

LCS (P2H0512-BS1) Prepared & Analyzed: 08/23/12 

Total Alkalinity 242 5.0 mg/L 250.0 97 90-110

LCS Dup (P2H0512-BSD1) Prepared & Analyzed: 08/23/12 

Total Alkalinity 246 5.0 mg/L 250.0 98 90-110 2 200

Batch P2H0513 - NO PREP

Blank (P2H0513-BLK1) Prepared & Analyzed: 08/23/12 

Carbonate Alkalinity BRL 5.0 mg/L

LCS (P2H0513-BS1) Prepared & Analyzed: 08/23/12 

Carbonate Alkalinity 242 5.0 mg/L 90-110

LCS Dup (P2H0513-BSD1) Prepared & Analyzed: 08/23/12 

Carbonate Alkalinity 246 5.0 mg/L 90-110 2 200

Batch P2H0514 - NO PREP

Blank (P2H0514-BLK1) Prepared & Analyzed: 08/23/12 

Bicarbonate Alkalinity BRL 5.0 mg/L

LCS (P2H0514-BS1) Prepared & Analyzed: 08/23/12 

Bicarbonate Alkalinity 242 5.0 mg/L 250.0 97 90-110

This report should not be reproduced, except in its entirety, without the written consent of Prism Laboratories, Inc.

449 Springbrook Road - P.O. Box 240543 - Charlotte, NC 28224-0543

Phone: 704/529-6364 - Toll Free Number: 1-800/529-6364 - Fax: 704/525-0409 Page 7 of 9

Page 36 of 39
8/27/12

Level II QC Report

13339 Hagers Ferry Road Project No: J12080348

Duke Energy Corporation (04) Project: Allen Shay/Miller Creek

Huntersville, NC 28078

Attn: Jay Perkins

Prism Work Order: 2080547

Time Submitted: 8/21/2012 5:03:00PM

Result Limit

Reporting

Units Level

Spike

Result

Source

%REC

%REC

Limits RPD

RPD

 Analyte Limit Notes 

General Chemistry Parameters - Quality Control

Batch P2H0514 - NO PREP

LCS Dup (P2H0514-BSD1) Prepared & Analyzed: 08/23/12 

Bicarbonate Alkalinity 246 5.0 mg/L 250.0 98 90-110 2 200

Batch P2H0516 - NO PREP

LCS (P2H0516-BS1) Prepared & Analyzed: 08/23/12 

pH 6.81 pH Units 6.860 99 99-101

This report should not be reproduced, except in its entirety, without the written consent of Prism Laboratories, Inc.

449 Springbrook Road - P.O. Box 240543 - Charlotte, NC 28224-0543

Phone: 704/529-6364 - Toll Free Number: 1-800/529-6364 - Fax: 704/525-0409 Page 8 of 9

Page 37 of 39
Page 9 of 9

Page 38 of 39
Page 39 of 39