Name: Bob Ohler

Title: Plant Manager

Company: Guardian Fiberglass, Inc.

Sign Date:

Name: Randal J. Tysar

Title: Senior Environmental Engineer

Company: BT Environmental Consulting, Inc.

Sign Date:

Permitted Facility Representative / Date Testing Company Representative / Date

Guardian Fiberglass Winnsboro South Carolina Residual Risk Data Collection Emissions Test

Final Test Report 

for

/ /

Based on information and belief formed 

after reasonable inquiry, I certify that the 

statements and information in this test 

report are true, accurate, and complete.

I have reviewed all testing details and 

results in this test report and hereby 

certify that the test report is authentic 

and accurate.

Wednesday, November 24, 2010 Page 1 of 15
Guardian Fiberglass Winnsboro South Carolina Residual Risk Data Collection Emissions Test

Facility Information:

Guardian Fiberglass; Winnsboro, South Car

1 Guardian Way

Winnsboro SC 29180

Contact: Jim Feeney

Phone: 803-712-6321

Fax: 803-635-8008

Email: jamesfeeney@bp.guardian.com

Testing Company:

BT Environmental Consulting INC

2615 Wolcott Ave

Ferndale MI 48220

Contact:Barry P. Boulianne

Phone: (248) 548-8070

Fax: (248) 548-8083

Email: bboulianne@sbcglobal.net

Air Permit Number:

1000-031-CA

Permitted Source ID/Name:

Guardian Fiberglass, Inc. - Winnsboro

Permitted Maximum Process Rate:

4.5 tph

Max. Normal Operation Process Rate:

4.0 tph

Target Process Test Rat

2 to 3 tph

Description of the source (including control equipment). Please see the attachments for source or

 process flow diagram:

Mixed batch is fed to an electric melter furnace and melted to be formed into fiberglass.

Industry/SCC/NAIS 327993

Sampling Location Information:

Test Parameter Information:

AFS #: 4503900031 FRS #: 110037142678

SCC / Description 30501203 Industrial Processes - Mineral Products - Fiberglass Manufacturing - Electric 

Furnace (Wool-type Fiber) 

The following state and federal regulations that apply to the proposed testing:

Location

Round

Duct

Diam.

Distance from 

upstream dist.

Distance from

downstream dist.

Number of

Traverse Ports

Equiv. 

Diam

Min.Travers 

Points

Rect. Duct 

Length /Width

Melter 

Furnace 

Baghouse

41.5 0 0 2 0 0 0 24

Target

Parameter 

Test

Method

Number of

Test Runs

Test Run

Duration

Sample 

Location Points Comments

Melter Chromium M0061 3 192 0 Hexavalent Chromium

Furnace 

Baghouse

Melter Hydrogen Fluoride Method 26A 3 60 0

Furnace 

Baghouse

Wednesday, November 24, 2010 Page 2 of 15
Guardian Fiberglass Winnsboro South Carolina Residual Risk Data Collection Emissions Test

The following describes any modifications and/or deviations to the applicable test 

methods. If alternative methods were requested, see the attachments for documentation 

of request AND approval, including dates.

Melter Hydrogen Chloride Method 26A 3 60 0

Furnace 

Baghouse

Melter Total PM2.5 OTM - 27/28 3 246 0

Furnace 

Baghouse

Organic Condensible 

Particulate

Melter OTM - 27/28 3 246 0

Furnace 

Baghouse

Inorganic (Aqueous) 

Condensible Part.

Melter OTM - 27/28 3 246 0

Furnace 

Baghouse

Melter Filterable PM2.5 OTM - 27/28 3 246 0

Furnace 

Baghouse

Melter Antimony Method 29 3 192 0

Furnace 

Baghouse

Melter Filterable Particulate Method 29 3 192 0

Furnace 

Baghouse

Melter Beryllium Method 29 3 192 0

Furnace 

Baghouse

Melter Arsenic Method 29 3 192 0

Furnace 

Baghouse

Melter Cadmium Method 29 3 192 0

Furnace 

Baghouse

Melter Cobalt Method 29 3 192 0

Furnace 

Baghouse

Melter Lead Method 29 3 192 0

Furnace 

Baghouse

Melter Manganese Method 29 3 192 0

Furnace 

Baghouse

Melter Nickel Method 29 3 192 0

Furnace 

Baghouse

Melter Selenium Method 29 3 192 0

Furnace 

Baghouse

Melter Chromium Method 29 3 192 0

Furnace 

Baghouse

Wednesday, November 24, 2010 Page 3 of 15
Guardian Fiberglass Winnsboro South Carolina Residual Risk Data Collection Emissions Test

Sampling / Stack Data Results Summary

Location Melter Furnace Baghouse - M0061

Average

Run Number 1 2 3 

Test Date 9/14/2010 9/14/2010 9/15/2010 

Run Start Time 7:44:00 AM 11:56:00 

AM

7:36:00 AM 

Run Finish Time 11:02:00 AM 3:22:00 PM 10:57:00 AM 

Net Run Time, minutes 192 192 192 

Dry Gas Meter Volume Sampled, dscf 124.652 123.945 125.993 124.863

Moisture Content of Stack Gas, % 4.20 3.69 4.60 4.163

Moisture Saturation at Stack Gas Temperature, % 13.27 17.57 12.85 14.563

Carbon Dioxide, % 0.6 0.8 0.6 0.667

Oxygen, % 20.5 20.5 20.5 20.500

Average Stack Gas Temperature, °F 125.35 135.90 124.27 128.507

Dry Volumetric Flow Rate, dry scfm 9,259.2 9,230.6 9,352.0 9,280.600

Actual Wet Volumetric Flue Gas Flow Rate, acfm 10,668.6 10,770.0 10,775.6 10,738.067

Percent Isokinetic of Sampling Rate, % 102.0 101.8 102.1 101.967

F-Factor, dscfm/MMBtu @ %O2 0 0 0 0.000

Fw 0 0 0 0.000

Fc 0 0 0 0.000

Wednesday, November 24, 2010 Page 4 of 15
Guardian Fiberglass Winnsboro South Carolina Residual Risk Data Collection Emissions Test

Location Melter Furnace Baghouse - Method 26A

Average

Run Number 1 2 3 

Test Date 9/14/2010 9/15/2010 9/15/2010 

Run Start Time 4:05:00 PM 11:32:00 

AM

12:47:00 PM 

Run Finish Time 5:05:00 PM 12:34:00 

PM

2:51:00 PM 

Net Run Time, minutes 60 60 60 

Dry Gas Meter Volume Sampled, dscf 48.281 47.737 47.342 47.787

Moisture Content of Stack Gas, % 2.29 2.69 2.61 2.530

Moisture Saturation at Stack Gas Temperature, % 16.60 14.69 16.80 16.030

Carbon Dioxide, % 0.8 1.3 1.3 1.133

Oxygen, % 20.5 20.5 20.5 20.500

Average Stack Gas Temperature, °F 133.75 129.21 134.29 132.417

Dry Volumetric Flow Rate, dry scfm 9,875.8 9,716.5 9,646.3 9,746.200

Actual Wet Volumetric Flue Gas Flow Rate, acfm 11,316.7 11,068.7 11,074.3 11,153.233

Percent Isokinetic of Sampling Rate, % 100.3 100.8 100.7 100.600

F-Factor, dscfm/MMBtu @ %O2 0 0 0 0.000

Fw 0 0 0 0.000

Fc 0 0 0 0.000

Wednesday, November 24, 2010 Page 5 of 15
Guardian Fiberglass Winnsboro South Carolina Residual Risk Data Collection Emissions Test

Location Melter Furnace Baghouse - Method 29

Average

Run Number 1 2 3 

Test Date 9/14/2010 9/14/2010 9/15/2010 

Run Start Time 7:44:00 AM 11:56:00 

AM

7:36:00 AM 

Run Finish Time 11:02:00 AM 3:22:00 PM 10:57:00 AM 

Net Run Time, minutes 192 192 192 

Dry Gas Meter Volume Sampled, dscf 156.853 151.203 157.231 155.096

Moisture Content of Stack Gas, % 2.35 2.22 2.60 2.390

Moisture Saturation at Stack Gas Temperature, % 11.91 16.02 11.60 13.177

Carbon Dioxide, % 0.6 0.8 0.6 0.667

Oxygen, % 20.5 20.5 20.5 20.500

Average Stack Gas Temperature, °F 121.42 132.38 120.54 124.780

Dry Volumetric Flow Rate, dry scfm 10,038.9 9,678.8 10,026.6 9,914.767

Actual Wet Volumetric Flue Gas Flow Rate, acfm 11,271.6 11,057.4 11,243.4 11,190.800

Percent Isokinetic of Sampling Rate, % 100.2 100.2 100.6 100.333

F-Factor, dscfm/MMBtu @ %O2 0 0 0 0.000

Fw 0 0 0 0.000

Fc 0 0 0 0.000

Wednesday, November 24, 2010 Page 6 of 15
Guardian Fiberglass Winnsboro South Carolina Residual Risk Data Collection Emissions Test

Location Melter Furnace Baghouse - OTM - 27/28

Average

Run Number 1 2 3 

Test Date 9/15/2010 9/16/2010 9/16/2010 

Run Start Time 12:09:00 PM 7:38:00 

AM

12:25:00 PM 

Run Finish Time 4:20:00 PM 11:58:00 

AM

4:29:00 PM 

Net Run Time, minutes 246.5 246.5 245 

Dry Gas Meter Volume Sampled, dscf 92.840 93.001 92.254 92.698

Moisture Content of Stack Gas, % 2.28 2.57 2.30 2.383

Moisture Saturation at Stack Gas Temperature, % 17.77 15.74 20.46 17.990

Carbon Dioxide, % 1.3 0.5 0.3 0.700

Oxygen, % 20.5 20.6 20.6 20.567

Average Stack Gas Temperature, °F 136.42 131.83 141.92 136.723

Dry Volumetric Flow Rate, dry scfm 9,600.3 9,639.3 9,528.3 9,589.300

Actual Wet Volumetric Flue Gas Flow Rate, acfm 11,023.6 11,012.4 11,040.5 11,025.500

Percent Isokinetic of Sampling Rate, % 99.9 99.7 100.7 100.100

F-Factor, dscfm/MMBtu @ %O2 0 0 0 0.000

Fw 0 0 0 0.000

Fc 0 0 0 0.000

Wednesday, November 24, 2010 Page 7 of 15
Guardian Fiberglass Winnsboro South Carolina Residual Risk Data Collection Emissions Test

Emissions Summary

Location: Melter Furnace Baghouse - M0061

Compound: Chromium

Average

RunNumber 1 2 3 

Mass_mg 0.0009 0.0004 0.0004 0.001

Elb/hr 8.84E-06 3.94E-06 3.93E-06 0.000

mg/dscm 2.55E-04 1.14E-04 1.12E-04 0.000

Location: Melter Furnace Baghouse - Method 26A

Compound: Hydrogen Chloride

Average

RunNumber 1 2 3 

Mass_mg 0.2 0.2 0.2 0.200

Elb/hr 5.41E-03 5.38E-03 5.39E-03 0.005

mg/dscm 1.46E-01 1.48E-01 1.49E-01 0.148

Compound: Hydrogen Fluoride

Average

RunNumber 1 2 3 

Mass_mg 0.2 0.2 0.2 0.200

Elb/hr 5.41E-03 5.38E-03 5.39E-03 0.005

mg/dscm 1.46E-01 1.48E-01 1.49E-01 0.148

Location: Melter Furnace Baghouse - Method 29

Compound: Antimony

Average

RunNumber 1 2 3 

Mass_mg 0.0008 0.0008 0.0008 0.001

Elb/hr 6.77E-06 6.77E-06 6.75E-06 0.000

mg/dscm 1.80E-04 1.87E-04 1.80E-04 0.000

Compound: Arsenic

Average

RunNumber 1 2 3 

Mass_mg 0.0036 0.00103 0.00091 0.002

Elb/hr 3.05E-05 8.72E-06 7.68E-06 0.000

mg/dscm 8.11E-04 2.41E-04 2.04E-04 0.000

Compound: Beryllium

Average

RunNumber 1 2 3 

Mass_mg 0.0002 0.0002 0.0002 0.000

Elb/hr 1.69E-06 1.69E-06 1.69E-06 0.000

mg/dscm 4.50E-05 4.67E-05 4.49E-05 0.000

Compound: Cadmium

Average

RunNumber 1 2 3 

Mass_mg 0.0017 0.00034 0.0002 0.001

Elb/hr 1.44E-05 2.88E-06 1.69E-06 0.000

Wednesday, November 24, 2010 Page 8 of 15
Guardian Fiberglass Winnsboro South Carolina Residual Risk Data Collection Emissions Test

mg/dscm 3.83E-04 7.94E-05 4.49E-05 0.000

Compound: Chromium

Average

RunNumber 1 2 3 

Mass_mg 0.0227 0.0066 0.00262 0.011

Elb/hr 1.92E-04 5.59E-05 2.21E-05 0.000

mg/dscm 5.11E-03 1.54E-03 5.89E-04 0.002

Compound: Cobalt

Average

RunNumber 1 2 3 

Mass_mg 0.0005 0.00077 0.00033 0.001

Elb/hr 4.23E-06 6.52E-06 2.78E-06 0.000

mg/dscm 1.13E-04 1.80E-04 7.41E-05 0.000

Compound: Filterable Particulate

Average

RunNumber 1 2 3 

Mass_mg 0.3 5.5 2.3 2.700

Elb/hr 2.54E-03 4.66E-02 1.94E-02 0.023

mg/dscm 6.76E-02 1.28E+00 5.17E-01 0.622

Compound: Lead

Average

RunNumber 1 2 3 

Mass_mg 0.0074 0.00404 0.00136 0.004

Elb/hr 6.26E-05 3.42E-05 1.15E-05 0.000

mg/dscm 1.67E-03 9.44E-04 3.06E-04 0.001

Compound: Manganese

Average

RunNumber 1 2 3 

Mass_mg 0.0034 0.0036 0.0039 0.004

Elb/hr 2.88E-05 3.05E-05 3.29E-05 0.000

mg/dscm 7.66E-04 8.41E-04 8.76E-04 0.001

Compound: Nickel

Average

RunNumber 1 2 3 

Mass_mg 0.0055 0.003 0.0075 0.005

Elb/hr 4.66E-05 2.54E-05 6.33E-05 0.000

mg/dscm 1.24E-03 7.01E-04 1.68E-03 0.001

Compound: Selenium

Average

RunNumber 1 2 3 

Mass_mg 0.0636 0.011 0.0047 0.026

Elb/hr 5.38E-04 9.31E-05 3.96E-05 0.000

mg/dscm 1.43E-02 2.57E-03 1.06E-03 0.006

Location: Melter Furnace Baghouse - OTM - 27/28

Wednesday, November 24, 2010 Page 9 of 15
Guardian Fiberglass Winnsboro South Carolina Residual Risk Data Collection Emissions Test

Compound: Filterable PM2.5

Average

RunNumber 1 2 3 

Mass_mg 5.07 0.3 1.72 2.363

Elb/hr 6.93E-02 4.11E-03 2.35E-02 0.032

mg/dscm 1.93E+00 1.14E-01 6.59E-01 0.901

Compound: Inorganic (Aqueous) Condensible P

Average

RunNumber 1 2 3 

Mass_mg 0.7 6.7 6 4.467

Elb/hr 9.57E-03 9.19E-02 8.20E-02 0.061

mg/dscm 2.66E-01 2.54E+00 2.30E+00 1.702

Compound: Organic Condensible Particulate

Average

RunNumber 1 2 3 

Mass_mg 3.8 1.4 4 3.067

Elb/hr 5.20E-02 1.92E-02 5.46E-02 0.042

mg/dscm 1.45E+00 5.32E-01 1.53E+00 1.171

Compound: Total PM2.5

Average

RunNumber 1 2 3 

Mass_mg 9.57 8.4 11.72 9.897

Elb/hr 1.31E-01 1.15E-01 1.60E-01 0.135

mg/dscm 3.64E+00 3.19E+00 4.49E+00 3.773

Wednesday, November 24, 2010 Page 10 of 15
Guardian Fiberglass Winnsboro South Carolina Residual Risk Data Collection Emissions Test

Process Run Data

APCD Run Data

Process Lab Run Data

Run Name Value UOM Comments

Target

Value

1 Material Processed tons/hr

2 Material Processed tons/hr

3 Material Processed tons/hr

4 Material Processed tons/hr

5 Material Processed tons/hr

6 Material Processed tons/hr

7 Material Processed tons/hr

Wednesday, November 24, 2010 Page 11 of 15
Guardian Fiberglass Winnsboro South Carolina Residual Risk Data Collection Emissions Test

Sampling / Stack Data Results Detail

Location Melter Furnace Baghouse - M0061

Average

Run Number 1 2 3 

Test Date 9/14/2010 9/14/2010 9/15/2010 

Run Start Time 7:44:00 AM 11:56:00 AM 7:36:00 AM 

Run Finish Time 11:02:00 AM 3:22:00 PM 10:57:00 AM 

Net Traversing Points 48 48 48 

Net Run Time, minutes 192 192 192 

Nozzle Diameter, inches 0.344 0.344 0.344 0.344

Pitot Tube Coefficient 0.84 0.84 0.84 0.840

Dry Gas Meter Calibration Factor 1.003 1.003 1.003 1.003

Barometric Pressure, inches of Mercury 30.05 30.05 30.12 30.073

Average Orifice Meter Differential, inches H2O 1.17 1.18 1.20 1.183

Dry Gas Meter Volume Sampled, cubic feet 127.990 129.970 129.760 129.240

Average Dry Gas Meter Temperature, °F 87.69 99.35 90.67 92.570

Dry Gas Meter Volume Sampled, dscf 124.652 123.945 125.993 124.863

Total Moisture Liquid collected, g 116 101 129 115.333

Volume of Water Vapor, standard cubic feet 5.46 4.75 6.07 5.427

Moisture Content of Stack Gas, % 4.20 3.69 4.60 4.163

Moisture Saturation at Stack Gas Temperature, % 13.27 17.57 12.85 14.563

Dry Mole Fraction 0.958 0.9631 0.954 0.958

Carbon Dioxide, % 0.6 0.8 0.6 0.667

Oxygen, % 20.5 20.5 20.5 20.500

Carbon Monoxide & Nitrogen, % 78.9 78.7 78.9 78.833

Fuel Factor 0.67 0.50 0.67 

Dry Molecular Weight, lb/lb-Mole 28.92 28.95 28.92 28.930

Wet Molecular weight, lb/lb-Mole 28.46 28.55 28.42 28.477

Flue Gas Static Pressure, inches of H2O -0.05 -0.05 -0.05 -0.050

Absolute Flue Gas Pressure, inches of Mercury 30.05 30.05 30.12 30.073

Average Stack Gas Temperature, °F 125.35 135.90 124.27 128.507

Average Velocity Head, inches of H2O 0.1015703825

75707

0.101983539

311833

0.1038800592

49161

0.102

Average Stack Gas Velocity, feet/second 18.93 19.11 19.12 19.053

Stack Cross-Sectional Area, square feet 9.393 9.393 9.393 9.393

Dry Volumetric Flow Rate, dry scfm 9,259.2 9,230.6 9,352.0 9,280.600

Actual Wet Volumetric Flue Gas Flow Rate, acfm 10,668.6 10,770.0 10,775.6 10,738.067

Percent Isokinetic of Sampling Rate, % 102.0 101.8 102.1 101.967

Percent Excess Air, % 6,219.7 7,406.1 6,219.7 6,615.167

F-Factor, dscfm/MMBtu @ %O2 0 0 0 0.000

Round Duct Diameter, inches 41.5 41.5 41.5 

Rectangular Duct Width, inches 0 0 0 

Rectangular Duct Length, inches 0 0 0 

Fw 0 0 0 0.000

Fc 0 0 0 0.000

Wednesday, November 24, 2010 Page 12 of 15
Guardian Fiberglass Winnsboro South Carolina Residual Risk Data Collection Emissions Test

Location Melter Furnace Baghouse - Method 26A

Average

Run Number 1 2 3 

Test Date 9/14/2010 9/15/2010 9/15/2010 

Run Start Time 4:05:00 PM 11:32:00 AM 12:47:00 PM 

Run Finish Time 5:05:00 PM 12:34:00 PM 2:51:00 PM 

Net Traversing Points 24 24 24 

Net Run Time, minutes 60 60 60 

Nozzle Diameter, inches 0.374 0.374 0.374 0.374

Pitot Tube Coefficient 0.84 0.84 0.84 0.840

Dry Gas Meter Calibration Factor 1.013 1.013 1.013 1.013

Barometric Pressure, inches of Mercury 30.05 30.12 30.12 30.097

Average Orifice Meter Differential, inches H2O 2.15 2.07 2.06 2.093

Dry Gas Meter Volume Sampled, cubic feet 50.420 49.250 49.040 49.570

Average Dry Gas Meter Temperature, °F 103.94 98.31 100.56 100.937

Dry Gas Meter Volume Sampled, dscf 48.281 47.737 47.342 47.787

Total Moisture Liquid collected, g 24 28 27 26.333

Volume of Water Vapor, standard cubic feet 1.13 1.32 1.27 1.240

Moisture Content of Stack Gas, % 2.29 2.69 2.61 2.530

Moisture Saturation at Stack Gas Temperature, % 16.60 14.69 16.80 16.030

Dry Mole Fraction 0.9771 0.9731 0.9739 0.975

Carbon Dioxide, % 0.8 1.3 1.3 1.133

Oxygen, % 20.5 20.5 20.5 20.500

Carbon Monoxide & Nitrogen, % 78.7 78.2 78.2 78.367

Fuel Factor 0.50 0.31 0.31 

Dry Molecular Weight, lb/lb-Mole 28.95 29.03 29.03 29.003

Wet Molecular weight, lb/lb-Mole 28.70 28.73 28.74 28.723

Flue Gas Static Pressure, inches of H2O -0.05 -0.05 -0.05 -0.050

Absolute Flue Gas Pressure, inches of Mercury 30.05 30.12 30.12 30.097

Average Stack Gas Temperature, °F 133.75 129.21 134.29 132.417

Average Velocity Head, inches of H2O 0.1135463940

57586

0.109867120

50658

0.1090728220

99919

0.111

Average Stack Gas Velocity, feet/second 20.08 19.64 19.65 19.790

Stack Cross-Sectional Area, square feet 9.393 9.393 9.393 9.393

Dry Volumetric Flow Rate, dry scfm 9,875.8 9,716.5 9,646.3 9,746.200

Actual Wet Volumetric Flue Gas Flow Rate, acfm 11,316.7 11,068.7 11,074.3 11,153.233

Percent Isokinetic of Sampling Rate, % 100.3 100.8 100.7 100.600

Percent Excess Air, % 7,406.1 14,157.5 14,157.5 11,907.033

F-Factor, dscfm/MMBtu @ %O2 0 0 0 0.000

Round Duct Diameter, inches 41.5 41.5 41.5 

Rectangular Duct Width, inches 0 0 0 

Rectangular Duct Length, inches 0 0 0 

Fw 0 0 0 0.000

Fc 0 0 0 0.000

Wednesday, November 24, 2010 Page 13 of 15
Guardian Fiberglass Winnsboro South Carolina Residual Risk Data Collection Emissions Test

Location Melter Furnace Baghouse - Method 29

Average

Run Number 1 2 3 

Test Date 9/14/2010 9/14/2010 9/15/2010 

Run Start Time 7:44:00 AM 11:56:00 AM 7:36:00 AM 

Run Finish Time 11:02:00 AM 3:22:00 PM 10:57:00 AM 

Net Traversing Points 48 48 48 

Net Run Time, minutes 192 192 192 

Nozzle Diameter, inches 0.374 0.374 0.374 0.374

Pitot Tube Coefficient 0.84 0.84 0.84 0.840

Dry Gas Meter Calibration Factor 1.013 1.013 1.013 1.013

Barometric Pressure, inches of Mercury 30.05 30.05 30.12 30.073

Average Orifice Meter Differential, inches H2O 2.15 2.05 2.16 2.120

Dry Gas Meter Volume Sampled, cubic feet 159.640 157.540 160.430 159.203

Average Dry Gas Meter Temperature, °F 89.61 102.51 92.29 94.803

Dry Gas Meter Volume Sampled, dscf 156.853 151.203 157.231 155.096

Total Moisture Liquid collected, g 80 73 89 80.667

Volume of Water Vapor, standard cubic feet 3.77 3.44 4.19 3.800

Moisture Content of Stack Gas, % 2.35 2.22 2.60 2.390

Moisture Saturation at Stack Gas Temperature, % 11.91 16.02 11.60 13.177

Dry Mole Fraction 0.9765 0.9778 0.974 0.976

Carbon Dioxide, % 0.6 0.8 0.6 0.667

Oxygen, % 20.5 20.5 20.5 20.500

Carbon Monoxide & Nitrogen, % 78.9 78.7 78.9 78.833

Fuel Factor 0.67 0.50 0.67 

Dry Molecular Weight, lb/lb-Mole 28.92 28.95 28.92 28.930

Wet Molecular weight, lb/lb-Mole 28.66 28.71 28.64 28.670

Flue Gas Static Pressure, inches of H2O -0.05 -0.05 -0.05 -0.050

Absolute Flue Gas Pressure, inches of Mercury 30.05 30.05 30.12 30.073

Average Stack Gas Temperature, °F 121.42 132.38 120.54 124.780

Average Velocity Head, inches of H2O 0.1148557369

7018

0.108667316

744563

0.1147179952

48277

0.113

Average Stack Gas Velocity, feet/second 20.00 19.62 19.95 19.857

Stack Cross-Sectional Area, square feet 9.393 9.393 9.393 9.393

Dry Volumetric Flow Rate, dry scfm 10,038.9 9,678.8 10,026.6 9,914.767

Actual Wet Volumetric Flue Gas Flow Rate, acfm 11,271.6 11,057.4 11,243.4 11,190.800

Percent Isokinetic of Sampling Rate, % 100.2 100.2 100.6 100.333

Percent Excess Air, % 6,219.7 7,406.1 6,219.7 6,615.167

F-Factor, dscfm/MMBtu @ %O2 0 0 0 0.000

Round Duct Diameter, inches 41.5 41.5 41.5 

Rectangular Duct Width, inches 0 0 0 

Rectangular Duct Length, inches 0 0 0 

Fw 0 0 0 0.000

Fc 0 0 0 0.000

Wednesday, November 24, 2010 Page 14 of 15
Guardian Fiberglass Winnsboro South Carolina Residual Risk Data Collection Emissions Test

Location Melter Furnace Baghouse - OTM - 27/28

Average

Run Number 1 2 3 

Test Date 9/15/2010 9/16/2010 9/16/2010 

Run Start Time 12:09:00 PM 7:38:00 AM 12:25:00 PM 

Run Finish Time 4:20:00 PM 11:58:00 AM 4:29:00 PM 

Net Traversing Points 24 24 24 

Net Run Time, minutes 246.5 246.5 245 

Nozzle Diameter, inches 0.26 0.26 0.26 0.260

Pitot Tube Coefficient 0.84 0.84 0.84 0.840

Dry Gas Meter Calibration Factor 1.003 1.003 1.003 1.003

Barometric Pressure, inches of Mercury 30.12 30.13 30.13 30.127

Average Orifice Meter Differential, inches H2O 0.40 0.39 0.39 0.393

Dry Gas Meter Volume Sampled, cubic feet 97.690 96.080 97.100 96.957

Average Dry Gas Meter Temperature, °F 101.52 91.48 101.85 98.283

Dry Gas Meter Volume Sampled, dscf 92.840 93.001 92.254 92.698

Total Moisture Liquid collected, g 46 52 46 48.000

Volume of Water Vapor, standard cubic feet 2.17 2.45 2.17 2.263

Moisture Content of Stack Gas, % 2.28 2.57 2.30 2.383

Moisture Saturation at Stack Gas Temperature, % 17.77 15.74 20.46 17.990

Dry Mole Fraction 0.9772 0.9743 0.977 0.976

Carbon Dioxide, % 1.3 0.5 0.3 0.700

Oxygen, % 20.5 20.6 20.6 20.567

Carbon Monoxide & Nitrogen, % 78.2 78.9 79.1 78.733

Fuel Factor 0.31 0.60 1.00 

Dry Molecular Weight, lb/lb-Mole 29.03 28.90 28.87 28.933

Wet Molecular weight, lb/lb-Mole 28.78 28.62 28.62 28.673

Flue Gas Static Pressure, inches of H2O -0.05 -0.05 -0.05 -0.050

Absolute Flue Gas Pressure, inches of Mercury 30.12 30.13 30.13 30.127

Average Stack Gas Temperature, °F 136.42 131.83 141.92 136.723

Average Velocity Head, inches of H2O 0.1078585715

97794

0.107858571

597794

0.1066137265

8534

0.107

Average Stack Gas Velocity, feet/second 19.56 19.54 19.59 19.563

Stack Cross-Sectional Area, square feet 9.393 9.393 9.393 9.393

Dry Volumetric Flow Rate, dry scfm 9,600.3 9,639.3 9,528.3 9,589.300

Actual Wet Volumetric Flue Gas Flow Rate, acfm 11,023.6 11,012.4 11,040.5 11,025.500

Percent Isokinetic of Sampling Rate, % 99.9 99.7 100.7 100.100

Percent Excess Air, % 14,157.5 8,972.1 7,294.6 10,141.400

F-Factor, dscfm/MMBtu @ %O2 0 0 0 0.000

Round Duct Diameter, inches 41.5 41.5 41.5 

Rectangular Duct Width, inches 0 0 0 

Rectangular Duct Length, inches 0 0 0 

Fw 0 0 0 0.000

Fc 0 0 0 0.000

Wednesday, November 24, 2010 Page 15 of 15