Order Online

Over 100,000 Products Available!

omega.com omega.com®

D-32 To download information and to order Industrial

pH Instrumentation and Electrode products online,

visit omega.com

PIPE & COUPLING

(NOT INCLUDED)

1/2" NPT

THREADED

CABLE ASSEMBLY

ELECTRODE CPVC 

AND

PVDF

MODELS

CPVC AND

PVDF MODELS

TEMPERATURE

COMPENSATED

Submersible Flat Surface pH/ORP Electrodes

Submersion Electrodes

The OMEGA® PHE/ORE-6510 submersible

electrodes are designed for use in drums,

open tanks and streams, etc. The flat

sensing surface is surrounded by a porous

polyethylene reference junction which

minimizes fouling. Electrode installation and

removal done by a simple 1⁄4 turn quick

disconnect. The cable assemblyʼs cap has 1

⁄2 MNPT thread, which can be connected to

a coupling and support pipe. The pipe

allows easy mechanical mounting and

protects the cable from the liquid measured.

The resulting assembly is lightweight for

convenient handling when maintenance is

needed. ATC is available as part of the

mounting assembly. 

The flat surface design is well suited for

applications with relatively high suspended

solids or where flocculation operations

require low velocities.

Where only low velocities are present and

coating is a problem, the electrode can be

mounted at a 45° angle facing the flow so

as to obtain increased scrubbing action

across the measuring surface. 

Maximum Temperature

Standard Models: 65°C (150°F)

“-ACRYL” Units: 87°C (190°F) CPVC body

100°C (212°F) PVDF body

Available Options:

“-ACRYL” Acrylamide gel, $30 additional

“-HF” Fluoride resistant, $50 additional

“-LC” Low conductivity, $30 additional

“-GL” Ground loop, $158 additional

See page D-34 for option compatibility.

Submersion/Electrodes

Note: Temperature compensation is not required with ORP electrodes. 

Order electrode and mounting assemblies separately. Materials of construction must be the same. 

Mounting Assemblies come with 3 m (10') cable with BNC connector. Cable lengths available up to

15 m (50') Add $2.00 per extra foot. Consult Engineering for ordering information.

Ordering Examples: PHE-6510-HF, CPVC electrode with HF resistant glass and

PHEH-65K-10-TH700 mounting assembly with 3 m (10') cable and 700 Series thermistor ATC, 

$134 + 50 + 206 = $390.

PHE-6510 CPVC pH electrode, and PHEH-65-20 mounting assembly with 6 m (20') cable,

$134 + 55 + [2 x (20-10)] = $209.

CPVC or PVDF Construction

Designed For Mounting in Tanks,

Flumes, etc

Easy Mounting in Tanks

MOST POPULAR MODELS HIGHLIGHTED!

PHE-6510 

$

134

Submersion Electrodes for use in

Open Tanks

PHE-6510, $134 and 

PHEH-65-10, $55,

shown larger than

actual size.

NOTE: At the time of initial purchase, please order

both electrode and mounting assembly. Mounting

assembly includes insertion assembly and cable. If

ordered with automatic temperature compensation

(ATC), the temperature compensator is built into the

insertion assembly. When electrode has expired, simply

re-order electrode.

Order

Code Description

PT100 100 Ω Pt RTD

PT1K 1000 Ω Pt RTD

TH700 Series 700 Thermistor

R3K 3000 Ω Balco

* Specify ATC Sensor

To Order (Specify Model Number)

Model No. Price Description

PHE-6510 $134 Submersion pH electrode, CPVC

PHE-6511 254 Submersion pH electrode, PVDF

ORE-6510 159 Submersion ORP electrode, CPVC

ORE-6511 270 Submersion ORP electrode, PVDF

Mounting Assemblies — Required for Installation

PHEH-65-10 $55 CPVC without automatic temperature compensation

PHEH-65-10-(*) 141 CPVC with ATC

PHEH-65K-10 78 PVDF without ATC

PHEH-65K-10-(*) 206 PVDF with ATC
One Omega Drive | Stamford, CT 06907 | 1-888-TC-OMEGA (1-888-826-6342) | info@ome

www.omega.com

More than 100,000 Products Availab

CANADA

www.omega.ca

Laval(Quebec)

1-800-TC-OMEGA

UNITED KINGDOM

www. omega.co.uk

Manchester, England

0800-488-488

GERMANY

www.omega.de

Deckenpfronn, Germany

0800-8266342

FRANCE

www.omega.fr

Guyancourt, France

088-466-342

BENELUX

www.omega.nl

Amstelveen, NL

0800-099-33-44

UNITED STATES

www.omega.com

1-800-TC-OMEGA

Stamford, CT.

CZECH REPUBLIC

www.omegaeng.cz

Karviná, Czech Republic

596-311-899

Temperature

Calibrators,Connectors,General Test and Measurement

Instruments,Glass Bulb Thermometers,Handheld Instruments

for Temperature Measurement, Ice Point References,

Indicating Labels,Crayons,Cements and Lacquers, Infrared

Temperature Measurement Instruments, Recorders Relative

Humidity Measurement Instruments, RTD Probes, Elements

and Assemblies, Temperature & Process Meters,Timers and

Counters, Temperature and Process Controllers and Power

Switching Devices,Thermistor Elements, Probes and

Assemblies,Thermocouples Thermowells and Head and Well

Assemblies,Transmitters,Wire

Pressure, Strain and Force

Displacement Transducers, Dynamic Measureme

Force Sensors, Instrumentation for Pressure and S

Measurements, Load Cells, Pressure Gauges, Pres

Reference Section, Pressure Switches, Pressure Tra

Proximity Transducers, Regulators,

Strain Gages, Torque Transducers, Valves

pH and Conductivity

Heaters

Band Heaters, Cartridge Heaters, Circulation He

Comfort Heaters, Controllers, Meters and Switch

lbll d 

Flow and Level

Air Velocity Indicators, Doppler Flowmeters, Level

Measurement, Magnetic Flowmeters, Mass Flowmeters,

Pitot Tubes, Pumps, Rotameters,Turbine and Paddle Wheel

Flowmeters, Ultrasonic Flowmeters, Valves,Variable Area

Flowmeters, Vortex Shedding Flowmeters

Data Acquisition

Auto-Dialers and Alarm Monitoring Systems,

Communication Products and Converters, Data

Acquisition and Analysis Software, Data Logge

Plug-in Cards, Signal Conditioners, USB, RS232

and Parallel Port Data Acquisition Systems, Wir

Transmitters and Receivers