# 

GLP10

- 01

- 02; Interim Report 28 - Analysis of PFBS, PFHS, and PFOS in Groundw ater

Samples Collected at the Former Sludge Incorporation Area (FSIA) located in Decatur, AL

, March/April 2012

Study Title

Analysis of Perfluorooctane Sulfonate (PFOS), Perfluorohexane Sulfonate (PFHS) and

Perfluorobutane Sulfonate (PFBS) in Groundwater, Soil and Sediment for the 3M Decatur Phase 3

Site

- Related Monitoring Program

Data Requirement

EPA TSCA Good Laboratory Practice Standards 40 CFR Part 792

Study Director

Jaisimh a Kesari P.E., DEE

Weston Solutions, Inc.

1400 Weston Way

West Chester, PA 19380

Phone: 610

- 701

- 3761

Author

Susan Wolf

3M Environmental Laboratory

Interim Report Completion Date

Date of signing

Performing Laboratory

3M Environmental Health and Safety O perations

Environmental Laboratory

3M Center, Bldg 260

- 05 -N- 17

St. Paul , MN 55144

Project Identification

GLP10

- 01

- 02 -28

Total Number of Pages

108

Testing Cert #2052.01 The testing reported herein meet the requirements of ISO/IEC 17025

- 200 5 “General Requirements for the

Competence of Testing and Calibration Laboratories”, in accordance with the A2LA Certificate #2052.01.

Testing that complies with this International Standard also operate in accordance with ISO 9001:2000.

### This page has been reserved for specific country requirement s.GLP10-01-02 ; Interi m Repor t 2 8

Analysi s o f PFBS

, PFHS

, an d PFO S i n Groundwate r

Forme r Sludg e Incorporatio n Are a - March/Apri l 201 2



### GL P COMPLIANC E STATEMEN T

Repor t Title : GLP10-01-02 ; Interi m Repor t 2 8

- Analysi s o f PFBS

, PFHS

, an d PFO S i n Groundwate r Sampile s Collecte d a

t th e Forme r Sludg e Incorporatio n Are a (FSIA ) i n Decatur

, AL

, March/Apri l 201 2

Study : Analysi s o f Perfluorooctan e Sulfonat e (PFOS)

, Perfluorohexan e Sulfonat e (PFHS ) an d

Perfluorobutan e Sulfonat e (PFBS ) i n Groundwater

, Soi l an d Sedimen t fo r th e 3 M Decatu r Phas e 3

Site-Relate d Monitorin g Program

.

Thi s analytica l phas e wa s conducte d i n complianc e wit h Toxi c Substance s Contro l Ac t (TSCA ) Goo d

Laborator y Practic e (GLP ) Standards

, 4 0 CF R 792

, wit h th e exception s liste d below :

• Thes e ar e environmenta l sample s wher e ther e i s n o specifi c tes t substance

, n o specifi c tes t

syste m an d n o dosin g o

f a tes t system

.

• Th e referenc e substance s hav e no t bee n characterize d unde r th e GLP s an d th e stabilit y unde r

storag e condition s a

t th e tes t sit e hav

e no t bee n determine d unde r GLPs

.

Gar y Hohefnsteir

y Sponso r Representativ e Dat e

Jaisimh a Kesari

, P.E.

, DEE

, Stud y Directo r Dat e

Pag e 3 o f 10 8GLP10-01-02 ; Interi m Repor t 2 8

Analysi s o f PFBS

, PFHS

, an d PFO S i n Groundwate r

Forme r Sludg e Incorporatio n Are a - March/Apri l 201 2



### QUALIT Y ASSURANC E STATEMEN T

Repor t Title : GLP10-01-02 ; Interi m Repor t 2 8

- Analysi s o f PFBS

, PFHS

, an d PFO S i n Groundwate r Sample s Collecte d a

t th e Forme r Sludg e Incorporatio n Are a (FSIA ) i n Decatur

, AL

, March/Apri l 201 2

Study : Analysi s o f Perfluorooctan e Sulfonat e (PFOS)

, Perfluorohexan e Sulfonat e (PFHS ) an d

Perfluorobutan e Sulfonat e (PFBS ) i n Groundwater

, Soi l an d Sedimen t fo r th e 3 M Decatu r Phas e 3

Site-Relate d Monitorin g Program

.

Thi s analytica l phas e wa s audite d b y th e 3 M Environmenta l Laborator y Qualit y Assuranc e Uni t (QAU)

,

a s indicate d i n th e followin g table

. Th e finding s wer e reporte d t o th e principa l investigato r (P.I.)

, laborator y managemen t an d stud y director

.

Inspectio n Date s Phas e Dat e R e porte d t o

Inspectio n Date s Phas e Testin g Facilit y IVIanagemen t Stud y Directo r

4/23/1 2

- 4/24/1 2 Dat a / Repor t 4/26/1 2 4/26/1 2

Pag e 4 o f 10 8

# T ABLE OF 

# C ONTENTS

GLP Compliance Statement

...................................................................................................................... 3

Quality Assurance Statement

.................................................................................................................... 4

Table of Contents

................................................................................................ ....................................... 5

List of Tables

................................................................................................ .............................................. 6

1 Study Information

................................................................................................ ................................ 8

2 Summary

................................................................................................ ............................................. 9

3 Introduction

................................................................................................ ........................................ 11

4 Test & Control Substances

.............................................................................................................. 11

5 Reference Substances

..................................................................................................................... 12

6 Test System

................................................................................................ ...................................... 13

7 Method Summary

............................................................................................................................. 13

7.1 Methods

............................................................................................................................ 13

7.2 Sample Collection

............................................................................................................. 13

7.3 Sample Preparation

.......................................................................................................... 13

7.4 Analysis

............................................................................................................................. 14

8 Analytical Results

.............................................................................................................................. 15

8.1 Calibration

......................................................................................................................... 15

8.2 System Suitability

............................................................................................................. 15

8.3 Limit of Quantitation (LOQ)

.............................................................................................. 16

8.4 Continuing Calibration

...................................................................................................... 16

8.5 Blanks

................................................................................................ ................................ 16

8.6 Lab Control Spikes (LCSs)

.............................................................................................. 16

8.7 Analytical Method Uncertainty

......................................................................................... 18

8.8 Field Matrix Spikes (FMS)

................................................................................................ 19

### 9 Data Summary and Discussion

....................................................................................................... 19

10 Conclusion

................................................................................................ ......................................... 33

11 Data/Sample Retention

.................................................................................................................... 33

12 Attachments

................................................................................................ ...................................... 33

13 Signatures

................................................................................................ ......................................... 34



# L IST OF 

# T ABLES

Table 1. S ummarized PFBS, PFHS, and PFOS Results (FSIA GW, March/April 2012).

.................... 10

Table 2. Sample Description Key Code.

.................................................................................................. 13

Table 3. Instrument Parameters.

.............................................................................................................. 14

Table 4. Liquid Chromatography Conditions.

.......................................................................................... 14

Table 5. Mass Transitions.

........................................................................................................................ 15

T able 6. Limit of Quantitation (LOQ).

........................................................................................................ 16

Table 7. Laboratory Control Spike Recovery.

.......................................................................................... 17

Table 8. Analytical Uncertainty.

................................................................................................................ 18

Table 9. Field Matrix Spike Levels.

........................................................................................................... 19

Table 10. DAL GW 130R 120329

............................................................................................................ 20

Table 11. DAL GW 130S 120329

............................................................................................................. 20

Table 12. DAL GW 130L 120329

............................................................................................................. 21

Table 13. DAL GW 131R 120328

............................................................................................................ 21

Table 14. DAL GW 131S 120328

............................................................................................................. 22

Table 15. DAL GW 131L 120328

............................................................................................................. 22

Table 16. DAL GW 133R 120328

............................................................................................................ 22

Table 17. DAL GW 133S 120328

............................................................................................................. 23

Table 18. DAL GW 133L 120328

............................................................................................................. 23

Table 19. DAL GW 134R 120328

............................................................................................................ 24

Table 20. DALGW 134S 120327

.............................................................................................................. 24

### Table 21. DAL GW 134L 120327

............................................................................................................. 25

Table 22. DAL GW 135R 120329

............................................................................................................ 25

Table 23. DAL GW 135S 120329

............................................................................................................. 25

Table 24. DAL GW 135L 120329

............................................................................................................. 26

Table 25. DAL GW 136S 120328

............................................................................................................. 26

Table 26. DAL GW 136L 120328

............................................................................................................. 27

Table 27. DAL GW 137S 120328

............................................................................................................. 27

Table 28. DAL GW 137L 120328

............................................................................................................. 28

Table 29. DAL GW 138S 120329

............................................................................................................. 28

Table 30. DAL GW 138L 120330

............................................................................................................. 29

Table 31. DAL GW 140R 120329

............................................................................................................ 29

Table 32. DAL GW 141R 120403

............................................................................................................. 30

Table 33. DAL GW 142R 120328

............................................................................................................. 30

Table 34. DAL GW 143R 120329

............................................................................................................. 31

Table 35. DAL GW 144R 120328

............................................................................................................. 31

Table 36. DAL GW 145R 120328

............................................................................................................. 31

Table 37. Trip Blank 1

................................................................................................ ................................ 32

Table 38. Rinseate Blanks

........................................................................................................................ 32

## 1 **Study Information**

Sponsor

3M Company

Sponsor Representative

Gary Hohenstein

3M EHS Operations

3M Building 224

- 5W

- 03

Saint Paul, MN 55144

- 1000

Phone: (651) 737

- 3570

Study Director

Jaisimha Kesari, P.E., DEE

Weston Solutions, Inc.

West Chester, PA 19380

Phone: (610) 701

- 3761

Fax: (610) 701

- 7401

j.kesari@westonsolutions.com

Study Location

Testing Facility

3M EHS Operations

3M Environmental La boratory

Building 260

- 5N

- 17

St. Paul, MN 55144

Study Personnel

William K. Reagen, Ph.D., 3M Laboratory Manager

Cleston Lange, Ph.D., Principal Analytical Investigator, ( clange@mmm.com ); phone (651)

- 733

- 9860

Susan Wolf, 3M Analyst

Chelsie Grochow, Analyst

Kelly Ukes, Analyst

Study Dates

Study Initiation: March 8, 2010

Interim 2 8 Experimental Termination: April 18, 2012

Interim Report Completion: Date of Interim Report Signing

Location of Archives

All original raw data an d the analytical report have been archived at the 3M Environmental

Laboratory according to 40 CFR Part 792. The test substance and analytical reference standard

reserve samples are archived at the 3M Environmental Laboratory according to 40 CFR Part 792

## 2 S **ummary**

The 3M Environmental Laboratory received groundwater samples from wells located at the Former

Sludge Incorporation Area (FSIA) in Decatur, AL, for pe rfluorobutane sulfonate (PFBS)

, perfluorohexane sulfonate (PFHS), and perfluoro octane sulfonate (PFOS) analysis under 3M project

GLP10

- 01

- 02

- 28

. A total of seventy

- seven sample bottles were prepared for thirty

- two different

sampling locations. Samples were not collected at five of the specified sampling locations; DAL GW

136R, DAL GW 137R, DAL GW 138R, DAL GW CW26C, and DAL GW CW26L. Samples were

collected March 27

- April 3, 2012 and received from Weston personnel on April 4, 2012.

Based on previous sampling of the FSIA (GLP10

- 01

- 02

- 07, GLP10

- 01

- 02

- 11, GLP10

- 01

- 02

- 18, and

GLP10

- 01

- 02

- 24), the 3M Environmental Laboratory has demonstrated through the use of field matrix

spikes, that 3M Environmental Laboratory Method ETS -8- 044.1 used for the analysis of the samples is

appropriate for the sample matrix. Therefore, for this sampling event, field matrix spike (FMS) samples

were prepared for six of the twenty

- seven sampling locations collected. In addition, a trip blank set

containing Milli

- QTM water and appropriate trip blank spikes were prepared, as well as two equipment

rinseate blanks. The equipment rinseate blanks did not have FMS samples prepared for determination

of PFBS, PFHS, or PFOS recovery.

All of the samples were prepared and analyzed for PFBS, PFHS, and PFOS following 3 M

Environmental Laboratory Method ETS -8- 044. 1. Many of the ground water samples required dilution to

attain PFBS, PFHS, and/or P FO S concentrations within the range of the curve, in some instances up to

100

- fold dilution were required. The average measured PFBS, PFHS, and PFOS concentrations are

summarized in Table 1. The equipment rinseate blanks and the trip blank sample were below the

lower limit of quantitation (LLOQ) for PFBS, PFHS, and PFOS , indicating adequate control of sample

contamination during shipping and sample collections.

The PFBS concentration results for all groundwater samples ranged from <0.0250 ng/mL to 67.3

ng/mL. The PFHS concentration results for all groundwater samples ranged from 0.0649 ng/mL to 765

ng/mL. The PFOS concentration results for all groundwater samples ranged from 0.355 ng/mL to 1120

n g/mL. The analytical method uncertainties associated with the reported results using external

calibration are: PFBS ± 21 %, PFHS ± 17 %, and PFOS ± 34 %. For sample DAL GW 13 3 L, the method

uncertainty was expanded for PFOS to ± 35%

. Samples analyzed using internal calibration for PFOS

have an analytical method uncertainty of ± 11 %.

### Table 1

. S ummarized PF BS, PFHS, and PFOS Results ( FSIA G W

, March/April 2012 ). (1)

PFBS PFHS PFOS

Sampl ing Location Avg. Conc

. (ng/mL) ± RPD Avg. Con c.

(ng/mL) ± RPD Avg. Conc

. (ng/mL) ± RPD



### DAL GW 130R 120329 14.6 ± 2.1% 119 ± 0.84% 4 05 ± 8.7 %

DAL GW 130S 120329 3.54 ± 3.1 % 31.0 ± 0.65 % 81.0 ± 1.7 %

DAL GW 130L 120329 1.53 ± 0 .0% 12.7 ± 0.79 % 50.7 ± 3.2 %

DAL GW 131R 120328 40.2 ± 3.0 % 234 ± 4.7 % 1120 ± 6.3 %

DAL GW 131S 120328 0.860 ± 6.3 % 4.75 ± 2.9 % 5.80 ± 6.4 % (2)



### DAL GW 131L 120328 3.13 ± 0.64 % 20.6 ± 3.4 % 27.3 ± 5.9 %

DAL GW 133R 120328 3.04 ± 4.6 % 19.6 ± 0.51 % 21.2 ± 2.8 %

DAL GW 133S 120328 8.57 ± 5.7 % 61.6 ± 1.8 % 160 ± 6 .9%

DAL GW 133L 1 20328 5.87 ± 0.51 % 39.1 ± 3.3 % 70.1 ± 0.57 % (3)



### DAL GW 134R 120328 55.9 ± 2.0 % 556 ± 4.9 % 1030 ± 16 %

DAL GW 134S 120327 0.428 ± 0.93 % 4.77 ± 0.42 % 13.7 ± 6.6 %

DAL GW 134L 120327 9.80 ± 6.2 % 103 ± 0.98 % 328 ± 5.2 %

DAL GW 135R 120329 67.3 ± 2.4 % 765 ± 4. 2 % 974 ± 7.4 %

DAL GW 135S 120329 2.17 ± 1.8 % 21.2 ± 2.4 % 13.5 ± 5.2 %

DAL GW 135L 120329 10.9 ± 0.92 % 79.4 ± 3.2 % 59.0 ± 19 %

DAL GW 136S 120328 2.58 ± 0.39 % 17.7 ± 3.4 % 66.1 ± 24 % (4)



### DAL GW 136L 120328 2.32 ± 0.43 % 16.0 ± 1.3 % 60.9 ± 2 .0%

DAL GW 137S 120328 0.109 ± 1.8 % 0.0922 ± 6.1 % 0.355 ± 6.2 % (2 )



### DAL GW 137L 120328 0.0679 ± 0.74 % 0.0876 ± 3.5 % 1.27 ± 3.1 % (2 )



### DAL GW 138S 120329 5.38 ± 9.3 % 26.4 ± 1.9 % 200 ± 15 %

DAL GW 138L 120330 <0.0250 0.0649 ± 5.2 % 0.396 ± 12 % (2 )



### DAL GW 140R 120329 1.59 ± 2 .5 % 17.7 ± 1.1 % 106 ± 16 %

DAL GW 141R 120403 0.0369 ± 2.7 % 0.303 ± 6.3 % 5.59 ± 11 % (2)



### DAL GW 142R 120328 2.08 ± 4.8 % 11.2 ± 2.7 % 55.4 ± 28 % (4)



### DAL GW 143R 120329 1.56 ± 0.64 % 5.69 ± 3.0 % 13.7 ± 9.5 %

DAL GW 144R 120328 1.99 ± 3.0 % 9.03 ± 3.0 % 13.4 ± 7 .5 %

DAL GW 145R 120328 1.50 ± 1.3 % 7.55 ± 0.79 % 49.1 ± 4.3 %

Trip Blank (Milli

- Q Water) <0.0250 <0.0250 <0.0232

DAL GW 130R RB 120329 <0.0250 <0.0250 <0.0232

DAL GW 135R RB 120329 <0.0250 <0.0250 <0.0232

(1) All samples reported using external st andard calibration unless noted otherwise. The analytical method

uncertainties associated with the reported results using external calibration are: PFBS ± 21%, PFHS ± 17%,

and PFOS ± 34 %.

(2) Sample results reported using internal standard calibration. The a nalytical method uncertainty associated with

the reported results using internal calibration is ± 11 % for PFOS

. (3) The FMS recovery did not meet acceptance criteria. The analytical uncertainty has been adjusted for PFOS

to ± 35 %.

(4) The RPD did not meet method acceptance criteria of ≤ 20%.

## 3 **Introduction**

This analytical study was conducted as part of the Phase 3 En vironmental M onitoring and A ssessment

P rogram for the 3M facility located in Decatur, Alabama. The objective of the overall program is to gain

information regarding concentrations of perfluorooctane sulfonate (PFOS), perfluorohexane sulfonate

(PFHS) and perfluorobutane sulfonate (PFBS)

, in various environmental media such as groundwater

, soils and sediments that are associated with and near the Decatur facility. This analytical study was

conducted to analyze groundwater samples collected from various wells located at the Former Sludge

Incorporation Area (FSIA) in Decatur, AL for PFBS, PFHS, and PFOS in an effort to characterize on

site groundwater conditions.

The 3M Environmental Laboratory prepared sample containers (250 mL high

- density polyethylene

bottles) which were shipped to Decatur, AL Weston personnel prior to field sampling. Sample bottle sets

for each groundwater sampling location included a field sample and field sample duplicate. A field

matrix spike samples was prepared for six of the twenty

- seven sampling locations collected. Each

empty container for groundwater samplings was marked with a “fill to here” line to produce a final

sample volume of 200 mL. Con tainers designated for field matrix samples were fortified with an

appropriate matrix spike solution containing PFBS (linear), PFHS (linear), and PFOS (linear and

branched ) prior to being sent to the field for sample collection. See section 8.8 of the rep ort for field

matrix spike levels

.

Samples were prepared and analyzed according to the procedure defined in 3M Environmental

Laboratory method ETS -8- 044.1 “Method of Analysis for the Determination of Perfluorinated

Compounds In Water by High Performance Liquid Chromatography/Mass Spectrometry Direct Injection

Analysis”. Sampling locations with PFOS results <10 ng/mL were reanlyzed with the addition of

internal standard 13 

### C 8 

### - PFOS to used to aid i n the data quality objectives. Sampling locations repor ted

for PFOS using internal standard calibraton included: DAL GW 131S, DAL GW 137S, DAL GW 137L,

DAL GW 138L, and DAL GW 141R.

Table 1 summarizes the average PFBS, PFHS, and PFOS concentrations for the duplicate samples

collected, the trip blank, and the aqueous equipment rinseate blank samples. Tables 10

- 38 summarize

the individual sample results and the associated field matrix spike recoveries. All results for the quality

control samples prepared and analyzed with the samples are reported and discussed elsewhere in this

report

.



## 4 **Test & Control Substances**

There was not a test substance or control substances in the classic sense of a GLP study. This study

was purely analytical in nature.

## 5 **Reference Substances**

Reference Substance PFBS

(predominantly lin ear

isomer) PFHS

(linear isomer) PFOS

(branched and linear

isomers)

Chemical Name Perfluorobutane sulfonate Perfluorohexane sulfonate Perfluorooctane sulfonate

Chemical Formula 

### C 4 

### F 9 

### SO 3 - 

### K + 

### C 6 

### F 13 

### SO 3 - 

### Na + 

### C 8 

### F 17 

### SO 3 - 

### K +

Identifier NA L- PFHXS Br- PFOSK

Use Tar get Analyte Reference

Standard Target Analyte Reference

Standard Target Analyte Reference

Standard

Source 3M Wellington Wellington

Expiration Date 1/10/2017 3/25/2018 3/17/2014 ; 12/01/2014

Storage Conditions Frozen Frozen Frozen

Chemical Lot Number 41

- 2600

- 8442 -5 LPFHxSAM08 brPFOSK0708 ;

brPFOSK1111

TCR Number TCR

- 121 TCR08

- 0018 TCR11

- 0010 ; TCR11

- 0041

Physical Description White Powder Crystalline Liquid

Purity 96.7% 100% 99.9%

Reference Substance PFOS

(branched and linear

isomers) 13 C

8 - PFOS 13 C

8 - PFO S

Chemical Name Perfluorooctane sulfonate Sodium

Perfluorooctanesulfonate Sodium

Perfluorooctanesulfonate

Chemical Formula 

### C 8 

### F 17 

### SO 3 - 

### K + 13 

### C 8 

### F 17 

### SO 3 - 

### Na + 13 

### C 8 

### F 17 

### SO 3 - 

### Na +

Identifier CAS # 2795

- 39 -3 MPFC -C- 0511 MPFC -C-0 112

Use FMS Reference Standard Internal Standard Internal Standard

Source Aldrich Wellington Wellington

Expiration Date 2/4/2014 05/25/2014 01/24/2015

Storage Conditions Room Temperature Frozen Frozen

Chemical Lot Number 1424328V 052411 012312

TCR Number TCR11

- 0028 TCR11

- 0016 TCR12

- 0004

P hysical Description White Powder Liquid Liquid

Purity 99.7% NA (1 ) NA (1 )

(1) Custom mixture of seven mass

- labeled ( 13 C) perfluoroalkylcarboxylic acids, two mass labeled ( 13 C) perfluoroalkylsulfonates

and one mass

- labeled ( 13 C ) perfluoro -1- octanesulfon amide at a nominal concentration of 5 μ g/mL

## 6 **Test System**

The test systems for this study are groundwater samples collected from wells located in Decatur, AL by

Weston Solutions, Inc. personnel. Samples for this study are “real world” samples, not dosed with a

specific lot of test substance.

Table 2 . Sample Description Key Code.

String Number String Descriptor Example

**Example DAL GW 131R**  **0 120328** 

**1 Sampling Location**  **DAL =**  **Decatur, Alabama**

**2 Well ID**  **GW = Groundwater**

**3 Sampl**  **ing Point**  **Example: 131** 

**4 Well Level**  **R = Residuum shallow water**

**- bearing zone** 

**L = Bedrock water**

**- bearing zone** 

**S = Epika rst middle water** 

**- bearing zone** 

**5 Sampling Date**  **120328**

**– March 28, 2012** 

**6 Sample Type**  **0=primary sample**

**DB =duplicate sample** 

**FMS = Field Matri x Spike** 



## 7 **Method Summary**

7.1 Methods

Analysis for all analytes was completed following 3M Environmental Laboratory method ETS -8- 044. 1

“ Method of Analysis for the Determination of Perfluorinated Compounds In Water by High Performance

Liquid Chromatography/Mass Spectrometry Direct Injection Analysis”.



## 7.2 Sample Collection

Samples were collected in 250 mL Nalgene TM 

### (high

- density polyethylene) bottles prepared at the 3M

Environmental Laboratory. Sample bottles associated with GLP10

- 01

- 02

- 28 were returned to the

laboratory at ambient conditions on April 4, 2012. Samples were stored refrigerated at the laboratory

after receipt. A set of laboratory prepared Trip Blank and Trip Blank field matrix spikes were sent with

the sample collection bottles.



## 7.3 Sample Preparat ion

Samples were prepared by removing an aliquot of the well mixed sample and placing it in an autovial for

analysis.

Samples that required dilution were prepared by diluting 1 mL sample with 9 mLs of Milli

- Q water

(dilution 1:10). Samples requiring a 1:100 dilution were prepared by diluting 0.1 mL of sample with 9.9

mLs of MilliQ water

Sampling locations with PFOS results <10 ng/mL were reported using internal standard calibraton.

Sampling locations DAL GW 131S, DAL GW 137S, DAL GW 137L, DAL GW 138L, and DAL GW 141R

were prepared by adding an aliquot of a separate internal standard spiking solution to 10 mLs of sample

(nominal concentration of 1 ng/mL). Laboratory control samples were prepared in a similar manner.

## 7.4 Analysis

All study samples and quality control samples were analyzed for PFBS, PFHS, and PFOS using high

performance liquid chromatography/ tandem mass spectrometry (HPLC/MS/MS). Detailed instrument

parameters, the liquid chromatography gradient program, and the specific mass transitions analyzed are

described in the raw data hard copies placed in the final data packet, and are briefly described below in

Table 3

, Table 4 and Table 5

.

Table 3 . Instrument Parameters

.

Instrument Name ETS Buster ETS Buster

Analytical M ethod Followed ETS

- 8

- 044.1 ETS

- 8

- 044.1

Analysis Date 4/11/12

- PFBS, PFHS, and PFOS

External standard calibration 4/17 /12

- PFOS

Internal standard calibration

Liquid Chromatograph Agilent 1100 Agilent 1100

Guard column Betasil C18 ( 4.6 mm X 100 mm), 5 μ Betasil C18 ( 4.6 mm X 100 mm), 5 μ

Analytical column Betasil C18 ( 4.6 mm X 100 mm), 5 μ Betasil C18 ( 4.6 mm X 100 mm), 5 μ

Injection Volume 10 μ L 5 0 μ L

Mass Spectrometer Applied Biosystems API 40 00 Applied Biosystems API 40 00

Ion Source Turbo Spray Turb o Spray

Electrode Turbo ion electrode Turbo ion electrode

Polarity Negative Negative

Software Analyst 1. 4 .2 Analyst 1. 4 .2



### Table 4. Liquid Chromatography Conditions

.

**Step**

**Number Total Time** 

**(min) Flow Rate** 

**(** μ

**L/min) Percent A** 

**(2 mM ammonium acetate)**  **Percent B**

**(Methanol)**

ETS

- 8

- 044.1

0 0 750 97.0 3 .0

1 2 .0 750 97 .0 3 .0

2 14.5 750 5. 0 9 5 .0

3 15.5 750 5. 0 95 .0

4 16.5 750 97 .0 3 .0

5 20 .0 750 97 .0 3 .0

### Table 5. Mass Transitions.

**Analyte Mass Transition** 

**Q1/Q3 Reference** 

**Material**

**Structure Internal** 

**Standard Mass Transition** 

**Q1/Q3**

**PFBS 299/80** 

**Linear NA**  **NA 299/99** 

**PFHS 399/80** 

**Linear NA**  **NA 399/99** 

**PFOS 499/80** 

**Branched and**

**Linear [**  **13 C**  **8 ]**  **PFOS (1)**  **507/80 499/99** 

**499/130**

Dwell time was 5 0 or 75 msec for each transition. The individual transitions were summed to produce a “total

ion chromatogram” (TIC), which was used for quantitation.

(1) In ternal standard was only used for PFOS for sampling locations DAL GW 13 1 S, DAL GW 137 S , DAL

GW 13 7 L, DAL GW 138L, and DAL GW 141 R.



## 8 **Analytical Results**

8.1 Calibration

Samples were analyzed against an external standard calibration curve. Calibration standards were

prepared by spiking known amounts of the stock solution containing the target analytes into prepared in

Milli Q water. A total of fifteen spiked standards ranging from 0.025 ng/mL to 100 ng/mL (nominal) were

analyzed. A quadratic, 1/x weighted, calibration curve of the peak area counts was used to fit the data

for each analyte. The data were not forced through zero during the fitting process. Calculating the

standard concentrations using the peak area confirmed accuracy of each curve point.

For sampling locations DAL GW 13 1 S, DAL GW 137 S , DAL GW 13 7 L, DAL GW 1 38L , and DAL GW

14 1R for PFOS: Samples were analyzed using a stable isotope internal standard calibration curve

. Calibration standards were prepared by spiking known amounts of the stock solution containing the

target analytes into a laboratory

- prepared synthetic groundwater containing calcium and magnesium. A

separate internal standard spiking solution was prepared and an aliquot was added at the same level to

all calibration standards at a nominal concentration of 1 ng/mL. A calibration curve ranging from 0.025

ng/mL to 100 ng/mL (nomina l) was analyzed

. A quadratic, 1/x weighted, calibration curve of the peak

area ratio was used to fit the data for PFOS . The data were not forced through zero during the fitting

process. Calculating the standard concentrations using the peak area ratios and the resultant calibration

curve confirmed accuracy of each curve point.

For both exter na l and internal standard calibration, each curve point was quantitated using the overall

calibration curve and reviewed for accuracy. Method calibration accuracy requirements of 100±25%

(100±30% for the lowest curve point) were met for all analytes. The correlation coefficient (r) was

greater than 0.995 for all analytes.



## 8.2 System Suitability

A calibration standard was analyzed four times at the beginning of each analytical sequence to

demonstrate overall system suitability. The acceptance criteria of less than or equal to 5% relative

standard deviation (RSD) for peak area or peak area ratio and retention time criteria of less than or

equal to 2% RSD was met fo r PFBS, PFHS, and PFOS

.

## 8.3 Limit of Quantitation (LOQ)

The LOQ for this analysis is the lowest non

- zero calibration standard in the curve that meets linearity

and accuracy requirements and for which the area counts or area ratio are at least twice those of t he

appropriate blanks. The LOQ for all analytes can be found in Table 6

.

Table 6 . Limit of Quantitation (LOQ).

**Analysis Date Dilution**  **PFBS**

**LOQ, ng/mL PFHS** 

**LOQ, ng/mL PFOS** 

**LOQ, ng/mL**

4/11/12 Analysis

–

external standard

calibration 1 0.0250 0.0250 0.0232

10 0.250 0.250 0.232

100 2.50 2.50 2.32

4/17 /12 Analysis

–

internal standard

calibration 1 NA NA 0.0 232

NA = Not Applicable



## 8.4 Continuing Calibration

During the course of each analytical sequence, continuing calibration verification samples (CCVs) were

analyzed to confirm that the instrument response and the initial calibration curve were still in control. All

CCVs met method criteria of 100% ± 25% for PFBS, PFHS, and PFOS

.



## 8.5 Blanks

Three types of blanks were prepared and analyzed with the samples: solvent blanks, trip blanks, and

equipment rinseate blanks. Solvent blank results were reviewed and used to evaluate method

performance to determi ne the LOQ for PFBS, PFHS, and PFOS . Trip blanks reflect the shipping and

sample collection conditons the sample bottles and samples experience. Equipment rinseate blanks

are aqueous samples that reflect the efficiency of equipment cleaning in the field between different

sample collections and are proof of no cross contamination of samples from the equipment.



## 8.6 Lab Control Spikes (LCSs)

Low, mid, and high

- level lab control spikes were prepared and analyzed in triplicate with each

preparation set. LCSs analyzed with the sample dilutions by exteranl standard calibration were

prepared by spiking known amounts of PFBS, PFHS, and PFOS into Milli Q water to produce the

desired concentration. LCSs analyzed by internal standard calibration were prepared by spiking known

amounts of PFOS into synthetic groundwater and adding an aliquot of a separa te internal standard

spiking solution at a nominal concentration of 1 ng/mL.

The method acceptance criteria, average of LCS at each level should be within 100% ± 20% with an

RSD ≤20%, were met for all LCS samples.

The following calculations were used to generate data in Table 7 for laboratory control spikes.

100% *

ion Concentrat Spike ion Concentrat Calculated Recovery Percent LC S

=

100% *

recovery LCS average replicates LCS deviation standard RSD LCS%

=

### Table 7. Laboratory Control Spike Recovery

.

ETS -8- 044.1

Analyzed 4/11/12

External standard

calibration **PFBS PFHS** 

**Lab ID Spiked** 

**Concentration**

**(ng/mL) Calculated** 

**Concentration**

**(ng/mL) %Recovery**  **Spiked**

**Concentration**

**(ng/mL) Calculated** 

**Concentration**

**(ng/mL) %Recovery** 

LCS

- 120405

- 1 0.498 0.494 99.2 0.497 0.476 95.7

LCS

- 120405

- 2 0.498 0.471 94.6 0.497 0.478 96.2

LCS

- 12 0405

- 3 0.498 0.475 95.3 0.497 0.498 100

Average ± %RSD 96.4 ± 2.6 % 97.3 % ± 2.4 %

LCS

- 120405

- 4 4.98 5.00 100 4.97 5.00 101

LCS

- 120405

- 5 4.98 4.99 100 4.97 4.92 98.9

LCS

- 120405

- 6 4.98 4.98 100 4.97 4.88 98.2

Average ± %RSD 100 % ± 0.0 % 99.4 % ± 1.5 %

LCS

- 1 20405

- 7 29.9 29.7 99.4 29.9 28.4 94.8

LCS

- 120405

- 8 29.9 29.2 97.6 29.9 27.6 92.2

LCS

- 120405

- 9 29.9 29.7 99.2 29.9 28.9 96.7

Average ± %RSD 98.7 % ± 1 .0 % 94.6 % ± 2.4 %

ETS -8- 044.1

Analyzed 4/11/12

External standard

calibration **PFOS (Branched and Linear)**

**Lab ID Spiked** 

**Concentration**

**(ng/mL) Calculated** 

**Concentration**

**(ng/mL) %Recovery** 

LCS

- 120405

- 1 0.462 0.386 83.6

LCS

- 120405

- 2 0.462 0.390 84.3

LCS

- 120405

- 3 0.462 0.367 79.5

Average ± %RSD 82.5 % ± 3.1 %

LCS

- 120405

- 4 4.62 4.55 98.4

LCS

- 120405

- 5 4.62 4 .48 97.0

LCS

- 120405

- 6 4.62 4.49 97.1

Average ± %RSD 97.5 % ± 0.80 %

LCS

- 120405

- 7 27.8 26.0 93.6

LCS

- 120405

- 8 27.8 26.4 95.0

LCS

- 120405

- 9 27.8 27.0 97.0

Average ± %RSD 95.2 % ± 1.8 %

### Table 7 continued

. Laboratory Control Spike Recovery

.

ETS -8- 044.1

Analyzed 4/17/12

Internal standard

calibration **PFOS (Branched and Linear)**

**Lab ID Spiked** 

**Concentration**

**(ng/mL) Calculated** 

**Concentration**

**(ng/mL) %Recovery** 

LCS

- 120416

- 1 0.184 0.177 96.3

LCS

- 120416

- 2 0.184 0.186 101

LCS

- 120416

- 3 0.184 0.196 106

Averag e ± %RSD 101 ± 4.8 %

LCS

- 120416

- 4 1.84 1.83 99.5

LCS

- 120416

- 5 1.84 1.84 100

LCS

- 120416

- 6 1.84 1.90 103

Average ± %RSD 101 % ± 1.9 %

LCS

- 120416

- 7 9.22 9.24 100

LCS

- 120416

- 8 9.22 8.60 93.3

LCS

- 120416

- 9 9.22 9.06 98.3

Average ± %RSD 97.2 % ± 3.6 %



## 8.7 Analyt ical Method Uncertainty

Analytical uncertainty is based on historical QC data that is control charted and used to evaluate

method accuracy and precision. The method uncertainty is calculated following ETS

- 12

- 012.2. The

standard deviation is calculated for the set of accuracy results (in %) obtained for the QC samples. The

method uncertainty is calculated by multiplying the standard deviation by a factor of 2, which

corresponds to a confidence level of 95%. The most recent 50 data points were used to generate the

method uncertainty values listed in Table 8.

Table 8 . Analytical Uncertainty.

Analyte Calibration Standard Deviation Method Uncertainty

PFBS External 10.7 ± 21%

PFHS External 8.69 ± 17%

PFOS External 17.2 ± 34%

PFOS Internal 5.49 ±11%

## 8.8 Field Matrix Spikes (FMS)

A field matrix spike (FMS) was collected at six sampling locations to verify that the analytical method is

applicable to the collected matrix. Field matrix spikes were generated by adding a measured volume of

field sample to a container spiked by the laboratory with PFBS (linear), PFHS (linear), and PFOS (linear

and branched) prior to shipping sample containers for sample collection. Field matrix spike recoveries

within method acceptance criteria of 100±30% confirm that “unknown” components in the sample matrix

do not significantly interfere with the extraction and analysis of the analytes of interest. Field matrix spike

concentrations must be 50% of the sample concentration to be considered an appropriate field spike.

Field matrix spikes are presented in section 9 of this report.

Table 9 . Field Matrix Spike Levels.

Sampling Location Spike Level PFBS , ng/mL PFHS , ng/mL PFOS , ng/mL

137S and 138L FMS 1.00 0.998 0.998

141R FMS 5.00 4.99 4.99

134S FMS 10.0 9.98 9.98

130L FMS 50.0 49.9 49.9

133L FMS 200 200 200

Trip Blank Low 1.00 0.998 0.998

Mid 10.0 9.98 9.98

Mid

- High 50.0 49.9 49.9

High 200 200 200

100%*

on Concentrat Spike Dup.) Sample Field & Sample Field:ion Concentrat Average FMS ofion Concentrat (Sample Recovery FMS −

=



## 9 **Data Summary and Discussion**

The tables below summarize the sample results and field matrix spike recoveries for the sampling

locations as well as the Trip Blanks and rinseate blanks. Results and average values are rounded to

three significant figures according to EPA rounding rules. Because of rounding, values may vary slightly

from those listed in the raw data. Field matrix spike recoveries meeting the method acceptance criteria

of ± 30%, demonstrate that the method was appropriate for the given matrix and their respective

quantitative ranges

.

DAL GW 13 3 L

– The recovery of the FMS sample for PFOS was 65.5 %. Based on the recovery of the

FMS sample, the analytical uncertainty has been expanded for PFOS to ± 35 %.

DAL GW 13 6 S

– The sample/sample duplicate RPD was 2 4 % for PFOS.

DAL GW 142R

– The sample/sample duplicate RPD was 2 8 % for PFOS.

DAL GW TRIP01

– The recovery of the MHS FMS sample for PFOS was 67.1%.

### Table 10 . DAL GW 130R 1 20329

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) Concentration** 

**(ng/mL) C**  **oncentration**

**(ng/mL)**

GLP10

- 01

- 02

- 28

- 001 DAL

- GW

- 130R

- 0

- 120329 14.4 119 387

GLP10

- 01

- 02

- 28

- 002 DAL

- GW

- 130R

- DB

- 120329 14.7 118 422

**Average Concentration (ng/mL) ±**  **%RPD 14.6**  **ng/mL ±**  **2.1% 119**  **ng/mL ±**  **0.84% 405**  **ng/mL ±**  **8.7%**

Samples diluted 1:100 and report ed by external standard calibration.



### Table 11 . DAL GW 130S 120329

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) Concentration** 

**(ng/mL) Concentration** 

**(ng/mL)**

GLP10

- 01

- 02

- 28

- 003 DAL

- GW

- 130S

- 0

- 120329 3.48 30.9 80. 3

GLP10

- 01

- 02

- 28

- 004 DAL

- GW

- 130S

- DB

- 120329 3.59 31.1 81.7

**Average Concentration (ng/mL) ±**  **%RPD 3.54**  **ng/mL ±**  **3.1% 31.0**  **ng/mL ±**  **0.65% 81.0**  **ng/mL ±**  **1.7%**

Sample s diluted 1:10 and reported by external standard calibration.

### Table 12 . DAL GW 130L 1 20329

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery** 

GLP10

- 01

- 02

- 28

- 005 DAL

- GW

- 130L

- 0

- 120329 1.53 NA 12.7 NA 51.5 NA

GLP10

- 01

- 02

- 28

- 006 DAL

- GW

- 13 0L

- DB

- 120329 1.53 NA 12.6 NA 49.9 NA

GLP10

- 01

- 02

- 28

- 007 DAL

- GW

- 130L

- FMS

- 120329 48.4 93.7 60.5 95.9 85.5 69.7

**Averag e Concentration (ng/mL)**  **± %RPD**  **1.53 ng/mL**  **± 0**  **.0% 12.7**  **ng/mL ±**  **0.79 %**  **50.7 ng/mL**  **± 3.2**  **%**

NA = Not Applicable

Samples diluted 1:10 and report ed by external standard calibration.



### Table 13 . DAL GW 131R 1 20328

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) Concentration** 

**(ng/mL) Concentration** 

**(ng/mL)**

GLP10

- 01

- 02

- 28

- 008 DAL

- GW

- 131R

- 0

- 120328 39.6 228 1080

GLP10

- 01

- 02

- 28

- 009 DAL

- GW

- 131R

- DB

- 120328 40.8 239 1150

**Average Concentration (ng/mL) ±**  **%RPD 40.2**  **ng/mL ±**  **3.0% 234**  **ng/mL ±**  **4.7% 1120**  **ng/mL ±**  **6.3%**

Samples diluted 1:100 and reported by external standard calibration.

### T able 14

. DAL GW 131S 1 20328

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) Concentration** 

**(ng/mL) Concentration** 

**(ng/mL)**

GLP10

- 01

- 02

- 28

- 010 DAL

- GW

- 131S

- 0

- 120328 0.833 4.82 5.98

GLP10

- 01

- 02

- 28

- 011 DAL

- GW

- 131S

- DB

- 120328 0.887 4.68 5.61

**Average Con centration (ng/mL)**  **± %RPD**  **0.860 ng/mL**  **± 6.3%**  **4.75 ng/mL**  **± 2.9%**  **5.80 ng/mL**  **± 6.4%** 

R esults for PFBS and PFHS are reported by external standard calibration. R esults for PFOS are reported by internal standard calibration.



### Table 15

. DAL GW 131L 1 20328

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) Concentration** 

**(ng/mL) Concentration** 

**(ng/mL)**

GLP10

- 01

- 02

- 28

- 012 DAL

- GW

- 131L

- 0

- 120328 3.14 20.9 28.1

GLP10

- 01

- 02

- 28

- 013 DAL

- GW

- 131L

- DB

- 120328 3.12 20.2 26.5

**Average Conc entration (ng/mL)**  **± %RPD**  **3.13 ng/mL**  **± 0.64%**  **20.6 ng/mL**  **± 3.4%**  **27.3 ng/mL**  **± 5.9%** 

Samples diluted 1:10 and reported by external standard calibration.



### Table 16 . DAL GW 133R 120328

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentrat ion** 

**(ng/mL) Concentration** 

**(ng/mL) Concentration** 

**(ng/mL)**

GLP10

- 01

- 02

- 28

- 014 DAL

- GW

- 133R

- 0

- 120328 3.11 19.5 20.9

GLP10

- 01

- 02

- 28

- 015 DAL

- GW

- 133R

- DB

- 120328 2.97 19.6 21.5

**Average Concentration (ng/mL) ±**  **%RPD 3.04**  **ng/mL ±**  **4.6% 19.6**  **ng/mL ±**  **0.51% 21.2**  **ng/m L**  **± 2.8%** 

Samples diluted 1:10 and reported by external standard calibration.

### Table 17 . DAL GW 133S 1 20328

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) Concentration** 

**(ng/mL) Concentration** 

**(ng/mL)**

GLP10

- 01

- 02

- 28

- 016 DAL

- GW

- 133S

- 0

- 120328 8.32 61.0 154

GLP10

- 01

- 02

- 28

- 017 DAL

- GW

- 133S

- DB

- 120328 8.81 62.1 165

**Average Concentration (ng/mL) ±**  **%RPD 8.57**  **ng/mL ±**  **5.7% 61.6**  **ng/mL ±**  **1.8% 160**  **ng/mL ±**  **6.9%**

Samples diluted 1:10 and reported by external standard calibrati on.



### Table 18 . DAL GW 133L 1 20328

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery** 

GLP10

- 01

- 02

- 28

- 018 DAL

- GW

- 133L

- 0

- 120328 5.85 NA 38.4 NA 70.3 NA

GLP10

- 01

- 02

- 28

- 019 DAL

- GW

- 133L

- DB

- 120328 5.88 NA 39.7 NA 69.9 NA

GLP10

- 01

- 02

- 28

- 020 DAL

- GW

- 133L

- FMS

- 120328 189 91.6 213 87.0 201 65.5 (1)

**Average Concentration (ng/mL) ±**  **%RPD 5.87**  **ng/mL ±**  **0.51 %**  **39.1 ng/mL**  **± 3.3**  **% 70.1**  **ng/mL ±**  **0.57 %**  **(2)**

NA = Not Applicable

Sample and sample duplicate diluted 1:10. FMS diluted 1:100. All samples reported by external standard calibration.

(1) FMS did not meet method acceptance criteria of 100 ± 30%.

(2) Method uncertainty is expanded to ± 35% for PFOS.

### Table 19 . DAL GW 134R 1 20328

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) Concentration** 

**(ng/mL) Concentration** 

**(ng/mL)**

GLP10

- 01

- 02

- 28

- 021 DAL

- GW

- 134R

- 0

- 120328 55.3 569 1110

GLP10

- 01

- 02

- 28

- 022 DAL

- GW

- 134R

- DB

- 120328 56. 4 542 947

**Average Concentration (ng/mL) ±**  **%RPD 55.9**  **ng/mL ±**  **2.0% 556**  **ng/mL ±**  **4.9% 1030**  **ng/mL ±**  **16%**

Samples diluted 1:100 and reported by external standard calibration.



### Table 20 . DALGW 134S 1 20327

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID De**  **scription Concentration** 

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery** 

GLP10

- 01

- 02

- 28

- 023 DAL

- GW

- 134S

- 0

- 120327 0.426 NA 4.78 NA 14.1 NA

GLP10

- 01

- 02

- 28

- 024 DAL

- GW

- 134S

- DB

- 120327 0.430 NA 4.76 NA 13.2 NA

GLP10

- 01

- 02

- 2 8

- 025 DAL

- GW

- 134S

- FMS

- 120327 9.99 95.6 14.4 96.5 20.9 72.6

**Average Concentration (ng/mL) ±**  **%RPD 0.428**  **ng/mL ±**  **0.93 %**  **4.77 ng/mL**  **± 0.42**  **% 13.7**  **ng/mL ±**  **6.6 %** 

NA = Not Applicable

Samples reported by external standard calibration.

### Table 21 . DAL GW 134L 1 20327

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) Concentration** 

**(ng/mL) Concentration** 

**(ng/mL)**

GLP10

- 01

- 02

- 28

- 026 DAL

- GW

- 134L

- 0

- 120327 10.1 103 336

GLP10

- 01

- 02

- 28

- 027 DAL

- GW

- 134L

- DB

- 120327 9.49 102 319

**Average C oncentration (ng/mL)**  **± %RPD**  **9.80 ng/mL**  **± 6.2%**  **103 ng/mL**  **± 0.98%**  **328 ng/mL**  **± 5.2%** 

Samples diluted 1:100 and reported by external standard calibration.



### Table 22 . DAL GW 135R 1 20329

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) Concentration** 

**(ng/mL) Concentration** 

**(ng/mL)**

GLP10

- 01

- 02

- 28

- 028 DAL

- GW

- 135R

- 0

- 120329 66.5 781 1010

GLP10

- 01

- 02

- 28

- 029 DAL

- GW

- 135R

- DB

- 120329 68.1 749 938

**Average Concentration (ng/mL) ±**  **%RPD 67.3**  **ng/mL ±**  **2.4% 765**  **ng/mL ±**  **4.2% 974**  **ng/mL ±**  **7.4%**

Samples diluted 1:100 and reported by external standard calibration.



### Table 23 . DAL GW 135S 1 20329

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) Concentration** 

**(ng/mL) Concentration** 

**(ng/mL)**

GLP10

- 01

- 02

- 28

- 030 DAL

- GW

- 135S

- 0

- 120329 2.19 21.4 13.1

GLP10

- 01

- 02

- 28

- 031 DAL

- GW

- 135S

- DB

- 120329 2.15 20.9 13.8

**Average Concentration (ng/mL) ±**  **%RPD 2.17**  **ng/mL ±**  **1.8% 21.2**  **ng/mL ±**  **2.4% 13.5**  **ng/mL ±**  **5.2%**

Results reported by external standard calibration.

### Table 24

. DAL GW 135L 1 20329

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) Concentration** 

**(ng/mL) Concentration** 

**(ng/mL)**

GLP10

- 01

- 02

- 28

- 032 DAL

- GW

- 135L

- 0

- 120329 10.8 80.6 64.6

GLP10

- 01

- 02

- 28

- 033 DAL

- GW

- 135L

- DB

- 120329 10.9 7 8.1 53.4

**Average Concentration (ng/mL) ±**  **%RPD 10.9**  **ng/mL ±**  **0.92% 79.4**  **ng/mL ±**  **3.2% 59.0**  **ng/mL ±**  **19%**

Samples diluted 1:10 and reported by external standard calibration.



### Table 25 . DAL GW 136S 1 20328

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID D**  **escription Concentration** 

**(ng/mL) Concentration** 

**(ng/mL) Concentration** 

**(ng/mL)**

GLP10

- 01

- 02

- 28

- 036 DAL

- GW

- 136S

- 0

- 120328 2.58 18.0 73.9

GLP10

- 01

- 02

- 28

- 037 DAL

- GW

- 136S

- DB

- 120328 2.57 17.4 58.2

**Average Concentration (ng/mL) ±**  **%RPD 2.58**  **ng/mL ±**  **0.39% 17.7**  **n g/mL**  **± 3.4%**  **66.1 ng/mL**  **± 24%**  **(1)**

Samples diluted 1:10 and reported by external standard calibration.

(1) Sample/sample duplicate RPD did not meet acceptance criteria of ≤ 20%.

### T able 26 . DAL GW 136L 1 20328

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) Concentration** 

**(ng/mL) Concentration** 

**(ng/mL)**

GLP10

- 01

- 02

- 28

- 038 DAL

- GW

- 136L

- 0

- 120328 2.32 16.1 60.3

GLP10

- 01

- 02

- 28

- 039 DAL

- GW

- 136L

- DB

- 120328 2.31 15.9 61.5

**Average Concentration (ng/mL) ±**  **%RPD 2.32**  **ng/mL ±**  **0.43% 16.0**  **ng/mL ±**  **1.3% 60.9**  **ng/mL ±**  **2.0%**

Samples diluted 1:10 and reported by external standard calibration.



### Table 27 . DAL GW 137S 1 20328

**PF B**  **S PF**  **HS PF**  **OS**

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery** 

GLP10

- 01

- 02

- 28

- 042 DAL

- GW

- 137S

- 0

- 120328 0.108 NA 0.0950 NA 0.366 NA

GLP10

- 01

- 02

- 28

- 043 DAL

- GW

- 137S

- DB

- 120328 0.110 NA 0 .0894 NA 0.344 NA

GLP10

- 01

- 02

- 28

- 044 DAL

- GW

- 137S

- FMS

- 120328 1.10 99.1 0.975 88.5 1.25 89.7

**Average Concentration (ng/mL) ±**  **%RPD 0.109**  **ng/mL ±**  **1.8 %**  **0.0922 ng/mL**  **± 6.1**  **% 0.355**  **ng/mL ±**  **6.2 %** 

NA = Not Applicable

R esults for PFBS and PFHS are reported by external standard calibration. R esults for PFOS are reported by internal standard calibration.

### Table 28 . DAL GW 137L 1 20328

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) Concentration** 

**(ng/mL) Concentration** 

**(n**

**g/mL)**

GLP10

- 01

- 02

- 28

- 045 DAL

- GW

- 137L

- 0

- 120328 0.0681 0.0860 1.25

GLP10

- 01

- 02

- 28

- 046 DAL

- GW

- 137L

- DB

- 120328 0.0676 0.0891 1.29

**Average Concentration (ng/mL) ±**  **%RPD 0.0679**  **ng/mL ±**  **0.74% 0.0876**  **ng/mL ±**  **3.5% 1.27**  **ng/mL ±**  **3.1%**

R esults for PFBS and PFHS are reported by external standard calibration. R esults for PFOS are reported by internal standard calibration.



### Table 29 . DAL GW 138S 1 20329

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) Concentration** 

**(ng/mL) Concentration** 

**(ng/mL)**

GLP10

- 01

- 02

- 28

- 049 DAL

- GW

- 138S

- 0

- 120329 5.63 26.6 214

GLP10

- 01

- 02

- 28

- 050 DAL

- GW

- 138S

- DB

- 120329 5.13 26.1 185

**Average Concentration (ng/mL) ±**  **%RPD 5.38**  **ng/mL ±**  **9.3% 26.4**  **ng/mL ±**  **1.9% 200**  **ng/mL ±**  **15%**

Samples diluted 1:10 and reported by external standard calibration.

### Table 30 . DAL GW 138 L 1 20330

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery** 

GLP10

- 01

- 02

- 28

- 051 DAL

- GW

- 1 3 8L

- 0

- 120330 <0.0250 NA 0.0666 NA 0.419 NA

GLP10

- 01

- 02

- 28

- 052 DAL

- GW

- 138L

- DB

- 120330 <0.0250 NA 0.0632 NA 0.373 NA

GLP10

- 01

- 02

- 28

- 053 DAL

- GW

- 138L

- FMS

- 120330 0.962 96.2 0.931 86.8 1.32 92.6

**Average Concentration (ng/mL) ±**  **%RPD <0.0250**  **ng/mL 0.0649**  **ng/mL ±**  **5 .2%**  **0.396 ng/mL**  **± 12**  **%**

NA = Not Applicable

R esults for PFBS and PFHS are reported by external standard calibration. R esults for PFOS are reported by internal standard calibration.



### Table 31 . DAL GW 1 40R 1 20329

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) Concentration** 

**(ng/mL) Concentration** 

**(ng/mL)**

GLP10

- 01

- 02

- 28

- 058 DAL

- GW

- 140R

- 0

- 120329 1.57 17.6 115

GLP10

- 01

- 02

- 28

- 059 DAL

- GW

- 140R

- DB

- 120329 1.61 17.8 97.8

**Average Concentration (ng/mL) ±**  **%RPD 1.59**  **ng/mL ±**  **2. 5%**  **17.7 ng/mL**  **± 1.1%**  **106 ng/mL**  **± 16%** 

Samples diluted 1:10 and reported by external standard calibration.

### Table 32. DAL GW 14 1 R 1 20403

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery** 

GLP10

- 01

- 02

- 28

- 060 DAL

- GW

- 141 R

- 0

- 120403 0.0364 NA 0.312 NA 5.89 NA

GLP10

- 01

- 02

- 28

- 061 DAL

- GW

- 141 R

- DB

- 120403 0.0374 NA 0.293 NA 5.28 NA

GLP10

- 01

- 02

- 28

- 062 DAL

- GW

- 141 R

- FMS

- 120403 4.70 93.3 4.51 84.3 9.70 82.5

**Average Concentrati on (ng/mL)**  **± %RPD**  **0.0369 ng/mL**  **± 2.7**  **% 0.303**  **ng/mL ±**  **6.3 %**  **5.59 ng/mL**  **± 11**  **%**

NA = Not Applicable

R esults for PFBS and PFHS are reported by external standard calibration. R esults for PFOS are reported by internal standard calibration.



### Table 33 . DAL GW 142 R 1 20328

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) Concentration** 

**(ng/mL) Concentration** 

**(ng/mL)**

GLP10

- 01

- 02

- 28

- 063 DAL

- GW

- 142R

- 0

- 120328 2.03 11.3 63.2

GLP10

- 01

- 02

- 28

- 064 DAL

- GW

- 142R

- DB

- 120328 2.13 11.0 47.6

**Average Concentration (n g/mL)**  **± %RPD**  **2.08 ng/mL**  **± 4.8%**  **11.2 ng/mL**  **± 2.7%**  **55.4 ng/mL**  **± 28%**  **(1)**

Samples diluted 1:10 and reported by external standard calibration.

(1) Sample/sample duplicate RPD did not meet acceptance criteria of ≤ 20%.

### Table 34. DAL GW 14 3 R 1 20329

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) Concentration** 

**(ng/mL) Concentration** 

**(ng/mL)**

GLP10

- 01

- 02

- 28

- 065 DAL

- GW

- 143R

- 0

- 120329 1.55 5.77 14.3

GLP10

- 01

- 02

- 28

- 066 DAL

- GW

- 143R

- DB

- 120329 1.56 5. 60 13.0

**Average Concentration (ng/mL) ±**  **%RPD 1.56**  **ng/mL ±**  **0.64% 5.69**  **ng/mL ±**  **3.0% 13.7**  **ng/mL ±**  **9.5%**

Results reported by external standard calibration.



### Table 35. DAL GW 144 R 1 20328

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng /mL)**  **Concentration**

**(ng/mL) Concentration** 

**(ng/mL)**

GLP10

- 01

- 02

- 28

- 067 DAL

- GW

- 144R

- 0

- 120328 1.96 9.16 13.9

GLP10

- 01

- 02

- 28

- 068 DAL

- GW

- 144R

- DB

- 120328 2.02 8.89 12.9

**Average Concentration (ng/mL) ±**  **%RPD 1.99**  **ng/mL ±**  **3.0% 9.03**  **ng/mL ±**  **3.0% 13.4**  **ng/mL ±**  **7.5%**

Samples diluted 1:10 and reported by external standard calibration.



### Table 36. DAL GW 14 5 R 120328

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) Concentration** 

**(ng/mL) Concentration** 

**(ng/mL)**

GLP10

- 01

- 02

- 28

- 069 DAL

- GW

- 145R

- 0

- 120328 1.49 7.52 50.1

GLP10

- 01

- 02

- 28

- 070 DAL

- GW

- 145R

- DB

- 120328 1.51 7.58 48.0

**Average Concentration (ng/mL) ±**  **%RPD 1.50**  **ng/mL ±**  **1.3% 7.55**  **ng/mL ±**  **0.79% 49.1**  **ng/mL ±**  **4.3%**

Samples diluted 1:10 and reported by external standard calibration

.

### Table 37. Trip Blank 1

**P F** 

**BS PF**  **HS PF**  **OS**

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery** 

GLP10

- 01

- 02

- 28

- 073 DAL

- GW

- TRIP01

- 0 <0.0250 NA <0.0250 NA <0.0232 NA

GLP10

- 01

- 02

- 28

- 074 DAL

- GW

- TRIP01

- LS 0.910 91.0 0. 880 88.2 0.695 69.6

GLP10

- 01

- 02

- 28

- 075 DAL

- GW

- TRIP01

- MS 9.67 96.7 9.47 94.9 8.91 89.3

GLP10

- 01

- 02

- 28

- 076 DAL

- GW

- TRIP01

- MHS 48.6 97.2 46.3 92.8 33.5 67.1 (1)

GLP10

- 01

- 02

- 28

- 077 DAL

- GW

- TRIP01

- HS 197 98.5 192 96.2 140 70.1

NA = Not Applicable

Results reported by external standard calibration. The MHS sample was diluted 1:10 and the HS sample was diluted 1:100.

(1) FMS did not meet acceptance criteria of 100 ± 30%.



### Table 38. Rinseate Blanks

**PF BS**  **PF HS**  **PF OS** 

**3M LIMS ID Description**  **Concentration**

**(ng/mL) Concentration** 

**(ng/mL) Concentration** 

**(ng/mL)**

GLP10

- 01

- 02

- 28

- 071 DAL

- GW

- 130R

- RB

- 120329 <0.0250 <0.0250 <0.0232

GLP10

- 01

- 02

- 28

- 072 DAL

- GW

- 135R

- RB

- 120329 <0.0250 <0.0250 <0.0232

Results reported by external standard calibration.

## 10 **Conclusion**

Laboratory control spikes and field matrix spikes were used to determine the analytical method

accuracy and precision for PFBS, PFHS, and PFOS. A nalysis was successfully completed following

3M Environmental Laboratory method ETS -8- 044.1 described herein.



## 11 **Data/Sample Reten tion** 

All remaining samples and associated project data (hardcopy and electronic) will be archived according

to 3M Environmental Laboratory standard operating procedures.



## 12 **Attachments**

Att achment A: Protocol Amendment 28 (General Project Outline)

Attachmen t B: Representative Chromatograms and Calibration Curves

Attachment C: Analytical Method

– ETS -8- 044.1

Attachment D: Method DeviationGLP10-01-02 ; Interi m Repor t 2 8

Analysi s o f PFBS

, PFHS

, an d PFO S i n Groundwate r

Forme r Sludg e Incorporatio n Are a - March/Apri l 201 2



## 13 Signatures



# 20)Z

Clesto n Lange

, Ph.D.

, 3 M Principa l Analytica l Investigato r Dat e

Willia m K

. Reagen

, Ph.D.

, 3 M Environmenta l Laborator y Departmen t Manage r Dat e

Gar y Hohensteinj^ponso r Representativ e Dat e



## I feate Jaisimh a Kesari

, Stud y Directo r

Pag e 3 4 o f 10 8

# A TTACHMENT 

# A: P ROTOCOL 

# A MENDMENT

# A TTACHMENT 

# B: R EPRESENTATIVE 

# S AMPLE 

# C HROMATOGRAMS AND



# C ALIBRATION 

# C URVE 

# ( S 

# )b120411a.rdb (PFOS): "Quadratic" Regression ("1 / x" weighting): y = -1.91e+003 x^2 + 5.73e+005 x + 1.68e+004 (r = 0.9996)0 5 10 15 20 25 30 35 40 45 50 55 60 65 70 75 80 85 90Concentration, ng/mL0.02.0e64.0e66.0e68.0e61.0e71.2e71.4e71.6e71.8e72.0e72.2e72.4e72.6e72.8e73.0e73.2e73.4e73.6e73.8e7

Area, countsWorkstation: ETSBUSTERBatch Name: b120411a.dab*Data worked up by KJUPage 1 of 1Printing Time: 10:01:44 AMPrinting Date: Tuesday, April 17, 2012b120411a.rdb (PFHS): "Quadratic" Regression ("1 / x" weighting): y = -3.87e+003 x^2 + 5.26e+005 x + 5.52e+003 (r = 0.9993)0 5 10 15 20 25 30 35 40 45 50 55 60 65 70 75 80 85 90 95 100Concentration, ng/mL0.01.0e62.0e63.0e64.0e65.0e66.0e67.0e68.0e69.0e61.0e71.1e71.2e71.3e71.4e71.5e71.6e71.7e71.8e71.9e72.0e72.1e72.2e72.3e72.4e72.5e72.6e7

Area, countsWorkstation: ETSBUSTERBatch Name: b120411a.dab*Data worked up by KJUPage 1 of 1Printing Time: 9:59:28 AMPrinting Date: Tuesday, April 17, 2012b120411a.rdb (PFBS): "Quadratic" Regression ("1 / x" weighting): y = -3.15e+003 x^2 + 4.91e+005 x + 8.42e+003 (r = 0.9992)0 5 10 15 20 25 30 35 40 45 50 55 60 65 70 75 80 85 90 95 100Concentration, ng/mL0.01.0e62.0e63.0e64.0e65.0e66.0e67.0e68.0e69.0e61.0e71.1e71.2e71.3e71.4e71.5e71.6e71.7e71.8e71.9e72.0e72.1e72.2e72.3e72.4e72.5e72.6e72.7e7

Area, countsWorkstation: ETSBUSTERBatch Name: b120411a.dab*Data worked up by KJUPage 1 of 1Printing Time: 10:01:26 AMPrinting Date: Tuesday, April 17, 2012Sample Name: "b120411a018" Sample ID: "11012-10-9" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "2.5 ng/mL FC std in Milli Q Water" Annotation: ""Sample Index: 18Sample Type: StandardConcentration: 2.50 ng/mLCalculated Conc: 2.71 ng/mLAcq. Date: 4/11/2012Acq. Time: 6:46:56 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.6 minArea: 1314309 countsHeight: 3.67e+005 cpsStart Time: 13.5 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e5

Intensity, cps13.56Sample Name: "b120411a028" Sample ID: "Solvent Blank" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "Milli Q Water TN12-0132" Annotation: ""Sample Index: 28Sample Type: UnknownConcentration: N/ACalculated Conc: < 0Acq. Date: 4/11/2012Acq. Time: 10:17:34 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.5 minArea: 1483 countsHeight: 5.08e+002 cpsStart Time: 13.5 minEnd Time: 13.6 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min01002003004005006007008009001000110012001300140015001600170018001900

Intensity, cpsSample Name: "b120411a029" Sample ID: "LCS-120405-1" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "0.5ppb LCS" Annotation: ""Sample Index: 29Sample Type: QCConcentration: 0.498 ng/mLCalculated Conc: 0.494 ng/mLAcq. Date: 4/11/2012Acq. Time: 10:38:41 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.6 minArea: 250124 countsHeight: 6.93e+004 cpsStart Time: 13.5 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e4

Intensity, cps13.5714.2415.4715.8711.7514.62Sample Name: "b120411a032" Sample ID: "LCS-120405-4" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "5ppb LCS" Annotation: ""Sample Index: 32Sample Type: QCConcentration: 4.98 ng/mLCalculated Conc: 5.00 ng/mLAcq. Date: 4/11/2012Acq. Time: 11:41:54 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.6 minArea: 2384031 countsHeight: 6.59e+005 cpsStart Time: 13.5 minEnd Time: 13.8 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e5

Intensity, cps13.58*** Buster J2930203Batch Name: b120411a.dabData printed by STWPrinting Time: 8:45:46 AMPrinting Date: Thursday, May 03, 2012Page 1 of 30Sample Name: "b120411a035" Sample ID: "LCS-120405-7" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "30ppb LCS" Annotation: ""Sample Index: 35Sample Type: QCConcentration: 29.9 ng/mLCalculated Conc: 29.7 ng/mLAcq. Date: 4/12/2012Acq. Time: 12:45:10 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.6 minArea: 11807788 countsHeight: 3.02e+006 cpsStart Time: 13.5 minEnd Time: 14.0 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e54.0e56.0e58.0e51.0e61.2e61.4e61.6e61.8e62.0e62.2e62.4e62.6e62.8e63.0e6

Intensity, cps13.57Sample Name: "b120411a042" Sample ID: "GLP10-01-02-28-071" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-130R-RB-" Annotation: ""Sample Index: 42Sample Type: UnknownConcentration: N/ACalculated Conc: < 0Acq. Date: 4/12/2012Acq. Time: 3:12:56 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.6 minArea: 1144 countsHeight: 4.19e+002 cpsStart Time: 13.5 minEnd Time: 13.6 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min050100150200250300350400450500550600650700750800850900950100010501100115012001250

Intensity, cpsSample Name: "b120411a045" Sample ID: "GLP10-01-02-28-073" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-TRIP01-0-" Annotation: ""Sample Index: 45Sample Type: UnknownConcentration: N/ACalculated Conc: < 0Acq. Date: 4/12/2012Acq. Time: 4:16:09 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.5 minArea: 1935 countsHeight: 3.68e+002 cpsStart Time: 13.5 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0501001502002503003504004505005506006507007508008509009501000105011001150120012501300

Intensity, cps14.2115.5315.4415.5014.3215.3711.7415.0811.7714.4212.3814.0415.0013.3213.5311.8213.83Sample Name: "b120411a048" Sample ID: "GLP10-01-02-28-076" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-TRIP01-MHS-" Annotation: ""Sample Index: 48Sample Type: UnknownConcentration: N/ACalculated Conc: 48.6 ng/mLAcq. Date: 4/12/2012Acq. Time: 5:19:21 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.6 minArea: 2317522 countsHeight: 6.36e+005 cpsStart Time: 13.4 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e5

Intensity, cps13.56*** Buster J2930203Batch Name: b120411a.dabData printed by STWPrinting Time: 8:45:46 AMPrinting Date: Thursday, May 03, 2012Page 2 of 30Sample Name: "b120411a057" Sample ID: "GLP10-01-02-28-001" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-130R-0-" Annotation: ""Sample Index: 57Sample Type: UnknownConcentration: N/ACalculated Conc: 14.4 ng/mLAcq. Date: 4/12/2012Acq. Time: 8:29:07 AMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 78977 countsHeight: 2.03e+004 cpsStart Time: 13.3 minEnd Time: 13.6 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.01000.02000.03000.04000.05000.06000.07000.08000.09000.01.0e41.1e41.2e41.3e41.4e41.5e41.6e41.7e41.8e41.9e42.0e4

Intensity, cpsSample Name: "b120411a059" Sample ID: "GLP10-01-02-28-003" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-130S-0-" Annotation: ""Sample Index: 59Sample Type: UnknownConcentration: N/ACalculated Conc: 3.48 ng/mLAcq. Date: 4/12/2012Acq. Time: 9:11:15 AMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 178969 countsHeight: 4.74e+004 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e43.4e43.6e43.8e44.0e44.2e44.4e44.6e4

Intensity, cps13.5314.2013.3915.40Sample Name: "b120411a064" Sample ID: "GLP10-01-02-28-005" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-130L-0-" Annotation: ""Sample Index: 64Sample Type: UnknownConcentration: N/ACalculated Conc: 1.53 ng/mLAcq. Date: 4/12/2012Acq. Time: 10:56:44 AMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 83684 countsHeight: 2.23e+004 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.01000.02000.03000.04000.05000.06000.07000.08000.09000.01.0e41.1e41.2e41.3e41.4e41.5e41.6e41.7e41.8e41.9e42.0e42.1e42.2e4

Intensity, cps13.53Sample Name: "b120411a066" Sample ID: "GLP10-01-02-28-007" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-130L-FMS-" Annotation: ""Sample Index: 66Sample Type: UnknownConcentration: N/ACalculated Conc: 48.4 ng/mLAcq. Date: 4/12/2012Acq. Time: 11:38:54 AMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 2311254 countsHeight: 6.40e+005 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e5

Intensity, cps13.5214.17*** Buster J2930203Batch Name: b120411a.dabData printed by STWPrinting Time: 8:45:46 AMPrinting Date: Thursday, May 03, 2012Page 3 of 30Sample Name: "b120411a067" Sample ID: "GLP10-01-02-28-008" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-131R-0-" Annotation: ""Sample Index: 67Sample Type: UnknownConcentration: N/ACalculated Conc: 39.6 ng/mLAcq. Date: 4/12/2012Acq. Time: 11:59:58 AMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 202288 countsHeight: 5.50e+004 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e43.4e43.6e43.8e44.0e44.2e44.4e44.6e44.8e45.0e45.2e45.4e4

Intensity, cps13.52Sample Name: "b120411a069" Sample ID: "GLP10-01-02-28-010" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-131S-0-" Annotation: ""Sample Index: 69Sample Type: UnknownConcentration: N/ACalculated Conc: 0.833 ng/mLAcq. Date: 4/12/2012Acq. Time: 12:42:12 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 415314 countsHeight: 1.11e+005 cpsStart Time: 13.3 minEnd Time: 13.6 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e51.05e51.10e5

Intensity, cps13.5213.37Sample Name: "b120411a071" Sample ID: "GLP10-01-02-28-012" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-131L-0-" Annotation: ""Sample Index: 71Sample Type: UnknownConcentration: N/ACalculated Conc: 3.14 ng/mLAcq. Date: 4/12/2012Acq. Time: 1:24:22 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 162400 countsHeight: 4.37e+004 cpsStart Time: 13.3 minEnd Time: 13.6 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e43.4e43.6e43.8e44.0e44.2e4

Intensity, cps13.52Sample Name: "b120411a076" Sample ID: "GLP10-01-02-28-014" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-133R-0-" Annotation: ""Sample Index: 76Sample Type: UnknownConcentration: N/ACalculated Conc: 3.11 ng/mLAcq. Date: 4/12/2012Acq. Time: 3:09:45 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 160913 countsHeight: 4.43e+004 cpsStart Time: 13.3 minEnd Time: 13.6 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e43.4e43.6e43.8e44.0e44.2e4

Intensity, cps13.52*** Buster J2930203Batch Name: b120411a.dabData printed by STWPrinting Time: 8:45:46 AMPrinting Date: Thursday, May 03, 2012Page 4 of 30Sample Name: "b120411a078" Sample ID: "GLP10-01-02-28-016" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-133S-0-" Annotation: ""Sample Index: 78Sample Type: UnknownConcentration: N/ACalculated Conc: 8.32 ng/mLAcq. Date: 4/12/2012Acq. Time: 3:51:58 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 414825 countsHeight: 1.15e+005 cpsStart Time: 13.3 minEnd Time: 13.6 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e51.05e51.10e5

Intensity, cps13.5113.37Sample Name: "b120411a080" Sample ID: "GLP10-01-02-28-018" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-133L-0-" Annotation: ""Sample Index: 80Sample Type: UnknownConcentration: N/ACalculated Conc: 5.85 ng/mLAcq. Date: 4/12/2012Acq. Time: 4:34:08 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 294338 countsHeight: 7.84e+004 cpsStart Time: 13.3 minEnd Time: 13.6 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e4

Intensity, cps13.52Sample Name: "b120411a082" Sample ID: "GLP10-01-02-28-020" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-133L-FMS-" Annotation: ""Sample Index: 82Sample Type: UnknownConcentration: N/ACalculated Conc: 189. ng/mLAcq. Date: 4/12/2012Acq. Time: 5:16:17 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 926052 countsHeight: 2.57e+005 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e51.9e52.0e52.1e52.2e52.3e52.4e52.5e5

Intensity, cps13.51Sample Name: "b120411a083" Sample ID: "GLP10-01-02-28-021" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-134R-0-" Annotation: ""Sample Index: 83Sample Type: UnknownConcentration: N/ACalculated Conc: 55.3 ng/mLAcq. Date: 4/12/2012Acq. Time: 5:37:25 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 279094 countsHeight: 7.63e+004 cpsStart Time: 13.3 minEnd Time: 13.6 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e4

Intensity, cps13.52*** Buster J2930203Batch Name: b120411a.dabData printed by STWPrinting Time: 8:45:47 AMPrinting Date: Thursday, May 03, 2012Page 5 of 30Sample Name: "b120411a088" Sample ID: "GLP10-01-02-28-023" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-134S-0-" Annotation: ""Sample Index: 88Sample Type: UnknownConcentration: N/ACalculated Conc: 0.426 ng/mLAcq. Date: 4/12/2012Acq. Time: 7:22:54 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 216775 countsHeight: 5.95e+004 cpsStart Time: 13.3 minEnd Time: 13.6 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e4

Intensity, cps13.52Sample Name: "b120411a090" Sample ID: "GLP10-01-02-28-025" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-134S-FMS-" Annotation: ""Sample Index: 90Sample Type: UnknownConcentration: N/ACalculated Conc: 9.99 ng/mLAcq. Date: 4/12/2012Acq. Time: 8:05:03 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 4599051 countsHeight: 1.28e+006 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e61.10e61.15e61.20e61.25e6

Intensity, cps13.5213.38Sample Name: "b120411a091" Sample ID: "GLP10-01-02-28-026" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-134L-0-" Annotation: ""Sample Index: 91Sample Type: UnknownConcentration: N/ACalculated Conc: 10.1 ng/mLAcq. Date: 4/12/2012Acq. Time: 8:26:11 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.5 minArea: 58103 countsHeight: 1.59e+004 cpsStart Time: 13.4 minEnd Time: 13.6 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.01000.02000.03000.04000.05000.06000.07000.08000.09000.01.0e41.1e41.2e41.3e41.4e41.5e41.6e4

Intensity, cpsSample Name: "b120411a093" Sample ID: "GLP10-01-02-28-028" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-135R-0-" Annotation: ""Sample Index: 93Sample Type: UnknownConcentration: N/ACalculated Conc: 66.5 ng/mLAcq. Date: 4/12/2012Acq. Time: 9:08:19 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 333283 countsHeight: 9.19e+004 cpsStart Time: 13.3 minEnd Time: 13.6 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e48.0e48.5e49.0e4

Intensity, cps13.5313.3814.1515.7815.6314.9414.69*** Buster J2930203Batch Name: b120411a.dabData printed by STWPrinting Time: 8:45:47 AMPrinting Date: Thursday, May 03, 2012Page 6 of 30Sample Name: "b120411a095" Sample ID: "GLP10-01-02-28-030" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-135S-0-" Annotation: ""Sample Index: 95Sample Type: UnknownConcentration: N/ACalculated Conc: 2.19 ng/mLAcq. Date: 4/12/2012Acq. Time: 9:50:33 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 1069216 countsHeight: 2.84e+005 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e5

Intensity, cps13.5213.37Sample Name: "b120411a100" Sample ID: "GLP10-01-02-28-032" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-135L-0-" Annotation: ""Sample Index: 100Sample Type: UnknownConcentration: N/ACalculated Conc: 10.8 ng/mLAcq. Date: 4/12/2012Acq. Time: 11:36:09 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 534213 countsHeight: 1.41e+005 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e5

Intensity, cps13.54Sample Name: "b120411a102" Sample ID: "GLP10-01-02-28-036" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-136S-0-" Annotation: ""Sample Index: 102Sample Type: UnknownConcentration: N/ACalculated Conc: 2.58 ng/mLAcq. Date: 4/13/2012Acq. Time: 12:18:16 AMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 134959 countsHeight: 3.74e+004 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e43.4e43.6e4

Intensity, cps13.54Sample Name: "b120411a104" Sample ID: "GLP10-01-02-28-038" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-136L-0-" Annotation: ""Sample Index: 104Sample Type: UnknownConcentration: N/ACalculated Conc: 2.32 ng/mLAcq. Date: 4/13/2012Acq. Time: 1:00:37 AMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 122044 countsHeight: 3.35e+004 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e4

Intensity, cps13.53*** Buster J2930203Batch Name: b120411a.dabData printed by STWPrinting Time: 8:45:47 AMPrinting Date: Thursday, May 03, 2012Page 7 of 30Sample Name: "b120411a106" Sample ID: "GLP10-01-02-28-042" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-137S-0-" Annotation: ""Sample Index: 106Sample Type: UnknownConcentration: N/ACalculated Conc: 0.108 ng/mLAcq. Date: 4/13/2012Acq. Time: 1:42:51 AMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 61330 countsHeight: 1.73e+004 cpsStart Time: 13.3 minEnd Time: 13.6 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.01000.02000.03000.04000.05000.06000.07000.08000.09000.01.0e41.1e41.2e41.3e41.4e41.5e41.6e41.7e4

Intensity, cps13.53Sample Name: "b120411a112" Sample ID: "GLP10-01-02-28-045" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-137L-0-" Annotation: ""Sample Index: 112Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0681 ng/mLAcq. Date: 4/13/2012Acq. Time: 3:49:33 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.5 minArea: 41830 countsHeight: 1.16e+004 cpsStart Time: 13.4 minEnd Time: 13.6 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.00500.001000.001500.002000.002500.003000.003500.004000.004500.005000.005500.006000.006500.007000.007500.008000.008500.009000.009500.001.00e41.05e41.10e41.15e4

Intensity, cps13.5314.1811.6515.40Sample Name: "b120411a114" Sample ID: "GLP10-01-02-28-049" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-138S-0-" Annotation: ""Sample Index: 114Sample Type: UnknownConcentration: N/ACalculated Conc: 5.63 ng/mLAcq. Date: 4/13/2012Acq. Time: 4:31:46 AMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 283867 countsHeight: 7.92e+004 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e4

Intensity, cps13.5314.1613.3715.4616.0515.18Sample Name: "b120411a116" Sample ID: "GLP10-01-02-28-051" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-138L-0-" Annotation: ""Sample Index: 116Sample Type: UnknownConcentration: N/ACalculated Conc: 0.00929 ng/mLAcq. Date: 4/13/2012Acq. Time: 5:13:55 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.5 minArea: 12978 countsHeight: 3.54e+003 cpsStart Time: 13.4 minEnd Time: 13.6 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min020040060080010001200140016001800200022002400260028003000320034003600

Intensity, cps*** Buster J2930203Batch Name: b120411a.dabData printed by STWPrinting Time: 8:45:47 AMPrinting Date: Thursday, May 03, 2012Page 8 of 30Sample Name: "b120411a118" Sample ID: "GLP10-01-02-28-053" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-138L-FMS-" Annotation: ""Sample Index: 118Sample Type: UnknownConcentration: N/ACalculated Conc: 0.962 ng/mLAcq. Date: 4/13/2012Acq. Time: 5:56:17 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.5 minArea: 477640 countsHeight: 1.30e+005 cpsStart Time: 13.4 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e51.05e51.10e51.15e51.20e51.25e5

Intensity, cps13.5314.1711.6513.38Sample Name: "b120411a119" Sample ID: "GLP10-01-02-28-058" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-140R-0-" Annotation: ""Sample Index: 119Sample Type: UnknownConcentration: N/ACalculated Conc: 1.57 ng/mLAcq. Date: 4/13/2012Acq. Time: 6:17:18 AMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 85449 countsHeight: 2.30e+004 cpsStart Time: 13.3 minEnd Time: 13.6 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.01000.02000.03000.04000.05000.06000.07000.08000.09000.01.0e41.1e41.2e41.3e41.4e41.5e41.6e41.7e41.8e41.9e42.0e42.1e42.2e42.3e4

Intensity, cps13.53Sample Name: "b120411a124" Sample ID: "GLP10-01-02-28-060" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-141R-0-" Annotation: ""Sample Index: 124Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0364 ng/mLAcq. Date: 4/13/2012Acq. Time: 8:02:54 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.5 minArea: 26274 countsHeight: 7.07e+003 cpsStart Time: 13.4 minEnd Time: 13.6 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min05001000150020002500300035004000450050005500600065007000

Intensity, cps13.5311.7014.17Sample Name: "b120411a127" Sample ID: "GLP10-01-02-28-063" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-142R-0-" Annotation: ""Sample Index: 127Sample Type: UnknownConcentration: N/ACalculated Conc: 2.03 ng/mLAcq. Date: 4/13/2012Acq. Time: 9:06:10 AMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 107839 countsHeight: 3.01e+004 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e4

Intensity, cps13.5311.7114.1915.9915.3713.3814.5912.3213.9212.89*** Buster J2930203Batch Name: b120411a.dabData printed by STWPrinting Time: 8:45:47 AMPrinting Date: Thursday, May 03, 2012Page 9 of 30Sample Name: "b120411a129" Sample ID: "GLP10-01-02-28-065" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-143R-0-" Annotation: ""Sample Index: 129Sample Type: UnknownConcentration: N/ACalculated Conc: 1.55 ng/mLAcq. Date: 4/13/2012Acq. Time: 9:48:24 AMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 759594 countsHeight: 2.09e+005 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e51.9e52.0e5

Intensity, cps13.5213.3711.7014.17Sample Name: "b120411a131" Sample ID: "GLP10-01-02-28-067" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-144R-0-" Annotation: ""Sample Index: 131Sample Type: UnknownConcentration: N/ACalculated Conc: 1.96 ng/mLAcq. Date: 4/13/2012Acq. Time: 10:30:40 AMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 104502 countsHeight: 2.83e+004 cpsStart Time: 13.3 minEnd Time: 13.6 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e4

Intensity, cps13.51Sample Name: "b120411a136" Sample ID: "GLP10-01-02-28-069" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-145R-0-" Annotation: ""Sample Index: 136Sample Type: UnknownConcentration: N/ACalculated Conc: 1.49 ng/mLAcq. Date: 4/13/2012Acq. Time: 12:16:01 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ManualRetention Time: 13.5 minArea: 81627 countsHeight: 2.16e+004 cpsStart Time: 13.3 minEnd Time: 13.6 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.01000.02000.03000.04000.05000.06000.07000.08000.09000.01.0e41.1e41.2e41.3e41.4e41.5e41.6e41.7e41.8e41.9e42.0e42.1e4

Intensity, cps13.5214.18Sample Name: "b120411a234" Sample ID: "GLP10-01-02-28-063" File: "b120411a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-142R-0-" Annotation: ""Sample Index: 234Sample Type: UnknownConcentration: N/ACalculated Conc: 1.83 ng/mLAcq. Date: 4/14/2012Acq. Time: 10:45:04 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.5 minArea: 98204 countsHeight: 2.64e+004 cpsStart Time: 13.4 minEnd Time: 13.6 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.01000.02000.03000.04000.05000.06000.07000.08000.09000.01.0e41.1e41.2e41.3e41.4e41.5e41.6e41.7e41.8e41.9e42.0e42.1e42.2e42.3e42.4e42.5e42.6e4

Intensity, cps13.52*** Buster J2930203Batch Name: b120411a.dabData printed by STWPrinting Time: 8:45:47 AMPrinting Date: Thursday, May 03, 2012Page 10 of 30Sample Name: "b120411a018" Sample ID: "11012-10-9" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "2.5 ng/mL FC std in Milli Q Water" Annotation: ""Sample Index: 18Sample Type: StandardConcentration: 2.50 ng/mLCalculated Conc: 2.70 ng/mLAcq. Date: 4/11/2012Acq. Time: 6:46:56 PMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 1396809 countsHeight: 4.08e+005 cpsStart Time: 15.0 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e5

Intensity, cps15.0917.5616.5315.57Sample Name: "b120411a028" Sample ID: "Solvent Blank" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "Milli Q Water TN12-0132" Annotation: ""Sample Index: 28Sample Type: UnknownConcentration: N/ACalculated Conc: < 0Acq. Date: 4/11/2012Acq. Time: 10:17:34 PMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.3 minArea: 1354 countsHeight: 2.98e+002 cpsStart Time: 15.2 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min05001000150020002500300035004000450050005500600065007000750080008500

Intensity, cpsSample Name: "b120411a029" Sample ID: "LCS-120405-1" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "0.5ppb LCS" Annotation: ""Sample Index: 29Sample Type: QCConcentration: 0.497 ng/mLCalculated Conc: 0.476 ng/mLAcq. Date: 4/11/2012Acq. Time: 10:38:41 PMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 254814 countsHeight: 7.55e+004 cpsStart Time: 15.0 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e4

Intensity, cps15.1017.5516.5315.5817.1916.39Sample Name: "b120411a032" Sample ID: "LCS-120405-4" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "5ppb LCS" Annotation: ""Sample Index: 32Sample Type: QCConcentration: 4.97 ng/mLCalculated Conc: 5.00 ng/mLAcq. Date: 4/11/2012Acq. Time: 11:41:54 PMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 2537144 countsHeight: 7.35e+005 cpsStart Time: 15.0 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e5

Intensity, cps15.0917.55*** Buster J2930203Batch Name: b120411a.dabData printed by STWPrinting Time: 8:45:47 AMPrinting Date: Thursday, May 03, 2012Page 11 of 30Sample Name: "b120411a035" Sample ID: "LCS-120405-7" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "30ppb LCS" Annotation: ""Sample Index: 35Sample Type: QCConcentration: 29.9 ng/mLCalculated Conc: 28.4 ng/mLAcq. Date: 4/12/2012Acq. Time: 12:45:10 AMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 11814663 countsHeight: 3.19e+006 cpsStart Time: 14.9 minEnd Time: 15.4 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.02.0e54.0e56.0e58.0e51.0e61.2e61.4e61.6e61.8e62.0e62.2e62.4e62.6e62.8e63.0e6

Intensity, cps15.08Sample Name: "b120411a042" Sample ID: "GLP10-01-02-28-071" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-130R-RB-" Annotation: ""Sample Index: 42Sample Type: UnknownConcentration: N/ACalculated Conc: < 0Acq. Date: 4/12/2012Acq. Time: 3:12:56 AMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 3069 countsHeight: 4.21e+002 cpsStart Time: 15.0 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0500100015002000250030003500400045005000550060006500700075008000

Intensity, cpsSample Name: "b120411a045" Sample ID: "GLP10-01-02-28-073" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-TRIP01-0-" Annotation: ""Sample Index: 45Sample Type: UnknownConcentration: N/ACalculated Conc: < 0Acq. Date: 4/12/2012Acq. Time: 4:16:09 AMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 3120 countsHeight: 3.32e+002 cpsStart Time: 14.8 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min050010001500200025003000350040004500500055006000650070007500

Intensity, cps17.5315.5716.4917.1716.6015.4116.3915.0816.0514.70Sample Name: "b120411a048" Sample ID: "GLP10-01-02-28-076" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-TRIP01-MHS-" Annotation: ""Sample Index: 48Sample Type: UnknownConcentration: N/ACalculated Conc: 46.3 ng/mLAcq. Date: 4/12/2012Acq. Time: 5:19:21 AMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 2359667 countsHeight: 6.87e+005 cpsStart Time: 14.9 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e5

Intensity, cps15.0817.54*** Buster J2930203Batch Name: b120411a.dabData printed by STWPrinting Time: 8:45:47 AMPrinting Date: Thursday, May 03, 2012Page 12 of 30Sample Name: "b120411a057" Sample ID: "GLP10-01-02-28-001" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-130R-0-" Annotation: ""Sample Index: 57Sample Type: UnknownConcentration: N/ACalculated Conc: 119. ng/mLAcq. Date: 4/12/2012Acq. Time: 8:29:07 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.1 minArea: 627877 countsHeight: 1.67e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e5

Intensity, cps15.06Sample Name: "b120411a059" Sample ID: "GLP10-01-02-28-003" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-130S-0-" Annotation: ""Sample Index: 59Sample Type: UnknownConcentration: N/ACalculated Conc: 30.9 ng/mLAcq. Date: 4/12/2012Acq. Time: 9:11:15 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.1 minArea: 1595855 countsHeight: 4.25e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e5

Intensity, cps15.0514.91Sample Name: "b120411a064" Sample ID: "GLP10-01-02-28-005" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-130L-0-" Annotation: ""Sample Index: 64Sample Type: UnknownConcentration: N/ACalculated Conc: 12.7 ng/mLAcq. Date: 4/12/2012Acq. Time: 10:56:44 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 664918 countsHeight: 1.74e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e5

Intensity, cps15.04Sample Name: "b120411a066" Sample ID: "GLP10-01-02-28-007" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-130L-FMS-" Annotation: ""Sample Index: 66Sample Type: UnknownConcentration: N/ACalculated Conc: 60.5 ng/mLAcq. Date: 4/12/2012Acq. Time: 11:38:54 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 3048495 countsHeight: 8.78e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e58.5e5

Intensity, cps15.0414.90*** Buster J2930203Batch Name: b120411a.dabData printed by STWPrinting Time: 8:45:47 AMPrinting Date: Thursday, May 03, 2012Page 13 of 30Sample Name: "b120411a067" Sample ID: "GLP10-01-02-28-008" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-131R-0-" Annotation: ""Sample Index: 67Sample Type: UnknownConcentration: N/ACalculated Conc: 228. ng/mLAcq. Date: 4/12/2012Acq. Time: 11:59:58 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 1183923 countsHeight: 3.14e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e5

Intensity, cps15.04Sample Name: "b120411a069" Sample ID: "GLP10-01-02-28-010" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-131S-0-" Annotation: ""Sample Index: 69Sample Type: UnknownConcentration: N/ACalculated Conc: 4.82 ng/mLAcq. Date: 4/12/2012Acq. Time: 12:42:12 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 2449540 countsHeight: 6.27e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e5

Intensity, cps15.0414.90Sample Name: "b120411a071" Sample ID: "GLP10-01-02-28-012" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-131L-0-" Annotation: ""Sample Index: 71Sample Type: UnknownConcentration: N/ACalculated Conc: 20.9 ng/mLAcq. Date: 4/12/2012Acq. Time: 1:24:22 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 1086120 countsHeight: 2.84e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e5

Intensity, cps15.0414.90Sample Name: "b120411a076" Sample ID: "GLP10-01-02-28-014" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-133R-0-" Annotation: ""Sample Index: 76Sample Type: UnknownConcentration: N/ACalculated Conc: 19.5 ng/mLAcq. Date: 4/12/2012Acq. Time: 3:09:45 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 1019125 countsHeight: 2.66e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e51.9e52.0e52.1e52.2e52.3e52.4e52.5e52.6e5

Intensity, cps15.0414.90*** Buster J2930203Batch Name: b120411a.dabData printed by STWPrinting Time: 8:45:47 AMPrinting Date: Thursday, May 03, 2012Page 14 of 30Sample Name: "b120411a078" Sample ID: "GLP10-01-02-28-016" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-133S-0-" Annotation: ""Sample Index: 78Sample Type: UnknownConcentration: N/ACalculated Conc: 61.0 ng/mLAcq. Date: 4/12/2012Acq. Time: 3:51:58 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 3071464 countsHeight: 7.88e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e5

Intensity, cps15.0314.90Sample Name: "b120411a080" Sample ID: "GLP10-01-02-28-018" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-133L-0-" Annotation: ""Sample Index: 80Sample Type: UnknownConcentration: N/ACalculated Conc: 38.4 ng/mLAcq. Date: 4/12/2012Acq. Time: 4:34:08 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 1966770 countsHeight: 5.07e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e54.4e54.6e54.8e55.0e5

Intensity, cps15.0414.90Sample Name: "b120411a082" Sample ID: "GLP10-01-02-28-020" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-133L-FMS-" Annotation: ""Sample Index: 82Sample Type: UnknownConcentration: N/ACalculated Conc: 213. ng/mLAcq. Date: 4/12/2012Acq. Time: 5:16:17 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 1110404 countsHeight: 3.23e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e5

Intensity, cps15.04Sample Name: "b120411a083" Sample ID: "GLP10-01-02-28-021" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-134R-0-" Annotation: ""Sample Index: 83Sample Type: UnknownConcentration: N/ACalculated Conc: 569. ng/mLAcq. Date: 4/12/2012Acq. Time: 5:37:25 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 2872121 countsHeight: 7.26e+005 cpsStart Time: 14.8 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e5

Intensity, cps15.0414.90*** Buster J2930203Batch Name: b120411a.dabData printed by STWPrinting Time: 8:45:47 AMPrinting Date: Thursday, May 03, 2012Page 15 of 30Sample Name: "b120411a088" Sample ID: "GLP10-01-02-28-023" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-134S-0-" Annotation: ""Sample Index: 88Sample Type: UnknownConcentration: N/ACalculated Conc: 4.78 ng/mLAcq. Date: 4/12/2012Acq. Time: 7:22:54 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 2433292 countsHeight: 5.95e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e5

Intensity, cps15.0414.90Sample Name: "b120411a090" Sample ID: "GLP10-01-02-28-025" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-134S-FMS-" Annotation: ""Sample Index: 90Sample Type: UnknownConcentration: N/ACalculated Conc: 14.4 ng/mLAcq. Date: 4/12/2012Acq. Time: 8:05:03 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 6765152 countsHeight: 1.82e+006 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.01.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e61.6e61.7e61.8e6

Intensity, cps15.0414.90Sample Name: "b120411a091" Sample ID: "GLP10-01-02-28-026" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-134L-0-" Annotation: ""Sample Index: 91Sample Type: UnknownConcentration: N/ACalculated Conc: 103. ng/mLAcq. Date: 4/12/2012Acq. Time: 8:26:11 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 544798 countsHeight: 1.39e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e48.0e48.5e49.0e49.5e41.0e51.1e51.1e51.2e51.2e51.3e51.3e51.4e5

Intensity, cps15.04Sample Name: "b120411a093" Sample ID: "GLP10-01-02-28-028" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-135R-0-" Annotation: ""Sample Index: 93Sample Type: UnknownConcentration: N/ACalculated Conc: 781. ng/mLAcq. Date: 4/12/2012Acq. Time: 9:08:19 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 3879320 countsHeight: 9.84e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e58.5e59.0e59.5e5

Intensity, cps15.0414.90*** Buster J2930203Batch Name: b120411a.dabData printed by STWPrinting Time: 8:45:47 AMPrinting Date: Thursday, May 03, 2012Page 16 of 30Sample Name: "b120411a095" Sample ID: "GLP10-01-02-28-030" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-135S-0-" Annotation: ""Sample Index: 95Sample Type: UnknownConcentration: N/ACalculated Conc: 21.4 ng/mLAcq. Date: 4/12/2012Acq. Time: 9:50:33 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 9495439 countsHeight: 2.26e+006 cpsStart Time: 14.7 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.01.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e61.6e61.7e61.8e61.9e62.0e62.1e62.2e6

Intensity, cps15.0414.90Sample Name: "b120411a100" Sample ID: "GLP10-01-02-28-032" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-135L-0-" Annotation: ""Sample Index: 100Sample Type: UnknownConcentration: N/ACalculated Conc: 80.6 ng/mLAcq. Date: 4/12/2012Acq. Time: 11:36:09 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.1 minArea: 3994704 countsHeight: 9.84e+005 cpsStart Time: 14.8 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e58.5e59.0e59.5e5

Intensity, cps15.0514.92Sample Name: "b120411a102" Sample ID: "GLP10-01-02-28-036" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-136S-0-" Annotation: ""Sample Index: 102Sample Type: UnknownConcentration: N/ACalculated Conc: 18.0 ng/mLAcq. Date: 4/13/2012Acq. Time: 12:18:16 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.1 minArea: 941385 countsHeight: 2.40e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e51.9e52.0e52.1e52.2e52.3e5

Intensity, cps15.05Sample Name: "b120411a104" Sample ID: "GLP10-01-02-28-038" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-136L-0-" Annotation: ""Sample Index: 104Sample Type: UnknownConcentration: N/ACalculated Conc: 16.1 ng/mLAcq. Date: 4/13/2012Acq. Time: 1:00:37 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.1 minArea: 842837 countsHeight: 2.19e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e51.9e52.0e52.1e5

Intensity, cps15.0514.91*** Buster J2930203Batch Name: b120411a.dabData printed by STWPrinting Time: 8:45:47 AMPrinting Date: Thursday, May 03, 2012Page 17 of 30Sample Name: "b120411a106" Sample ID: "GLP10-01-02-28-042" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-137S-0-" Annotation: ""Sample Index: 106Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0950 ng/mLAcq. Date: 4/13/2012Acq. Time: 1:42:51 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 55462 countsHeight: 1.39e+004 cpsStart Time: 14.8 minEnd Time: 15.1 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.0500.01000.01500.02000.02500.03000.03500.04000.04500.05000.05500.06000.06500.07000.07500.08000.08500.09000.09500.01.0e41.1e41.1e41.2e41.2e41.3e41.3e41.4e4

Intensity, cps15.05Sample Name: "b120411a112" Sample ID: "GLP10-01-02-28-045" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-137L-0-" Annotation: ""Sample Index: 112Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0860 ng/mLAcq. Date: 4/13/2012Acq. Time: 3:49:33 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 50760 countsHeight: 1.31e+004 cpsStart Time: 14.8 minEnd Time: 15.1 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.0500.01000.01500.02000.02500.03000.03500.04000.04500.05000.05500.06000.06500.07000.07500.08000.08500.09000.09500.01.0e41.1e41.1e41.2e41.2e41.3e41.3e4

Intensity, cps15.0417.4815.5114.9117.1316.42Sample Name: "b120411a114" Sample ID: "GLP10-01-02-28-049" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-138S-0-" Annotation: ""Sample Index: 114Sample Type: UnknownConcentration: N/ACalculated Conc: 26.6 ng/mLAcq. Date: 4/13/2012Acq. Time: 4:31:46 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 1376042 countsHeight: 3.57e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e5

Intensity, cps15.0414.9117.4815.50Sample Name: "b120411a116" Sample ID: "GLP10-01-02-28-051" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-138L-0-" Annotation: ""Sample Index: 116Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0666 ng/mLAcq. Date: 4/13/2012Acq. Time: 5:13:55 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 40526 countsHeight: 1.02e+004 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.00500.001000.001500.002000.002500.003000.003500.004000.004500.005000.005500.006000.006500.007000.007500.008000.008500.009000.009500.001.00e4

Intensity, cps*** Buster J2930203Batch Name: b120411a.dabData printed by STWPrinting Time: 8:45:47 AMPrinting Date: Thursday, May 03, 2012Page 18 of 30Sample Name: "b120411a118" Sample ID: "GLP10-01-02-28-053" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-138L-FMS-" Annotation: ""Sample Index: 118Sample Type: UnknownConcentration: N/ACalculated Conc: 0.931 ng/mLAcq. Date: 4/13/2012Acq. Time: 5:56:17 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 491722 countsHeight: 1.45e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e5

Intensity, cps15.0417.4715.5114.9017.12Sample Name: "b120411a119" Sample ID: "GLP10-01-02-28-058" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-140R-0-" Annotation: ""Sample Index: 119Sample Type: UnknownConcentration: N/ACalculated Conc: 17.6 ng/mLAcq. Date: 4/13/2012Acq. Time: 6:17:18 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 921408 countsHeight: 2.36e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e51.9e52.0e52.1e52.2e52.3e5

Intensity, cps15.0414.9017.47Sample Name: "b120411a124" Sample ID: "GLP10-01-02-28-060" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-141R-0-" Annotation: ""Sample Index: 124Sample Type: UnknownConcentration: N/ACalculated Conc: 0.312 ng/mLAcq. Date: 4/13/2012Acq. Time: 8:02:54 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 169041 countsHeight: 4.54e+004 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e43.4e43.6e43.8e44.0e44.2e44.4e4

Intensity, cps15.04Sample Name: "b120411a127" Sample ID: "GLP10-01-02-28-063" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-142R-0-" Annotation: ""Sample Index: 127Sample Type: UnknownConcentration: N/ACalculated Conc: 11.3 ng/mLAcq. Date: 4/13/2012Acq. Time: 9:06:10 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 592988 countsHeight: 1.45e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e5

Intensity, cps15.0414.9117.4715.51*** Buster J2930203Batch Name: b120411a.dabData printed by STWPrinting Time: 8:45:47 AMPrinting Date: Thursday, May 03, 2012Page 19 of 30Sample Name: "b120411a129" Sample ID: "GLP10-01-02-28-065" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-143R-0-" Annotation: ""Sample Index: 129Sample Type: UnknownConcentration: N/ACalculated Conc: 5.77 ng/mLAcq. Date: 4/13/2012Acq. Time: 9:48:24 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 2910075 countsHeight: 7.52e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e5

Intensity, cps15.0414.90Sample Name: "b120411a131" Sample ID: "GLP10-01-02-28-067" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-144R-0-" Annotation: ""Sample Index: 131Sample Type: UnknownConcentration: N/ACalculated Conc: 9.16 ng/mLAcq. Date: 4/13/2012Acq. Time: 10:30:40 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 484160 countsHeight: 1.21e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e51.05e51.10e51.15e51.20e5

Intensity, cps15.03Sample Name: "b120411a136" Sample ID: "GLP10-01-02-28-069" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-145R-0-" Annotation: ""Sample Index: 136Sample Type: UnknownConcentration: N/ACalculated Conc: 7.52 ng/mLAcq. Date: 4/13/2012Acq. Time: 12:16:01 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 399014 countsHeight: 1.06e+005 cpsStart Time: 14.8 minEnd Time: 15.1 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e51.05e5

Intensity, cps15.0314.9017.46Sample Name: "b120411a234" Sample ID: "GLP10-01-02-28-063" File: "b120411a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-142R-0-" Annotation: ""Sample Index: 234Sample Type: UnknownConcentration: N/ACalculated Conc: 11.1 ng/mLAcq. Date: 4/14/2012Acq. Time: 10:45:04 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.0 minArea: 582557 countsHeight: 1.53e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e5

Intensity, cps15.0314.89*** Buster J2930203Batch Name: b120411a.dabData printed by STWPrinting Time: 8:45:47 AMPrinting Date: Thursday, May 03, 2012Page 20 of 30Sample Name: "b120411a018" Sample ID: "11012-10-9" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "2.5 ng/mL FC std in Milli Q Water" Annotation: ""Sample Index: 18Sample Type: StandardConcentration: 2.32 ng/mLCalculated Conc: 2.43 ng/mLAcq. Date: 4/11/2012Acq. Time: 6:46:56 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.1 minArea: 1398390 countsHeight: 3.22e+005 cpsStart Time: 15.7 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e5

Intensity, cps16.0515.91Sample Name: "b120411a028" Sample ID: "Solvent Blank" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "Milli Q Water TN12-0132" Annotation: ""Sample Index: 28Sample Type: UnknownConcentration: N/ACalculated Conc: < 0Acq. Date: 4/11/2012Acq. Time: 10:17:34 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 95Base. Sub. Window: 1.00 minPeak-Split. Factor: 8Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: Base To BaseRetention Time: 16.1 minArea: 1955 countsHeight: 7.33e+002 cpsStart Time: 16.0 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0500100015002000250030003500400045005000550060006500

Intensity, cpsSample Name: "b120411a029" Sample ID: "LCS-120405-1" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "0.5ppb LCS" Annotation: ""Sample Index: 29Sample Type: QCConcentration: 0.462 ng/mLCalculated Conc: 0.386 ng/mLAcq. Date: 4/11/2012Acq. Time: 10:38:41 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.1 minArea: 237973 countsHeight: 5.57e+004 cpsStart Time: 15.7 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e4

Intensity, cps16.0515.9116.4417.7815.7517.6417.2716.7814.3818.0818.28Sample Name: "b120411a032" Sample ID: "LCS-120405-4" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "5ppb LCS" Annotation: ""Sample Index: 32Sample Type: QCConcentration: 4.62 ng/mLCalculated Conc: 4.55 ng/mLAcq. Date: 4/11/2012Acq. Time: 11:41:54 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 2584026 countsHeight: 5.90e+005 cpsStart Time: 15.7 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e5

Intensity, cps16.0515.9115.7316.47*** Buster J2930203Batch Name: b120411a.dabData printed by STWPrinting Time: 8:45:47 AMPrinting Date: Thursday, May 03, 2012Page 21 of 30Sample Name: "b120411a035" Sample ID: "LCS-120405-7" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "30ppb LCS" Annotation: ""Sample Index: 35Sample Type: QCConcentration: 27.8 ng/mLCalculated Conc: 26.0 ng/mLAcq. Date: 4/12/2012Acq. Time: 12:45:10 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 13637286 countsHeight: 2.73e+006 cpsStart Time: 15.6 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e61.6e61.7e61.8e61.9e62.0e62.1e62.2e62.3e62.4e62.5e62.6e62.7e6

Intensity, cps16.0415.9015.72Sample Name: "b120411a042" Sample ID: "GLP10-01-02-28-071" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-130R-RB-" Annotation: ""Sample Index: 42Sample Type: UnknownConcentration: N/ACalculated Conc: < 0Acq. Date: 4/12/2012Acq. Time: 3:12:56 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 6327 countsHeight: 1.87e+003 cpsStart Time: 16.0 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min020040060080010001200140016001800200022002400260028003000320034003600380040004200440046004800500052005400

Intensity, cpsSample Name: "b120411a045" Sample ID: "GLP10-01-02-28-073" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-TRIP01-0-" Annotation: ""Sample Index: 45Sample Type: UnknownConcentration: N/ACalculated Conc: < 0Acq. Date: 4/12/2012Acq. Time: 4:16:09 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 95Base. Sub. Window: 1.00 minPeak-Split. Factor: 8Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: Base To BaseRetention Time: 16.0 minArea: 3223 countsHeight: 6.64e+002 cpsStart Time: 16.0 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min05001000150020002500300035004000450050005500

Intensity, cps16.4517.7516.2917.6516.0217.3615.5415.6716.8318.0814.3418.2014.4114.9215.42Sample Name: "b120411a048" Sample ID: "GLP10-01-02-28-076" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-TRIP01-MHS-" Annotation: ""Sample Index: 48Sample Type: UnknownConcentration: N/ACalculated Conc: 33.5 ng/mLAcq. Date: 4/12/2012Acq. Time: 5:19:21 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 1913454 countsHeight: 3.77e+005 cpsStart Time: 15.7 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e5

Intensity, cps16.0315.8815.7216.4317.75*** Buster J2930203Batch Name: b120411a.dabData printed by STWPrinting Time: 8:45:47 AMPrinting Date: Thursday, May 03, 2012Page 22 of 30Sample Name: "b120411a057" Sample ID: "GLP10-01-02-28-001" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-130R-0-" Annotation: ""Sample Index: 57Sample Type: UnknownConcentration: N/ACalculated Conc: 387. ng/mLAcq. Date: 4/12/2012Acq. Time: 8:29:07 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 2207959 countsHeight: 2.69e+005 cpsStart Time: 15.6 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e51.9e52.0e52.1e52.2e52.3e52.4e52.5e52.6e5

Intensity, cps16.0115.8515.71Sample Name: "b120411a059" Sample ID: "GLP10-01-02-28-003" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-130S-0-" Annotation: ""Sample Index: 59Sample Type: UnknownConcentration: N/ACalculated Conc: 80.3 ng/mLAcq. Date: 4/12/2012Acq. Time: 9:11:15 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.9 minArea: 4497694 countsHeight: 5.00e+005 cpsStart Time: 15.5 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e54.4e54.6e54.8e55.0e5

Intensity, cps15.8616.0115.70Sample Name: "b120411a064" Sample ID: "GLP10-01-02-28-005" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-130L-0-" Annotation: ""Sample Index: 64Sample Type: UnknownConcentration: N/ACalculated Conc: 51.5 ng/mLAcq. Date: 4/12/2012Acq. Time: 10:56:44 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 2919225 countsHeight: 4.07e+005 cpsStart Time: 15.5 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e5

Intensity, cps15.9915.8415.69Sample Name: "b120411a066" Sample ID: "GLP10-01-02-28-007" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-130L-FMS-" Annotation: ""Sample Index: 66Sample Type: UnknownConcentration: N/ACalculated Conc: 85.5 ng/mLAcq. Date: 4/12/2012Acq. Time: 11:38:54 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 4778058 countsHeight: 7.55e+005 cpsStart Time: 15.5 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e5

Intensity, cps15.9915.8415.69*** Buster J2930203Batch Name: b120411a.dabData printed by STWPrinting Time: 8:45:47 AMPrinting Date: Thursday, May 03, 2012Page 23 of 30Sample Name: "b120411a067" Sample ID: "GLP10-01-02-28-008" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-131R-0-" Annotation: ""Sample Index: 67Sample Type: UnknownConcentration: N/ACalculated Conc: 1080. ng/mLAcq. Date: 4/12/2012Acq. Time: 11:59:58 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.8 minArea: 5983050 countsHeight: 6.84e+005 cpsStart Time: 15.5 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e5

Intensity, cps15.8416.0015.69Sample Name: "b120411a069" Sample ID: "GLP10-01-02-28-010" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-131S-0-" Annotation: ""Sample Index: 69Sample Type: UnknownConcentration: N/ACalculated Conc: 4.66 ng/mLAcq. Date: 4/12/2012Acq. Time: 12:42:12 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.8 minArea: 2648430 countsHeight: 3.54e+005 cpsStart Time: 15.5 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e5

Intensity, cps15.8415.9915.69Sample Name: "b120411a071" Sample ID: "GLP10-01-02-28-012" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-131L-0-" Annotation: ""Sample Index: 71Sample Type: UnknownConcentration: N/ACalculated Conc: 28.1 ng/mLAcq. Date: 4/12/2012Acq. Time: 1:24:22 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.8 minArea: 1613784 countsHeight: 2.16e+005 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e51.9e52.0e52.1e5

Intensity, cps15.8415.9915.69Sample Name: "b120411a076" Sample ID: "GLP10-01-02-28-014" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-133R-0-" Annotation: ""Sample Index: 76Sample Type: UnknownConcentration: N/ACalculated Conc: 20.9 ng/mLAcq. Date: 4/12/2012Acq. Time: 3:09:45 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 95Base. Sub. Window: 1.00 minPeak-Split. Factor: 8Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: Base To BaseRetention Time: 15.8 minArea: 1208231 countsHeight: 1.34e+005 cpsStart Time: 15.5 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e48.0e48.5e49.0e49.5e41.0e51.1e51.1e51.2e51.2e51.3e51.3e5

Intensity, cps15.8415.9915.69*** Buster J2930203Batch Name: b120411a.dabData printed by STWPrinting Time: 8:45:47 AMPrinting Date: Thursday, May 03, 2012Page 24 of 30Sample Name: "b120411a078" Sample ID: "GLP10-01-02-28-016" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-133S-0-" Annotation: ""Sample Index: 78Sample Type: UnknownConcentration: N/ACalculated Conc: 154. ng/mLAcq. Date: 4/12/2012Acq. Time: 3:51:58 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 8405226 countsHeight: 9.26e+005 cpsStart Time: 15.5 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e58.5e59.0e5

Intensity, cps15.9915.8415.68Sample Name: "b120411a080" Sample ID: "GLP10-01-02-28-018" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-133L-0-" Annotation: ""Sample Index: 80Sample Type: UnknownConcentration: N/ACalculated Conc: 70.3 ng/mLAcq. Date: 4/12/2012Acq. Time: 4:34:08 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.8 minArea: 3951385 countsHeight: 4.57e+005 cpsStart Time: 15.5 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e54.4e5

Intensity, cps15.8315.9915.68Sample Name: "b120411a082" Sample ID: "GLP10-01-02-28-020" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-133L-FMS-" Annotation: ""Sample Index: 82Sample Type: UnknownConcentration: N/ACalculated Conc: 201. ng/mLAcq. Date: 4/12/2012Acq. Time: 5:16:17 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 1159836 countsHeight: 1.76e+005 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e5

Intensity, cps15.9915.84Sample Name: "b120411a083" Sample ID: "GLP10-01-02-28-021" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-134R-0-" Annotation: ""Sample Index: 83Sample Type: UnknownConcentration: N/ACalculated Conc: 1110. ng/mLAcq. Date: 4/12/2012Acq. Time: 5:37:25 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.8 minArea: 6125824 countsHeight: 7.31e+005 cpsStart Time: 15.5 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e5

Intensity, cps15.8315.9915.68*** Buster J2930203Batch Name: b120411a.dabData printed by STWPrinting Time: 8:45:47 AMPrinting Date: Thursday, May 03, 2012Page 25 of 30Sample Name: "b120411a088" Sample ID: "GLP10-01-02-28-023" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-134S-0-" Annotation: ""Sample Index: 88Sample Type: UnknownConcentration: N/ACalculated Conc: 14.1 ng/mLAcq. Date: 4/12/2012Acq. Time: 7:22:54 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.8 minArea: 7706894 countsHeight: 9.16e+005 cpsStart Time: 15.5 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e58.5e59.0e5

Intensity, cps15.8315.9815.68Sample Name: "b120411a090" Sample ID: "GLP10-01-02-28-025" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-134S-FMS-" Annotation: ""Sample Index: 90Sample Type: UnknownConcentration: N/ACalculated Conc: 20.9 ng/mLAcq. Date: 4/12/2012Acq. Time: 8:05:03 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 11172061 countsHeight: 1.26e+006 cpsStart Time: 15.5 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e61.10e61.15e61.20e61.25e6

Intensity, cps15.9915.8315.68Sample Name: "b120411a091" Sample ID: "GLP10-01-02-28-026" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-134L-0-" Annotation: ""Sample Index: 91Sample Type: UnknownConcentration: N/ACalculated Conc: 336. ng/mLAcq. Date: 4/12/2012Acq. Time: 8:26:11 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 95Base. Sub. Window: 1.00 minPeak-Split. Factor: 8Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 1920209 countsHeight: 2.42e+005 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e51.9e52.0e52.1e52.2e52.3e52.4e5

Intensity, cps15.9915.83Sample Name: "b120411a093" Sample ID: "GLP10-01-02-28-028" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-135R-0-" Annotation: ""Sample Index: 93Sample Type: UnknownConcentration: N/ACalculated Conc: 1010. ng/mLAcq. Date: 4/12/2012Acq. Time: 9:08:19 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.8 minArea: 5618193 countsHeight: 7.38e+005 cpsStart Time: 15.5 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e5

Intensity, cps15.8315.9915.68*** Buster J2930203Batch Name: b120411a.dabData printed by STWPrinting Time: 8:45:48 AMPrinting Date: Thursday, May 03, 2012Page 26 of 30Sample Name: "b120411a095" Sample ID: "GLP10-01-02-28-030" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-135S-0-" Annotation: ""Sample Index: 95Sample Type: UnknownConcentration: N/ACalculated Conc: 13.1 ng/mLAcq. Date: 4/12/2012Acq. Time: 9:50:33 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.8 minArea: 7217688 countsHeight: 9.18e+005 cpsStart Time: 15.5 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e58.5e59.0e5

Intensity, cps15.8315.9915.68Sample Name: "b120411a100" Sample ID: "GLP10-01-02-28-032" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-135L-0-" Annotation: ""Sample Index: 100Sample Type: UnknownConcentration: N/ACalculated Conc: 64.6 ng/mLAcq. Date: 4/12/2012Acq. Time: 11:36:09 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 95Base. Sub. Window: 1.00 minPeak-Split. Factor: 8Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.8 minArea: 3639551 countsHeight: 4.56e+005 cpsStart Time: 15.5 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e54.4e5

Intensity, cps15.8416.0015.70Sample Name: "b120411a102" Sample ID: "GLP10-01-02-28-036" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-136S-0-" Annotation: ""Sample Index: 102Sample Type: UnknownConcentration: N/ACalculated Conc: 73.9 ng/mLAcq. Date: 4/13/2012Acq. Time: 12:18:16 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 4151608 countsHeight: 5.94e+005 cpsStart Time: 15.5 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e5

Intensity, cps16.0015.8415.69Sample Name: "b120411a104" Sample ID: "GLP10-01-02-28-038" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-136L-0-" Annotation: ""Sample Index: 104Sample Type: UnknownConcentration: N/ACalculated Conc: 60.3 ng/mLAcq. Date: 4/13/2012Acq. Time: 1:00:37 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 3406553 countsHeight: 4.91e+005 cpsStart Time: 15.6 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e54.4e54.6e54.8e5

Intensity, cps16.0015.8415.69*** Buster J2930203Batch Name: b120411a.dabData printed by STWPrinting Time: 8:45:48 AMPrinting Date: Thursday, May 03, 2012Page 27 of 30Sample Name: "b120411a106" Sample ID: "GLP10-01-02-28-042" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-137S-0-" Annotation: ""Sample Index: 106Sample Type: UnknownConcentration: N/ACalculated Conc: 0.110 ng/mLAcq. Date: 4/13/2012Acq. Time: 1:42:51 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 80000 countsHeight: 1.22e+004 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.00500.001000.001500.002000.002500.003000.003500.004000.004500.005000.005500.006000.006500.007000.007500.008000.008500.009000.009500.001.00e41.05e41.10e41.15e41.20e4

Intensity, cpsSample Name: "b120411a112" Sample ID: "GLP10-01-02-28-045" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-137L-0-" Annotation: ""Sample Index: 112Sample Type: UnknownConcentration: N/ACalculated Conc: 0.384 ng/mLAcq. Date: 4/13/2012Acq. Time: 3:49:33 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 236760 countsHeight: 4.01e+004 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e43.4e43.6e43.8e44.0e4

Intensity, cps15.9915.8416.4115.6917.7217.35Sample Name: "b120411a114" Sample ID: "GLP10-01-02-28-049" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-138S-0-" Annotation: ""Sample Index: 114Sample Type: UnknownConcentration: N/ACalculated Conc: 214. ng/mLAcq. Date: 4/13/2012Acq. Time: 4:31:46 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 11400010 countsHeight: 1.84e+006 cpsStart Time: 15.6 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e61.6e61.7e61.8e6

Intensity, cps15.9915.8515.69Sample Name: "b120411a116" Sample ID: "GLP10-01-02-28-051" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-138L-0-" Annotation: ""Sample Index: 116Sample Type: UnknownConcentration: N/ACalculated Conc: 0.137 ng/mLAcq. Date: 4/13/2012Acq. Time: 5:13:55 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 95286 countsHeight: 1.41e+004 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01000.02000.03000.04000.05000.06000.07000.08000.09000.01.0e41.1e41.2e41.3e41.4e4

Intensity, cps*** Buster J2930203Batch Name: b120411a.dabData printed by STWPrinting Time: 8:45:48 AMPrinting Date: Thursday, May 03, 2012Page 28 of 30Sample Name: "b120411a118" Sample ID: "GLP10-01-02-28-053" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-138L-FMS-" Annotation: ""Sample Index: 118Sample Type: UnknownConcentration: N/ACalculated Conc: 0.563 ng/mLAcq. Date: 4/13/2012Acq. Time: 5:56:17 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 338909 countsHeight: 5.74e+004 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e4

Intensity, cps15.9915.8516.3915.6817.71Sample Name: "b120411a119" Sample ID: "GLP10-01-02-28-058" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-140R-0-" Annotation: ""Sample Index: 119Sample Type: UnknownConcentration: N/ACalculated Conc: 115. ng/mLAcq. Date: 4/13/2012Acq. Time: 6:17:18 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.8 minArea: 6353552 countsHeight: 7.16e+005 cpsStart Time: 15.5 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e5

Intensity, cps15.8415.9915.6916.40Sample Name: "b120411a124" Sample ID: "GLP10-01-02-28-060" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-141R-0-" Annotation: ""Sample Index: 124Sample Type: UnknownConcentration: N/ACalculated Conc: 1.70 ng/mLAcq. Date: 4/13/2012Acq. Time: 8:02:54 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 985662 countsHeight: 1.82e+005 cpsStart Time: 15.6 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e5

Intensity, cps15.9915.84Sample Name: "b120411a127" Sample ID: "GLP10-01-02-28-063" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-142R-0-" Annotation: ""Sample Index: 127Sample Type: UnknownConcentration: N/ACalculated Conc: 73.0 ng/mLAcq. Date: 4/13/2012Acq. Time: 9:06:10 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 4101145 countsHeight: 7.16e+005 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e5

Intensity, cps15.9915.8415.68*** Buster J2930203Batch Name: b120411a.dabData printed by STWPrinting Time: 8:45:48 AMPrinting Date: Thursday, May 03, 2012Page 29 of 30Sample Name: "b120411a129" Sample ID: "GLP10-01-02-28-065" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-143R-0-" Annotation: ""Sample Index: 129Sample Type: UnknownConcentration: N/ACalculated Conc: 14.3 ng/mLAcq. Date: 4/13/2012Acq. Time: 9:48:24 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.8 minArea: 7850648 countsHeight: 8.94e+005 cpsStart Time: 15.5 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e58.5e5

Intensity, cps15.8315.9915.68Sample Name: "b120411a131" Sample ID: "GLP10-01-02-28-067" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-144R-0-" Annotation: ""Sample Index: 131Sample Type: UnknownConcentration: N/ACalculated Conc: 13.9 ng/mLAcq. Date: 4/13/2012Acq. Time: 10:30:40 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 15.8 minArea: 812595 countsHeight: 1.11e+005 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e51.05e51.10e5

Intensity, cps15.8215.98Sample Name: "b120411a136" Sample ID: "GLP10-01-02-28-069" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-145R-0-" Annotation: ""Sample Index: 136Sample Type: UnknownConcentration: N/ACalculated Conc: 50.1 ng/mLAcq. Date: 4/13/2012Acq. Time: 12:16:01 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 2844113 countsHeight: 5.02e+005 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e54.4e54.6e54.8e55.0e5

Intensity, cps15.9815.8315.6816.39Sample Name: "b120411a234" Sample ID: "GLP10-01-02-28-063" File: "b120411a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-142R-0-" Annotation: ""Sample Index: 234Sample Type: UnknownConcentration: N/ACalculated Conc: 63.2 ng/mLAcq. Date: 4/14/2012Acq. Time: 10:45:04 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ManualRetention Time: 16.0 minArea: 3565414 countsHeight: 6.10e+005 cpsStart Time: 15.6 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e5

Intensity, cps15.9815.83*** Buster J2930203Batch Name: b120411a.dabData printed by STWPrinting Time: 8:45:48 AMPrinting Date: Thursday, May 03, 2012Page 30 of 30b120417a.rdb (PFOS): "Quadratic" Regression ("1 / x" weighting): y = -0.0134 x^2 + 1.58 x + 0.00786 (r = 0.9998)0 1 2 3 4 5 6 7 8 9 10 11 12 13 14 15 16 17 18 19 20 21 22 23 24Analyte Conc. / IS Conc.024681012141618202224262830

Analyte Area / IS AreaWorkstation: ETSBUSTERBatch Name: b120417a.dab*Data worked up by KJUPage 1 of 1Printing Time: 2:22:04 PMPrinting Date: Wednesday, April 18, 2012Sample Name: "b120417a017" Sample ID: "11012-62-8" File: "b120417a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "1.0 ng/mL FC std in Synth. Water" Annotation: ""Sample Index: 17Sample Type: StandardConcentration: 0.960 ng/mLCalculated Conc: N/AAcq. Date: 4/17/2012Acq. Time: 10:15:29 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: NoMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 0 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.1 minArea: 925821 countsHeight: 3.29e+005 cpsStart Time: 16.0 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e5

Intensity, cps16.06Sample Name: "b120417a025" Sample ID: "12001-1" File: "b120417a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "Method Blank" Annotation: ""Sample Index: 25Sample Type: UnknownConcentration: 0.960 ng/mLCalculated Conc: N/AAcq. Date: 4/18/2012Acq. Time: 1:07:26 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: NoMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 0 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.1 minArea: 1053326 countsHeight: 3.53e+005 cpsStart Time: 16.0 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e5

Intensity, cps16.06Sample Name: "b120417a026" Sample ID: "LCS-120416-1" File: "b120417a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "0.2ppb LCS" Annotation: ""Sample Index: 26Sample Type: QCConcentration: 0.951 ng/mLCalculated Conc: N/AAcq. Date: 4/18/2012Acq. Time: 1:28:55 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: NoMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 0 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.1 minArea: 881211 countsHeight: 3.05e+005 cpsStart Time: 16.0 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e5

Intensity, cps16.05Sample Name: "b120417a029" Sample ID: "LCS-120416-4" File: "b120417a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "2ppb LCS" Annotation: ""Sample Index: 29Sample Type: QCConcentration: 0.951 ng/mLCalculated Conc: N/AAcq. Date: 4/18/2012Acq. Time: 2:33:24 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: NoMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 0 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: Base To BaseRetention Time: 16.1 minArea: 922978 countsHeight: 3.11e+005 cpsStart Time: 16.0 minEnd Time: 16.3 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e5

Intensity, cps16.05*** Buster J2930203Batch Name: b120417a.dabData printed by STWPrinting Time: 8:48:40 AMPrinting Date: Thursday, May 03, 2012Page 1 of 7Sample Name: "b120417a032" Sample ID: "LCS-120416-7" File: "b120417a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "10ppb LCS" Annotation: ""Sample Index: 32Sample Type: QCConcentration: 0.954 ng/mLCalculated Conc: N/AAcq. Date: 4/18/2012Acq. Time: 3:37:52 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: NoMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 0 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: Base To BaseRetention Time: 16.0 minArea: 953240 countsHeight: 3.24e+005 cpsStart Time: 16.0 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e5

Intensity, cps16.05Sample Name: "b120417a039" Sample ID: "GLP10-01-02-28-010" File: "b120417a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "DAL-GW-131S-0-" Annotation: ""Sample Index: 39Sample Type: UnknownConcentration: 0.955 ng/mLCalculated Conc: N/AAcq. Date: 4/18/2012Acq. Time: 6:08:10 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: NoMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 0 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: Base To BaseRetention Time: 16.1 minArea: 1134185 countsHeight: 3.88e+005 cpsStart Time: 16.0 minEnd Time: 16.3 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e5

Intensity, cps16.05Sample Name: "b120417a041" Sample ID: "GLP10-01-02-28-042" File: "b120417a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "DAL-GW-137S-0-" Annotation: ""Sample Index: 41Sample Type: UnknownConcentration: 0.955 ng/mLCalculated Conc: N/AAcq. Date: 4/18/2012Acq. Time: 6:51:04 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: NoMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 0 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: Base To BaseRetention Time: 16.1 minArea: 1035149 countsHeight: 3.52e+005 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e5

Intensity, cps16.04Sample Name: "b120417a043" Sample ID: "GLP10-01-02-28-044" File: "b120417a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "DAL-GW-137S-FMS-" Annotation: ""Sample Index: 43Sample Type: UnknownConcentration: 0.955 ng/mLCalculated Conc: N/AAcq. Date: 4/18/2012Acq. Time: 7:34:03 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: NoMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 0 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: Base To BaseRetention Time: 16.1 minArea: 564920 countsHeight: 1.88e+005 cpsStart Time: 16.0 minEnd Time: 16.3 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e5

Intensity, cps16.05*** Buster J2930203Batch Name: b120417a.dabData printed by STWPrinting Time: 8:48:40 AMPrinting Date: Thursday, May 03, 2012Page 2 of 7Sample Name: "b120417a044" Sample ID: "GLP10-01-02-28-045" File: "b120417a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "DAL-GW-137L-0-" Annotation: ""Sample Index: 44Sample Type: UnknownConcentration: 0.955 ng/mLCalculated Conc: N/AAcq. Date: 4/18/2012Acq. Time: 7:55:39 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: NoMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 0 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: Base To BaseRetention Time: 16.1 minArea: 923570 countsHeight: 3.15e+005 cpsStart Time: 16.0 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e5

Intensity, cps16.05Sample Name: "b120417a049" Sample ID: "GLP10-01-02-28-051" File: "b120417a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "DAL-GW-138L-0-" Annotation: ""Sample Index: 49Sample Type: UnknownConcentration: 0.955 ng/mLCalculated Conc: N/AAcq. Date: 4/18/2012Acq. Time: 9:43:08 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: NoMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 0 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: Base To BaseRetention Time: 16.0 minArea: 1037981 countsHeight: 3.67e+005 cpsStart Time: 15.9 minEnd Time: 16.3 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e5

Intensity, cps16.04Sample Name: "b120417a051" Sample ID: "GLP10-01-02-28-053" File: "b120417a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "DAL-GW-138L-FMS-" Annotation: ""Sample Index: 51Sample Type: UnknownConcentration: 0.955 ng/mLCalculated Conc: N/AAcq. Date: 4/18/2012Acq. Time: 10:26:02 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: NoMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 0 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: Base To BaseRetention Time: 16.0 minArea: 859514 countsHeight: 2.86e+005 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e5

Intensity, cps16.04Sample Name: "b120417a052" Sample ID: "GLP10-01-02-28-060" File: "b120417a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "DAL-GW-141R-0-" Annotation: ""Sample Index: 52Sample Type: UnknownConcentration: 0.955 ng/mLCalculated Conc: N/AAcq. Date: 4/18/2012Acq. Time: 10:47:32 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: NoMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 0 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: Base To BaseRetention Time: 16.0 minArea: 744592 countsHeight: 2.50e+005 cpsStart Time: 16.0 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e51.9e52.0e52.1e52.2e52.3e52.4e52.5e5

Intensity, cps16.04*** Buster J2930203Batch Name: b120417a.dabData printed by STWPrinting Time: 8:48:40 AMPrinting Date: Thursday, May 03, 2012Page 3 of 7Sample Name: "b120417a054" Sample ID: "GLP10-01-02-28-062" File: "b120417a.wiff"Peak Name: "M8PFOS(IS)" Mass(es): "507.0/80.0 amu"Comment: "DAL-GW-141R-FMS-" Annotation: ""Sample Index: 54Sample Type: UnknownConcentration: 0.955 ng/mLCalculated Conc: N/AAcq. Date: 4/18/2012Acq. Time: 11:30:34 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: NoMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 0 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 714337 countsHeight: 2.37e+005 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e51.9e52.0e52.1e52.2e52.3e5

Intensity, cps16.04Sample Name: "b120417a017" Sample ID: "11012-62-8" File: "b120417a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "1.0 ng/mL FC std in Synth. Water" Annotation: ""Sample Index: 17Sample Type: StandardConcentration: 0.928 ng/mLCalculated Conc: 0.923 ng/mLAcq. Date: 4/17/2012Acq. Time: 10:15:29 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.1 minArea: 1402111 countsHeight: 3.32e+005 cpsStart Time: 15.7 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e5

Intensity, cps16.0615.92Sample Name: "b120417a025" Sample ID: "12001-1" File: "b120417a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "Method Blank" Annotation: ""Sample Index: 25Sample Type: UnknownConcentration: N/ACalculated Conc: 0.00119 ng/mLAcq. Date: 4/18/2012Acq. Time: 1:07:26 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 15.9 minArea: 10336 countsHeight: 1.50e+003 cpsStart Time: 15.8 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0100200300400500600700800900100011001200130014001500160017001800190020002100

Intensity, cpsSample Name: "b120417a026" Sample ID: "LCS-120416-1" File: "b120417a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "0.2ppb LCS" Annotation: ""Sample Index: 26Sample Type: QCConcentration: 0.184 ng/mLCalculated Conc: 0.177 ng/mLAcq. Date: 4/18/2012Acq. Time: 1:28:55 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.1 minArea: 266059 countsHeight: 6.51e+004 cpsStart Time: 15.7 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e4

Intensity, cps16.0515.9216.4315.7417.7417.8617.6317.3515.4916.5618.3816.7914.8814.3614.64*** Buster J2930203Batch Name: b120417a.dabData printed by STWPrinting Time: 8:48:41 AMPrinting Date: Thursday, May 03, 2012Page 4 of 7Sample Name: "b120417a029" Sample ID: "LCS-120416-4" File: "b120417a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "2ppb LCS" Annotation: ""Sample Index: 29Sample Type: QCConcentration: 1.84 ng/mLCalculated Conc: 1.83 ng/mLAcq. Date: 4/18/2012Acq. Time: 2:33:24 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.1 minArea: 2768956 countsHeight: 6.53e+005 cpsStart Time: 15.7 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e5

Intensity, cps16.0515.9115.74Sample Name: "b120417a032" Sample ID: "LCS-120416-7" File: "b120417a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "10ppb LCS" Annotation: ""Sample Index: 32Sample Type: QCConcentration: 9.22 ng/mLCalculated Conc: 9.24 ng/mLAcq. Date: 4/18/2012Acq. Time: 3:37:52 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 13401069 countsHeight: 2.85e+006 cpsStart Time: 15.6 minEnd Time: 16.3 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e54.0e56.0e58.0e51.0e61.2e61.4e61.6e61.8e62.0e62.2e62.4e62.6e62.8e6

Intensity, cps16.0515.9115.73Sample Name: "b120417a039" Sample ID: "GLP10-01-02-28-010" File: "b120417a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-131S-0-" Annotation: ""Sample Index: 39Sample Type: UnknownConcentration: N/ACalculated Conc: 5.98 ng/mLAcq. Date: 4/18/2012Acq. Time: 6:08:10 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 15.9 minArea: 10629872 countsHeight: 1.47e+006 cpsStart Time: 15.5 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e6

Intensity, cps15.9016.0515.74Sample Name: "b120417a041" Sample ID: "GLP10-01-02-28-042" File: "b120417a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-137S-0-" Annotation: ""Sample Index: 41Sample Type: UnknownConcentration: N/ACalculated Conc: 0.366 ng/mLAcq. Date: 4/18/2012Acq. Time: 6:51:04 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.1 minArea: 632255 countsHeight: 1.17e+005 cpsStart Time: 15.7 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e51.05e51.10e51.15e5

Intensity, cps16.05*** Buster J2930203Batch Name: b120417a.dabData printed by STWPrinting Time: 8:48:41 AMPrinting Date: Thursday, May 03, 2012Page 5 of 7Sample Name: "b120417a043" Sample ID: "GLP10-01-02-28-044" File: "b120417a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-137S-FMS-" Annotation: ""Sample Index: 43Sample Type: UnknownConcentration: N/ACalculated Conc: 1.25 ng/mLAcq. Date: 4/18/2012Acq. Time: 7:34:03 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.1 minArea: 1162596 countsHeight: 2.14e+005 cpsStart Time: 15.7 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e51.9e52.0e52.1e5

Intensity, cps16.0615.9115.74Sample Name: "b120417a044" Sample ID: "GLP10-01-02-28-045" File: "b120417a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-137L-0-" Annotation: ""Sample Index: 44Sample Type: UnknownConcentration: N/ACalculated Conc: 1.25 ng/mLAcq. Date: 4/18/2012Acq. Time: 7:55:39 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.1 minArea: 1897428 countsHeight: 3.75e+005 cpsStart Time: 15.7 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e5

Intensity, cps16.0615.90Sample Name: "b120417a049" Sample ID: "GLP10-01-02-28-051" File: "b120417a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-138L-0-" Annotation: ""Sample Index: 49Sample Type: UnknownConcentration: N/ACalculated Conc: 0.419 ng/mLAcq. Date: 4/18/2012Acq. Time: 9:43:08 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 725512 countsHeight: 1.26e+005 cpsStart Time: 15.6 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e51.05e51.10e51.15e51.20e51.25e5

Intensity, cps16.0515.90Sample Name: "b120417a051" Sample ID: "GLP10-01-02-28-053" File: "b120417a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-138L-FMS-" Annotation: ""Sample Index: 51Sample Type: UnknownConcentration: N/ACalculated Conc: 1.32 ng/mLAcq. Date: 4/18/2012Acq. Time: 10:26:02 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 1865827 countsHeight: 3.33e+005 cpsStart Time: 15.6 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e5

Intensity, cps16.0415.8815.73*** Buster J2930203Batch Name: b120417a.dabData printed by STWPrinting Time: 8:48:41 AMPrinting Date: Thursday, May 03, 2012Page 6 of 7Sample Name: "b120417a052" Sample ID: "GLP10-01-02-28-060" File: "b120417a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-141R-0-" Annotation: ""Sample Index: 52Sample Type: UnknownConcentration: N/ACalculated Conc: 5.89 ng/mLAcq. Date: 4/18/2012Acq. Time: 10:47:32 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 6887313 countsHeight: 1.33e+006 cpsStart Time: 15.6 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e58.5e59.0e59.5e51.0e61.1e61.1e61.2e61.2e61.3e61.3e6

Intensity, cps16.0415.8915.73Sample Name: "b120417a054" Sample ID: "GLP10-01-02-28-062" File: "b120417a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-141R-FMS-" Annotation: ""Sample Index: 54Sample Type: UnknownConcentration: N/ACalculated Conc: 9.70 ng/mLAcq. Date: 4/18/2012Acq. Time: 11:30:34 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 10484305 countsHeight: 1.92e+006 cpsStart Time: 15.6 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e61.6e61.7e61.8e61.9e6

Intensity, cps16.0415.8815.73*** Buster J2930203Batch Name: b120417a.dabData printed by STWPrinting Time: 8:48:41 AMPrinting Date: Thursday, May 03, 2012Page 7 of 7

# A TTACHMENT 

# C: A NALYTICAL 

# M ETHOD 

# ( S 

# )

### ETS -8- 044. 1 Page 2 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS;

Direct Injection Analysis 

## 1 Scope and Application

This method describes the direct injection analysis of perfluorinated compounds (PFCs) from

water matrices using high

- performance liquid chromatography tandem mass spectrometry

(HPLC/MS/MS). The method is generally applicable but not limited to the measurement of

perfluoroalkyl sulfonamides and perfluorinated alkyl acids (PFAAs) such as perfluorosulfonic

acids (PFSAs) and perfluorocarboxylic acids (PFCAs) (Table 1). Water samples containing

heavy particulate may require preparation by an alternate method such as ETS -8- 154

“ Determination of Perfluorinated Acids, Alcohols, Amides, and Sulfonates In Water By Solid

Phase Extraction and High Performance Liquid Chromatography/Mass Spectrometry ”. The

method is applicable to both external standard and internal standard calibration 1



### .

Table 1. Representative Tar get Analytes

**Acronym Analyte**  **Chemical Abstract Services**

**Registry Number (CASRN)**

PFBA (C4 Acid) Perfluorobutanoic acid 375

- 22

- 4

PFPeA (C5 Acid) Perfluoropentanoic acid 2706

- 90

- 3

PFHxA (C6 Acid) Perfluorohexanoic acid 307

- 24

- 4

PFHpA (C7 Acid) Perfluoroheptanoic acid 375

- 85

- 9

PFOA (C8 Acid) Perfluorooctanoic acid 335

- 67

- 1

PFNA (C9 Acid) Perfluorononanoic acid 375

- 95

- 1

PFDA (C10 Acid) Perfluorodecanoic acid 335

- 76

- 2

PFUnA (C11 Acid) Perfluoroundecanoic acid 2058

- 94

- 8

PFDoA (C12 Acid) Perfluorododecanoic acid 307

- 55

- 1

PF TrD A (C1 3 Acid) Perfluorotridecanoic acid 72629

- 94

- 8

PFBS (C4 Sulfonate) Perfluorobutanesulfonic acid 375

- 73

- 5

PFHS (C6 Sulfonate) Perfluorohexanesulfonic acid 355

- 46

- 4

PFOS (C8 Sulfonate) Perfluorooctanesulfonic acid 1763

- 23

- 1

FBSA (C4 Sulfonamide Perfluorobutanesulfonamide 30334

- 69

- 1

FOSA (C8 Sulfonamide) Pefluorooctanesulfonamide 754

- 91

- 6



### The Minimum Reporting Level (MRL) is the Limit of Quantitation (LOQ) that meets Data

Quality Objectives (DQOs) that are developed based on the intended use of this method.

Method Flexibility

– This is a performance

- based method and may be generally applied to the

determination of perfluorinated compounds in water matrices when analysis batch quality

c ontrol ( QC ) criteria are met 2



### . Each set of samples are prepared in an analysis batch with

calibration standards, LCS s, blanks, and continuing calibration check standards analyzed on

the same instrument during a time period that begins and ends with the analysis of the

appropriate continuing calibration check standards. The laboratory is permitted to modify the

LC column, mobile phase composition, LC conditions, and MS/MS conditions. Method

modifications should be considered to improve method performance or to meet data quality

objectives for the study. In all cases where method modifications are implemented, the batch

1 The method is supported by validation with internal standard calibration for C4

- C13 PFCAs, C4, C6, and C8 PFSAs,

and C8 perfluoroalkane sulfonamide in laboratory control samples under 3M method validation E11

- 0667.

2 Guidance for establishing method QC Criteria based on a.) FDA May 2001, “Guidance for Industry, Bioanalytical

Method Vali dation”, b.) EPA Method 537, and c.) European Commission: Guidance for Generating and Reporting

Methods of Analysis in Support of Pre

- registration Data Requirements for Annex II (Part A, section 4) and Annex III

(Part A,section 5) of Directive 91/414, SANCO/3029/99 rev. 4 (11/07/00).

### ETS -8- 044. 1 Page 3 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS;

Direct Injection Analysis analytical QCs ( section 9) must be completed and pass QC acceptance criteria ( section 13 ) if

the data from the analytical batch are to be reported

.



## 2 Method Summary

Water samples are analyzed as neat aqueous sample or as solvent diluted aqueous samples

by direct injection using LC/ MS/MS. Samples containing heavy particulate may not be

suitable for analysis by this method. Samples containing suspended particulate should be

centrifuge d or filtered prior to removing a sample aliquot or diluting with solvent

. The water

sample is mixe d well prior to removing an aliquot or diluting, if necessary, with ASTM Type I

water, HPLC water, other suitable water, or solvent (methanol).

Quantitation is by stable isotope internal standard calibration in laboratory reagent water. All

perfluorinat ed compounds ( PFC s) target analyte concentrations of perfluorosulfonic acids

(PFSAs) and perfluorocarboxylic acids (PFCAs) are reported as anions and corrected for

their salt or free acid forms. Alternatively, quantitation may be performed by external

st andard calibration.

This is a performance

- based method. Method uncertainty for each target analyte is

determined for each analytical batch using multiple laboratory control spikes at multiple

concentrations. This method also requires that the precision and accuracy for each sample

be determined using field matrix spikes to verify that the method is applicable to each sample

matrix.

Calibration standards for PFUnA, PFDoA, PFTrDA, and FOSA have been found to be

un stable for more than 2 days in 100% water. Samples requiring analysis for these

compounds by this method should be diluted 1:1 with methanol and analyzed against a

calibration curve prepared in 1:1 synthetic groundwater:MeOH.



## 3 Definitions

3.1 Analysis Batch

A set of study samples that are prepared with calibration standards, laboratory control

samples, and procedural blanks , and analyzed on the same instrument during a time period

that begins and ends with the analysis of the appropriate continuing calibration check

standards.



## 3.2 Analytical Sample

A portion of a laboratory sample prepared for analysis.



## 3.3 Calibration Standard

A solution prepared by spiking a known volume of the Working Standard (WS) into a

predete rmined amount of ASTM Type I, HPLC grade water, or other suitable water (i.e.

matrix water), and analyzed according to this method. Calibration standards are used to

calibrate the instrument response with respect to analyte concentration.



## 3.4 Laboratory Duplicate Sample (LDS, or Lab Dup)

A laboratory duplicate sample is a separate aliquot of a sample taken in the analytical

laboratory that is analyzed separately with identical procedures. Analysis of LDSs compared

to that of the first aliquot give a measure of the precision associated with laboratory

procedures, but not with sample collection, preservation, or storage procedures.

### ETS -8- 044. 1 Page 4 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS;

Direct Injection Analysis 

## 3.5 Field Blank (FB)/Trip Blank (TB)

ASTM Type I, HPLC grade water, or other suitable water, placed in a sample container in the

laboratory and treated as a sample in all respects, including exposure to samp ling site

conditions, storage, preservation and all analytical procedures. The purpose of the TB is to

determine if test substances or other interferences are present in the field environment. This

sample is also referred to as a Trip Blank.



## 3.6 Field Duplicate Sample (FDS, Field Dup)

A sample collected in duplicate at the same time from the same location as the sample. The

FDS is handled under identical circumstances and treated exactly the same throughout field

and laboratory procedures. Analysis of the FDS compared to that of the first sample gives a

measure of the precision associated with sample collection, preservation and storage, as well

as with laboratory procedures.



## 3.7 Field Matrix Spike (FMS)

A sample to which known quantities of the target analyte s , ISs and SRSs are added to the

sample bottle in the laboratory before the bottles are sent to the field for collection of aqueous

samples. A known, specific volume of sample must be added to the sample container

without rinsing. This may be accomplished by making a “fill to this level” line on the outside of

the sample container. The FMS is analyzed to ascertain if any matrix effects, interferences,

or stability issues may complicate the interpretation of the sample analysis.



## 3.8 Trip Blank Matrix Spike ( T BMS )

An aliquot of ASTM Type I, HPLC grade water, or other suitable water, to which known

quantities of the target analytes , ISs and SRSs are added in the laboratory prior to the

shipment of the collection bottles. The TBMS is analyzed exactly like a study sample to help

determine if the method is in control and whether a loss of analyte or analytical bias could be

attributed to sample holding time, sample storage and/or shipment issues. A low and high

TBMS are appropriate when expected sample concentrations are not known or may vary.



## 3.9 Internal Standard (IS)

A compound added to each study sample, calibration standard, laboratory control samples,

and procedural blanks at a consistent level (typically around 1 ng/mL). T he internal

standard (s) are stable isotope labeled versions of the target analyte s . The area count ratio of

the target analyte to the internal standard is used for calibration. Surrogate ISs are applied

when stable isotope ISs of target analytes are unavailable. A surrogate IS is not necessar ily

a stable isotope labeled version of the target analyte, but is treated as an internal standard for

quantitation.



## 3.10 Laboratory Control Sample (LCS)

An aliquot of control matrix to which known quantities of the target analytes , ISs and SRSs

(when applicabl e) are added in the laboratory at the time when samples are aliquotted . At

least three levels (two levels for SRSs) in triplicate are included, one generally at the low end

of the calibration curve and one near the mid range and the upper end of the curve. The

LCSs are analyzed exactly like a laboratory sample to determine whether the stability of the

standards . LCSs should be prepared each day samples are aliquoted

.



## 3.11 Laboratory Matrix Spike (LMS)

A laboratory matrix spike is an aliquot of a sample to which known quantities of target

analytes , ISs and SRSs (when applicable) are added in the laboratory. The LMS is analyzed

exactly like a laboratory sample to determine whether the sample matrix contributes bias to

the analytical results. The endogenous concentrations of the analytes in the sample matrix

must be determined in a separate aliquot and the measured values in the LMS corrected for

these concentrations. LMSs are optional for analysis of aqueous samples.

### ETS -8- 044. 1 Page 5 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS;

Direct Injection Analysis 

## 3.12 Laboratory Sample

A portion or aliquot of a sample received from the field for testing.



## 3.13 Limit of Quantitation (LOQ)

The lower limit of quantitation (LLOQ) for an analytical batch is the lowest concentration that

can be reliably quantitated within the specified limits of precision and accuracy. The LLOQ is

generally selected as the lowest non

- zero standard in the calibration curve that meets method

acceptance criteria. The LLOQ for each target analyte is established for each analysis batch

as the lowest calibration standard with area counts at least twice that of the average area

counts of the procedural blanks.

The upper limit of quantitation (ULOQ) for an analytical batch is the highest concentration that

can be reliably quantitated within the specified limits of precision and accuracy. The highest

standard in the calibration curve that meets method acceptance criteria is defined as the

ULOQ.



## 3.14 Method /Procedural Blank

An aliquot of control matrix that is treated exactly like a laboratory sample including exposure

to all glassware, equipment, solvents, and reagents that are used with other laboratory

samples. The method blank is used to determine if test substances or other interferences are

present in the laboratory environment, the reagents, or the apparatus.



## 3.15 Sample

A sample is an aliquot removed from a larger quantity of material intended to represent the

original source material.



## 3.16 Stock Standard Solution (SSS)

A concentrated solution of a single

- analyte prepared in the laboratory with an assayed

reference compound.



## 3.17 Surrogate Internal Standard

An IS that is not necessarily a stable isotopically labeled target analyte, but is treated as an

internal standard for quantitation. Surrogate ISs are used when isotopically labeled

counter parts of the target analyte are not commercially or re adily available.



## 3.18 Surrogate Recovery Standard (SRS)

An isotopically labeled standard , not used as an internal standard, that is added to each

sample and appropriate QC sample as a means to evaluate the method performance for a

chemical class of compounds (e .g., PFSAs, PFCAs)

.



## 3.19 Working Standard (WS)

A solution of several analytes prepared in the laboratory from SSSs and diluted as needed to

prepare calibration standards and other required analyte solutions.



## 4 Warnings and Cautions

4.1 Health and Safety

The acute and chronic toxicity of the standards for this method have not been precisely

determined; however, each should be treated as a potential health hazard. The analyst

should wear gloves, a lab coat, and safety glasses to prevent exposure to chemicals that

might be present.

### ETS -8- 044. 1 Page 6 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS;

Direct Injection Analysis The laboratory is responsible for maintaining a safe work environment and a current

awareness of local regulations regarding the handling of the chemicals used in this method.

A reference file of material safety data sheets (MSDS) should be available to all personnel

involved in these analyses.



## 4.2 Cautions

The analyst must be familiar with the laboratory equipment and potential hazards including,

but not limited to, the use of solvents, pressurized gas and solvent lines, high voltage, and

vacuum systems. Refer to the appropriate equipment procedure or operator manual for

additional information and cautions.



## 5 Interferences

During sample preparation and analysis, major potential contaminant sources are reagents

and glassware. All materials used in the analyses shall be demonstrated to be free from

interferences under conditions of analysis by running method blanks.

Parts and supplies that contain Teflon® should be avoided or minimized due to the possibility

of interference and/or contamination. These may include, but are not limited to: wash bottles,

Teflon ® 

### lined caps, autovial caps, HPLC parts, etc.

The use of disposable micropipettes or pipettes to aliquot standard solutions is recommended

to make calibration standards and matrix spikes.



## 6 Instrumen tation, Supplies, and Materials

6.1 Instrumentation

Analytical balance capable of reading to 0.0001g

HPLC/MS/MS or HPLC/MS system, as described in Section 10

.



## 6.2 Supplies and Materials

Sample collection bottles

— HDPE (e.g., NalgeneTM) wide

- mouth bottles with screw cap.

Note: Do not use fluorinated or Teflon® bottles or lined caps.

Coolers or boxes for sample shipment.

15

- mL and 50

- mL disposable polypropylene centrifuge tubes.

Class A pipettes and volumetric flasks, various.

2 mL HPLC auto vial s

Disposable pipettes, polypropylene or glass as appropriate

Centrifuge capable of spinning 15

- mL and 50

- mL polypropylene tubes at 3000 rpm.



## 7 Reagents and Standards

Note: Suppliers and catalog numbers are for illustrative purposes only. Equivalent

performance may be ach ieved using chemicals obtained from other suppliers. Do not use a

lesser grade of chemical than those listed.



## 7.1 Chemicals

Water

– Milli

- Q, HPLC grade, or other suitably appropriate sources

Calcium Acetate

- A.C.S. Reagent Grade

### ETS -8- 044. 1 Page 7 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS;

Direct Injection Analysis Magnesium Acetate

- A.C.S. Re agent Grade

Methanol

– HPLC grade

Ammonium Acetate

– A.C.S. Reagent Grade



## 7.2 Representative Target Analytes, ISs, and SRSs

PFBA, Heptafluorobutyric Acid, (C 4 

### Perfluorinated Acid)

PFPeA , Nonafluoropentanoic Acid (C 5 

### Perfluorinated Acid)

PFHxA, Perfluorohexanoi c Acid (C 6 

### Perfluorinated Acid)

PFHpA, Tridecafluoroheptanoic Acid, (C 7 

### Perfluorinated Acid)

PFOA, Ammonium perfluorooctanoate, (C 8 

### Perfluorinated Acid)

PFNA, Heptadecafluorononanoic Acid, (C 9 

### Perfluorinated Acid)

PFDA, Nonadecafluorodecanoic Acid (C 10 

### Per fluorinated Acid)

PFUnA, Perfluoroundecanoic Acid, (C 11 

### Perfluorinated Acid)

PFDoA, Perfluorododecanoic Acid, (C 12 

### Perfluorinated Acid)

PFTrDA, Perfluorotridecanoic Acid, (C 13 

### Perfluorinated Acid)

FBSA, Perfluorobutanesulfonamide

FOSA, Perfluorooctanesulfo nylamide

PFBS, Potassium Perfluorobutane sulfonate

PFHS, Perfluorohexanesulfonate

PFOS, Potassium perfluorooctanesulfonate

PFOA [1,2, 3, 4

- 13 

### C], 13 

### C 4 

### - isotopically labeled perfluorooctanoic acid (SRS)

PFOS [1,2, 3, 4

- 13 

### C], 13 

### C 4 

### - isotopically labeled Perfluorooctanesulfonate (SRS)

PFUnA [1,2

- 13 

### C], 13 

### C 2 

### - isotopically labeled Perfluoroundecanoic acid (SRS)

A custom mix of ISs in a methanolic solution containing ([1,2,3,4

- 13 

### C 4 

### ]PFBA, [1,2

-

13 

### C 2 

### ]PFHxA, [1,2,3,4 ,5,6,7,8

- 13 

### C 8 

### ]PFOA, [1,2,3,4,5 ,6,7,8,9

- 13 

### C 9 

### ]PFN A, [1,2

- 13 

### C 2 

### ]PFDA,

[1,2 ,3,4,5,6,7

- 13 

### C 7 

### ]PFUnA, [1,2

- 13 

### C 2 

### ]PFDoA, [1,2 ,3

- 13 

### C 3 

### ]PFHS, [1,2,3,4 ,5,6,7,8

- 13 

### C 8 

### ]PFOS

, and [1,2,3,4,5,6,7,8

- 13 

### C 8 

### ]PFOSA (Wellington Laboratories, Guelph, ON) in combination with

added ([1,2,3,4 ,5

- 13 

### C 5 

### ]PFPe A

, ([1,2,3,4

- 13 

### C 4 

### ]PFHpA, and [ 18 

### O 2 

### ]PFBS can be used to prepare

a stock IS solution. Alternatively, individual stable isotope ISs can be used to prepare a stock

IS mixture.

Other ISs can be applied.



## 7.3 Reagent Preparation

2 mM A mmonium acetate solution (Analysis)

— Weigh 0.3 g of A mmonium acetate and

dissolve in 2.0 L of reagent water.

Synthetic Groundwater (containing 25 ppm Ca and Mg)

– Weigh 0.61 g of Calcium Acetate

and 0.92 g of Magnesium Acetate and dissolve in 6.0 L of reagent water.

Note: Alternative volumes may be prepared as long as the ratios of the solvent to solute ratios

are maintained.

### ETS -8- 044. 1 Page 8 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS;

Direct Injection Analysis 

## 7.4 Stock Standard Solution (SSS) and Working Standard Solution

Preparation

The following standard preparation procedure serves as an example. Weighed amounts and

final volumes may be changed to suit the needs of a particular study. For example, μL

volumes may be spiked into volumetric flasks when diluting stock solutions to appropriate

levels.

100 μg/mL target analyte SSSs

— Weigh out 10 mg of analytical standard **(corrected for**

**percen t salt**  **, acid [ETS -4-**  **031] and purity**  ) and dilute to 100 mL with methanol or other

suitable solvent, in a 100 mL volumetric flask. Transfer to a 125 mL LDPE bottle or other

suitable container. Prepare a separate solution for each analyte. Expiration dates and

storage conditions of stock solutions should be assigned in accordance with laboratory

standard operating procedure. An example of purity and salt correction is given below for

PFOS.



### salt of weight moclecular anion of weight molecular factor correction salt

=

9275.0

538 499 factor correction )salt (K PFOS

== +



### 10 mg C 8 

### F 17 

### S0 3 - 

### K + 

### with purity 90% = 8.35 mg C 8 

### F 17 

### S0 3 – 

### (10 mg*0.90*0.9275=8.35 mg)

10 μg/mL (10,000 ng/mL) mixed working standard

— Add 5.0 mL each of the 100 μg/mL

SSSs to a 50 mL volumetric flask and bring up to volume with solvent.

1 μg/mL (1,000 ng/mL) mixed working stan dard

— Add 0.5 mL of the 100 μg/mL SSSs to a

50 mL volumetric flask and bring up to volume with solvent.

0.1 μg/mL (100 ng/mL) mixed standard

— Add 0.05 mL of the 100 μg/mL SSSs to a 50 mL

volumetric flask and bring up to volume with solvent.

Storage Conditi ons

— Store all SSSs and working standards in accordance with laboratory

standard operating procedure or in a refrigerator at 4°±2°C for a maximum period of 6 months

from the date of preparation.



## 7.5 Calibration Standards

Calibration can be performed by IS or external calibration. Using the working standards

described above, prepare calibration solutions in ASTM Type I water, HPLC water, other

suitable water, or a mixture of solvent and water using the information in Table 2 as a

guideline. Note: Volumes of wa ter or water/solvent mixtures and working standards may be

adjusted to meet the data quality objectives addressed in the general project outline.

Calibration levels other than those listed below can be prepared as needed.

For the quantitation of PFOA and PFOS, reference materials of certified mixed linear and

branched isomer are preferred. Alternately, reference materials of primarily linear isomers of

PFOA and/or PFOS may be used, however, when quantitating with predominantly linear

reference standards, additional LCS samples containing both linear and branched isomers of

PFOA and PFOS are required 3



### .

7.5.1 Internal Standard (IS) and Surrogate Recovery Standard (SRS)

For IS calibration, stable isotope internal standards of each target analyte or appropriate

surrogate ISs should be spiked at the same level in all calibration standards. Once the

calibration standards have been prepared as stated above in Section 7.5, all calibration

standards are spiked with a separate internal standard spiking solution. Typically the

3 A report summarizing an assessment of the use of reference standards containing certified linear and branched isomers of

PFOA/PFOS can be found in 3M report E11

- 0560.

### ETS -8- 044. 1 Page 9 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS;

Direct Injection Analysis concentration of the internal standard is consistent with the internal standard concentration

expected in the samples being prepared, usually 1 ng/mL. The concentration of the internal

standard spiking solution is typically 2 μg/ mL. A separate zero point or method blank is

typically prepared at the same time as the calibration standards, using the same solution

used to prepare the standards (ASTM Type I water, HPLC water, other suitable water, or a

solvent/water mixture), and is spiked with the internal standard at the same concentration as

the calibration curve, typically at 1 ng/mL.

If the sample s being analzyed w ere pre

- spiked with SRSs , the calibration curve prepared in

Section 7.5 is spiked with a separate SRS spiking solution. Typically, the sample bottles are

spiked with a SRS at 0.1 ng/mL. The final calibration curve must consist of at least six

calibration points after analysis. The following table provides an example of spike

concentrations and volumes used to achieve a multi

- point extracted calibration curve with

internal standard and surrogate standard.

Table 1 lists recommended stable isotope internal standards for several PFSA and PFCA

target compounds. A custom mix of isotopically labeled target analytes in a methanolic

solution containing ([1,2,3,4

- 13 

### C 4 

### ]PFBA, [1,2

- 13 

### C 2 

### ]PFHxA, [1,2,3,4,5,6,7,8

- 13 

### C 8 

### ]PFOA,

[1,2,3,4,5,6,7,8,9

- 13 

### C 9 

### ]PFNA, [1,2,3,4,5,6

- 13 

### C 6 

### ]PFDA, [1,2,3,4,5,6,7

- 13 

### C 7 

### ]PFUnA, [1,2

-

13 

### C 2 

### ]PFDoA, [ 1,2,3

- 13 

### C 3 

### ]PFHS, [1,2,3,4,5,6,7,8

- 13 

### C 8 

### ]PFOS, and [1,2,3,4,5,6,7,8

- 13 

### C 8 

### ]FOSA

(Wellington Laboratories, Guelph, ON) in combination with added ([1,2,3,4 ,5

- 13 

### C 5 

### ]PFPe A

, ([1,2,3,4

- 13 

### C 4 

### ]PFHp A , and [ 18 

### O 2 

### ]PFBS can be used to prepare a stock IS solution.

Alternative sources of certified stable isotope labeled target analytes are applicable.

Alternatively, individual stable isotope ISs can be used to prepare a stock IS mixture. The

table below lists the recommended stable isotope ISs and SRSs applied in the method.

Other stable isotope ISs and SRSs of target analytes not listed in the table may be used if

supported by validation and/or analysis batch QCs meeting method acceptance criteria (e.g.,

[ 13 

### C 2 

### ]- PFOA) . The same internal standard should be used for a given analyte throughout the

entire project/study. Note: some of the compounds listed below are appropriate to use as

surrogate ISs when a stable isotope IS of a target analyte is not available. Generally,

surrogate isotopically labeled PFCAs are used for PFCAs, and surrogate i sotopically labeled

PFSA s are used for PFSAs.

Table 2 provides examples of spike concentrations and volumes used to achieve a multi

- point

calibration curve with ISs and SRSs

.

### ETS -8- 044. 1 Page 10 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS;

Direct Injection Analysis Table 1 . Stable Isotope PFCAs and PFSAs used for ISs and SRSs

Compound Name Synonym or Acronym Analytical Purpose Reference Standard

Source

13 C 4 - Perfluorobutanoic acid [1,2,3,4

- 13 C 4 ]PFBA IS for PFBA Wellington Labs

(Mix or Individual)

13 C 4 - Perfluoro pentanoic acid [1,2,3,4 ,5

- 13 C 5 ]PFPe A IS for PFPeA Wellington Labs

(Mix or Individual)

13 C 2 - Perfluorohexanoic acid [1,2

- 13 C 2 ]PFHxA IS for PFHxA Wellington Labs

(Mix or Individual)

13 C 4 - Perfluoro hept anoic acid [1,2,3,4

- 13 C 4 ]PFHp A IS for PF HpA Wellington Labs

(Mix or Individual)

13 C 8 - Perfluorooctanoic acid [1,2,3,4,5,6,7,8

- 13 C 8 ]PFOA IS for PFOA and [1,2,3,4

13 C 4 ]PFOA Wellington Labs

(Mix or Individual)

13 C 9 - Perfluorononanoic acid [1,2,3,4,5,6,7,8,9

- 13 C 9 ]PFNA IS for PFNA Wellington Labs

(Mix or Individual)

13 C 6 - Perfluorodecanoic acid [1,2,3,4,5,6

- 13 C 6 ]PFDA IS for PFDA Wellington Labs

(Mix or Individual)

13 C 7 - Perfluoroundecanoic acid [1,2,3,4,5,6,7

- 13 C 7 ]PFUnA IS for PFUnA Wellington Labs

(Mix or Individual)

13 C 2 - Perfluorododecanoic acid [1,2

- 13 C 2 ]PFDoA IS for PFDoA, *PFTA Wellington Labs

(Mix or Individual)

18 O 2 - Ammonium Perfluorobutane sulfonate [ 18 O 2 ]PFBS IS for PFBS RTI International

(Individual)

13 C 3 - Ammonium Perfluorohexane sulfonate [ 1,2,3

- 13 C 3 ]PFHS IS for PFHS Wellington Labs

(Mix or Individual)

13 C 8 - Sodium Perfluorooctane sulfonate [1,2,3,4,5,6,7,8

- 13 C 8 ]PFOS IS for PFOS and

PFOS[1,2,3,4 13 C 4 ], Wellington Labs

(Mix or Individual)

13 C 8 - Perfluorooctanesulfonamide [1,2,3,4,5,6,7,8

- 13 C 8 ]FOSA IS for FOSA Wellington Labs (mix)

RTI International

(Individual)

13 C 4 - Perfluorooctanoic acid [1,2,3,4

- 13 C 4 ]PFOA SRS for all PFCAs : C4

- C8 Wellington

13 C 2 - Perfluoroundecanoic acid [1,2

- 13 C 2 ]PFUnA SRS for all PFCAs C9

- C13 Wellington

13 C 8 - Perfluorooctane sulfonate [1,2,3,4

- 13 C 4 ]PFOS SRS for all PFSA s: C4,

C6, and C8 Wellington

### ETS -8- 044. 1 Page 11 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Water by LC/MS/MS; Direct Injection Analysis Table 2 . Example Preparation of Calibration Curve with ISs and SRSs

**Sample Description Concentration** 

**of WS, μ**  **g/mL Volume of** 

**WS,** μ **L Volume of IS** 

**(2 μ**  **g/mL),** μ **L Concentration of** 

**Surrogate, μ**  **g/mL Volume of** 

**Surrogate,** μ **L Volume of ASTM Type I**  **Water, or**

**other suitable solvent**  **(1) , mL** 

0.025 ng/mL curve point 0.10 25 50 0.20 12.5 100

0.030 ng/mL curve point 0.10 30 50 0.20 15 100

0.04 ng/mL curve point 0.10 40 50 0.20 20 100

0.05 ng/mL curve point 0.10 50 50 0.20 25 100

0.1 ng/mL curve point 0.10 100 50 0.20 50 100

0.25 ng/mL curve point 0.10 250 50 0.20 125 100

0.5 ng/mL curve point 1.0 50 50 0.20 250 100

1 ng/mL curve point 1.0 100 50 0.20 500 100

2.5 ng/mL curve point 10.0 25 50 10.0 25 100

5.0 ng/mL curve point 10.0 50 50 10.0 50 100

10.0 ng/mL curve point 10.0 100 50 10.0 100 100

25.0 ng/mL curve point 10.0 250 50 10.0 NA 100

50.0 ng/mL curve point 10.0 500 50 10.0 NA 100

75.0 ng/mL curve point 10.0 750 50 10.0 NA 100

100 ng/mL curve point 10.0 1000 50 10.0 NA 100

N/A

- Not Applicable

(1) Samples requiring analysis for PFUnA, PFDoA, PFTrDA, and FOSA should be analyzed against a calibration curve prepared in 1:1 synthetic groundwa ter:MeOH.

### ETS -8- 044. 1 Page 12 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS; Direct Injection

Analysis 

## 8 Sample Collection and Bottle Preparation

Sample collection bottles are prepared by 3M Environmental Laboratory (or subcontract supplier) personnel for

shipment at ambient temperature to the collection site. Typically, four separate collection bottles are

associated with a single collection site: sample, field duplicate sample, low field matrix spike, and high field

matrix spike. Alternatively, the sample and field duplicate sample may contain SRSs in lieu of additional target

analyte low field matrix spike and target analyte high field matrix spike samples. Depending on the scope of

the project, additional replicates of the field sample and field matrix spikes may be added. Also, it is not

uncommon for additional mid

- level field matrix spikes to be collected if the expected sample concentrations are

truly unknown or could span a large concentration range.

High

- density polyethylene (HDPE) wide

- mouth Nalgene bottles are used for the sample collection containers.

(Volumes of the bottles may vary depending on how much sample is required to meet data quality objectives.)

Sample collection volumes are project specific and based on data quality objectives. The Nalgene bottles do

not require any pretreatment prior to use. Typically, placement of a sample bottle volumetric “fill to here” line is

done by using a sample bottle marker template. Alternatively, bottles may be weighed prior to bottle

preparation and weighed again after samples have been collected.

All bottles should be clearly labeled to indicate its intended use as a sample, field sample duplicate, low field

matrix spike, high field matrix spike, sample/SRS field matrix spike, field duplicate sample/SRS field matrix

spike, trip blank, or trip blank matrix spike. If each location has different designated spike levels, the label

should also clearly indicate the sample location designation. Generally, a set of bottles for a given collection

site are prepared then grouped together in plastic bags for organizational purposes. For each sample

collection event, at least one set of trip blank and trip blank matrix spikes are prepared.

Bottle preparation should be documented in a Note to File or on a sample preparation worksheet and should

include the following information: date prepared, total number of bottles prepared, number of sample sites, the

standard identification numbers and spike volumes used to prepare spiked bottles, the “fill to here” volume, and

any other pertinent information needed for reconstructibility of the data. The Note to File will be included in the

final data package for the project.

Samples are collected in the field and shipped to the laboratory at ambient temperature.



## 8.1 Field Matrix Spike Sample (FMS)

Field matrix spike samples are a requirement of the method. A FMS sample is defined as a QC sample to

which known quantities of appropriate target analytes are added to the sample bottle in the field or in the

laboratory before the bottles are sent to the field. The sample and field duplicate sample may contain

appropriate SRSs in lieu of target analyte FMS samples. Sample quantities are determined vol umetrically or

gravimetrically. A known, specific volume or weight of sample is added to the sample container without

rinsing. Volumetric sample measurements may be acquired by a laboratory applied “fill to this level” line on the

outside of the sample container. Target analyte FMS samples should be spiked at approximately 0.5

- 10 times

the expected analyte concentration in the sample. If the expected range of analyte concentrations is unknown,

multiple spikes at varying levels may be prepared to increase the likelihood that a spike at an appropriate level

is made. Typically a low and a high target analyte spike are prepared for each sampling location. In those

instances where SRSs are to be used in lieu of target analyte FMS samples, the sample and field duplicate

sample are spiked at approximately 2

- 5 times the target LOQ. The FMS is analyzed to ascertain if matrix

effects or sample holding time contributes bias to the analytical results. For the sample bottles designated for

matrix spikes, an appropriate volume of matrix spiking solution is added to the empty bottle prior to sampling.

The volume of spike solution added should produce the desired final concentration of target analytes once the

bottle is filled with sample to the “fill to here line”. The matrix spiking solution(s) should be prepared in a

suitable solvent and contain all of the appropriate target analytes, ISs, and SRSs. The target analyte matrix

spiking solution is often the same as the working standards used to create the calibration standards. An

example of a bottle spike is given below.

“Fill to here” volume = 200 mL (A 250 mL Nalgene bottle is used)

Desired Field Spike Concentration = 0.25 ng/mL

### ETS -8- 044. 1 Page 13 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS; Direct Injection

Analysis 500 μ L of a 0.1 μ g/mL spiking solution (containing the target analytes) is added to the bottle and the bottle cap

promptly sealed.



## 8.2 Internal Standard and Surrogate Recovery Standard

If analysis of a surrogate recovery standard (SRS) is included in the project objectives, an appropriate volume

of a surrogate standard solution is added to all the bottles prior to sampling and SPE. Typically sample bottles

are spiked with surrogate recovery standards at a final desired spike concentration of 0.1 ng/mL.

If quantitation by internal standard (IS) is included in the project objective, an appropriate volume of internal

standard solution is added to all the bottles prior to sampling and SPE . Typically sample bottles are spiked

with internal standard at a final desired spike concentration of 1 ng/mL.

For the trip blank, the SRS spike and IS spike i s added to the bottle and then ASTM Type I water (HPLC grade

reagent water or other suitable water may used) is added to the “fill to here” line. The bottle is capped and

sealing tape may be placed around the outer edge of the cap. Trip blank matrix spikes are prepared by adding

the appropriate volume of target analyte spiking solution, IS, and SRS spiking solutions and filling the bottle to

the desired volume with the appropriate water and capping and sealing the cap.



## 9 Quality Control and Data Quality Objectives

9.1 Data Quality Objectives

This method and required quality control samples is designed to generate data accurate to ±30% with a

targeted LOQ of 0.025 ng/mL. Any deviations from the quality control measures spelled out below will be

documented in the raw data and footnoted in the final report.



## 9.2 Method/Procedural Blanks

The method/procedural blank is zero point calibration standard (which includes ISs) analyzed in a regular basis

with each analysis batch. At a minimum, method blanks are analyzed prior to instrument calibration, prior to

the analysis of CCV samples, after every 10 sample injections, and at the end of the analytical run.

The mean area count or area ratios when using internal standard calibration, for each analyte in the method

blanks m ust be less than 50% of the area count counts or area ratios when using internal standard calibration,

of the LOQ standard. The standard deviation of the area counts, or area ratios when using internal standard

calibration, of these method blanks should be calculated. A specific %RSD acceptance criteria is not specified

but is assessed on an analytical batch basis. If the mean area counts or area ratios when using internal

standard calibration, of the method blanks exceed 50% of the LOQ standard, then the LOQ must be raised to

the first standard level in the curve that meets criteria. Method blanks may be eliminated if technical

justification can be provided (e.g. the procedural blank was analyzed after an unexpectedly high level sample).

If any procedural blanks are removed from the LOQ determination, document in the raw data and report as

appropriate. Laboratory Sample Replicates / Field Duplicate Sample

Typically, samples are collected in duplicates in the field. The relative percent difference (RPD) of duplicate

samples should be ≤20% for the precision of sample preparation and analysis to be considered in control.

Replicate samples not meeting the ≤20% RPD criteria are flagged and reported as outside of QC acceptance

criteria.



## 9.3 Laboratory Matrix S pikes (LMSs)

LMSs may be performed in lieu of FMSs if FMSs have previously been performed for the sample matrix.

Additionally, LMSs may be performed in lieu of FMSs for a sample matrix if the FMS levels were not

appropriate for determining spike recoveries relative to endogenous levels of target analytes and appropriate

SRSs. Generally, each sample location represents a different sample and sample matrix. LMSs are prepared

for each sample and analyzed to determine the matrix effect on spike recovery efficiency of each target analyte

and appropriate SRSs. LMSs should be prepared at a minimum of one level and in duplicate. LMS

concentrations should be prepared at approximately 0.5

- 10 times the endogenous concentration or

approximately 4

- 10 times the LOQ concentration of each target analyte.

### ETS -8- 044. 1 Page 14 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS; Direct Injection

Analysis Lab matrix spike recoveries should fall within ± 30% of expected values. Sample data with LMS recovery

outside of ±30% but within ±50% of the expected value are flagged and reported as outside of QC acceptance

criter ia. Data with LMS recovery outside of ±50% of the expected value are reported as NR, where NR is

defined as “Not Reportable” data outside of QC acceptance criteria.



## 9.4 Lab Control Sample

Lab control spikes are prepared for each analysis batch to determine method accuracy and precision. LCSs

should be prepared at three levels in triplicate for each target analyte and at a minimum of two levels in

triplicate for appropriate SRSs. Low lab control spikes should be prepared at a concentration in the range of

a pproximately four to ten times higher than the targeted lower LOQ, the mid lab control spikes should be

prepared at a concentration near the mid

- point of the calibration curve and the high lab control spikes at

approximately 80% of the upper LOQ. For each target analyte and SRSs, the percent relative standard

deviation (method precision) for each control spike level must be less than or equal to 20% and the average

recovery (method accuracy) for each control spike level must be 80

- 120%. Sample data for target analytes

outside of the laboratory control spike acceptance criteria will be handled as follows:

If the average recovery of a spiking level falls outside method acceptance, but at least 67% (6 out of 9 ) o f LCS

samples are within 20 % of their respective nominal value (33% of the QC samples, not all replicates at the

same concentration, may be outside 20% of nominal value), the average recovery will be flagged as outside

method acceptance criteria. All LCS samples will be control charted as per ETS -4- 02 6. If the average

recovery of one of the spiking levels exceeded the analytical method uncertainty as determined by ETS

- 12

- 012, that analytical batch uncertainty will be expanded for that particular study.

If more than 67% of the LCS samples fail to meet method acceptance criteria, the data will not be reported.

Calibration standards consisting of mixed branched and linear isomer PFOS/PFOA are preferred. However,

for PFOS/PFOA target analytes, if the calibration standards are comprised of predominantly linear isomers

only, at least one level of triplicate LCSs should be prepared using PFOS/PFOA which contains a mix of linear

and branched isomers. These LCSs will be used to demonstrate quantitative equivalency (or quantitative bias)

of the isomeric mix when using a predominantly linear standard for calibration. The mixed linear and branched

isomer PFOS/PFOA LCSs recoveries should fall within ± 30% of expected values. Alternatively, in lieu of

mixed branched and linear isomer PFOS/PFOA LCSs, mixed branched and linear isomer PFOS/PFOA

TBMSs may be applied to demonstrate method accuracy and precision.



## 9.5 Field Matrix Spikes (FMSs) / Surrogate Recovery Standards (SRSs)

FMSs are prepared for each sampling location and analyzed to determine the matrix effect and sample holding

time on the spike recovery of each target analyte and/or appropriate SRS. Generally, each sample location

represents a different sample and sample matrix.

FMSs are QC samples to which known quantities of appropriate target analytes are added to the sample bottle

in the laboratory before the bottles are sent to the field. Typically a low and a high target analyte FMS are

prepared for each sampling location. The sample and field duplicate sample may contain appropriate SRSs in

lieu of target analyte low field matrix spike and target analyte high field matrix spike samples.

Field matrix spike method acceptance criteria are recoveries within ±30% of the expected value. If FMS

recovery (target analyte or SRS spike) is outside of ±30% of the expected value or could not be assessed

because the FMS (target analyte) was spiked at an inappropriate level, the sample result is reported as follows:

1.) If target analyte FMS recovery could not be assessed because the FMS's were at an inappropriate level,

then Laboratory Matrix Spikes (LMS) may be substituted. If LMS recoveries are within ±30% the data are

reportable and flagged to indicate that the FMS spikes levels were inappropriate.

2.) If multiple target analyte FMS's were prepared on a sample and the closest FMS level to the reported

sample meets the ±30% acceptance criteria but additional FMS's are outside the ±30% acceptance range, the

data are reportable and flagged to indicate that while there were failing FMS’s, the uncertainty will not be

expanded since the most appropriate spike level passed.

### ETS -8- 044. 1 Page 15 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS; Direct Injection

Analysis 3.) If the target analyte FMS recoveries are outside of the ±30% acceptance range but at least 30 acceptable

historical reportable FMS sample results are available, the data may be reported but flagged with an expanded

uncertainty and as not meeting FMS criteria.

4.) Sample data with FMS recovery outside of ±30% but within ±50% of the expected value are flagged and

reported as outside of QC acceptance criteria with an expanded uncertainty.

5.) If FMS recovery is outside of ±50%, the sample result is reported as NR, where NR is defined as “Not

Reportable” due to noncompliant QC results.

The targeted fortification levels should be at least 50% of the endogenous level and less than 10 times the

e ndogenous level to be used without justification to determine the statement of accuracy for analytical results.

Note: It is possible for bottles utilized for Field Matrix Spike samples to be under

- filled or over

- filled during

sample collection. Since this scenario will effect the actual concentration of the FMS sample (surrogate and

internal standard concentrations will also be effected, if used), it is important that any obvious under

- filling or

over

- filling of sample bottles be documented in the data package and taken into account in the FMS , ISs, or

SRSs recovery calculations. Samples over

- filled or under

- filled by more than 10% will be require recalculation

of the FMS, ISs, and SRS true values.

The average of the sample and the field duplicate should be used to calculate the recovery.



## 10 Procedures

10.1 Water Sample Preparation

This method is applicable to water samples. Samples containing heavy particulate may not be suitable for

analysis by this method. Samples containing suspended particulate should be centrifuge prior to removing a

sample aliquot, or filtered.

• Thoroughly mix sample before removing a n aliquot and placing in a labeled autovial.

• Dilute sample, if necessary, with ASTM Type I water , HPLC water, other suitable water , or solvent

(methanol)

.

• L ab control spikes are prepared for each analysis batch to determine method accuracy and precision.

LCSs should be prepared at three levels in triplicate for each target analyte and at a minimum of two

levels in triplicate for appropriate SRSs. Low lab control spikes should be prepared at a concentration

in the range of approximately four to ten times higher than the targeted lower LOQ, the mid lab control

spikes should be prepared at a concentration near the mid

- point of the calibration curve and the high

lab control spikes at approximately 80% of the upper LOQ. For IS quantitation, stable isotope internal

standards of each target analyte or appropriate surrogate ISs should be spiked at the same level as the

samples being analyzed, in all LCSs

.

• If LCSs are being prepared us ing synthetic groundwater, allow the LCSs samples to equilibrate for a

minimum of 4 hours before aliquoting for analysis or diluting with solvent (methanol).



## 11 Sample Analysis

- LC/MS/MS

11.1 Instrument Setup

Note: In this example, an Applied Biosystems Sciex API 4000 (API 5000 or API 5500) Tandem Mass

Spectrometer (LC/MS/MS) is used. Other brands/models of LC/MS/MS instruments as well as single

quadrupole mass spectrometers (LC/MS) may be used as long as the method acceptance criteria are met.

Brand names, suppliers, part numbers, and models are for illustrative purposes only. Equivalent performance

may be achieved using apparatus and materials other than those specified here, but demonstration of

equivalent performance that meets the requirements of this method is the responsibility of the laboratory. The

operator must optimize and document the equipment and settings used.

### ETS -8- 044. 1 Page 16 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS; Direct Injection

Analysis Establish the LC/MS/MS system and operating conditions equivalent to the following:

Mass Spec: Applied Biosystems API 4000, API 5000, or API 5500

Ion Source: Turbo Ion Spray (ABS)

Mode: Electrospray Negative

Scan Type: MRM (Multiple Reaction Monitoring)

Computer: Dell DHM

Software: Windows 2000 or Windows XP, Analyst 1.4.2 or higher versi ons

HPLC: Agilent Series 1100,1200, or 1290

Agilent Quaternary Pump

Agilent Vacuum Degasser

Agilent Autosampler

Agilent Column Oven

Note: One or more C18 HPLC analytical columns (2.1 mm x 100 mm, 5 μ m or 2.1 mm x 50 mm, 5 μ m) may be

attached on

- line after the purge valve and before the sample injection port to retard and separate any residue

contaminants that may be in the mobile phase and/or HPLC system.

HPLC Column: Betasil C18, 4.6mm x 100mm, 5 μ m (ThermoElectron Corporation)

Column Temperature: 35 ° C

Inje ction Volume: 5μL

Mobile Phase (A): 2mM Ammonium Acetate in ASTM Type I water (See 7.3)

Mobile Phase (B): Methanol

Table 3

. Liquid Chromatography Gradient Program

.

**Step**

**Number Total Time** 

**(min) Flow Rate** 

**(** μ

**L/min) Percent A** 

**(2 mM ammonium** 

**acetate) Percent B** 

**(Methanol)**

0 0 750 97.0 3.0

1 2.0 750 97.0 3.0

2 14.5 750 5.0 95.0

3 15.5 750 5.0 95.0

4 16.5 750 97.0 3.0

5 20.0 750 97.0 3.0



### Note: Other HPLC gradients may be used as long as the method criteria and project data quality objectives are

met.

It may be necessary to adjust the HPLC gradient in order to optimize instrument performance. Columns with

different dimensions (e.g. 2.1mm x 30mm) and columns from different manufacturers (Keystone Betasil C18

etc.) may be used.

### ETS -8- 044. 1 Page 17 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS; Direct Injection

Analysis Table 4 Suggested MRM Transitions for Target Analytes, Surrogates, and Internal Standards

Analyte Analyte Description Mass Transition

Q1 (amu) Mass Transition

Q3 (amu)

PFBA (C4 Acid) Target 213 169

PFPeA (C5 Acid) Target 263 219

PFHxA (C6 Acid) Target 313 269, 119

PFHpA (C7 Acid) Target 363 319, 169

PFOA (C8 Acid) Target 413 369, 219, 169

PFNA (C9 Acid) Target 463 419, 169, 219

PFDA (C10 Acid) Target 513 469, 269, 219

PFUnA (C11 Acid) Target 563 519, 269, 219

PFDoA (C12 Acid) Target 613 569, 169, 319

PFTA (C13 Acid) Target 663 619, 369, 319

FBSA (C4 Sulfonamide) Target 298 78

FOSA (C8 Sulfonamide) Target 498 78

PFBS (C4 Sulfonate) Target 299 99, 80

PFHS (C6 Sulfonate) Target 399 99, 80

PFOS (C8 Sulfonate) Target 499 80, 99, 130

[1,2,3,4

- 13 C 4 ]PFBA IS for PFBA 217 172

[1,2 ,3,4,5

- 13 C 5 ]PFPe A IS for PFPeA 268 223

[1,2

- 13 C 2 ]PFHxA IS for PFHxA 315 270

[1,2,3,4

- 13 C 4 ]PFHpA IS for PFHpA 367 322

[1,2,3,4,5,6,7,8

- 13 C 8 ]PFOA IS for PFOA 421 376

[1,2,3,4,5,6,7,8,9

- 13 C 9 ]PFNA IS for PFNA 472 427

[1,2,3,4,5,6

- 13 C 6 ]PFDA IS for PFDA 519 474

[1,2,3,4,5,6,7

- 13 C 7 ]PFUnA IS for PFUnA 570 525

[1,2

- 13 C 2 ]PFDoA IS for PFDoA and PFTA 615 570

[ 18 O 2 ]PFBS IS for PFBS 303 84

[ 1,2,3

- 13 C 3 ]PFHS IS for PFHS 402 80

[1,2,3,4

- 13 C 4 ]PFOS IS for PFOS 503 80

[1,2,3,4,5,6,7,8

- 13 C 8 ]FOSA IS for FOSA 507 80

[1,2,3,4

- 13 C 4 ]PFOA Surrogate (C4

- C8 Acids) 417 372

[1,2,3,4

- 13 C 4 ]PFOS Surrogate(Sulfonates, FOSA) 503 80

[1,2

- 13 C 2 ]PFUnA Surrogate (C9

- C13 Acids) 565 520



### Multiple transitions for monitoring the analytes is an option. The use of one daughter ion is acceptable if data

sensitivity and selectivity is achieved and provided that retention time criteria are met to assure adequate

specificity. While the daughter ions may be chosen at the discretion of the analyst, mass transition 99 is

suggested for PFOS. Quantitation may be performed using the total ion chromatogram (TIC, or summed

MRMs) for a given analyte. For example, the PFOA TIC would sum all three of the monitored transitions. Use

of the suggested primary ion is recommended. Retention times may vary slightly, on a day

- to

- day basis,

depending on the batch of mobile phase and the gradient, column, guard column(s) used etc. Drift in retention

times is acce ptable within an analytical run, as long as the drift continues through the entire analysis and the

standards are interspersed throughout the analytical run.



## 11.2 Calibration Curve

Quantitation is by internal standard or external standard calibration. Calibration standards may be prepared in

ASTM Type I, HPLC water, other suitable water, or a solvent/water mixture. If internal standard calibration

does not meet calibration acceptance criteria, external calibration can be applied. See Table 1 for

### ETS -8- 044. 1 Page 18 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS; Direct Injection

Analysis recom mended application of available internal standards. Quantitation of PFOA and PFOS is by summed

analyte

- specific mass transitions.

Analyze the standard curve prior to each set of samples. If internal standards were added to the calibration

standards area ratios are used to generate the calibration curve. The standard curve may be plotted using a

linear regression (y = mx + b), weighted 1/x or unweighted, or by quadratic fi t (y = ax 2 

### + bx + c), weighted 1/x or

unweighted

, using suitable software

. The mathematical method used to calculate the calibration curve should

be applied consistently throughout a study. Any change should be thoroughly documented in the raw data.

High and/or low points may be excluded from the calibration curves to provide a better fit over the range

appropriate to the data or because they did not meet the pre

- determined acceptance criteria. Low

- level curve

points should also be excluded if their area counts (or area ratio if quantitating by IS) are not at least twice that

of the ave rage area counts (or area ratio if quantitating by IS) of method and/or solvent blanks. The coefficient

of determination (r 2 

### ) value for the calibration curve must be greater than or equal to 0.990 (or a correlation

coefficient (r) of 0.995). Each point in the curve must be within ± 25% of the theoretical concentration with the

exception of the LLOQ, which may be within ± 30%. Justification for exclusion of calibration curve points will be

noted in the raw data. A minimum of 6 points will be used to construct the calibration curve.

If the calibration curve does not meet acceptance criteria, perform routine maintenance or prepare a new

standard curve (if necessary) and reanalyze.



## 11.3 Continuing Calibration Verification (CCV)

Continuing calibration verifications (CCV) are analyzed to verify the accuracy of the calibration curve. Analyze

a mid

- range calibration standard, one of the same standards used to construct the calibration curve, at a

minimum after every tenth sample, not including solvent blanks, with a minimum of one per sample set.

Calibration verification injections must be within ± 25% to be considered acceptable. The calibration curve and

the last passing CCV will then bracket acceptable samples. Multiple CCV levels may be used. Samples must

be bracketed by passing CCVs or the calibration curve and a passing CCV to be reportable.



## 11.4 System Suitability

A minimum of three system suitability samples should be injected at the beginning of each analytical run, prior

to th e analysis of the calibration curve. Typically these samples are at a concentration near the mid

- level of the

calibration curve and are repeated injections from one autosampler vial. It is suggested that the system

suitability injections have area counts or area ratios when using internal standard calibration, with a target RSD

of ≤5% and a target retention time RSD of ≤2%. There is no defined acceptability limit on these results as the

%RSD value is dependent on the number of MRM transitions being monitored in the LC/MS/MS run or time

period. Ultimately, any effects on these parameters for the System Suitability samples will also be evident on

all standards and QC samples analyzed as part of the analysis batch. Any effect of system suitability is

incorp orated within QC acceptance criteria. 4



## 11.5 Sample Analysis and QCs

For each analysis batch, the instrument analysis run sequence should include an initial calibration curve,

samples, FDSs, interspersed blanks, interspersed CCVs, appropriate QCs (i.e., LCSs, LMSs, FMSs, TBMSs,

and TBs), and a final CCV or calibration curve bracketing samples and appropriate QCs

Inject the same volume (between 5

- 100μL) of each standard, analytical sample and blank into the instrument

(unless an on

- instrument sample dilution is desired).

Samples containing analytes that are quantitated above the concentration of the highest standard in the curve

should be further diluted and reanalyzed.

4 3M Environmental Laboratory study E08

- 0096 evaluated the effect on these results as a function of the number of MRMs being monitored.

### ETS -8- 044. 1 Page 19 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS; Direct Injection

Analysis 

## 12 Data Analysis and Calculations

The chromatography analysis software will typically calculate the amount of target analyte in the sample

extracts using the established calibration curve.

Calculate the percent recovery of the LCS using the following equation:



### 100%*

)

mL ng (ion Concentrat Spike )

mL ng (ion Concentrat LCS

recovery LCS%

=

Calculate the percent recovery of the LMS using the following equation:

100%*

)

mL ng ( ion Concentrat Spike )

mL ng ( Sample of ion Concentrat -)

mL ng ( ion Concentrat LMS

recovery % LMS

=



### For samples fortified with known amounts of analyte prior to extraction, use the following equation to calculate

the percent recovery.



### 100 (ng/mL) added Analyte (ng/mL) sample in found analyte Average- (ng/mL) found analyte Total Recovery × =



## 13 Analysis Batch Method Performance Criteria

Any method performance parameters that are not achieved must be considered in the evaluation of the data.

Nonconformance to any specified parameters must be described and discussed in the final report if the

Technical Manager (non

- GLP study) or Study Director (GLP study) chooses to report the data.

If criteria listed in this method performance section are not met, maintenance may be performed on the system

and samples reanalyzed, or other actions taken as appropriate. Document all actions in the raw data.

If data are to be reported when performance criteria have not been met, the data must be footnoted on tables

and discussed in the text of the report.



## 13.1 System Suitability

– Analysis Batch

A minimum of three system suitability samples should be injected at the beginning of each analytical run.

These samples are run prior to the calibration curve. It is suggested that the system suitability injections have

area counts with a target RSD of ≤5% and a target retention time RSD of ≤2%. There is no defined

acceptability limit on these results as the %RSDs are dependent on the number of MRM transitions being

monitored in the LC/MS/MS run or time period. Any effect of system suitability is incorporated in the QC

acceptance criteria.



## 13.2 Calibration and Limit of Quantitation (LOQ)

– Analysis Batch

Calibration Curve: The coefficient of determination (r 2 

### ) value for the calibration curve must be greater than or

equal to 0 .990 corresponding to a correlation coefficient (r) = 0.995. Each point in the curve must be within

± 25% of the theoretical concentration with the exception of the LLOQ, which may be within ± 30%.

CCV Performance: The calibration standards that are interspersed throughout the analytical sequence are

evaluated as continuing calibration verifications in addition to being part of the calibration curve. The accuracy

### ETS -8- 044. 1 Page 20 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS; Direct Injection

Analysis of each curve point must be within 25% of the theoretical value ( within 30% for lowest curve point ) . Samples

that are bracketed by CCVs not meeting these criteria must be reanalyzed.

Limits of Quantitation (LOQ): The lower LOQ (LLOQ) is the lowest non

- zero active standard in the

calibration curve; the peak area of the LLOQ must be at least 2X that of the average area counts for all

prepared procedural blank(s). By definition, the measured value of the LLOQ must be within 30% of the

theoretical value.

Demonstration of Specificity: Specificity is demonstrated by chromatographic retention time (within 4% of

standard) and the mass spectral response of unique ions.



## 13.3 Blanks

– Method/Procedural Blanks and Trip

Method/ Procedural Blanks: Multiple procedural blanks should be interspersed throughout the analysis batch

and the analytical sequence. At a minimum, method blanks are analyzed prior to instrument calibration, prior

to the analysis of CCV samples, after every 10 sample injections, and at the end of the analytical run.

The mean area counts (or area ratios when using IS calibration) for each analyte must be less than 50% of the

area count of the LOQ standard. If the area counts of the procedural blanks exceed 50% of the LOQ standard,

then the LOQ must be raised to the first standard level that meets criteria.

Trip Blank: A t rip blank of ASTM Type I water (or lab equivalent) is prepared in a sample container in the

laboratory and treated as a sample, including exposure to shipping, sampling site conditions, storage,

preservation and all analytical procedures. The trip blanks results for each analyte are included with the

repo rted sample results.



## 13.4 Data Accuracy and Precision

– Analysis Batch

Lab Control Spikes: The average recovery at each LCS level for each target analyte and appropriate SRS

should be within 80

- 120% and the percent relative standard deviation of the recoveries must be less than or

equal to 20%. If the average recovery of a spiking level falls outside method acceptance, but at least 67% (6

out of 9 ) o f LCS samples are within 20 % of their respective nominal value (33% of the QC samples, not all

replicates at the same concentration, may be outside 20% of nominal value), the average recovery will be

flagged as outside method acceptance criteria. All LCS samples will be control charted as per ETS

- 12

- 012. If

the average recovery of one of the spiking levels exceeded the analytical method uncertainty as determined by

ETS

- 12

- 012, that analytical batch uncertainty will be expanded for that particular study. The average recovery

at each LCS level for mixed branched/linear isomer PFOA and PFOS should be within 70

- 130% and the

percent relative standard deviation of the recoveries must be less than or equal to 20%.

Field Duplicates: The relative percent difference (RPD) of duplicate samples should be less than 20% for the

precision of sample preparation and analysis to be considered in control. Replicate samples not meeting the

20% RPD criteria are flagged and reported as outside of QC acceptance criteria.

Field Matrix Spikes: FMS acceptance criteria are recoveries within ±30% of the expected value for each

target analyte and appropriate SRS. Sample data with FMS recovery outside of ±30% but within ±50% of the

expected value are flagged and reported as outside of QC acceptance criteria. Data with FMS recovery

outside of ±50% of the expected value are reported as NR, where NR is defined as “Not Reportable” data

outside of QC acceptance criteria. If FMS recovery could not be assessed because FMSs were at an

inappropriate level, then Laboratory Matrix Spikes (LMSs) may be substituted. If LMS recoveries are within

±30% for each target analyte and SRSs the data are reportable but flagged as not meeting the FMS method

acceptance criteria.



## 13.5 Analytical Method Uncertainty

Analytical method uncertainty for each target analyte and SRS is determined with control charted historica l

analysis batch LCS data for the method and reported with each analysis batch. 5 

### Uncertainty determinations

5 Method uncertainty based on INTERNATIONAL ANS/ISO/IED STANDARD 17025 reference (GUM, Guide to the Expression of

Uncert ainty in Measurement). Method application demonstrated in ETS

- 12

- 012, citing references: a.) EURACHEM/CITAC Guide,

“Quantifying Uncertainty in Analytical Measurement,” Second Edition; Editors: S.L.R. Ellison, M. Rosslein, and A. Williams

. b.)Georgian, Thomas, “Estimation of Laboratory Analytical Uncertainty Using Laboratory Control Samples,” Environmental Testing &

### ETS -8- 044. 1 Page 21 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS; Direct Injection

Analysis are based on INTERNATIONAL ANS/ISO/IED STANDARD 17025 reference (GUM, Guide to the Expression

of Uncertainty in Measurement) and described in ETS -12- 012 . At least thirty data points are required for

determining analytical method uncertainty. The method uncertainty is defined as 2x the standard deviation of

the percent recoveries of the pooled lab control spikes. While all LCS data points are control charted, only the

most recent fifty data points are used for determining the method uncertainty.

When less than thirty LCS data points have been generated for a given analyte, the analysis batch LCSs are

used to determine the data uncertainty. If FMSs meet the ±30% recovery criteria at a level appropriate to the

endogenous level, and the LCS meet the ±20% recovery criteria, then the uncertainty of the data is determined

as within 100±20%.

Analysis batch sample data with FMS recovery outside of ±30% but within ±50% of the expected value are

flagged and reported as outside of QC acceptance criteria with expanded uncertainties. Data with FMS

recovery outside of ±50% of the expected value are reported as NR, where NR is defined as “Not Reportable”

data outside of QC acceptance criteria. If FMS recovery could not be assessed because FMSs were at an

inappropriate level, then Laboratory Matrix Spikes (LMSs) may be substituted. If LMS recoveries are within

±30% for each target analyte and appropriate SRSs the data are reportable but flagged as not meeting the

FMS method acceptance criteria with uncertainties of ±30%. If FMS do not meet the ±30% recovery criteria,

and historical FMS data does not exist, the analytical uncertainty is evaluated on a sample

- by -s ample basis,

the data may be reported with expanded uncertainty and are flagged.



## 13.6 Quantitation of PFOA/PFOS

– Analysis Batch

Calibration standards consisting of mixed branched and linear isomer PFOS/PFOA are preferred.

Quantitation is performed by integrating the linear and branched isomers together. Alternately, the linear and

branched isomers can be integrated separately, applying the appropriate true value to each calibration curve

po int for each isomer. The LCS and samples are then quantitated by integrating the linear and branched

isomers separately (requires separate analytical results files) and quantitating the resulting peak against the

linear or branched calibration curve

. The results from both integrations are then summed to produce the final

result. Integrating the linear and branched isomers separately may be helpful for those samples where the

linear/branched ratios do not closely match those of the reference standards.

Ho wever, for PFOS/PFOA target analytes, if the calibration standards are comprised of predominantly linear

isomers only t he method requires the addition of LCS s of mixed branched/linear isomer PFOS/PFOA. The

purpose of including these LCSs is to demonstrate quantitative equivalency (or quantitative bias) of the

isomeric mix when using a predominantly linear PFOS or PFOA standard for calibration. Alternatively, in lieu

of mixed branched and linear isomer PFOS/PFOA LCSs, mixed branched and linear isomer PFOS/ PFOA

TBMSs may be applied to demonstrate method accuracy and precision.

An alternate method of quantitation can be performed whereby only the linear isomer of PFOS/PFOA is

integrated and used for generating the calibration curve. The LCS and samples are then quantitated by

integrating the linear and branched isomers separately (requires separate analytical results files) and

quantitating the resulting peak against the linear calibration curve. The results from both integrations are then

summed to produce the final result. Integrating the linear and branched isomers separately reduces the on

column concentration for those samples that contain both linear and branched isomers of PFOA/PFOS. This

ensures that the concentration detected is within the a range of the calibration curve that is comparable

regardless of whether the calibration curve was generated using predominantly linear isomers of PFOS/PFOA

or linear plus branched isomers of PFOS/PFOA.



## 14 Pollution Prevention and Waste Management

Waste generated when performing this method will be disposed of appropriately. The original samples will be

archived at the 3M Environmental Laboratory in accordance with internal procedures.

Analysis, November/December 2000. c.)Taylor, B.N. and CE. Kuyatt, NIST Technical Note 1297, 1994 Edition: “Guidelines for

Evaluating and Expressing the Uncertainty of NIST Measurement Results.”d.)Adams, T.M., “A2LA Guide for the Estimation of

Measurement Uncertainty in Testing”, July 2002.

### ETS -8- 044. 1 Page 22 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS; Direct Injection

Analysis 

## 15 Records

Each data package generated for a study must include all supporting information for reconstruction of the data.

Information for the data package must include, but is not limited to the following items: study or project

number, sample and standard prep sheets/records, instrument run log (instrument batch records, instrument

ac quisition method, summary pages), instrument results files, chromatograms, calibration curves, and data

calculations.



## 16 Affected Documents

None.



## 17 Revisions

Revision

Number Summary of Changes

1 Section 1

. Included the use of internal standard calibration by this method.

Section 2. Included the use of internal standard calibration by this method. Included the

use of a solvent/water mixture when analyzing for PFUnA, PFDoA, PFTrDA, and FOSA.

Section 3

. Added definitions for internal standard, surrogate internal standard, and

surrogate recovery standard.

Section 6.Removed the details regarding the instrument parameters to section 10 of the

method

. Section 7. Updated reference standards to include internal standards and surrogates.

Changed concentration levels for working standards and included the use of internal

standards and surrogates.

Section 8. Inserted a new section on sample bottle preparation.

Se ction 9 Quality Control. This section was previously section 10 in ETS -8- 044.0. Updated

QC criteria to be consistent with method ETS -8- 154.4.

Section 10 Procedures . This section was previously section 8 (Sample Handling) in E TS -8-

044.0. Added detail regarding the preparation of LCSs. Included the use of methanol as a

dilution solvent.

Section 11 Samp le Analysis

. This section was previously section 10 in ETS -8- 044.0.

Included the details regarding the instrument parameters.

Section 12 Data Analysis and Calculations. This section was previously section 11 in ETS

8- 044.0. Removed the equation for calculating the analytes concentration, indicating that

this is done by the instrument software.

Section 13 Method Performance. This section was previously section 12 in ETS -8- 044.0.

Updated QC criteria to be consistent with ETS -8- 154.4. Added information on the

determination of analytical method uncertainty and quantitation of PFOA/PFOS.

Section 14 Pollution Prevention. This section was previously section 13 in ETS -8- 044.0.

Section 15 Records. This section was previously section 14 in ETS -8- 044.0.

Section 16 Affected Documents. This section was previously section 15 in ETS -8- 044.0.

Section 17 Revisions. This section was previously section 16 in ETS

- 8

- 044.0.

# A TTACHMENT 

# D: D EVIATION 

# ( S 

# )

# R ECORD OF 

# D EVIATION 

# /N ONCONFORMANCE

ETS -4- 008.7

Documentation of Deviations and Control of Nonconforming Testing Page 1 of 1 

## I. Identification

Study / Project No.

GLP10

- 01

- 0 2

- 2 8 Date(s) of Occurrence:

4/11/12 Document Number:

ETS

- 8

- 044. 1

Deviation type  SOP  Equipment Procedure  Method

(Check one)  Protocol  GPO  Other:



## II. Description (attach extra pages as needed)



## Method Requirements:

1. FMS re covery within ±30% (section 13.4 ).

2. RPD values ≤20% (section 13.4).



## Actual procedure/process:

1. The sampling location DAL GW 13 3L had a PFOS FMS recovery of 65.5 %. Trip Blank MHS FMS

recovery was 67.1%.

2. The sampling location DAL GW 136S had a sample/sample duplicate RPD of 24 % for PFOS and DAL

GW 1 42R had a RPD of 28% for PFOS

.



## III. Actions Taken

(such as amendment issued, SOP revision, etc.)



## Corrective Action (  Yes  No) Reference :

Acceptability of the nonconforming work:

1. The sampling location and trip blank with a FMS recovery outside method acceptance criteria will be

flagged in the report and the analytical uncertainty for sampling location DAL GW 133L will be expanded

to ± 35 % for PFOS.

2. The non

- compliant RPDs will be flagged in the final report.



### Actions:  Halting of Work  Client Notification  Work Recall  Withholding of Report

 Other: Deviations will be noted in final report.



### Project Lead /PAI Approval: Date:

Study Director (if GLP): Date:

Sponsor Approval (for GLP protocol deviations): NA Date: NA

T echnical Reviewer (optional): NA Dat e: NA

Laboratory Department Manager Approval: Date:



## IV. Authorization to Resume Work

Where halting of work occurred, resumption of work must first be approved by Laboratory Management



### Laboratory Department Manager Approval: NA Date: NA



## Deviation No. ______________

(assigned by Study Director or Team Leader at the end of study or project)