PROCEDURE DD: QUALITY CONTROL AND 

QUALITY ASSURANCE REQUIREMENTS FOR 

HYDROCHLORIC ACID CONTINUOUS EMISSION 

MONITORING SYSTEMS AT STATIONARY 

SOURCES 

 

 November 2005

Revision: 1.0 

 

 
EXECUTIVE SUMMARY 

Procedure DD outlines the continuing quality control and quality assurance requirements for a 

hydrochloric acid (HCl) continuous emission monitoring system (CEMS). Procedure DD provides the

procedures and performance criteria for daily checks of the high-level and zero-level drift. This

procedure also provides the procedures and performance criteria for quarterly absolute calibration

audits as well as annual accuracy audits. 

i 
TABLE OF CONTENTS 

1.0 USING PROCEDURE DD......................................................................................................3 

1.1 WHAT IS THE PURPOSE AND APPLICABILITY OF PROCEDURE DD? ....................3 

1.2 WHAT ARE THE BASIC REQUIREMENTS OF PROCEDURE DD? .............................3 

1.3 WHAT SPECIAL DEFINITIONS APPLY TO PROCEDURE DD?...................................3 

2.0 INTERFERENCES, SAFETY, AND EQUIPMENT REQUIREMENTS ..................................6 

2.1 WHAT DO I NEED TO KNOW TO ENSURE THE SAFETY OF PERSONS

USING PROCEDURE DD? .............................................................................................6 

2.2 WHAT EQUIPMENT AND SUPPLIES DO I NEED?.......................................................6 

2.3 WHAT REGENTS AND STANDARDS DO I NEED? ......................................................6 

3.0 QUALITY CONTROL, CALIBRATION AND STANDARDIZATION .......................................8 

3.1 WHAT QUALITY CONTROL MEASURES ARE REQUIRED BY

PROCEDURE DD?..........................................................................................................8 

3.2 WHAT CALIBRATION CHECKS AND AUDIT PROCEDURES MUST I

PERFORM FOR MY HCL CEMS?..................................................................................8 

4.0 PERFORMANCE CRITERIA AND CALCULATIONS..........................................................12 

4.1 WHAT ARE THE PERFORMANCE CRITERIA FOR MY HCl CEMS?.........................12 

4.2 WHAT CALCULATIONS, EQUATIONS, AND DATA ANALYSES ARE

NEEDED?......................................................................................................................13

5.0 OTHER REQUIREMENTS AND INFORMATION................................................................19 

5.1 WHAT POLLUTION PREVENTION PROCEDURES MUST I FOLLOW? 

[RESERVED].................................................................................................................19 

5.2 WHAT REFERENCE TABLES AND FIGURES ARE RELEVANT TO 

PROCEDURE DD?........................................................................................................19 

6.0 RELEVANT REFERENCES.................................................................................................22 

 

Table 1. Ongoing Quality Assurance/Quality Control Calibration Gas Ranges 

Figure 1. Dynamic Spiking Overview 

ii 
1.0 USING PROCEDURE DD

1.1 WHAT IS THE PURPOSE AND APPLICABILITY OF PROCEDURE DD? 

The purpose of Procedure DD is to establish the minimum requirements for quality control (QC) and 

quality assurance (QA) procedures for the HCl CEMS. These quality control and quality assurance

procedures are in place to ensure the accuracy and validity of the data reported by the HCl CEMS for 

emission compliance purposes. Any HCl CEMS used for reporting purposes to the Environmental

Protection Agency (EPA), state, or local environmental agencies must comply with Procedure DD 

immediately after the completion of the performance specification test outlined in Performance 

Specification ZI

. 

1.2 WHAT ARE THE BASIC REQUIREMENTS OF PROCEDURE DD? 

Procedure DD includes procedures and performance criteria for conducting daily calibration drift tests,, 

quarterly absolute calibration audits, and annual accuracy determinations. Procedure DD also covers 

the requirements and timelines associated with periods when the HCl is out of control.

1.3 WHAT SPECIAL DEFINITIONS APPLY TO PROCEDURE DD? 

Calibration Drift: The difference in the CEMS output readings from the established reference value after 

a stated period of operation during which no unscheduled maintenance or adjustment took place. 

Calibration Error: The mean difference between the concentration indicated by the CEMS and the

known concentration generated by a calibration source at three levels when the entire CEMS, including

the sampling interface is challenged. A CE test is performed to document the accuracy and linearity of

the CEMS over the entire measurement range.

Centroidal Area: means a concentric area that is geometrically similar to the stack or duct cross section 

and is no greater than 1 percent of the stack or duct cross-sectional area. 

Continuous Emission Monitoring System: means the total equipment required for the determination of

a gas concentration or emission rate. The sample interface, pollutant analyzer, diluent analyzer, and data

recorder are the major subsystems of the CEMS. 

3 
Correlation Coefficient: determines the extent of a linear relationship between two fields over a given

period of time.

Data Recorder: The portion of the CEMS that provides a record of analyzer output, flags which indicate

normal operation, and flags indicating abnormal operation. The data recorder may record other pertinent

data such as effluent flow rates, and various instrument temperatures. 

Diluent Analyzer: means that portion of the CEMS that senses the diluent gas (i.e. O2) and generates an

output proportional to the gas concentration. 

Dynamic Spiking: a procedure used to document the accuracy, precision, and bias of the monitoring

system by quantitatively spiking a certified gas into the pollutant gas stream.

High-Level Drift: means the absolute difference between a high-level calibration gas and the monitor 

response, in units of the applicable standard.

Instrument Measurement Range: The range of HCl concentrations the instrument can reliably measure 

from the lowest concentration to the highest. 

Intercept: value of the Y variable when the X variable is equal to zero-level. 

Linear Regression: a methodology used to find a formula that can be used to relate two variables that

are linearly related. 

Path Sampling CEMS: A CEMS that samples the source effluent along a path greater than 10 percent of

the equivalent diameter of the stack or duct cross section. 

Point Sampling CEMS: A CEMS that samples the source effluent at a single point. 

Pollutant Analyzer: means that portion of the CEMS that senses the pollutant gas and generates an

output proportional to the gas concentration. 

Relative Accuracy (RA): means the absolute mean difference between the gas concentration or 

emission rate determined by the CEMS and the value determined by the Reference Method (RM), plus

4 
the 2.5 percent error confidence coefficient of a series of tests, divided by the mean of the RM tests or the 

applicable emission limit. 

Response Time: The time interval between the start of a step change in the system input and when the 

pollutant analyzer output reached 95% of the final value. 

Sample Interface: The portion of the CEMS used for one or more of the following: sample acquisition, 

sample transport, sample conditioning, or protection of the monitor from the effects of stack gas. 

Slope: The rate of change of Y relative to the change in X. 

Zero-level-level drift: means the absolute difference between a high-level calibration gas and the

monitor response, in units of the applicable standard. 

5 
2.0 INTERFERENCES, SAFETY, AND EQUIPMENT REQUIREMENTS 

2.1 WHAT DO I NEED TO KNOW TO ENSURE THE SAFETY OF PERSONS USING PROCEDURE 

DD? 

People using Procedure DD may be exposed to hazardous material, operational hazards, and hazardous 

site conditions. Procedure DD does not address all the safety issues associated with its use. It is your 

responsibility to ensure the safety of persons using Procedure DD. Some helpful references may include 

the CEMS manual, the CEMS manufacturer, other reference methods, and on-site safety regulations. 

2.2 WHAT EQUIPMENT AND SUPPLIES DO I NEED? 

2.2.1 Equipment for the CEMS 

1. Sample Extraction System: This portion of the CEMS must present a sample of source

effluent to the sampling module that is directly representative or can be corrected so as to be

representative of source. The sample extraction system typically consists of a sample probe

and a heated umbilical line.

2. Pressure Regulation Module: The pressure regulation module is designed to remove both 

free particulates and water-soluble aerosols from the gas stream prior to analysis by the

sampling module. This module also provides a sample gas to the analyzer at a constant

pressure (optional).

3. Analyzer module: The portion of the CEMS that quantitates stack gas concentrations of HCl. 

4. Diluent Module: This portion of the CEMS quantifies stack gas concentrations of oxygen or

CO2. For systems with a multi-component analyzer, the same analyzer quantifies the

concentration of all measured analytes.

5. System Controller: This portion of the CEMS provides control of the analyzer, sample

probe, pressure regulation module and the sample interface. 

6. Data recorder: Your HCl CEMS must be able to record HCl concentrations and instrument

status signals (flags). 

2.3 WHAT REGENTS AND STANDARDS DO I NEED? 

6 
2.3.1 Reference Gases, Gas Cells, or Optical Filters 

See Table 1 for calibration gas concentration ranges. 

 

7 
3.0 QUALITY CONTROL, CALIBRATION AND STANDARDIZATION 

3.1 WHAT QUALITY CONTROL MEASURES ARE REQUIRED BY PROCEDURE DD? 

You must develop and implement a quality control program that, at a minimum includes detailed written

procedures for all of the following activities. 

1. Procedures for performing drift checks on a daily basis, including but not limited to, zero-level 

drift, high-level drift, and sample volume measurement drift (when applicable). 

2. Procedures and methods of adjusting the HCl CEMS in response to the results of the drift 

checks.

3. Preventative maintenance of the HCl CEMS. 

4. Data recording, calculations, and reporting. 

5. Procedures for required audits, including absolute calibration audits and accuracy audits. 

6. Procedures for adjusting your CEMS based on audit results. 

7. A program of corrective action and stack operation procedures in case of a CEMS 

malfunction and an out of control period. 

You are required to keep written documentation of your QA/QC procedures on record and available for 

inspection for the life of the CEMS or until you are no longer subject to the requirements of this

procedure. If you fail two consecutive audits you must revise your QA/QC procedures. 

3.2 WHAT CALIBRATION CHECKS AND AUDIT PROCEDURES MUST I PERFORM FOR MY HCL 

CEMS? 

Ongoing quality control will include daily calibration drift tests, quarterly absolute calibration audits, and

annual accuracy test audits, which are discussed below: 

3.2.1 Daily Calibration Drift Checks 

The zero-level and high-level drift shall be checked once daily. The calibration gas concentration ranges 

are provided in Table 1. To determine the zero-level drift or calibration drift, introduce to the CEMS the

reference gases, gas cells, or optical filters (these need not be certified) for the zero-level and high-level 

values. Record the CEMS response and subtract this value from the reference value. The calibration drift 

shall not exceed 5% of the instrument span. Zero-level and high-level calibration drifts shall be adjusted,

at a minimum, whenever the 24-hour zero-level drift exceeds the limits of the calibration drift specification.

8 
The amount of excess zero-level and high-level drift measured at the 24-hour interval checks will be

recorded. 

3.2.2 Quarterly Absolute Calibration Audit (Calibration Error Test) 

The quarterly Absolute Calibration Audit will follow the Calibration Error procedure outlined in

Performance Specification Z. The HCl CEMS shall be challenged three non-consecutive times with zerolevel, mid-level, and high-level certified gases. The cylinder gases will be certified gases (EPA Protocol 1

gases are not available for HCl). The calibration gases will be injected into the sample system as close to 

the sampling probe as practical and will pass through all CEMS components used during normal 

monitoring. The absolute difference between the instrument response and the reference value (certified 

gas) will be calculated after each injection and the resulting three differences will be averaged to

determine the calibration error at each measurement point. The calibration error for each of the three

levels will not exceed 5% of span. 

An absolute calibration audit will be performed quarterly, except during the quarter the annual accuracy is 

performed. 

3.2.3 Annual Accuracy Test Audit 

The annual accuracy audit of the HCl CEMS will consist of a calibration drift test (seven day drift) and an 

accuracy determination by performing either a relative accuracy determination, or by dynamic spiking. 

3.2.3.1 Seven-Day Drift Test (seven day drift) 

Prior to the start of the accuracy test (either the Relative Accuracy or Dynamic Spiking) you must perform

a calibration drift test for a period of seven consecutive days. The seven-day calibration drift test must be

conducted when the facility is under normal operations. During the calibration drift test period you must

determine the magnitude of the zero-level calibration drift and the high-level calibration drift at least once

each day. During the stability tests, no adjustments or calibrations may be made to the CEMS. If 

periodic automatic or manual adjustments are made to the CEMS zero-level and calibration settings, 

conduct the CD test immediately before these adjustments, or conduct it in such a way that the CD can

be determined. The zero-level and high-level drift each must be less than 5% of the instrument span for 

each of seven consecutive days. 

3.2.3.2 Relative Accuracy Determination 

The relative accuracy determination will be conducted while the affected facility is operating during normal

operation, or as specified in an applicable subpart. 

Reference Methods (RM). Unless otherwise specified in an applicable subpart of the regulations,

Methods 26 or 26A, or their approved alternatives, are the reference methods for HCl. Other reference

methods for moisture, oxygen, etc. may be necessary. 

9 
Sampling Strategy for RM Tests. Conduct the RM tests in such a way that they will yield results 

representative of the emissions from the source and can be correlated to the CEMS data. It is preferable

to conduct the diluent (if applicable), moisture (if needed), and pollutant measurements simultaneously. 

However, diluent and moisture measurements that are taken within an hour of the pollutant 

measurements may be used to calculate dry pollutant concentration and emission rates. In order to 

correlate the CEMS and RM data properly, note the beginning and end of each RM test period of each 

run (including the exact time of day) on the CEMS chart recordings or other permanent record of output.

For integrated samples make a sample traverse of at least 21 minutes, sampling for an equal time at

each traverse point. 

Number of RM Tests. Conduct a minimum of nine sets of all necessary RM test runs. 

NOTE: More than nine sets of RM tests may be performed. If this option is chosen, a maximum of three

sets of the test results may be rejected so long as the total number of test results used to determine the 

RA is greater than or equal to nine. However, all data must be reported, including the rejected data. 

Correlation of RM and CEMS Data. Correlate the CEMS and the RM test data as to the time and

duration by first determining from the CEMS final output (the one used for reporting) the integrated 

average pollutant concentration or emission rate for each pollutant RM test period. Consider system

response time, if important, and confirm that the pair of results are on a consistent moisture, temperature, 

and diluent concentration basis. Then, compare each integrated CEMS value against the corresponding 

average RM value. If the RM has an integrated sampling technique, make a direct comparison of the RM 

results and CEMS integrated average value. 

Calculate the mean difference between the RM and CEMS values in the units of the emission standard,

the standard deviation, the confidence coefficient, and the relative accuracy according to the calculations 

in Section 4.0. The RA of the CEMS must not be greater than 20% of the mean value of the reference 

method (RM) test data, or not greater than 10% in terms of the emission standard (ppmv, dry), or have an

absolute difference of less than 5 ppmv between the mean reference value and the mean CEMS value. 

3.2.3.3 Dynamic Spiking 

Dynamic spiking can be used in lieu of the Relative Accuracy Determination as an alternative method for 

documenting the accuracy, precision, and bias of the HCl CEMS. Figure 1 provides an overview of the

dynamic spiking arrangement. 

While the HCl CEMS is sampling flue gas, HCl reference gas is introduced into the CEMS sample

interface by using a mass flow controller (or equivalent). The target ratio of actual flue gas to the HCl

reference gas is 9:1, but must be kept at a minimum ratio of 1:1. The HCl concentration is quantitated by 

10 
the HCl CEMS. One minute averaged values, in units of the applicable standard, are collected and

archeived. 

An element of opportunity shall be used to calculate the dynamically spiked HCl concentration

(represents the reference HCl concentration). The element of opportunity is an analyte, which would 

normally be present in the flue gas at steady concentrations. Examples of prospective elements of 

opportunity would be O2, NOx, SO2, CO2, or moisture. This analyte is measured using a certified 

reference CEMS, separate from the HCL CEMS. 

In addition, a calibrated Laminar Flow Element (flow as an element of opportunity) can be used as a

secondary indicator for calculating the dynamically spiked HCl concentration (represents the reference 

HCl concentration). While the Laminar Flow Element is mainly for verification of the dynamic spiking 

system operation, the use of flow as an element of opportunity can be used to calculate the concentration

of the HCl dynamic spike. 

In addition, HCl concentration of the flue gas (baseline HCL) shall be determined prior to each spiking

trial. The baseline HCl concentration is based upon a series of at least 10, one-minute averaged data

points. Baseline data must be collected immediately prior to each test run. The data points must include

the HCl concentration, element of opportunity concentration(s) and the flow recorded by the laminar flow

element. A site-specific standard operating procedure for the HCL dynamic spiking is required.

Linear regression is used to establish the accuracy, precision, and bias of the HCl CEMS. The criteria for 

the HCl dynamic spiking are: 

1. The correlation coefficient (r) must be greater than or equal to 0.90

2. The slope must be 1.0, +/- 0.15 

3. The Intercept must be equal to or less than 15% of the instrument span 

 

. 

11 
4.0 PERFORMANCE CRITERIA AND CALCULATIONS

4.1 WHAT ARE THE PERFORMANCE CRITERIA FOR MY HCl CEMS? 

You must demonstrate that your HCl CEMS has adequate stability, precision, accuracy, response time

and data reporting capabilities to determine if your facility is in compliance with HCl emission standards or

operating permit limits as specified in applicable regulations or permits. You will demonstrate this

capacity by showing that your HCl CEMS meets the following performance criteria. 

4.1.1 Daily Calibration Check Criteria 

The daily zero-level drift or calibration drift each shall not exceed 5% of span. The calculations are 

provided in Section 4.2.2. 

4.1.2 Quarterly Absolute Audit 

The absolute mean difference between the CEMS and the reference values, at all three test points, must

be no greater than 5% of span. The calculations are provided in Section 4.2.3. 

4.1.3 Relative Accuracy 

The RA of the CEMS must not be greater than 20% of the mean value of the reference method (RM) test 

data, or not greater than 10% in terms of the emission standard (ppmv, dry), or have an absolute

difference of less than 5 ppmv between the mean reference value and the mean CEMS value. The 

calculations are provided in Sections 4.2.5 - 4.2.7. 

4.1.4 Dynamic Spiking 

Dynamically spiking may be used in lieu of a Relative Accuracy Test as a means to document the

accuracy, precision, and bias of the HCl CEMS. 

The criteria for the HCl dynamic spiking is (1) The correlation coefficient, r, must be greater than or equal 

to 0.90 (2) The slope must be 1.0, +/- 0.15 (3) The Intercept must be equal to or less than 15% of the 

instrument span. 

The calculations are provided in Section 4.2.8. 

4.1.4.1 Out of Control 

The HCl CEMS will be considered out of control if the zero-level (low-level) or high level calibration drift 

exceeds two times the applicable calibration drift specification (i.e. the calibration drift is greater than ± 10

12 
% of span.) In addition, the HCl CEMS will also be considered out of control if the CEMS fails the

quarterly absolute calibration audit or the annual accuracy test audit. 

When the CEMS is out of control, the owner or operator shall take the necessary corrective action and

shall repeat all necessary tests, which indicate that the system is out of control. The owner or operator 

shall take corrective action and conduct retesting until the performance requirements are below the 

applicable criteria. 

The out of control period begins immediately after the failed drift test or audit. The out of control period

ends immediately following the successful completion of required procedures associated with the failure 

of the audit test. During the out of control period you may not use the data generated by the HCl CEMS

for compliance purposes. 

4.2 WHAT CALCULATIONS, EQUATIONS, AND DATA ANALYSES ARE NEEDED? 

4.2.1 Arithmetic Mean 

Calculate the arithmetic mean of a data set as follows:

= ∑xi

n

x

1

(Equation 1) 

where : 

x = Arithmetic mean 

n = Number of data points 

xi = Value of each data point 

4.2.2 Daily High-level and Zero-level Drift 

 CD =(⎪R - A⎪/ S ) * 100 (Equation 2) 

where: 

 CD = Percent calibration drift (%), 

R = Reference value of zero-level or high level calibration gas introduced into the 

monitoring system (ppmv), 

13 
A = Actual monitor response to calibration gas (ppmv), 

S = Span of the instrument (ppmv). 

4.2.3 Calibration Error 

 CE = ⎪davg / S⎪ * 100 (Equation 3) 

where: 

 CE = Percent calibration error (%), 

 davg = Mean difference between CEMS response and the known reference gas (ppmv), 

S = Span of the monitor (ppmv). 

4.2.4 Mean Difference 

Calculate the arithmetic mean difference as follows: 

= ∑ i avg d

n

d 1 (Equation 4) 

where : 

davg = Arithmetic mean of differences

n = Number of data sets 

di = difference of each data set 

And 

di = xi − yi (Equation 5)

where: 

xi = Value at data set x 

yi = Value at data set y 

14 
4.2.5 Relative Accuracy

The RA for the monitors will be calculated using the least restrictive of equation 6 or equation 7:

RA = (⎪davg⎪ + ⎪CC⎪) (Equation 6)

 or 

RA = ((⎪davg⎪ + ⎪CC⎪) / TMavg ) * 100 (Equation 7)

where: 

RA = Relative accuracy, either percent or ppmv or % absolute, 

 davg = Arithmetic mean of differences between value measured by the installed CEMS and the

reference method, 

CC = Confidence coefficient, 

 TMavg = Average value measured by the reference test method monitors. 

4.2.6 Confidence Coefficient 

The confidence coefficient will be calculated using equation 8: 

 CC = t0.975 * (Sd / (n1⁄2)) (Equation 8)

where: 

 CC = Confidence coefficient, 

 t0.975 = t-value obtained from Table 2.1 in Performance Specification 2, 40 CFR Part 60

Appendix B 

4.2.7 Standard Deviation (of Differences) 

Sd = Standard deviation of differences measured between the installed CEMS and the

reference methods calculated using equation 8: 

1 / 2 2

1

1

2

1

⎥

⎥

⎥

⎥

⎥

⎥

⎦

⎤

⎢

⎢

⎢

⎢

⎢

⎢

⎣

⎡

−

⎟

⎟

⎠

⎞

⎜

⎜

⎝

⎛

−

=

∑ ∑ =

=

n

n

d

S

n

i i n

i i

d

d (Equation 9)

where: n = Number of values in this data set. 

15 
4.2.8 Dynamic Spiking 

For baseline data, determine the average concentration for the HCl CEMS, the reference method

elements or opportunity, the Laminar Flow Element flow value and the mass flow control settings. For 

each specific test run, determine the average value of the CEMS HCl data, the average value of the

reference values for the elements of opportunity, including the laminar flow. The average CEMS HCl 

value is actual CEMS HCl value for the specific test run. The reference HCl value for the test run is

calculated as follows: 

HCl ref = CalGas BaselineHCl

Opp pre

Opp post

avg

avg + −

− 1− * (Equation 10)

where: 

 HCl ref = Reference HCl value for run “X”

 Opp-post avg = average reference element of opportunity value during test (downstream of spike) 

 Opp-pre avg = average reference element of opportunity value during test (upstream of spike) 

CalGas = HCl calibration gas concentration 

Baseline HCl = CEMS HCl average baseline value 

Using data from each of the three test runs, perform a linear regression of the CEMS HCl average values

against the Reference HCl values using the following equations given below: 

y b b x o 1 ˆ = + (Equation 11)

where: 

bo = The y intercept 

b1 = The slope 

16 
The intercept is calculated according to the following equation: 

b y b x o = − 1 (Equation 12)

where: 

∑=

=

n

i

i x

n

x

1

1 

∑=

=

n

i

i y

n

y

1

1 

The slope of the line is calculated according to equation below: 

∑

∑

=

=

−

− −

= n

i 1

2

i

n

i 1

i i

1

( x x )

( x x )( y y )

b (Equation 13) 

The linear correlation coefficient is calculated according to the following equation. 

∑ ∑

∑

= =

=

− −

⎟

⎠

⎞ ⎜

⎝

⎛ − −

= n

i

i

n

i

i

n

i

i i

x x y y

x x y y

r

1

2

1

2

2

2 1

( ) ( )

( )( )

 (Equation 14) 

4.2.8.1 Dynamic Spiking Corrected Values 

a). If the HCl CEMS fails to meet both the slope and intercept criteria, the following correction factor must

be applied to the one-minute average HCl data: 

 

1

0

b

y b CC i − = (Equation 15)

17 
Where:

= Corrected CEMS HCL concentration C C

yi = CEMS reported HCL concentration 

b0 = the intercept of the least squares linear regression line 

b1 = the slope of the least squares linear regression line 

b). If the HCl CEMS fails to meet the slope criteria, but meets the intercept criteria, the following

correction factor must be applied to the one-minute average HCl data: 

 

1 b

y CC i = (Equation 16)

c). If the HCl CEMS fails to meet the intercept criteria, but meets the slope criteria, the following correction

factor must be applied to the one-minute average HCl data: 

 (Equation 17) 0 C y b i

C = −

18 
5.0 OTHER REQUIREMENTS AND INFORMATION 

5.1 DATA RECORDING AND REPORTING

In general, all data and records associated with the HCl CEMS must be retained for 5 years. At a

minimum, at least 2 years of data and records must be readily available at the site. The balance of the

records may be retained offsite. 

HCl CEMS calibration drift tests and audit results shall be reported at intervals specified in the applicable 

regulation. 

5.2 WHAT POLLUTION PREVENTION PROCEDURES MUST I FOLLOW? [RESERVED] 

5.3 WHAT REFERENCE TABLES AND FIGURES ARE RELEVANT TO PROCEDURE DD? 

Table 1. Ongoing Quality Assurance/Quality Control Calibration Gas Ranges 

Figure 1. Dynamic Spiking Overview 

 

19 
Table 1. Ongoing Quality Assurance/Quality Control Calibration Gas Ranges 

HCl Calibration Gas Concentrations a

Test Units Zero-level Mid-Level High-Level

Daily Calibration Drift % of Span 0-30 NA 50-100 

Absolute Calibration

Audit 

% of Span 0-29 30-49 50-100 

Dynamic Spiking % of Span 0-50 25-75 50-100 

a

A copy of the supplier’s certificate of analysis must be provided for each gas cylinder. Calibration gases 

do not need to be Protocol 1 gases. 

20 
Figure 1. Dynamic Spiking Apparatus 

Dynamic Spiking Apparatus

HCL

CEMS

900-1000 ppm HCL in N2

Calibration Manifold

(Mass flow meters)

Sample Probe

•Heated Filter

• Pressure Control

•Laminar Flow

Element

Heated sample line

Flue Gas

Gas

Conditioner

Reference

Measurement(s)

Heated line inside

sample umbillical

21 
6.0 RELEVANT REFERENCES 

I. Performance Specification Z: Specifications and Test Procedures for Hydrochloric Acid 

Continuous Emission Monitoring Systems at Stationary Sources. 2005. 

22