13339 Hagers Ferry Road

Huntersville, NC 28078-7929

McGuire Nuclear Complex - MG03A2

Phone: 980-875-5245 Fax: 980-875-4349

Order Summary Report

Analytical Laboratory

Order Number: J12110302

Project Name: Flex Fuel WW

Lab Contact: Jason C Perkins

Date: 12/7/2012

Customer Address: 3195 Pine Hall Rd

Customer Name(s): Bill Kennedy, Melonie Martin, Wayne Chapman, Tom Johnson

Mailcode: Belews Steam Station

Belews Creek, NC 28012

Phone: 980-875-5348

Report Authorized By:

(Signature)

Program Comments:

Please contact the Program Manager (Jason C Perkins) with any questions regarding this report.

Data Flags & Calculations:

Any analytical tests or individual analytes within a test flagged with a Qualifier indicate a deviation from the method quality 

system or quality control requirement. The qualifier description is found at the end of the Certificate of Analysis (sample results) 

under the qualifiers heading. All results are reported on a dry weight basis unless otherwise noted. Subcontracted data 

included on the Duke Certificate of Analysis is to be used as information only. Certified vendor results can be found in the 

subcontracted lab final report. Duke Energy Analytical Laboratory subcontracts analyses to other vendor laboratories that have 

been qualified by Duke Energy to perform these analyses except where noted.

Data Package:

This data package includes analytical results that are applicable only to the samples described in this narrative. An estimation of 

the uncertainty of measurement for the results in the report is available upon request. This report shall not be reproduced, except 

in full, without the written consent of the Analytical Laboratory. Please contact the Analytical laboratory with any questions. The 

order of individual sections within this report is as follows:

Job Summary Report, Sample Identification, Technical Validation of Data Package, Analytical Laboratory Certificate of Analysis, 

Analytical Laboratory QC Reports, Sub-contracted Laboratory Results, Customer Specific Data Sheets, Reports & 

Documentation, Customer Database Entries, Test Case Narratives, Chain of Custody (COC)

Certification:

The Analytical Laboratory holds the following State Certifications : North Carolina (DENR) Certificate #248, South Carolina 

(DHEC) Laboratory ID # 99005. Contact the Analytical Laboratory for definitive information about the certification status of 

specific methods. 

Page 1
Sample ID's & Descriptions:

Sample ID Plant/Station

Collection 

Date and Time Collected By Sample Description

2012024777 BELEWS 16-Nov-12 7:30 AM TRAVIS THORNTON FGD Purge Eff 

2012024778 BELEWS 16-Nov-12 7:35 AM TRAVIS THORNTON EQ TANK 

2012024779 BELEWS 16-Nov-12 7:40 AM TRAVIS THORNTON BIOREACTOR 1 INF 

2012024780 BELEWS 16-Nov-12 7:40 AM TRAVIS THORNTON biOREACTOR 1 INF HG BLK 

2012024781 BELEWS 16-Nov-12 7:45 AM TRAVIS THORNTON BIOREACTOR 2 INF. 

2012024782 BELEWS 16-Nov-12 7:45 AM TRAVIS THORNTON BIOREACTOR 2 INF. HG BLANK 

2012024783 BELEWS 16-Nov-12 7:50 AM TRAVIS THORNTON BIOREACTOR 2 EFF. 

2012024784 BELEWS 16-Nov-12 7:50 AM TRAVIS THORNTON BIOREACTOR 2 EFF. HG BLANK 

2012024785 BELEWS 16-Nov-12 7:55 AM TRAVIS THORNTON FILTER BLANK 

9 Total Samples

Page 2
COC and .pdf report are in agreement with sample totals 

and analyses (compliance programs and procedures).

All Results are less than the laboratory reporting limits.

All laboratory QA/QC requirements are acceptable.

Yes No

Technical Validation Review

Checklist:

Yes No

Yes No

Report Sections Included:

Job Summary Report Sub-contracted Laboratory Results

Sample Identification Customer Specific Data Sheets, Reports, & Documentation

Technical Validation of Data Package Customer Database Entries

Analytical Laboratory Certificate of Analysis

Analytical Laboratory QC Report

Chain of Custody

Reviewed By: DBA Account Date: 12/7/2012

Electronic Data Deliverable (EDD) Sent Separately

Page 3
Certificate of Laboratory Analysis

This report shall not be reproduced, except in full.

Order # J12110302

2012024777

Collection Date: 16-Nov-12 7:30 AM

Site: FGD Purge Eff 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

DF Analyst

INORGANIC IONS BY IC

Bromide 96 mg/L 5 50 EPA 300.0 11/27/2012 20:35 JAHERMA

Chloride 6300 mg/L 100 1000 EPA 300.0 11/27/2012 20:35 JAHERMA

Sulfate 1400 mg/L 100 1000 EPA 300.0 11/27/2012 20:35 JAHERMA

MERCURY (COLD VAPOR) IN WATER

Mercury (Hg) 190 ug/L 5 100 EPA 245.1 11/29/2012 14:28 AGIBBS

DISSOLVED METALS BY ICP

Manganese (Mn) 7.76 mg/L 0.05 10 EPA 200.7 11/29/2012 10:13 MHH7131

TOTAL RECOVERABLE METALS BY ICP

Boron (B) 179 mg/L 0.5 10 EPA 200.7 11/28/2012 13:43 MHH7131

Calcium (Ca) 4130 mg/L 0.1 10 EPA 200.7 11/28/2012 13:43 MHH7131

Iron (Fe) 147 mg/L 0.1 10 EPA 200.7 11/28/2012 13:43 MHH7131

Magnesium (Mg) 797 mg/L 0.05 10 EPA 200.7 11/28/2012 13:43 MHH7131

Manganese (Mn) 9.07 mg/L 0.05 10 EPA 200.7 11/28/2012 13:43 MHH7131

DISSOLVED METALS BY ICP-MS

Selenium (Se) 250 ug/L 10 10 EPA 200.8 11/29/2012 13:12 DJSULL1

TOTAL RECOVERABLE METALS BY ICP-MS

Arsenic (As) 291 ug/L 10 10 EPA 200.8 11/30/2012 15:04 KRICHAR

Cadmium (Cd) < 10 ug/L 10 10 EPA 200.8 11/30/2012 15:04 KRICHAR

Chromium (Cr) 279 ug/L 10 10 EPA 200.8 11/30/2012 15:04 KRICHAR

Copper (Cu) 159 ug/L 10 10 EPA 200.8 11/30/2012 15:04 KRICHAR

Nickel (Ni) 227 ug/L 10 10 EPA 200.8 11/30/2012 15:04 KRICHAR

Selenium (Se) 5640 ug/L 20 20 EPA 200.8 11/30/2012 15:04 KRICHAR

Silver (Ag) < 10 ug/L 10 10 EPA 200.8 11/30/2012 15:04 KRICHAR

Zinc (Zn) 274 ug/L 10 10 EPA 200.8 11/30/2012 15:04 KRICHAR

SELENIUM SPECIATION - (Analysis Performed by Applied Speciation and Consulting, LLC)

Vendor Parameter Complete Vendor Method V_AS&C

TOTAL DISSOLVED SOLIDS

TDS 16000 mg/L 200 1 SM2540C 11/26/2012 16:22 SWILLI3

TOTAL SUSPENDED SOLIDS

TSS 3800 mg/L 250 1 SM2540D 11/26/2012 10:13 SWILLI3

Page 4
Certificate of Laboratory Analysis

This report shall not be reproduced, except in full.

Order # J12110302

2012024778

Collection Date: 16-Nov-12 7:35 AM

Site: EQ TANK 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

DF Analyst

MERCURY (COLD VAPOR) IN WATER

Mercury (Hg) 147 ug/L 2.5 50 EPA 245.1 11/29/2012 14:30 AGIBBS

DISSOLVED METALS BY ICP

Manganese (Mn) 7.07 mg/L 0.05 10 EPA 200.7 11/29/2012 10:17 MHH7131

TOTAL RECOVERABLE METALS BY ICP

Boron (B) 187 mg/L 0.5 10 EPA 200.7 11/28/2012 13:47 MHH7131

Calcium (Ca) 4040 mg/L 0.1 10 EPA 200.7 11/28/2012 13:47 MHH7131

Iron (Fe) 114 mg/L 0.1 10 EPA 200.7 11/28/2012 13:47 MHH7131

Magnesium (Mg) 804 mg/L 0.05 10 EPA 200.7 11/28/2012 13:47 MHH7131

Manganese (Mn) 8.00 mg/L 0.05 10 EPA 200.7 11/28/2012 13:47 MHH7131

DISSOLVED METALS BY ICP-MS

Selenium (Se) 143 ug/L 10 10 EPA 200.8 11/29/2012 13:16 DJSULL1

TOTAL RECOVERABLE METALS BY ICP-MS

Arsenic (As) 215 ug/L 10 10 EPA 200.8 11/30/2012 14:25 KRICHAR

Cadmium (Cd) < 10 ug/L 10 10 EPA 200.8 11/30/2012 14:25 KRICHAR

Chromium (Cr) 222 ug/L 10 10 EPA 200.8 11/30/2012 14:25 KRICHAR

Copper (Cu) 119 ug/L 10 10 EPA 200.8 11/30/2012 14:25 KRICHAR

Nickel (Ni) 219 ug/L 10 10 EPA 200.8 11/30/2012 14:25 KRICHAR

Selenium (Se) 4840 ug/L 10 10 EPA 200.8 11/30/2012 14:25 KRICHAR

Silver (Ag) < 10 ug/L 10 10 EPA 200.8 11/30/2012 14:25 KRICHAR

Zinc (Zn) 243 ug/L 10 10 EPA 200.8 11/30/2012 14:25 KRICHAR

2012024779

Collection Date: 16-Nov-12 7:40 AM

Site: BIOREACTOR 1 INF 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

DF Analyst

MERCURY 1631 - (Analysis Performed by Brooks Rand Labs LLC)

Vendor Parameter Complete Vendor Method V_BRAND

DISSOLVED METALS BY ICP

Manganese (Mn) 1.44 mg/L 0.05 10 EPA 200.7 11/29/2012 10:21 MHH7131

TOTAL RECOVERABLE METALS BY ICP

Boron (B) 178 mg/L 0.5 10 EPA 200.7 11/28/2012 13:50 MHH7131

Calcium (Ca) 3380 mg/L 0.1 10 EPA 200.7 11/28/2012 13:50 MHH7131

Iron (Fe) < 0.1 mg/L 0.1 10 EPA 200.7 11/28/2012 13:50 MHH7131

Magnesium (Mg) 729 mg/L 0.05 10 EPA 200.7 11/28/2012 13:50 MHH7131

Manganese (Mn) 1.52 mg/L 0.05 10 EPA 200.7 11/28/2012 13:50 MHH7131

Page 5
Certificate of Laboratory Analysis

This report shall not be reproduced, except in full.

Order # J12110302

2012024779

Collection Date: 16-Nov-12 7:40 AM

Site: BIOREACTOR 1 INF 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

DF Analyst

DISSOLVED METALS BY ICP-MS

Selenium (Se) 127 ug/L 10 10 EPA 200.8 11/29/2012 13:19 DJSULL1

TOTAL RECOVERABLE METALS BY ICP-MS

Arsenic (As) < 10 ug/L 10 10 EPA 200.8 11/30/2012 14:28 KRICHAR

Cadmium (Cd) < 10 ug/L 10 10 EPA 200.8 11/30/2012 14:28 KRICHAR

Chromium (Cr) < 10 ug/L 10 10 EPA 200.8 11/30/2012 14:28 KRICHAR

Copper (Cu) 10.9 ug/L 10 10 EPA 200.8 11/30/2012 14:28 KRICHAR

Nickel (Ni) 28.2 ug/L 10 10 EPA 200.8 11/30/2012 14:28 KRICHAR

Selenium (Se) 95.8 ug/L 10 10 EPA 200.8 11/30/2012 14:28 KRICHAR

Silver (Ag) < 10 ug/L 10 10 EPA 200.8 11/30/2012 14:28 KRICHAR

Zinc (Zn) < 10 ug/L 10 10 EPA 200.8 11/30/2012 14:28 KRICHAR

SELENIUM SPECIATION - (Analysis Performed by Applied Speciation and Consulting, LLC)

Vendor Parameter Complete Vendor Method V_AS&C

2012024780

Collection Date: 16-Nov-12 7:40 AM

Site: biOREACTOR 1 INF HG BLK 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

DF Analyst

MERCURY 1631 - (Analysis Performed by Brooks Rand Labs LLC)

Vendor Parameter Complete Vendor Method V_BRAND

2012024781

Collection Date: 16-Nov-12 7:45 AM

Site: BIOREACTOR 2 INF. 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

DF Analyst

MERCURY 1631 - (Analysis Performed by Brooks Rand Labs LLC)

Vendor Parameter Complete Vendor Method V_BRAND

DISSOLVED METALS BY ICP

Manganese (Mn) 1.50 mg/L 0.05 10 EPA 200.7 11/29/2012 10:25 MHH7131

TOTAL RECOVERABLE METALS BY ICP

Boron (B) 178 mg/L 0.5 10 EPA 200.7 11/28/2012 13:54 MHH7131

Calcium (Ca) 3380 mg/L 0.1 10 EPA 200.7 11/28/2012 13:54 MHH7131

Iron (Fe) 0.119 mg/L 0.1 10 EPA 200.7 11/28/2012 13:54 MHH7131

Magnesium (Mg) 725 mg/L 0.05 10 EPA 200.7 11/28/2012 13:54 MHH7131

Manganese (Mn) 1.54 mg/L 0.05 10 EPA 200.7 11/28/2012 13:54 MHH7131

Page 6
Certificate of Laboratory Analysis

This report shall not be reproduced, except in full.

Order # J12110302

2012024781

Collection Date: 16-Nov-12 7:45 AM

Site: BIOREACTOR 2 INF. 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

DF Analyst

DISSOLVED METALS BY ICP-MS

Selenium (Se) 17.2 ug/L 10 10 EPA 200.8 11/29/2012 13:22 DJSULL1

TOTAL RECOVERABLE METALS BY ICP-MS

Arsenic (As) < 10 ug/L 10 10 EPA 200.8 11/30/2012 14:31 KRICHAR

Cadmium (Cd) < 10 ug/L 10 10 EPA 200.8 11/30/2012 14:31 KRICHAR

Chromium (Cr) < 10 ug/L 10 10 EPA 200.8 11/30/2012 14:31 KRICHAR

Copper (Cu) < 10 ug/L 10 10 EPA 200.8 11/30/2012 14:31 KRICHAR

Nickel (Ni) < 10 ug/L 10 10 EPA 200.8 11/30/2012 14:31 KRICHAR

Selenium (Se) 10.8 ug/L 10 10 EPA 200.8 11/30/2012 14:31 KRICHAR

Silver (Ag) < 10 ug/L 10 10 EPA 200.8 11/30/2012 14:31 KRICHAR

Zinc (Zn) < 10 ug/L 10 10 EPA 200.8 11/30/2012 14:31 KRICHAR

SELENIUM SPECIATION - (Analysis Performed by Applied Speciation and Consulting, LLC)

Vendor Parameter Complete Vendor Method V_AS&C

2012024782

Collection Date: 16-Nov-12 7:45 AM

Site: BIOREACTOR 2 INF. HG BLANK 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

DF Analyst

MERCURY 1631 - (Analysis Performed by Brooks Rand Labs LLC)

Vendor Parameter Complete Vendor Method V_BRAND

2012024783

Collection Date: 16-Nov-12 7:50 AM

Site: BIOREACTOR 2 EFF. 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

DF Analyst

INORGANIC IONS BY IC

Bromide 100 mg/L 5 50 EPA 300.0 11/27/2012 20:54 JAHERMA

Chloride 7100 mg/L 100 1000 EPA 300.0 11/27/2012 20:54 JAHERMA

Sulfate 1600 mg/L 100 1000 EPA 300.0 11/27/2012 20:54 JAHERMA

MERCURY 1631 - (Analysis Performed by Brooks Rand Labs LLC)

Vendor Parameter Complete Vendor Method V_BRAND

DISSOLVED METALS BY ICP

Manganese (Mn) 1.35 mg/L 0.05 10 EPA 200.7 11/29/2012 10:29 MHH7131

Page 7
Certificate of Laboratory Analysis

This report shall not be reproduced, except in full.

Order # J12110302

2012024783

Collection Date: 16-Nov-12 7:50 AM

Site: BIOREACTOR 2 EFF. 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

DF Analyst

TOTAL RECOVERABLE METALS BY ICP

Boron (B) 180 mg/L 0.5 10 EPA 200.7 11/28/2012 13:58 MHH7131

Calcium (Ca) 3440 mg/L 0.1 10 EPA 200.7 11/28/2012 13:58 MHH7131

Iron (Fe) < 0.1 mg/L 0.1 10 EPA 200.7 11/28/2012 13:58 MHH7131

Magnesium (Mg) 747 mg/L 0.05 10 EPA 200.7 11/28/2012 13:58 MHH7131

Manganese (Mn) 1.41 mg/L 0.05 10 EPA 200.7 11/28/2012 13:58 MHH7131

DISSOLVED METALS BY ICP-MS

Selenium (Se) 7.08 ug/L 5 5 EPA 200.8 11/29/2012 13:25 DJSULL1

TOTAL RECOVERABLE METALS BY ICP-MS

Arsenic (As) < 5 ug/L 5 5 EPA 200.8 11/30/2012 14:34 KRICHAR

Cadmium (Cd) < 5 ug/L 5 5 EPA 200.8 11/30/2012 14:34 KRICHAR

Chromium (Cr) < 5 ug/L 5 5 EPA 200.8 11/30/2012 14:34 KRICHAR

Copper (Cu) < 5 ug/L 5 5 EPA 200.8 11/30/2012 14:34 KRICHAR

Nickel (Ni) < 5 ug/L 5 5 EPA 200.8 11/30/2012 14:34 KRICHAR

Selenium (Se) < 5 ug/L 5 5 EPA 200.8 11/30/2012 14:34 KRICHAR

Silver (Ag) < 5 ug/L 5 5 EPA 200.8 11/30/2012 14:34 KRICHAR

Zinc (Zn) < 5 ug/L 5 5 EPA 200.8 11/30/2012 14:34 KRICHAR

SELENIUM SPECIATION - (Analysis Performed by Applied Speciation and Consulting, LLC)

Vendor Parameter Complete Vendor Method V_AS&C

2012024784

Collection Date: 16-Nov-12 7:50 AM

Site: BIOREACTOR 2 EFF. HG BLANK 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

DF Analyst

MERCURY 1631 - (Analysis Performed by Brooks Rand Labs LLC)

Vendor Parameter Complete Vendor Method V_BRAND

2012024785

Collection Date: 16-Nov-12 7:55 AM

Site: FILTER BLANK 

Matrix: OTHER

Analyte Result Units Qualifiers RDL Method Analysis Date/Time

Sample #:

DF Analyst

DISSOLVED METALS BY ICP

Manganese (Mn) 0.023 mg/L 0.005 1 EPA 200.7 11/29/2012 10:01 MHH7131

DISSOLVED METALS BY ICP-MS

Selenium (Se) 1.62 ug/L 1 1 EPA 200.8 11/29/2012 13:03 DJSULL1

Page 8
 

December 4, 2012 

Jay Perkins 

Duke Energy Analytical Laboratory 

Mail Code MGO3A2 (Building 7405) 

13339 Hagers Ferry Rd. 

Huntersville, NC 28078 

(704) 875-5245 

Project: Belews Creek (Flex Fuel) - WW (LIMS #J12110302) 

Dear Mr. Perkins, 

Attached is the report associated with four (4) aqueous samples submitted for selenium 

speciation on November 19, 2012. The samples were received in a sealed cooler at 0.0°C on 

November 20, 2012. Selenium speciation analysis was performed via ion chromatography 

inductively coupled plasma collision reaction cell mass spectrometry (IC-ICP-CRC-MS). Any 

issues associated with the analysis are addressed in the following report. 

If you have any questions, please feel free to contact me at your convenience. 

 Sincerely, 

 

 Russell Gerads 

Vice President 

Applied Speciation and Consulting, LLC 

 

 

Page 9
Applied Speciation and Consulting, LLC 

Report prepared for: 

Jay Perkins 

Duke Energy Analytical Laboratory 

Mail Code MGO3A2 (Building 7405) 

13339 Hagers Ferry Rd. 

Huntersville, NC 28078 

Project: Belews Creek (Flex Fuel) - WW (LIMS #J12110302) 

December 4, 2012 

1. Sample Reception 

Four (4) aqueous samples in 125mL HDPE bottles (provided by Applied Speciation and 

Consulting) were submitted for selenium speciation analysis on November 19, 2012. The 

samples were received on November 20, 2012 in a sealed container at 0.0°C. 

The samples were received in a laminar flow clean hood, void of trace metals contamination 

and ultra-violet radiation, and were designated discrete sample identifiers. An aliquot of 

each sample was filtered (0.45μm) and each filtrate was stored in a secure, monitored 

cryofreezer (maintained at a temperature of -80oC) until selenium speciation analysis could 

be performed via ion chromatography inductively coupled plasma collision reaction cell 

mass spectrometry (IC-ICP-CRC-MS). 

2. Sample Preparation 

All sample preparation is performed in laminar flow clean hoods known to be free from trace 

metals contamination. All applied water for dilutions and sample preservatives are 

monitored for contamination to account for any biases associated with the sample results. 

Selenium Speciation Analysis by IC-ICP-CRC-MS Prior to analysis, an aliquot of each 

sample was filtered with a syringe filter (0.45μm) and injected directly into an autosampler 

vial. No further sample preparation was performed as any chemical alteration of a sample 

may shift the equilibrium of the system, resulting in changes in speciation ratios. 

3. Sample Analysis 

All sample analysis is preceded by a minimum of a five-point calibration curve spanning the 

entire concentration range of interest. Calibration curves are performed at the beginning of 

Page 10
each analytical day. All calibration curves, associated with each species of interest, are 

standardized by linear regression resulting in a response factor. All sample results are 

instrument blank corrected to account for any operational biases associated with the 

analytical platform. 

Prior to sample analysis, all calibration curves are verified using second source standards 

which are identified as initial calibration verification standards (ICV). 

Ongoing instrument performance is identified by the analysis of continuing calibration 

verification standards (CCV) and continuing calibration blanks (CCB) at a minimum interval 

of every ten analytical runs. 

Selenium Speciation Analysis by IC-ICP-CRC-MS Each sample for selenium speciation 

analysis was analyzed by ion chromatography inductively coupled plasma collision reaction 

cell mass spectrometry (IC-ICP-CRC-MS) on November 30, 2012. An aliquot of each 

sample is injected onto an anion exchange column and mobilized by a basic (pH > 7) 

gradient. The eluting selenium species are then introduced into a radio frequency (RF) 

plasma where energy-transfer processes cause desolvation, atomization, and ionization. The 

ions are extracted from the plasma through a differentially-pumped vacuum interface and 

travel through a pressurized chamber (CRC) containing a reaction gas which preferentially 

reacts with interfering ions of the same target mass to charge ratios (m/z). A solid-state 

detector detects ions transmitted through the mass analyzer and the resulting current is 

processed by a data handling system. 

Retention times for each eluting species are compared to known standards for species 

identification. 

4. Analytical Issues 

The overall analyses went well and no significant analytical issues were encountered. All 

quality control parameters associated with the samples were within acceptance limits. 

The estimated method detection limits (eMDLs) for selenite, selenate, and selenocyanate are 

generated from replicate analyses of the lowest standard in the calibration curve. Not all 

selenium species are present in preparation blanks; therefore, eMDL calculations based on 

preparation blanks are artificially biased low. 

The eMDL for methylseleninic acid and selenomethionine is calculated from the average 

eMDL of selenite, selenate, and selenocyanate. The calibration does not contain 

methylseleninic acid or selenomethionine due to impurities in these standards which would 

bias the results for other selenium species. 

Page 11
If you have any questions or concerns regarding this report, please feel free to contact me. 

Sincerely, 

Russell Gerads 

Vice President 

Applied Speciation and Consulting, LLC 

Page 12
Sample Results

Sample ID Se(IV) Se(VI) SeCN MeSe(IV) SeMe

Unknown Se 

Species (n)

FGD Purge Eff 172 63.4 ND (<2.0) 3.5 ND (<1.8) 0.0 (0)

BioReactor 1 Inf 32.4 55.6 ND (<0.51) 3.09 ND (<0.45) 6.31 (1)

BioReactor 2 Inf 2.06 1.57 ND (<0.51) ND (<0.45) ND (<0.45) 0.0 (0)

BioReactor 2 Eff 0.35 ND (<0.63) ND (<0.51) ND (<0.45) ND (<0.45) 0.0 (0)

All results reflect the applied dilution and are reported in μg/L

ND = Not detected at the applied dilution

SeCN = Selenocyanate

MeSe(IV) = Methylseleninic acid

SeMe = Selenomethionine

Unknown Se Species = Total concentration of all unknown Se species observed by IC-ICP-MS

Report Generated by: Russell Gerads

Applied Speciation and Consulting, LLC

Selenium Speciation Results for Duke Energy

Project Name: Belews Creek (Flex Fuel) - WW

Contact: Jay Perkins

Date: December 4, 2012

LIMS #J12110302

Page 13
Quality Control Summary - Preparation Blank Summary

Analyte (μg/L) PBW1 PBW2 PBW3 PBW4 Mean StdDev eMDL* eMDL 250x eMDL 1000x

Se(IV) 0.000 0.000 0.000 0.000 0.000 0.000 0.001 0.22 0.86

Se(VI) 0.000 0.000 0.000 0.000 0.000 0.000 0.003 0.63 2.5

SeCN 0.000 0.000 0.000 0.000 0.000 0.000 0.002 0.51 2.0

MeSe(IV) 0.000 0.000 0.000 0.000 0.000 0.000 0.002 0.45 1.8

SeMe 0.000 0.000 0.000 0.000 0.000 0.000 0.002 0.45 1.8

eMDL = Estimated Method Detection Limit

*Please see narrative regarding eMDL calculations

Quality Control Summary - Certified Reference Materials

Analyte (μg/L) CRM True Value Result Recovery

Se(IV) LCS 9.57 9.57 100.0

Se(VI) LCS 9.48 9.23 97.3

SeCN LCS 8.92 8.78 98.4

MeSe(IV) LCS 6.47 6.15 95.1

SeMe LCS 9.32 8.78 94.2

Applied Speciation and Consulting, LLC

Selenium Speciation Results for Duke Energy

Project Name: Belews Creek (Flex Fuel) - WW

Contact: Jay Perkins

LIMS #J12110302

Date: December 4, 2012

Report Generated by: Russell Gerads

Page 14
Quality Control Summary - Matrix Duplicates

Analyte (μg/L) Sample ID Rep 1 Rep 2 Mean RPD

Se(IV) Batch QC 199.5 190.2 194.9 4.7

Se(VI) Batch QC 63.5 60.6 62.0 4.6

SeCN Batch QC ND (<2.0) ND (<2.0) NC NC

MeSe(IV) Batch QC 4.6 4.1 4.3 12.4

SeMe Batch QC ND (<1.8) ND (<1.8) NC NC

ND = Not detected at the applied dilution

NC = Value was not calculated due to one or more concentrations below the eMDL

Quality Control Summary - Matrix Spike/ Matrix Spike Duplicate

Analyte (μg/L) Sample ID Spike Conc MS Result Recovery Spike Conc MSD Result Recovery RPD

Se(IV) Batch QC 5560 5852 101.7 5560 5836 101.5 0.3

Se(VI) Batch QC 5045 5169 101.2 5045 5197 101.8 0.5

SeCN Batch QC 4575 4576 100.0 4575 4598 100.5 0.5

Applied Speciation and Consulting, LLC

Selenium Speciation Results for Duke Energy

Project Name: Belews Creek (Flex Fuel) - WW

Contact: Jay Perkins

LIMS #J12110302

Date: December 4, 2012

Report Generated by: Russell Gerads

Page 15
Page 16
 

3958 6th Ave NW • Seattle, WA 98107 • T: 206-632-6206 • F: 206-632-6017 • www.brooksrand.com • brl@brooksrand.com 

November 30, 2012 

Duke Energy 

ATTN: Jay Perkins 

Scientific Support-Laboratory

13339 Hagers Ferry Road 

Huntersville NC 28078 

jcperkins@duke-energy.com 

labcustomer@duke-energy.com 

RE: Project DUK-HV1201 Client Project: J12110302 

Dear Mr. Perkins, 

On November 20, 2012, Brooks Rand Labs (BRL) received three (3) wastewater samples and 

three (3) corresponding field blanks. An aliquot was removed from each sample bottle and 

filtered into a separate container designed for dissolved mercury (Hg) analysis. The sample 

volume from the original container was logged-in for total Hg analysis. All samples were 

received, prepared, analyzed, and stored according to BRL SOPs and EPA methodology. 

Data used for regulatory purposes has a 24 hour filtration holding time requirement. Nonregulatory purposed data has a 48 hour filtration holding time. The samples for filtration were 

received outside of the non-regulatory requirement holding time and were qualified H. 

The results were blank-corrected as described in the calculations section of the relevant SOP 

and may have been evaluated using reporting limits that have been adjusted to account for 

sample aliquot size. Please refer to the Sample Results page for sample-specific MDLs, MRLs, 

and other details. Aside from concentration qualifiers, all data was reported without further 

qualification and all associated quality control sample results met the acceptance criteria. 

BRL, an accredited laboratory, certifies the reported results of all analyses for which BRL is 

NELAP accredited meet all NELAP requirements. For more details, see the Report Information

page of the report. Please feel free to contact us if you have any questions regarding this report. 

Sincerely, 

 

Lydia Greaves 

Project Manager 

lydia@brooksrand.com 

Page 17
Project ID: DUK-HV1201

PM: Tiffany Stilwater

Client PM: Jay Perkins

 Client PO: 141391

Definition of Data Qualifiers

(Effective 9/23/09)

Laboratory Accreditation

BRL is accredited by the National Environmental Laboratory Accreditation Program (NELAP) through the State of Florida 

Department of Health, Bureau of Laboratories (E87982) and is certified to perform many environmental analyses. BRL is 

also certified by many other states to perform environmental analyses. For a current list of our

accreditations/certifications, please visit our website at <http://www.brooksrand.com/default.asp?contentID=586>. Results

reported relate only to the samples listed in the report.

Report Information

BLK

BRL

BS

CAL

CCV

CRM

D

DUP

ICV

MSD

ND

NR

PS

REC

RPD

RSD

SCV

SOP

method blank 

Brooks Rand Labs

laboratory fortified blank

calibration standard

continuing calibration verification

certified reference material

dissolved fraction

duplicate

initial calibration verification

matrix spike duplicate

non-detect

non-reportable

post preparation spike

percent recovery

relative percent difference

relative standard deviation

secondary calibration verification

standard operating procedure

MDL

MRL

MS

method detection limit

method reporting limit

matrix spike

SRM

T

COC

standard reference material

total recoverable fraction

chain of custody record 

Common Abbreviations

These qualifiers are based on those previously utilized by Brooks Rand Labs, those found in the EPA SOW ILM 03.0, 

Exhibit B, Section III, pg. B-18, and the USEPA Contract Laboratory Program National Functional Guidelines for 

Inorganic Superfund Data Review ; USEPA ; January 2010. These supersede all previous qualifiers ever employed by 

BRL.

B Detected by the instrument, the result is > the MDL but ≤ the MRL. Result is reported and considered an estimate.

E An estimated value due to the presence of interferences. A full explanation is presented in the narrative.

H Holding time and/or preservation requirements not met. Result is estimated.

J Estimated value. A full explanation is presented in the narrative.

J-M Duplicate precision (RPD) for associated QC sample was not within acceptance criteria. Result is estimated.

J-N Spike recovery for associated QC sample was not within acceptance criteria. Result is estimated.

M Duplicate precision (RPD) was not within acceptance criteria. Result is estimated.

N Spike recovery was not within acceptance criteria. Result is estimated.

R Rejected, unusable value. A full explanation is presented in the narrative.

U Result is ≤ the MDL or client requested reporting limit (CRRL). Result reported as the MDL or CRRL.

X Result is not BLK-corrected and is within 10x the absolute value of the highest detectable BLK in the batch. 

Result is estimated.

Field Quality Control Samples

Please be notified that certain EPA methods require the collection of field quality control samples of an appropriate type

and frequency; failure to do so is considered a deviation from some methods and for compliance purposes should only be

done with the approval of regulatory authorities. Please see the specific EPA methods for details regarding required field

quality control samples.

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.com

Page 18
Project ID: DUK-HV1201

PM: Tiffany Stilwater

Client PM: Jay Perkins

 Client PO: 141391

Sample Information

Sample Lab ID Report Matrix Type Sampled Received

BioReactor 1 Inf 1247007-01 Influent Sample 11/16/2012 11/20/2012

BioReactor 1 Inf 1247007-02 Influent Sample 11/16/2012 11/20/2012

BioReactor 1 Inf Hg Blk 1247007-03 DIW Field Blank 11/16/2012 11/20/2012

BioReactor 1 Inf Hg Blk 1247007-04 DIW Field Blank 11/16/2012 11/20/2012

BioReactor 2 Inf 1247007-05 Influent Sample 11/16/2012 11/20/2012

BioReactor 2 Inf 1247007-06 Influent Sample 11/16/2012 11/20/2012

BioReactor 2 Inf Hg Blk 1247007-07 DIW Field Blank 11/16/2012 11/20/2012

BioReactor 2 Inf Hg Blk 1247007-08 DIW Field Blank 11/16/2012 11/20/2012

BioReactor 2 Eff 1247007-09 Effluent Sample 11/16/2012 11/20/2012

BioReactor 2 Eff 1247007-10 Effluent Sample 11/16/2012 11/20/2012

BioReactor 2 Eff Hg Blk 1247007-11 DIW Field Blank 11/16/2012 11/20/2012

BioReactor 2 Eff Hg Blk 1247007-12 DIW Field Blank 11/16/2012 11/20/2012

Batch Summary

Analyte Lab Matrix Method Prepared Analyzed Batch Sequence

Hg Water EPA 1631 11/21/2012 11/26/2012 B122176 1200884

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.com

Page 19
Project ID: DUK-HV1201

PM: Tiffany Stilwater

Client PM: Jay Perkins

 Client PO: 141391

Sample Results

Sample Analyte Report Matrix Basis Result Qualifier MDL MRL Unit Batch Sequence

BioReactor 1 Inf

1247007-01 Hg Influent 10.1 T 119 3.79 ng/L B122176 1200884

1247007-02 Hg Influent 2.02 D H 70.3 0.76 ng/L B122176 1200884

BioReactor 1 Inf Hg Blk

1247007-03 Hg DIW T U 0.15 0.15 0.41 ng/L B122176 1200884

1247007-04 Hg DIW D H 0.15 , U 0.15 0.41 ng/L B122176 1200884

BioReactor 2 Eff

1247007-09 Hg Effluent 0.40 T 6.14 0.15 ng/L B122176 1200884

1247007-10 Hg Effluent D H 0.86 0.15 0.40 ng/L B122176 1200884

BioReactor 2 Eff Hg Blk

1247007-11 Hg DIW T U 0.15 0.15 0.40 ng/L B122176 1200884

1247007-12 Hg DIW D H 0.15 , U 0.15 0.40 ng/L B122176 1200884

BioReactor 2 Inf

1247007-05 Hg Influent 1.02 T 33.4 0.38 ng/L B122176 1200884

1247007-06 Hg Influent D H 3.02 0.15 0.40 ng/L B122176 1200884

BioReactor 2 Inf Hg Blk

1247007-07 Hg DIW T U 0.15 0.15 0.41 ng/L B122176 1200884

1247007-08 Hg DIW D H 0.15 , U 0.15 0.41 ng/L B122176 1200884

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.com

Page 20
Project ID: DUK-HV1201

PM: Tiffany Stilwater

Client PM: Jay Perkins

 Client PO: 141391

Accuracy & Precision Summary

Batch: B122176

Analyte Native Spike Result Units REC & Limits RPD & Limits

Method: EPA 1631

Lab Matrix: Water

Sample

B122176-SRM1 Certified Reference Material (1245026, NIST 1641d 1000x dilution)

Hg 15.68 101% 15.90 ng/L 85-115

B122176-MS1 Matrix Spike (1247007-01)

Hg 119.4 505.1 107% 662.3 ng/L 71-125

B122176-MSD1 Matrix Spike Duplicate (1247007-01)

Hg 662.4 119.4 505.1 108% ng/L 71-125 24 0.02%

Method Blanks & Reporting Limits

Batch: B122176

Method: EPA 1631

Matrix: Water

Analyte: Hg

Sample Result Units

B122176-BLK1 0.09 ng/L

B122176-BLK2 0.07 ng/L

B122176-BLK3 0.08 ng/L

B122176-BLK4 0.07 ng/L

Average: 0.08 Standard Deviation: 0.01 MDL: 0.15

Limit: 0.50 Limit: 0.10 MRL: 0.40

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.com

Page 21
Project ID: DUK-HV1201

PM: Tiffany Stilwater

Client PM: Jay Perkins

 Client PO: 141391

Instrument Calibration

Sequence: 1200884

Instrument: THG-05 Method: EPA 1631

Date: 11/26/2012

Total Mercury and Mercury Speciation by CVAFS

Analyte: Hg

Lab ID True Value Result Units REC & Limits

1200884-IBL1 2.05 pg of Hg

1200884-IBL2 4.46 pg of Hg

1200884-IBL3 4.03 pg of Hg

1200884-IBL4 4.39 pg of Hg

1200884-CAL1 10.00 9.93 pg of Hg 99%

1200884-CAL2 25.00 25.49 pg of Hg 102%

1200884-CAL3 100.0 97.50 pg of Hg 97%

1200884-CAL4 500.0 479.9 pg of Hg 96%

1200884-CAL5 2500 2603 pg of Hg 104%

1200884-CAL6 10000 10160 pg of Hg 102%

1200884-ICV1 1568 1590 pg of Hg 101% 85-115

1200884-CCB1 8.21 pg of Hg

1200884-CCV1 500.0 529.5 pg of Hg 106% 77-123

1200884-CCB2 4.98 pg of Hg

1200884-CCB3 4.10 pg of Hg

1200884-CCB4 4.36 pg of Hg

1200884-CCV2 500.0 510.9 pg of Hg 102% 77-123

1200884-CCB5 6.48 pg of Hg

1200884-CCV3 500.0 535.4 pg of Hg 107% 77-123

1200884-CCB6 4.51 pg of Hg

1200884-CCV4 500.0 548.3 pg of Hg 110% 77-123

1200884-CCB7 4.14 pg of Hg

1200884-CCV5 500.0 542.6 pg of Hg 109% 77-123

1200884-CCB8 4.33 pg of Hg

1200884-CCV6 500.0 538.5 pg of Hg 108% 77-123

1200884-CCB9 4.26 pg of Hg

1200884-CCV7 500.0 534.2 pg of Hg 107% 77-123

1200884-CCBA 4.58 pg of Hg

1200884-CCV8 500.0 531.9 pg of Hg 106% 77-123

1200884-CCBB 4.40 pg of Hg

1200884-CCV9 500.0 540.0 pg of Hg 108% 77-123

1200884-CCBC 4.56 pg of Hg

1200884-CCVA 500.0 535.5 pg of Hg 107% 77-123

1200884-CCBD 5.26 pg of Hg

1200884-CCVB 500.0 538.2 pg of Hg 108% 77-123

1200884-CCBE 4.16 pg of Hg

1200884-CCVC 500.0 542.7 pg of Hg 109% 77-123

1200884-CCBF 3.74 pg of Hg

1200884-CCVD 500.0 541.2 pg of Hg 108% 77-123

1200884-CCBG 4.35 pg of Hg

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.com

Page 22
Project ID: DUK-HV1201

PM: Tiffany Stilwater

Client PM: Jay Perkins

 Client PO: 141391

Sample Containers

Lab ID: 1247007-01 Report Matrix: Influent

Sample: BioReactor 1 Inf Sample Type: Sample Received: 11/20/2012

Collected: 11/16/2012

Des Container Size Lot Preservation P-Lot pH Ship. Cont.

A 71666330

10

Bottle FLPE Hg-T 500 mL none n/a Cooler

Lab ID: 1247007-02 Report Matrix: Influent

Sample: BioReactor 1 Inf Sample Type: Sample Received: 11/20/2012

Collected: 11/16/2012

Comments: Qualify H

Des Container Size Lot Preservation P-Lot pH Ship. Cont.

A 71691270

10

Bottle FLPE Hg-T 250 mL none n/a Cooler

Lab ID: 1247007-03 Report Matrix: DIW

Sample: BioReactor 1 Inf Hg Blk Sample Type: Field Blank Received: 11/20/2012

Collected: 11/16/2012

Des Container Size Lot Preservation P-Lot pH Ship. Cont.

A 71666330

10

Bottle FLPE Hg-T 500 mL none n/a Cooler

Lab ID: 1247007-04 Report Matrix: DIW

Sample: BioReactor 1 Inf Hg Blk Sample Type: Field Blank Received: 11/20/2012

Collected: 11/16/2012

Comments: Qualify H

Des Container Size Lot Preservation P-Lot pH Ship. Cont.

A 71691270

10

Bottle FLPE Hg-T 250 mL none n/a Cooler

Lab ID: 1247007-05 Report Matrix: Influent

Sample: BioReactor 2 Inf Sample Type: Sample Received: 11/20/2012

Collected: 11/16/2012

Des Container Size Lot Preservation P-Lot pH Ship. Cont.

A 71666330

10

Bottle FLPE Hg-T 500 mL none n/a Cooler

Lab ID: 1247007-06 Report Matrix: Influent

Sample: BioReactor 2 Inf Sample Type: Sample Received: 11/20/2012

Collected: 11/16/2012

Comments: Qualify H

Des Container Size Lot Preservation P-Lot pH Ship. Cont.

A 71691270

10

Bottle FLPE Hg-T 250 mL none n/a Cooler

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.com

Page 23
Project ID: DUK-HV1201

PM: Tiffany Stilwater

Client PM: Jay Perkins

 Client PO: 141391

Sample Containers

Lab ID: 1247007-07 Report Matrix: DIW

Sample: BioReactor 2 Inf Hg Blk Sample Type: Field Blank Received: 11/20/2012

Collected: 11/16/2012

Des Container Size Lot Preservation P-Lot pH Ship. Cont.

A 71666330

10

Bottle FLPE Hg-T 500 mL none n/a Cooler

Lab ID: 1247007-08 Report Matrix: DIW

Sample: BioReactor 2 Inf Hg Blk Sample Type: Field Blank Received: 11/20/2012

Collected: 11/16/2012

Comments: Qualify H

Des Container Size Lot Preservation P-Lot pH Ship. Cont.

A 71691270

10

Bottle FLPE Hg-T 250 mL none n/a Cooler

Lab ID: 1247007-09 Report Matrix: Effluent

Sample: BioReactor 2 Eff Sample Type: Sample Received: 11/20/2012

Collected: 11/16/2012

Des Container Size Lot Preservation P-Lot pH Ship. Cont.

A 71666330

10

Bottle FLPE Hg-T 500 mL none n/a Cooler

Lab ID: 1247007-10 Report Matrix: Effluent

Sample: BioReactor 2 Eff Sample Type: Sample Received: 11/20/2012

Collected: 11/16/2012

Comments: Qualify H

Des Container Size Lot Preservation P-Lot pH Ship. Cont.

A 71691270

10

Bottle FLPE Hg-T 250 mL none n/a Cooler

Lab ID: 1247007-11 Report Matrix: DIW

Sample: BioReactor 2 Eff Hg Blk Sample Type: Field Blank Received: 11/20/2012

Collected: 11/16/2012

Des Container Size Lot Preservation P-Lot pH Ship. Cont.

A 71666330

10

Bottle FLPE Hg-T 500 mL none n/a Cooler

Lab ID: 1247007-12 Report Matrix: DIW

Sample: BioReactor 2 Eff Hg Blk Sample Type: Field Blank Received: 11/20/2012

Collected: 11/16/2012

Comments: Qualify H

Des Container Size Lot Preservation P-Lot pH Ship. Cont.

A 71691270

10

Bottle FLPE Hg-T 250 mL none n/a Cooler

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.com

Page 24
Project ID: DUK-HV1201

PM: Tiffany Stilwater

Client PM: Jay Perkins

 Client PO: 141391

Shipping Containers

Cooler

Tracking No: 535305195917 via FedEx

Temperature: -0.4 °C

Coolant Type: Ice

Description: Cooler

Damaged in transit? No

Returned to client? No

Custody seals present? No

Custody seals intact? No

COC present? Yes

Received: November 20, 2012 8:50

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.com

Page 25
Page 26
Page 27