 GLP10-01-02, Interim Report 31: Analysis of PFBS, PFHS, and PFOS in Groundwater 

Samples from New On-Site Wells - Adjacent to the 3M Decatur Wastewater Treatment Plant 

(WWTP) April 2012

Study Title

Analysis of Perfluorooctane Sulfonate (PFOS), Perfluorohexane Sulfonate (PFHS) and 

Perfluorobutane Sulfonate (PFBS) in Groundwater, Soil and Sediment for the 3M Decatur Phase 3 

Site-Related Monitoring Program 

Data Requirement 

EPA TSCA Good Laboratory Practice Standards 40 CFR Part 792

Study Director

Jaisimha Kesari P.E., DEE

Weston Solutions, Inc.

1400 Weston Way

West Chester, PA 19380

Phone: 610-701-3761

Author

Susan Wolf

3M Environmental Laboratory

Interim Report Completion Date 

Date of signing

Performing Laboratory

3M Environmental Health and Safety Operations

Environmental Laboratory

3M Center, Bldg 260-05-N-17

St. Paul, MN 55144

Project Identification

GLP10-01-02-31 

 

Total Number of Pages

89

Testing Cert #2052.01

The testing reported herein meet the requirements of ANSI/ISO/IEC 17025:2005 “General Requirements for 

the Competence of Testing and Calibration Laboratories”, in accordance with the A2LA Testing Certificate # 

2052.01. Testing that complies with this International Standard also meets principles of ISO 9001:2000. 
This page has been reserved for specific country requirements. 

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 2 of 89
GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS In Groundwater 

New On-site VWVTP Wells - April 2012 

GLP COMPLIANCE STATEMENT 

Report Title: GLP10-01-02, Interim Report 31: Analysis of PFBS, PFHS, and PFOS in Groundwater 

Samples from New On-Site Wells - Adjacent to the 3M Decatur Wastewater Treatment Plant (WWTP) 

April 2012. 

Study: Analysis of Perfluorooctane Sulfonate (PFOS), Perfluorohexane Sulfonate (PFHS) and 

Perfluorobutane Sulfonate (PFBS) in Groundwater, Soil and Sediment for the 3M Decatur Phase 3 

Site-Related Monitoring Program. 

This analytical phase was conducted in compliance with Toxic Substances Control Act (TSCA) Good 

Laboratory Practice (GLP) Standards, 40 CFR 792, with the exceptions listed below: 

• These are environmental samples where there is no specific test substance, no specific test 

system and no dosing of a test system. 

• The reference substances have not been characterized under the GLPs and the stability under 

storage conditions at the test site have not been detemnined under GLPs. 

Gary Hohehsteirf, Sponsor Representative Date 

Jaisimha Kesari, P.E., DEE, Study Director ' Date 

Page 3 of 89 
GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS In Groundwater 

New On-Site WWTP Wells - April 2012 

QUALITY ASSURANCE STATEMENT 

Report Title: GLP10-01-02, Interim Report 31: Analysis of PFBS, PFHS, and PFOS in Groundwater 

Samples from New On-Site Wells - Adjacent to the 3M Decatur Wastewater Treatment Plant (WWTP) 

April 2012. 

Study: Analysis of Perfluorooctane Sulfonate (PFOS), Perfluorohexane Sulfonate (PFHS) and 

Perfluorobutane Sulfonate (PFBS) in Groundwater, Soil and Sediment for the 3M Decatur Phase 3 

Site-Related Monitoring Program. 

This analytical phase was audited by the 3M Environmental Laboratory Quality Assurance Unit (QAU), 

as indicated in the following table. The findings were reported to the principal investigator (P.I.), 

laboratory management and study director. 

Inspection Dates Phase 

Date Re ported to 

Inspection Dates Phase Testing Facility 

Management 

Study Director 

4/26-27/12, 5/1/12 Data and Report 5/4/12 5/4/12 

QAU Representative 

Page 4 of 89 
TABLE OF CONTENTS 

GLP Compliance Statement...................................................................................................................... 3

Quality Assurance Statement.................................................................................................................... 4

Table of Contents....................................................................................................................................... 5

List of Tables .............................................................................................................................................. 6

1 Study Information................................................................................................................................ 7

2 Summary ............................................................................................................................................. 8

3 Introduction.......................................................................................................................................... 8

4 Test & Control Substances ................................................................................................................ 9

5 Reference Substances .....................................................................................................................10

6 Test System ......................................................................................................................................11

7 Method Summary .............................................................................................................................11

7.1 Method ..............................................................................................................................11

7.2 Sample Collection.............................................................................................................11

7.3 Sample Preparation..........................................................................................................11

7.4 Analysis.............................................................................................................................12

8 Analytical Results..............................................................................................................................13

8.1 Calibration .........................................................................................................................13

8.2 System Suitability .............................................................................................................13

8.3 Limit of Quantitation (LOQ)..............................................................................................13

8.4 Continuing Calibration ......................................................................................................13

8.5 Blanks................................................................................................................................14

8.6 Lab Control Spikes (LCSs) ..............................................................................................14

8.7 Analytical Method Uncertainty .........................................................................................17

8.8 Field Matrix Spikes (FMS)................................................................................................17

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 5 of 89
8.9 Lab Matrix Spikes (LMS)..................................................................................................17

9 Data Summary and Discussion .......................................................................................................18

10 Conclusion.........................................................................................................................................23

11 Data/Sample Retention ....................................................................................................................23

12 Attachments ......................................................................................................................................23

13 Signatures .........................................................................................................................................24

LIST OF TABLES

Table 1. Summarized PFBS, PFHS, and PFOS Results (Residuum Monitoring Wells)................. 8

Table 2. Sample Description Key Code............................................................................................ 11

Table 3. Instrument Parameters........................................................................................................ 12

Table 4. Liquid Chromatography Conditions. ................................................................................... 12

Table 5. Mass Transitions.................................................................................................................. 12

Table 6. Limit of Quantitation (LOQ). ................................................................................................ 13

Table 7. Laboratory Control Spike Recovery.................................................................................... 14

Table 8. Analytical Uncertainty. ......................................................................................................... 17

Table 9. Field Matrix Spike Levels. ................................................................................................... 17

Table 10. DAL GW TW1R 120405 .................................................................................................... 19

Table 11. DAL GW TW2R 120405 .................................................................................................... 19

Table 12. DAL GW TW3R 120405 .................................................................................................... 20

Table 13. DAL GW TW4R 120405 .................................................................................................... 20

Table 14. DAL GW TW5R 120405 ................................................................................................... 21

Table 15. DAL GW TW6R 120405 .................................................................................................... 21

Table 16. DAL GW TRIP01 120405 .................................................................................................. 22

Table 17. Rinseate Blank.................................................................................................................... 22

 

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 6 of 89
1 Study Information

Sponsor

3M Company

Sponsor Representative

Gary Hohenstein

3M EHS Operations

3M Building 224-5W-03

Saint Paul, MN 55144-1000

Phone: (651) 737-3570

Study Director

Jaisimha Kesari, P.E., DEE

Weston Solutions, Inc.

West Chester, PA 19380

Phone: (610) 701-3761

Fax: (610) 701-7401

j.kesari@westonsolutions.com

Study Location

Testing Facility

3M EHS Operations

3M Environmental Laboratory

Building 260-5N-17

St. Paul, MN 55144

Study Personnel

William K. Reagen, Ph.D., 3M Laboratory Manager

Cleston Lange, Ph.D., Principal Analytical Investigator, (clange@mmm.com); phone (651)-733-9860

Susan Wolf, 3M Analyst

Chelsie Grochow, Analyst

Kelly Ukes, Analyst

Study Dates

Study Initiation: March 8, 2010

Interim 31 Experimental Termination: April 28, 2012

Interim Report Completion: Date of Interim Report Signing

Location of Archives

All original raw data and the analytical report have been archived at the 3M Environmental 

Laboratory according to 40 CFR Part 792. The test substance and analytical reference standard 

reserve samples are archived at the 3M Environmental Laboratory according to 40 CFR Part 792

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 7 of 89
2 Summary

The 3M Environmental Laboratory received groundwater samples from six new on-site temporary 

residuum monitoring wells in the vicinity of the 3M Decatur facility Wastewater Treatment Plant 

(WWTP) in Decatur, AL. A total of twenty-eight sample bottles were received at the 3M Environmental 

Laboratory for perfluorooctane sulfonate (PFOS), perfluorohexane sulfonate (PFHS) and 

perfluorobutane sulfonate (PFBS), and included duplicate groundwater samples and two field matrix 

spike (FMS) samples from each sampling location. Samples also included one trip blank containing 

Milli-QTM water and appropriate trip blank spikes and one equipment rinseate blank. The equipment 

rinseate blank did not have FMS samples prepared for determination of PFOS, PFHS and PFBS 

recovery. All samples were logged into the laboratory information management system (LIMS) under 

project GLP10-01-02-31. The groundwater samples and trip blanks were received from Weston 

personnel on April 9, 2012. All of the samples were prepared and analyzed for PFBS, PFHS, and 

PFOS following 3M Environmental Laboratory Method ETS-8-044.1 “Method of Analysis for the 

Determination of Perfluorinated Compounds in Water by LC/MS/MS; Direct Injection Analysis”. 

The average measured PFBS, PFHS, and PFOS concentrations are summarized in Table 1. The trip 

blank sample was below the lower limit of quantitation (LLOQ) for all analytes, indicating adequate 

control of sample contamination during shipping and sample collections. The analytical method 

uncertainties associated with the reported results are: PFBS + 12%, PFHS + 12% and PFOS + 35%.

Table 1. Summarized PFBS, PFHS, and PFOS Results (Residuum Monitoring Wells, April 

2012).

PFBS PFHS PFOS

Sampling Location Avg. Conc. 

(ng/mL) ± RPD

Avg. Conc. 

(ng/mL) ± RPD

Avg. Conc. 

(ng/mL) ± RPD

DAL GW TW1R 120405 33.8 ± 0.30% 57.3 ± 2.1% 29.0 ± 9.0% 

DAL GW TW2R 120405 117 ± 5.1% 579 ± 5.5% 1090 ± 17%

DAL GW TW3R 120405 304 ± 7.6% 1310 ± 15% 3660 ± 17%

DAL GW TW4R 120405 184 ± 4.9% 19.1 ± 1.0% 247 ± 2.4%

DAL GW TW5R 120405 522 ± 4.4% 43.0 ± 1.6% 696 ± 2.3%

DAL GW TW6R 120405 156 ± 0.64% 46.7 ± 5.6% 281 ± 4.6%

DAL GW TW3R Rinseate Blank <0.250 <0.250 <0.232

DAL-GW-TRIP01 Blank <0.250 <0.250 <0.232

The analytical method uncertainties associated with the reported results are: PFBS ± 12%, PFHS ± 12%, and 

PFOS ± 35%.

3 Introduction 

This analytical study was conducted as part of the Phase 3 Environmental Monitoring and Assessment 

Program for the 3M facility located in Decatur, Alabama. The objective of the overall program is to gain 

information regarding concentrations of perfluorooctane sulfonate (PFOS), perfluorohexane sulfonate 

(PFHS) and perfluorobutane sulfonate (PFBS), in various environmental media such as groundwater, 

soils and sediments that are associated with and near the Decatur facility. This analytical study was 

conducted to analyze groundwater samples collected from six new temporary on-site residuum 

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 8 of 89
monitoring wells located adjacent to the 3M Decatur WWTP for PFOS, PFHS and PFBS in an effort to 

characterize regional groundwater conditions. 

The 3M Environmental Laboratory prepared sample containers (250 mL high-density polyethylene 

bottles) which were shipped to Decatur, AL Weston personnel prior to field sampling. Sample 

containers for each sampling location included a field sample, field sample duplicate, and two field 

matrix spike samples. Each empty container was marked with a “fill to here” line to produce a final 

sample volume of 200 mL. Containers designated for field matrix samples were fortified with an 

appropriate matrix spike solution containing PFBS (linear isomer), PFHS (linear isomer), and PFOS 

(linear and branched isomers) prior to being sent to the field for sample collection. All sample bottles 

included the addition of 18O2-PFBS, 13C3-PFHS, and 13C8-PFOS (internal standard) at a nominal 

concentration of 1 ng/mL and 13C4-PFOS (surrogate recovery standard) at a nominal concentration of 

0.1 ng/mL. Due to the level of PFBS, PFHS, and PFOS detected in the samples, neither the internal 

standards nor the surrogate recovery standard was utilized during sample analysis. Laboratory matrix 

spikes were prepared for sampling locations TW2, TW3, TW4, TW5, and TW6 since the field matrix 

spike levels were not appropriate for these sampling locations for at least one of the target analytes. 

See section 8.9 of the report for laboratory matrix spike levels. See section 8.8 of the report for field 

matrix spike levels. See section 8.9 of the report for laboratory matrix spike levels.

Samples were prepared and analyzed according to the procedure defined in 3M Environmental 

Laboratory method ETS-8-044.1 “Method of Analysis for the Determination of Perfluorinated 

Compounds In Water by LC/MS/MS; Direct Injection Analysis”.

Table 1 summarizes the average PFBS, PFHS, and PFOS concentrations for the duplicate surface 

water samples collected and the trip blank sample. Tables 10-17 summarize the individual sample 

results and the associated field matrix spike recoveries. All results for the quality control samples 

prepared and analyzed with the samples are reported and discussed elsewhere in this report.

4 Test & Control Substances

There was not a test substance or control substances in the classic sense of a GLP study. This study 

was purely analytical in nature.

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 9 of 89
5 Reference Substances

Reference 

Substance

PFBS 

(Linear)

PFHS 

(Linear)

Chemical Name Potassium Perfluorobutane 

sulfonate

Sodium Perfluorohexane 

sulfonate

Chemical Formula C4F9SO3

-

K+ C6F13SO3

-

Na+

Identifier NA L-PFHXS

Use Target Analyte Reference 

Standard

Target Analyte 

Reference Standard

Source 3M Wellington

Expiration Date 01/10/2017 03/25/2018

Storage Conditions Frozen Frozen

Chemical Lot 

Number 41-2600-8442-5 LPFHxSAM08

TCR Number TCR-121 TCR08-0018

Physical Description White Powder Crystalline

Purity 96.7% 100%

Reference 

Substance

PFOS

(Linear + Branched)

PFOS

(Linear + Branched)

Chemical Name Potassium Perfluorooctane 

sulfonate

Potassium 

Perfluorooctane sulfonate

Chemical Formula C8F17SO3

-

K+ C8F17SO3

-

K+

Identifier Br-PFOSK CAS # 2795-39-3 

Use Target Analyte Reference 

Standard FMS Reference Standard

Source Wellington Sigma-Aldrich

Expiration Date 03/17/2014; 12/01/2014 02/04/2014

Storage Conditions Frozen Ambient

Chemical Lot 

Number

brPFOSK0708; 

brPFOSK1111 1424328V

TCR Number TCR11-0010; TCR11-0041 TCR11-0028

Physical Description Liquid White Powder

Purity 99.9% 99.7%

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 10 of 89
6 Test System

The test systems for this study are groundwater samples collected from wells located in Decatur, AL by 

Weston Solutions, Inc. personnel. Samples for this study are “real world” samples, not dosed with a 

specific lot of test substance.

Table 2. Sample Description Key Code.

String Number String Descriptor Example

Example DAL-GW-TW1R-0-120405

1 Sample Location DAL= Decatur, Alabama

2 Sample Type GW = Ground Water

3 Well ID Example: TW1R

4 Sample Type 0=primary sample 

DB=duplicate sample

LS = low spike

HS = high spike

5 Sampling Date 120405– April 5, 2012

7 Method Summary

7.1 Method

Analysis for all analytes was completed following 3M Environmental Laboratory method ETS-8-044.1 

“Method of Analysis for the Determination of Perfluorinated Compounds In Water by High Performance 

Liquid Chromatography/Mass Spectrometry Direct Injection Analysis”. 

7.2 Sample Collection

Samples were collected in 250 mL NalgeneTM (high-density polyethylene) bottles prepared at the 3M 

Environmental Laboratory. Sample bottles associated with GLP10-01-02-31 were returned to the 

laboratory at ambient conditions on April 9, 2012. Samples were stored refrigerated at the laboratory 

after receipt. A set of laboratory prepared Trip Blank and Trip Blank field matrix spikes were sent with 

the sample collection bottles.

7.3 Sample Preparation

All samples were initially diluted 1:10 by diluting 1 mL of a well mixed sample with 9 mL of Milli Q water 

and analyzed on 4/13/12. 

Sampling locations TW2, TW3, TW4, TW5, and TW6 were re-analyzed and included the preparation of 

a laboraotry matrix spike with each sample set. Location TW2R was diluted 1:50 by diluting 0.2 mL of 

the well mixed sample with 9.8 mL of Milli Q water and included a 1000 ppb LMS for PFHS and PFOS. 

Location TW3R was diluted 1:100 by diluting 0.1 mL of the well mixed sample with 9.9 mL of Milli Q 

water and included a 2000 ppb LMS and 5000ppb LMS for PFBS, PFHS, and PFOS. Locations TW4R, 

TW5R and TW6R were diluted 1:10 by diluting 1.0 mL of the well mixed sample with 9.0 mL of Milli Q 

water. Location TW4R included a 200 ppb LMS for PFOS. Location TW5R included a 500 ppb LMS for 

PFBS and PFOS. Location TW6R included a 200 ppb LMS for PFOS.

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 11 of 89
7.4 Analysis

All study samples and quality control samples were analyzed for PFBS, PFHS, and PFOS using high 

performance liquid chromatography/ tandem mass spectrometry (HPLC/MS/MS). Detailed instrument 

parameters, the liquid chromatography gradient program, and the specific mass transitions analyzed are 

described in the raw data hard copies placed in the final data packet, and are briefly described below.

Table 3. Instrument Parameters.

Instrument Name ETS Ginger ETS Buster

Analytical Method Followed ETS-8-044.1 ETS-8-044.1

Analysis Date 4/13/12 4/16/12, 4/27/12

Liquid Chromatograph Agilent 1100 Agilent 1100

Guard column Betasil C18 (4.6 mm X 100 mm), 5μ Betasil C18 (4.6 mm X 100 mm), 5μ

Analytical column Betasil C18 (4.6 mm X 100 mm), 5μ Betasil C18 (4.6 mm X 100 mm), 5μ

Injection Volume 2 or 10 μL 2 or 10 μL

Mass Spectrometer Applied Biosystems API 5000 Applied Biosystems API 4000

Ion Source Turbo Spray Turbo Spray

Electrode Turbo ion electrode Turbo ion electrode

Polarity Negative Negative

Software Analyst 1.4.2 Analyst 1.4.2

Table 4. Liquid Chromatography Conditions.

Step 

Number

Total Time 

(min)

Flow Rate 

(μL/min)

Percent A

(2 mM ammonium acetate)

Percent B

(methanol)

ETS-8-044.1

0 0.0 750 97.0 3.0

1 2.0 750 97.0 3.0

2 14.5 750 5.0 95.0

3 15.5 750 5.0 95.0

4 16.5 750 97.0 3.0

5 20.0 750 97.0 3.0

Table 5. Mass Transitions. 

Analyte

Mass Transition

Q1/Q3

Reference Material 

Structure

PFBS

299/80

Linear

299/99

PFHS

399/80

Linear

399/99

PFOS

499/80

499/99 Linear + Branched

499/130

Dwell time was 50 or 75 msec for each transition. The individual transitions were 

summed to produce a “total ion chromatogram” (TIC), which was used for quantitation. 

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 12 of 89
8 Analytical Results

8.1 Calibration

Samples were analyzed against an external standard calibration curve. Calibration standards were 

prepared by spiking known amounts of the stock solution containing PFBS, PFHS, PFOS (reference 

standard for PFOS containing both linear and branched isomers)into Milli Q water. Standards ranging 

from 0.0250 ng/mL to 100 ng/mL (nominal) were analyzed. Low and or high curve points may have 

been disabled to meet method criteria. A quadratic, 1/x weighted, calibration curve of the peak area 

counts was used to fit the data for each analyte. The data were not forced through zero during the fitting 

process. Calculating the standard concentrations using the peak area confirmed accuracy of each curve 

point. Each curve point was quantitated using the overall calibration curve and reviewed for accuracy. 

Method calibration accuracy requirements of 100±25% (100±30% for the lowest curve point) were met 

for all analytes. The correlation coefficient (r) was greater than 0.995 for all analytes. Each curve point 

was quantitated using the overall calibration curve and reviewed for accuracy. Method calibration 

accuracy requirements of 100±25% (100±30% for the lowest curve point) were met for all analytes. The 

correlation coefficient (r) was greater than 0.995 for PFBS, PFHS and PFOS for each analysis. 

8.2 System Suitability

A calibration standard was analyzed four times at the beginning of each analytical sequence to 

demonstrate overall system suitability. The acceptance criteria of less than or equal to 5% relative 

standard deviation (RSD) for peak area and retention time criteria of less than or equal to 2% RSD was 

met for PFBS, PFHS and PFOS for each analysis.

8.3 Limit of Quantitation (LOQ)

The LOQ for this analysis is the lowest non-zero calibration standard in the curve that meets linearity 

and accuracy requirements and for which the area counts or area ratio are at least twice those of the 

appropriate blanks. The LOQ for all analytes can be found in Table 6.

Table 6. Limit of Quantitation (LOQ).

Analysis Date Dilution 

Factor

PFBS

LOQ, ng/mL

PFHS

LOQ, ng/mL

PFOS

LOQ, ng/mL

4/13/12

10 0.250 0.250 NA

50 1.25 1.25 NA

4/16/12

10 0.250 0.250 0.232

50 1.25 1.25 1.16

100 2.50 2.50 2.32

4/27/12

10 10.0 9.98 9.28

50 50.0 49.9 46.4

100 100 99.8 92.8

250 250 250 232

NA = Not Applicable

8.4 Continuing Calibration 

During the course of each analytical sequence, continuing calibration verification samples (CCVs) were 

analyzed to confirm that the instrument response and the initial calibration curve were still in control. All 

CCVs met method criteria of 100% ± 25% for PFBS, PFHS and PFOS for each analysis. 

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 13 of 89
8.5 Blanks

Three types of blanks were prepared and analyzed with the samples: method procedural blanks, a trip 

blank, and an equipment rinseate blank. Method procedural blank results were reviewed and used to 

evaluate method performance and to determine the LOQ for PFBS, PFHS and PFOS. The trip blank 

reflects the shipping and sample collection conditions the sample bottles and samples experience. The 

equipment rinseate blank is an aqueous sample that reflect the efficiency of equipment cleaning in the 

field between different sample collections and are proof of no cross contamination of samples from the 

equipment.

8.6 Lab Control Spikes (LCSs) 

Low, mid, and high lab control spikes were prepared for the target analytes and analyzed in triplicate. 

LCSs were prepared by spiking known amounts of the analyte into Milli-Q water to produce the desired 

concentration. The spiked water samples were then prepared and analyzed in the same manner as the 

samples. The method acceptance criteria, average of LCS at each level should be within 100% ± 20% 

with an RSD ≤20%, was met for all analytes except for the low level LCSs for PFOS analyzed on 

4/16/12 which had an average recovery of 75.2%. All LCS samples were used in the determination of 

the analytical method uncertainty on section 8.7. A method deviation is included in Attachment D. 

The following calculations were used to generate data in Table 7 for laboratory control spikes.

* 100%

Spike Concentration

Calculated Concentration LCSPercent Recovery =

 

* 100%

average LCS recovery

standard deviation LCS replicates LCS% RSD =

Table 7. Laboratory Control Spike Recovery. 

ETS-8-044.1

Analyzed 4/13/12 PFBS PFHS

Lab ID

Spiked 

Concentration 

(ng/mL)

Calculated 

Concentration 

(ng/mL) %Recovery

Spiked 

Concentration 

(ng/mL)

Calculated 

Concentration 

(ng/mL) %Recovery

LCS-120411-1 0.498 0.479 96.2 0.497 0.472 95.0

LCS-120411-2 0.498 0.466 93.7 0.497 0.464 93.3

LCS-120411-3 0.498 0.476 95.6 0.497 0.469 94.3

Average ± %RSD 95.2% ± 1.4% 94.2% ± 0.91%

LCS-120411-4 4.98 4.69 94.1 4.97 4.73 95.2

LCS-120411-5 4.98 4.77 95.8 4.97 4.76 95.7

LCS-120411-6 4.98 4.71 94.5 4.97 4.69 94.4

Average ± %RSD 94.8% ± 0.94% 95.1% ± 0.69%

LCS-120411-7 29.9 NA (1) NA (1) 29.9 NA (1) NA (1)

LCS-120411-8 29.9 NA (1) NA (1) 29.9 NA (1) NA (1)

LCS-120411-9 29.9 NA (1) NA (1) 29.9 NA (1) NA (1)

Average ± %RSD NA NA

NA = Not Applicable

(1) LCS spiked concentration exceeded the calibration range.

(2) LCS did not meet acceptance criteria of 100 ± 20%.

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 14 of 89
Table 7 continued. Laboratory Control Spike Recovery. 

ETS-8-044.1

Analyzed 4/16/12 PFBS PFHS

Lab ID

Spiked 

Concentration 

(ng/mL)

Calculated 

Concentration 

(ng/mL) %Recovery

Spiked 

Concentration 

(ng/mL)

Calculated 

Concentration 

(ng/mL) %Recovery

LCS-120411-1 0.498 0.505 101 0.497 0.489 98.3

LCS-120411-2 0.498 0.511 103 0.497 0.491 98.9

LCS-120411-3 0.498 0.500 100 0.497 0.493 99.1

Average ± %RSD 101% ± 1.5% 98.8% ± 0.42%

LCS-120411-4 4.98 5.12 103 4.97 4.96 99.7

LCS-120411-5 4.98 5.19 104 4.97 4.79 96.4

LCS-120411-6 4.98 4.95 99.4 4.97 4.95 99.6

Average ± %RSD 102% ± 2.4% 98.6% ± 1.9%

LCS-120411-7 29.9 29.3 98.0 29.9 27.0 90.3

LCS-120411-8 29.9 28.8 96.3 29.9 27.7 92.5

LCS-120411-9 29.9 29.0 97.0 29.9 28.3 94.8

Average ± %RSD 97.1% ± 0.88% 92.5% ± 2.4%

ETS-8-044.1

Analyzed 4/16/12 PFOS (Linear and Branched)

Lab ID

Spiked 

Concentration 

(ng/mL)

Calculated 

Concentration 

(ng/mL) %Recovery

LCS-120411-1 0.462 0.348 75.3

LCS-120411-2 0.462 0.351 75.9

LCS-120411-3 0.462 0.344 74.5

Average ± %RSD 75.2% ± 0.93% (2)

LCS-120411-4 4.62 4.20 91.0

LCS-120411-5 4.62 4.17 90.3

LCS-120411-6 4.62 4.26 92.2

Average ± %RSD 91.2% ± 1.1%

LCS-120411-7 27.8 26.5 95.2

LCS-120411-8 27.8 25.8 92.8

LCS-120411-9 27.8 26.8 96.3

Average ± %RSD 94.8% ± 1.9%

NA = Not Applicable

(1) LCS spiked concentration exceeded the calibration range.

(2) LCS did not meet acceptance criteria of 100 ± 20%.

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 15 of 89
Table 7 continued. Laboratory Control Spike Recovery. 

ETS-8-044.1

Analyzed 4/27/12 PFBS PFHS

Lab ID

Spiked 

Concentration 

(ng/mL)

Calculated 

Concentration 

(ng/mL) %Recovery

Spiked 

Concentration 

(ng/mL)

Calculated 

Concentration 

(ng/mL) %Recovery

LCS-120426-1 4.98 4.95 99.5 4.97 4.85 97.6

LCS-120426-2 4.98 5.09 102 4.97 5.00 101

LCS-120426-3 4.98 5.30 107 4.97 5.12 103

Average ± %RSD 103% ± 3.7% 101% ± 2.7%

LCS-120426-4 20.0 18.6 93.0 19.9 17.7 88.8

LCS-120426-5 20.0 18.5 92.6 19.9 18.5 93.1

LCS-120426-6 20.0 18.2 90.8 19.9 18.8 94.6

Average ± %RSD 92.1% ± 1.3% 92.2% ± 3.3%

LCS-120426-7 49.8 50.4 101 49.7 49.1 98.8

LCS-120426-8 49.8 47.5 95.4 49.7 46.6 93.7

LCS-120426-9 49.8 47.1 94.6 49.7 47.0 94.6

Average ± %RSD 97.0% ± 3.6% 95.7% ± 2.8%

ETS-8-044.1

Analyzed 4/27/12 PFOS (Linear and Branched)

Lab ID

Spiked 

Concentration 

(ng/mL)

Calculated 

Concentration 

(ng/mL) %Recovery

LCS-120426-1 4.62 4.26 92.1

LCS-120426-2 4.62 4.02 87.0

LCS-120426-3 4.62 4.21 91.1

Average ± %RSD 90.1% ± 3.0%

LCS-120426-4 18.5 15.7 85.0

LCS-120426-5 18.5 16.6 89.6

LCS-120426-6 18.5 16.8 90.6

Average ± %RSD 88.4% ± 3.4%

LCS-120426-7 46.2 46.3 100

LCS-120426-8 46.2 43.8 94.9

LCS-120426-9 46.2 43.5 94.1

Average ± %RSD 96.3% ± 3.3%

NA = Not Applicable

(1) LCS spiked concentration exceeded the calibration range.

(2) LCS did not meet acceptance criteria of 100 ± 20%.

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 16 of 89
8.7 Analytical Method Uncertainty

Analytical uncertainty is based on historical QC data that is control charted and used to evaluate 

method accuracy and precision. The method uncertainty is calculated following ETS-12-012.2. The 

standard deviation is calculated for the set of accuracy results (in %) obtained for the QC samples. The 

expanded uncertainty is calculated by multiplying the standard deviation by a factor of 2, which 

corresponds to a confidence level of 95%. 

 

Table 8. Analytical Uncertainty.

Analyte Standard Deviation Method Uncertainty

PFBS 5.91 ±12%

PFHS 5.88 ±12%

PFOS 17.6 ±35%

8.8 Field Matrix Spikes (FMS)

Low, mid, and high field matrix spikes (FMS) were collected at each sampling point to verify that the 

analytical method is applicable to the collected matrix. Field matrix spikes were generated by adding a 

measured volume of field sample to a container spiked by the laboratory with PFBS (linear), PFHS 

(linear), and PFOS (linear and branched) prior to shipping sample containers for sample collection. 

Field matrix spike recoveries within method acceptance criteria of 100±30% confirm that “unknown” 

components in the sample matrix do not significantly interfere with the extraction and analysis of the 

analytes of interest. Field matrix spike concentrations must be 50% of the sample concentration to be 

considered an appropriate field spike. Field matrix spikes are presented in section 9 of this report. 

Table 9. Field Matrix Spike Levels. 

Sampling Location Spike Level PFBS, 

ng/mL

PFHS, 

ng/mL

PFOS, 

ng/mL

All locations and Trip Blank Low 10 9.98 9.98

High 100 99.8 99.8

*100%

 Spike Concentraton

 ( Sample Concentration of FMS Average Concentration Field: Sample Field & Sample Dup.) FMS Recovery − =

8.9 Lab Matrix Spikes (LMS)

Due to the high levels of PFBS, PFHS, and/or PFOS detected in sampling locations TW2, TW3, TW4, 

TW5, and TW6, the FMS spike levels were not appropriate for at least one of the analytes. Therefore,

laboratory matrix spikes (LMS) samples were prepared to verify that the analytical method is applicable 

to the collected matrix. Laboratory matrix spikes were generated by adding a measured volume of 

PFBS, PFHS, and PFOS (reference standard containing both linear and branched isomers) to an aliquot 

of the primary sample. Since all samples required dilution prior to sample analysis, the laboratory matrix 

spike added was based on the on-column instrument concentration.

Sampling locationTW2; the primary sample (GLP10-01-02-31-005) was diluted 1:50, to which 20 ng/mL 

(nominal) of target analytes were added for a LMS concentration of approximately 1000 ng/mL.

Sampling location TW3; the primary sample (GLP10-01-02-31-009) was diluted 1:100, to which the low 

LMS had 20 ng/mL (nominal) of target analytes added for a LMS concentration of approximately 2000

ng/mL, and the high LMS had 50 ng/mL (nominal) of target analytes added for a LMS concentration of 

approximately 5000 ng/mL. 

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 17 of 89
Sampling location TW4; the primary sample (GLP10-01-02-31-013) was diluted 1:10, to which 20 ng/mL 

(nominal) of target analytes were added for a LMS concentration of approximately 200 ng/mL.

Sampling location TW5; the primary sample (GLP10-01-02-31-017) was diluted 1:50, to which 10 ng/mL 

(nominal) of target analytes were added for a LMS concentration of approximately 500 ng/mL.

Sampling location TW6; the primary sample (GLP10-01-01-32-017) was diluted 1:10, to which 20 ng/mL 

(nominal) of target analytes were added for a LMS concentration of approximately 200 ng/mL.

Lab matrix spike recoveries within method acceptance criteria of 100±30% confirm that “unknown” 

components in the sample matrix do not significantly interfere with the extraction and analysis of the 

analytes of interest. Lab matrix spike concentrations must be 50% of the sample concentration to be 

considered an appropriate field spike. Lab matrix spikes are presented in section 9 of this report. 

The following calculation was used to calculate the lab matrix spike recovery in Section 9 of the report:

*100%

 Spike Concentraton

 ( Sample Concentration of LMS Average Concentration Field: Sample Field & Sample Dup.) LMS Recovery − =

9 Data Summary and Discussion

The tables below summarize the sample results and field or lab matrix spike recoveries for the 

sampling locations as well as the Trip Blank. Results and average values are rounded to three 

significant figures according to EPA rounding rules. Because of rounding, values may vary slightly from 

those listed in the raw data. Field matrix spike and lab matrix spike recoveries meeting the method 

acceptance criteria of ± 30%, demonstrate that the method was appropriate for the given matrix and 

their respective quantitative ranges. 

DAL GW TW1R 120405; The recovery of the high field matix spike for PFOS was 65.8%. Since the 

analytical method uncertainty was calculated at ± 35% for PFOS, no further adjustment to the analytical 

uncertainty is needed for DAL GW TW1R.

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 18 of 89
Table 10. DAL GW TW1R 120405

PFBS PFHS PFOS

3M LIMS ID Description

Concentration 

(ng/mL) %Recovery

Concentration 

(ng/mL) %Recovery

Concentration 

(ng/mL) %Recovery

GLP10-01-02-31-001 DAL-GW-TW1R-0-120405 33.8 NA 56.7 NA 27.7 NA

GLP10-01-02-31-002 DAL-GW-TW1R-DB-120405 33.7 NA 57.9 NA 30.3 NA

GLP10-01-02-31-003 DAL-GW-TW1R-LS-120405 42.8 NC 63.3 NC 35.4 NC

GLP10-01-02-31-004 DAL-GW-TW1R-HS-120405 120 86.3 147 89.9 94.7 65.8 (1) 

Average Concentration (ng/mL) ± %RPD 33.8 ng/mL ± 0.30% 57.3 ng/mL ± 2.1 29.0 ng/mL ± 9.0%

NA = Not Applicable

NC = Not Calculated; the sample concentration is greater than 2x the spike level. 

Samples analyzed using a 1:10 dilution factor

PFBS and PFHS reported from 4/13/12 analysis. PFOS results reported from 4/16/12 analysis.

(1) FMS recovery did not meet acceptance criteria.

Table 11. DAL GW TW2R 120405

PFBS PFHS PFOS

3M LIMS ID Description

Concentration 

(ng/mL) %Recovery

Concentration 

(ng/mL) %Recovery

Concentration 

(ng/mL) %Recovery

GLP10-01-02-31-005 DAL-GW-TW2R-0-120405 120 NA 563 NA 995 NA

GLP10-01-02-31-006 DAL-GW-TW2R-DB-120405 114 NA 595 NA 1180 NA

GLP10-01-02-31-007 DAL-GW-TW2R-LS-120405 125 NC NA (1) NA (1) NA (1) NA (1)

GLP10-01-02-31-008 DAL-GW-TW2R-HS-120405 217 100 NA (1) NA (1) NA (1) NA (1)

GLP10-01-02-31-005; 1000ppb LMS DAL-GW-TW2R-0-LMS NA (1) NA (1) 1520 94.5 2020 101

Average Concentration (ng/mL) ± %RPD 117 ng/mL ± 5.1% 579 ng/mL ± 5.5% 1090 ng/mL ± 17%

NA = Not Applicable

NC = Not Calculated; the sample concentration is greater than 2x the spike level. 

Samples analyzed on 4/13/12 for PFBS using a 1:10 dilution factor), and on 4/27/12 for PFHS and PFOS using a 1:50 dilution factor.

(1) Sample not analyzed for this analyte due to inappropriate spike level.

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 19 of 89
Table 12. DAL GW TW3R 120405

PFBS PFHS PFOS

3M LIMS ID Description

Concentration 

(ng/mL) %Recovery

Concentration 

(ng/mL) %Recovery

Concentration 

(ng/mL) %Recovery

GLP10-01-02-31-009 DAL-GW-TW3R-0-120405 292 NA 1210 NA 3340 NA

GLP10-01-02-31-010 DAL-GW-TW3R-DB-120405 315 NA 1410 NA 3970 NA

GLP10-01-02-31-011 DAL-GW-TW3R-LS-120405 NA (1) NA (1) NA (1) NA (1) NA (1) NA (1)

GLP10-01-02-31-012 DAL-GW-TW3R-HS-120405 NA (1) NA (1) NA (1) NA (1) NA (1) NA (1)

GLP10-01-02-31-009; 2000ppb LMS DAL-GW-TW3R-0-LMS 2280 98.8 3160 93.0 5240 85.7

GLP10-01-02-31-009; 5000ppb LMS DAL-GW-TW3R-0-LMS 4960 93.5 5940 93.2 8480 104

Average Concentration (ng/mL) ± %RPD 304 ng/mL ± 7.6% 1310 ng/mL ± 15% 3660 ng/mL ± 17%

NA = Not Applicable

Samples analyzed on 4/27/12 using a 1:100 dilution factor; 5000ppb LMS analyzed using a 1:250 dilution factor.

(1) Sample not analyzed for this analyte due to inappropriate spike level.

Table 13. DAL GW TW4R 120405

PFBS PFHS PFOS

3M LIMS ID Description

Concentration 

(ng/mL) %Recovery

Concentration 

(ng/mL) %Recovery

Concentration 

(ng/mL) %Recovery

GLP10-01-02-31-013 DAL-GW-TW4R-0-120405 179 NA 19.0 NA 244 NA

GLP10-01-02-31-014 DAL-GW-TW4R-DB-120405 188 NA 19.2 NA 250 NA

GLP10-01-02-31-015 DAL-GW-TW4R-LS-120405 NA (1) NA (1) 28.2 91.2 NA (1) NA (1)

GLP10-01-02-31-016 DAL-GW-TW4R-HS-120405 278 94.5 102 83.1 NA (1) NA (1)

GLP10-01-02-31-013; 200ppb LMS DAL-GW-TW4R-0-LMS NA (1) NA (1) NA (1) NA (1) 450 110

Average Concentration (ng/mL) ± %RPD 184 ng/mL ± 4.9% 19.1 ng/mL ± 1.0% 247 ng/mL ± 2.4%

NA = Not Applicable

Samples analyzed on 4/16/12 for PFBS using a 1:50 dilution factor and on 4/27/12 for PFHS and PFOS using a 1:10 dilution factor.

(1) Sample not analyzed for this analyte due to inappropriate spike level.

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 20 of 89
Table 14. DAL GW TW5R 120405

PFBS PFHS PFOS

3M LIMS ID Description

Concentration 

(ng/mL) %Recovery

Concentration 

(ng/mL) %Recovery

Concentration 

(ng/mL) %Recovery

GLP10-01-02-31-017 DAL-GW-TW5R-0-120405 533 NA 43.3 NA 704 NA

GLP10-01-02-31-018 DAL-GW-TW5R-DB-120405 510 NA 42.6 NA 688 NA

GLP10-01-02-31-019 DAL-GW-TW5R-LS-120405 NA (1) NA (1) 51.1 NC NA (1) NA (1)

GLP10-01-02-31-020 DAL-GW-TW5R-HS-120405 NA (1) NA (1) 136 93.2 NA (1) NA (1)

GLP10-01-02-31-017; 500ppb LMS DAL-GW-TW5R-0-LMS 988 93.3 NA (1) NA (1) 696 91.4

Average Concentration (ng/mL) ± %RPD 522 ng/mL ± 4.4% 43.0 ng/mL ± 1.6% 696 ng/mL ± 2.3%

NA = Not Applicable

NC = Not Calculated; the sample concentration is greater than 2x the spike level. 

Samples analyzed on 4/27/12 for PFBS and PFOS using a 1:50 dilution factor and on 4/13/12 for PFHS using a 1:10 dilution factor. 

(1) Sample not analyzed for this analyte due to inappropriate spike level.

Table 15. DAL GW TW6R 120405

PFBS (1) PFHS (2) PFOS (1)

3M LIMS ID Description

Concentration 

(ng/mL) %Recovery

Concentration 

(ng/mL) %Recovery

Concentration 

(ng/mL) %Recovery

GLP10-01-02-31-021 DAL-GW-TW6R-0-120405 156 NA 48.0 NA 287 NA

GLP10-01-02-31-022 DAL-GW-TW6R-DB-120405 155 NA 45.4 NA 274 NA

GLP10-01-02-31-023 DAL-GW-TW6R-LS-120405 NA (1) NA (1) 54.6 NC NA (1) NA (1)

GLP10-01-02-31-024 DAL-GW-TW6R-HS-120405 238 82.5 131 84.5 NA (1) NA (1)

GLP10-01-02-31-021; 200ppb LMS DAL-GW-TW6R-0-LMS NA (1) NA (1) NA (1) NA (1) 456 94.9

Average Concentration (ng/mL) ± %RPD 156 ng/mL ± 0.64% 46.7 ng/mL ± 5.6% 281 ng/mL ± 4.6%

NA = Not Applicable

NC = Not Calculated; the sample concentration is greater than 2x the spike level. 

Samples analyzed on 4/27/12 for PFBS and PFOS using a 1:50 dilution factor and on 4/13/12 for PFHS using a 1:10 dilution factor.

(1) Sample not analyzed for this analyte due to inappropriate spike level.

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 21 of 89
Table 16. DAL GW TRIP01 120405

PFBS PFHS PFOS

3M LIMS ID Description

Concentration 

(ng/mL) %Recovery

Concentration 

(ng/mL) %Recovery

Concentration 

(ng/mL) %Recovery

GLP10-01-02-31-026 DAL-GW-TRIP01-0-120405 <0.250 NA <0.250 NA <0.232 NA

GLP10-01-02-31-027 DAL-GW-TRIP01-LS-120405 8.51 85.1 8.60 86.2 6.96 69.7

GLP10-01-02-31-028 DAL-GW-TRIP01-HS-120405 92.0 92.0 88.4 88.6 70.1 70.2

NA = Not Applicable

Samples analyzed for PFBS and PFHS on 4/13/12 and on 4/16/12 for PFOS using a 1:10 dilution factor.

Table 17. Rinseate Blank

PFBS PFHS PFOS

3M LIMS ID Description

Concentration 

(ng/mL)

Concentration 

(ng/mL)

Concentration 

(ng/mL)

GLP10-01-02-31-028 DAL-GW-TW3R-RB-120405 <0.250 <0.250 <0.232

Samples analyzed for PFBS and PFHS on 4/13/12 and on 4/16/12 for PFOS using a 1:10 dilution factor.

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 22 of 89
10 Conclusion

Laboratory control spikes, field matrix spikes, and lab matrix spikes were used to determine the 

analytical method accuracy and precision for PFBS, PFHS, and PFOS. Analysis was successfully 

completed following 3M Environmental Laboratory method ETS-8-044.1 described herein. 

11 Data/Sample Retention

All remaining samples and associated project data (hardcopy and electronic) will be archived according 

to 3M Environmental Laboratory standard operating procedures. 

12 Attachments

Attachment A: Protocol Amendment 31 (General Project Outline)

Attachment B: Representative Chromatograms and Calibration Curves

Attachment C: Analytical Method-ETS-8-044.1 

Attachment D: Method Deviation

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 23 of 89
GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS In Groundwater 

New On-Site WWTP Wells - April 2012 

13 Signatures 

Cleston Lange, Ph.D., 3M Principal Analytical Investigator Date 

William K. Reagen, Ph.D., 3M Environmental Laboratory Department Manager Date 

Hi /2 

Gary Hohenstein, Sponsor Representative Date 

Jaisimha Keg^ri, Study Director 

Page 24 of 89 
ATTACHMENT A: PROTOCOL AMENDMENT

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 25 of 89
GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 26 of 89
GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 27 of 89
GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 28 of 89
GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 29 of 89
GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 30 of 89
GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 31 of 89
ATTACHMENT B: REPRESENTATIVE SAMPLE CHROMATOGRAMS AND 

CALIBRATION CURVE(S) 

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 32 of 89
g120413a.rdb (PFHS): "Quadratic" Regression ("1 / x" weighting): y = -1.61e+004 x^2 + 1.24e+006 x + 6.41e+003 (r = 0.9999)

0 5 10 15 20 25 30 35 40 45 50 55 60 65 70 75 80 

Concentration, ng/mL

0.0

2.0e64.0e66.0e68.0e61.0e71.2e71.4e71.6e71.8e72.0e72.2e72.4e72.6e72.8e73.0e73.2e73.4e73.6e73.8e74.0e74.2e74.4e74.6e7

Area, counts

*** Ginger AG01330509 Results Name: g120413a.rdbData printed by KJUPrinting Time: 4:46:58 PMPrinting Date: Monday, April 16, 2012

Page 1 of 1

G

Analysis of PFBS, PFH

New On-
g120413a.rdb (PFBS): "Quadratic" Regression ("1 / x" weighting): y = -1.84e+004 x^2 + 1.27e+006 x + 4.06e+003 (r = 0.9998)

0 5 10 15 20 25 30 35 40 45 50 55 60 65 70 75 80 

Concentration, ng/mL

0.0

2.0e64.0e66.0e68.0e61.0e71.2e71.4e71.6e71.8e72.0e72.2e72.4e72.6e72.8e73.0e73.2e73.4e73.6e73.8e74.0e74.2e74.4e7

Area, counts

*** Ginger AG01330509 Results Name: g120413a.rdbData printed by KJUPrinting Time: 4:46:30 PMPrinting Date: Monday, April 16, 2012

Page 1 of 1

G

Analysis of PFBS, PFH

New On-
Sample Name: "g120413a017" Sample ID: "11012-10-11" File: "g120413a.wiff"

Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "10 ng/mL FC std in Milli Q water" Annotation: ""

Sample Index: 17 Sample Type: Standard Concentration: 10.0 ng/mL Calculated Conc: 9.83 ng/mL Acq. Date: 4/14/2012 Acq. Time: 12:18:40 AM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 50 Base. Sub. Window: 1.00 minPeak-Split. Factor: 2 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.0 minUse Relative RT: No Int. Type: Valley Retention Time: 13.3 minArea: 10731882 countsHeight: 2970000. cpsStart Time: 13.2 minEnd Time: 13.7 min

11.0 11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5

1714 1792 1870 1947 2025 2103 2181 2259 2337 2415

Time, min

0.0

2.0e54.0e56.0e58.0e51.0e61.2e61.4e61.6e61.8e62.0e62.2e62.4e62.6e62.8e6

Intensity, cps

13.27

13.12

Sample Name: "g120413a024" Sample ID: "TN12-0139" File: "g120413a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "Method Blank" Annotation: ""

Sample Index: 24 Sample Type: Unknown Concentration: N/A Calculated Conc: < 0 Acq. Date: 4/14/2012 Acq. Time: 2:46:48 AM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 50 Base. Sub. Window: 1.00 minPeak-Split. Factor: 2 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.0 minUse Relative RT: No Int. Type: Valley Retention Time: 13.0 minArea: 2284 countsHeight: 566. cpsStart Time: 12.9 minEnd Time: 13.1 min

11.0 11.5 12.0 12.5 

1714 1792 1870 1947 

T

0

200400600800

100012001400160018002000220024002600280030003200

Intensity, cps

Sample Name: "g120413a025" Sample ID: "LCS-120411-1" File: "g120413a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "0.5ppb LCS" Annotation: ""

Sample Index: 25 Sample Type: QC Concentration: 0.498 ng/mL Calculated Conc: 0.479 ng/mL Acq. Date: 4/14/2012 Acq. Time: 3:08:04 AM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 50 Base. Sub. Window: 1.00 minPeak-Split. Factor: 2 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.0 minUse Relative RT: No Int. Type: Valley Retention Time: 13.1 minArea: 609352 countsHeight: 187000. cpsStart Time: 13.0 minEnd Time: 13.3 min

11.0 11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5

1714 1792 1870 1947 2025 2103 2181 2259 2337 2415

Time, min

0.0

1.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e5

Intensity, cps

13.11

13.75 12.95 15.22

Sample Name: "g120413a028" Sample ID: "LCS-120411-4" File: "g120413a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "5ppb LCS" Annotation: ""

Sample Index: 28 Sample Type: QC Concentration: 4.98 ng/mL Calculated Conc: 4.69 ng/mL Acq. Date: 4/14/2012 Acq. Time: 4:11:21 AM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 50 Base. Sub. Window: 1.00 minPeak-Split. Factor: 2 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.0 minUse Relative RT: No Int. Type: Valley Retention Time: 13.1 minArea: 5563369 countsHeight: 1630000. cpsStart Time: 13.0 minEnd Time: 13.6 min

11.0 11.5 12.0 12.5 

1714 1792 1870 1947 

T

0.0

1.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e61.6e6

Intensity, cps

*** Ginger AG01330509 Results Name: g120413a.rdbData printed by STWPrinting Time: 11:15:42 AMPrinting Date: Wednesday, May 09, 2012

Page 1 of 9

G

Analysis of PFBS, PFH

New On-
Sample Name: "g120413a031" Sample ID: "LCS-120411-7" File: "g120413a.wiff"

Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "30ppb LCS" Annotation: ""

Sample Index: 31 Sample Type: QC Concentration: 29.9 ng/mL Calculated Conc: No Intercept Acq. Date: 4/14/2012 Acq. Time: 5:14:51 AM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 50 Base. Sub. Window: 1.00 minPeak-Split. Factor: 2 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.0 minUse Relative RT: No Int. Type: Valley Retention Time: 13.0 minArea: 23742500 countsHeight: 5920000. cpsStart Time: 12.9 minEnd Time: 13.5 min

11.0 11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5

1714 1792 1870 1947 2025 2103 2181 2259 2337 2415

Time, min

0.0

5.0e51.0e61.5e62.0e62.5e63.0e63.5e64.0e64.5e65.0e65.5e6

Intensity, cps

13.01

Sample Name: "g120413a043" Sample ID: "GLP10-01-02-31-001" File: "g120413a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-TW1R-0 (dil 1:10)" Annotation: ""

Sample Index: 43 Sample Type: Unknown Concentration: N/A Calculated Conc: 33.8 ng/mL Acq. Date: 4/14/2012 Acq. Time: 9:28:50 AM Modified: Yes RT Window: 30.0 secExpected RT: 13.0 minUse Relative RT: No Int. Type: Manual Retention Time: 13.0 minArea: 4094319 countsHeight: 1180000. cpsStart Time: 12.8 minEnd Time: 13.4 min

11.0 11.5 12.0 12.5 1

1714 1792 1870 1947 2

Tim

0.00

5.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e61.10e61.15e6

Intensity, cps

1

Sample Name: "g120413a046" Sample ID: "GLP10-01-02-31-004" File: "g120413a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-TW1R-HS (dil 1:10)" Annotation: ""

Sample Index: 46 Sample Type: Unknown Concentration: N/A Calculated Conc: 120. ng/mL Acq. Date: 4/14/2012 Acq. Time: 10:32:24 AM Modified: Yes RT Window: 30.0 secExpected RT: 13.0 minUse Relative RT: No Int. Type: Manual Retention Time: 13.0 minArea: 12616952 countsHeight: 3430000. cpsStart Time: 12.8 minEnd Time: 13.4 min

11.0 11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5

1714 1792 1870 1947 2025 2103 2181 2259 2337 2415

Time, min

0.0

2.0e54.0e56.0e58.0e51.0e61.2e61.4e61.6e61.8e62.0e62.2e62.4e62.6e62.8e63.0e63.2e63.4e6

Intensity, cps

13.01

Sample Name: "g120413a050" Sample ID: "GLP10-01-02-31-005" File: "g120413a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-TW2R-0 (dil 1:10)" Annotation: ""

Sample Index: 50 Sample Type: Unknown Concentration: N/A Calculated Conc: 116. ng/mL Acq. Date: 4/14/2012 Acq. Time: 11:57:29 AM Modified: Yes RT Window: 30.0 secExpected RT: 13.0 minUse Relative RT: No Int. Type: Manual Retention Time: 13.3 minArea: 2851509 countsHeight: 801000. cpsStart Time: 13.0 minEnd Time: 13.6 min

11.0 11.5 12.0 12.5 1

1714 1792 1870 1947 2

Tim

0.0

5.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e5

Intensity, cps

*** Ginger AG01330509 Results Name: g120413a.rdbData printed by STWPrinting Time: 11:15:42 AMPrinting Date: Wednesday, May 09, 2012

Page 2 of 9

G

Analysis of PFBS, PFH

New On-
Sample Name: "g120413a081" Sample ID: "GLP10-01-02-31-005" File: "g120413a.wiff"

Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-TW2R-0 (dil 1:10)" Annotation: ""

Sample Index: 81 Sample Type: Unknown Concentration: N/A Calculated Conc: 120. ng/mL Acq. Date: 4/14/2012 Acq. Time: 10:53:53 PM Modified: Yes RT Window: 30.0 secExpected RT: 13.0 minUse Relative RT: No Int. Type: Manual Retention Time: 13.0 minArea: 12579452 countsHeight: 3350000. cpsStart Time: 12.8 minEnd Time: 13.4 min

11.0 11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5

1714 1792 1870 1947 2025 2103 2181 2259 2337 2415

Time, min

0.0

2.0e54.0e56.0e58.0e51.0e61.2e61.4e61.6e61.8e62.0e62.2e62.4e62.6e62.8e63.0e63.2e6

Intensity, cps

13.01

12.86

Sample Name: "g120413a055" Sample ID: "GLP10-01-02-31-009" File: "g120413a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-TW3R-0 (dil 1:10)" Annotation: ""

Sample Index: 55 Sample Type: Unknown Concentration: N/A Calculated Conc: 309. ng/mL Acq. Date: 4/14/2012 Acq. Time: 1:42:48 PM Modified: Yes RT Window: 30.0 secExpected RT: 13.0 minUse Relative RT: No Int. Type: Manual Retention Time: 13.3 minArea: 7168230 countsHeight: 2040000. cpsStart Time: 13.0 minEnd Time: 13.6 min

11.0 11.5 12.0 12.5 1

1714 1792 1870 1947 2

Tim

0.0

1.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e61.6e61.7e61.8e61.9e62.0e6

Intensity, cps

Sample Name: "g120413a062" Sample ID: "GLP10-01-02-31-013" File: "g120413a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-TW4R-0 (dil 1:10)" Annotation: ""

Sample Index: 62 Sample Type: Unknown Concentration: N/A Calculated Conc: 185. ng/mL Acq. Date: 4/14/2012 Acq. Time: 4:10:50 PM Modified: Yes RT Window: 30.0 secExpected RT: 13.0 minUse Relative RT: No Int. Type: Manual Retention Time: 13.0 minArea: 17265648 countsHeight: 4550000. cpsStart Time: 12.8 minEnd Time: 13.5 min

11.0 11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5

1714 1792 1870 1947 2025 2103 2181 2259 2337 2415

Time, min

0.0

2.0e54.0e56.0e58.0e51.0e61.2e61.4e61.6e61.8e62.0e62.2e62.4e62.6e62.8e63.0e63.2e63.4e63.6e63.8e64.0e64.2e64.4e6

Intensity, cps

13.04

Sample Name: "g120413a064" Sample ID: "GLP10-01-02-31-015" File: "g120413a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-TW4R-LS (dil 1:10)" Annotation: ""

Sample Index: 64 Sample Type: Unknown Concentration: N/A Calculated Conc: 195. ng/mL Acq. Date: 4/14/2012 Acq. Time: 4:53:12 PM Modified: Yes RT Window: 30.0 secExpected RT: 13.0 minUse Relative RT: No Int. Type: Manual Retention Time: 13.1 minArea: 17807061 countsHeight: 4820000. cpsStart Time: 12.8 minEnd Time: 13.6 min

11.0 11.5 12.0 12.5 1

1714 1792 1870 1947 2

Tim

0.0

2.0e54.0e56.0e58.0e51.0e61.2e61.4e61.6e61.8e62.0e62.2e62.4e62.6e62.8e63.0e63.2e63.4e63.6e63.8e64.0e64.2e64.4e64.6e6

Intensity, cps

*** Ginger AG01330509 Results Name: g120413a.rdbData printed by STWPrinting Time: 11:15:43 AMPrinting Date: Wednesday, May 09, 2012

Page 3 of 9

G

Analysis of PFBS, PFH

New On-
Sample Name: "g120413a067" Sample ID: "GLP10-01-02-31-017" File: "g120413a.wiff"

Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-TW5R-0 (dil 1:10)" Annotation: ""

Sample Index: 67 Sample Type: Unknown Concentration: N/A Calculated Conc: No Intercept Acq. Date: 4/14/2012 Acq. Time: 5:57:06 PM Modified: Yes RT Window: 30.0 secExpected RT: 13.0 minUse Relative RT: No Int. Type: Manual Retention Time: 13.1 minArea: 33576423 countsHeight: 7580000. cpsStart Time: 12.9 minEnd Time: 13.5 min

11.0 11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5

1714 1792 1870 1947 2025 2103 2181 2259 2337 2415

Time, min

0.0

5.0e51.0e61.5e62.0e62.5e63.0e63.5e64.0e64.5e65.0e65.5e66.0e66.5e67.0e67.5e6

Intensity, cps

13.08

Sample Name: "g120413a074" Sample ID: "GLP10-01-02-31-021" File: "g120413a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-TW6R-0 (dil 1:10)" Annotation: ""

Sample Index: 74 Sample Type: Unknown Concentration: N/A Calculated Conc: 156. ng/mL Acq. Date: 4/14/2012 Acq. Time: 8:25:30 PM Modified: Yes RT Window: 30.0 secExpected RT: 13.0 minUse Relative RT: No Int. Type: Manual Retention Time: 13.3 minArea: 15369048 countsHeight: 4190000. cpsStart Time: 13.0 minEnd Time: 13.7 min

11.0 11.5 12.0 12.5 1

1714 1792 1870 1947 2

Tim

0.0

2.0e54.0e56.0e58.0e51.0e61.2e61.4e61.6e61.8e62.0e62.2e62.4e62.6e62.8e63.0e63.2e63.4e63.6e63.8e64.0e6

Intensity, cps

Sample Name: "g120413a038" Sample ID: "GLP10-01-02-31-025" File: "g120413a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-RW3R-RB (dil 1:10)" Annotation: ""

Sample Index: 38 Sample Type: Unknown Concentration: N/A Calculated Conc: < 0 Acq. Date: 4/14/2012 Acq. Time: 7:43:05 AM Modified: Yes Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 50 Base. Sub. Window: 1.00 minPeak-Split. Factor: 2 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.3 minUse Relative RT: No Int. Type: Base To Base Retention Time: 13.3 minArea: 1643 countsHeight: 558. cpsStart Time: 13.2 minEnd Time: 13.4 min

11.0 11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5

1714 1792 1870 1947 2025 2103 2181 2259 2337 2415

Time, min

0

200400600800

100012001400160018002000220024002600280030003200

Intensity, cps

Sample Name: "g120413a039" Sample ID: "GLP10-01-02-31-026" File: "g120413a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-TRIP01-0 (dil 1:10)" Annotation: ""

Sample Index: 39 Sample Type: Unknown Concentration: N/A Calculated Conc: < 0 Acq. Date: 4/14/2012 Acq. Time: 8:04:09 AM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 50 Base. Sub. Window: 1.00 minPeak-Split. Factor: 2 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.0 minUse Relative RT: No Int. Type: Valley Retention Time: 13.1 minArea: 1065 countsHeight: 281. cpsStart Time: 13.0 minEnd Time: 13.1 min

11.0 11.5 12.0 12.5 

1714 1792 1870 1947 

T

0

200400600800

100012001400160018002000220024002600280030003200

Intensity, cps

11.17 12.93

*** Ginger AG01330509 Results Name: g120413a.rdbData printed by STWPrinting Time: 11:15:43 AMPrinting Date: Wednesday, May 09, 2012

Page 4 of 9

G

Analysis of PFBS, PFH

New On-
Sample Name: "g120413a040" Sample ID: "GLP10-01-02-31-027" File: "g120413a.wiff"

Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-TRIP01-LS (dil 1:10)" Annotation: ""

Sample Index: 40 Sample Type: Unknown Concentration: N/A Calculated Conc: 8.51 ng/mL Acq. Date: 4/14/2012 Acq. Time: 8:25:25 AM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 50 Base. Sub. Window: 1.00 minPeak-Split. Factor: 2 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.0 minUse Relative RT: No Int. Type: Valley Retention Time: 13.0 minArea: 1073895 countsHeight: 318000. cpsStart Time: 12.9 minEnd Time: 13.3 min

11.0 11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5

1714 1792 1870 1947 2025 2103 2181 2259 2337 2415

Time, min

0.0

2.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e5

Intensity, cps

13.05

13.70 12.90

Sample Name: "g120413a041" Sample ID: "GLP10-01-02-31-028" File: "g120413a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-TRIP01-HS (dil 1:10)" Annotation: ""

Sample Index: 41 Sample Type: Unknown Concentration: N/A Calculated Conc: 92.0 ng/mL Acq. Date: 4/14/2012 Acq. Time: 8:46:37 AM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 50 Base. Sub. Window: 1.00 minPeak-Split. Factor: 2 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.0 minUse Relative RT: No Int. Type: Valley Retention Time: 13.1 minArea: 10148792 countsHeight: 2920000. cpsStart Time: 13.0 minEnd Time: 13.4 min

11.0 11.5 12.0 12.5 

1714 1792 1870 1947 

T

0.0

2.0e54.0e56.0e58.0e51.0e61.2e61.4e61.6e61.8e62.0e62.2e62.4e62.6e62.8e6

Intensity, cps

Sample Name: "g120413a017" Sample ID: "11012-10-11" File: "g120413a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "10 ng/mL FC std in Milli Q water" Annotation: ""

Sample Index: 17 Sample Type: Standard Concentration: 9.98 ng/mL Calculated Conc: 9.91 ng/mL Acq. Date: 4/14/2012 Acq. Time: 12:18:40 AM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 50 Base. Sub. Window: 1.00 minPeak-Split. Factor: 1 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 14.6 minUse Relative RT: No Int. Type: Valley Retention Time: 14.8 minArea: 10664568 countsHeight: 3180000. cpsStart Time: 14.7 minEnd Time: 15.1 min

12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0

1947 2025 2103 2181 2259 2337 2415 2492 2570 2648

Time, min

0.0

2.0e54.0e56.0e58.0e51.0e61.2e61.4e61.6e61.8e62.0e62.2e62.4e62.6e62.8e63.0e6

Intensity, cps

14.81

Sample Name: "g120413a024" Sample ID: "TN12-0139" File: "g120413a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "Method Blank" Annotation: ""

Sample Index: 24 Sample Type: Unknown Concentration: N/A Calculated Conc: < 0 Acq. Date: 4/14/2012 Acq. Time: 2:46:48 AM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 50 Base. Sub. Window: 1.00 minPeak-Split. Factor: 1 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 14.6 minUse Relative RT: No Int. Type: Valley Retention Time: 14.6 minArea: 4389 countsHeight: 1260. cpsStart Time: 14.5 minEnd Time: 14.6 min

12.5 13.0 13.5 14.0 1

1947 2025 2103 2181 2

T

0

500

100015002000250030003500400045005000550060006500

Intensity, cps

*** Ginger AG01330509 Results Name: g120413a.rdbData printed by STWPrinting Time: 11:15:43 AMPrinting Date: Wednesday, May 09, 2012

Page 5 of 9

G

Analysis of PFBS, PFH

New On-
Sample Name: "g120413a025" Sample ID: "LCS-120411-1" File: "g120413a.wiff"

Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "0.5ppb LCS" Annotation: ""

Sample Index: 25 Sample Type: QC Concentration: 0.497 ng/mL Calculated Conc: 0.472 ng/mL Acq. Date: 4/14/2012 Acq. Time: 3:08:04 AM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 50 Base. Sub. Window: 1.00 minPeak-Split. Factor: 1 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 14.6 minUse Relative RT: No Int. Type: Valley Retention Time: 14.6 minArea: 586064 countsHeight: 185000. cpsStart Time: 14.5 minEnd Time: 14.8 min

12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0

1947 2025 2103 2181 2259 2337 2415 2492 2570 2648

Time, min

0.0

1.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e5

Intensity, cps

14.65

15.18

Sample Name: "g120413a028" Sample ID: "LCS-120411-4" File: "g120413a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "5ppb LCS" Annotation: ""

Sample Index: 28 Sample Type: QC Concentration: 4.97 ng/mL Calculated Conc: 4.73 ng/mL Acq. Date: 4/14/2012 Acq. Time: 4:11:21 AM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 50 Base. Sub. Window: 1.00 minPeak-Split. Factor: 1 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 14.6 minUse Relative RT: No Int. Type: Valley Retention Time: 14.6 minArea: 5490381 countsHeight: 1710000. cpsStart Time: 14.5 minEnd Time: 14.9 min

12.5 13.0 13.5 14.0 1

1947 2025 2103 2181 2

T

0.0

1.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e61.6e61.7e6

Intensity, cps

Sample Name: "g120413a031" Sample ID: "LCS-120411-7" File: "g120413a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "30ppb LCS" Annotation: ""

Sample Index: 31 Sample Type: QC Concentration: 29.9 ng/mL Calculated Conc: 37.1 ng/mL Acq. Date: 4/14/2012 Acq. Time: 5:14:51 AM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 50 Base. Sub. Window: 1.00 minPeak-Split. Factor: 1 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 14.6 minUse Relative RT: No Int. Type: Valley Retention Time: 14.5 minArea: 23673721 countsHeight: 6390000. cpsStart Time: 14.4 minEnd Time: 14.9 min

12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0

1947 2025 2103 2181 2259 2337 2415 2492 2570 2648

Time, min

0.0

5.0e51.0e61.5e62.0e62.5e63.0e63.5e64.0e64.5e65.0e65.5e66.0e6

Intensity, cps

14.55

Sample Name: "g120413a043" Sample ID: "GLP10-01-02-31-001" File: "g120413a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-TW1R-0 (dil 1:10)" Annotation: ""

Sample Index: 43 Sample Type: Unknown Concentration: N/A Calculated Conc: 56.7 ng/mL Acq. Date: 4/14/2012 Acq. Time: 9:28:50 AM Modified: Yes RT Window: 30.0 secExpected RT: 14.6 minUse Relative RT: No Int. Type: Manual Retention Time: 14.6 minArea: 6486895 countsHeight: 1730000. cpsStart Time: 14.3 minEnd Time: 14.9 min

12.5 13.0 13.5 14.0 14

1947 2025 2103 2181 22

Tim

0.0

1.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e61.6e61.7e6

Intensity, cps

1

*** Ginger AG01330509 Results Name: g120413a.rdbData printed by STWPrinting Time: 11:15:43 AMPrinting Date: Wednesday, May 09, 2012

Page 6 of 9

G

Analysis of PFBS, PFH

New On-
Sample Name: "g120413a046" Sample ID: "GLP10-01-02-31-004" File: "g120413a.wiff"

Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-TW1R-HS (dil 1:10)" Annotation: ""

Sample Index: 46 Sample Type: Unknown Concentration: N/A Calculated Conc: 147. ng/mL Acq. Date: 4/14/2012 Acq. Time: 10:32:24 AM Modified: Yes RT Window: 30.0 secExpected RT: 14.6 minUse Relative RT: No Int. Type: Manual Retention Time: 14.5 minArea: 14709560 countsHeight: 3990000. cpsStart Time: 14.3 minEnd Time: 14.9 min

12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0

1947 2025 2103 2181 2259 2337 2415 2492 2570 2648

Time, min

0.0

2.0e54.0e56.0e58.0e51.0e61.2e61.4e61.6e61.8e62.0e62.2e62.4e62.6e62.8e63.0e63.2e63.4e63.6e63.8e6

Intensity, cps

14.54

14.40

Sample Name: "g120413a050" Sample ID: "GLP10-01-02-31-005" File: "g120413a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-TW2R-0 (dil 1:10)" Annotation: ""

Sample Index: 50 Sample Type: Unknown Concentration: N/A Calculated Conc: 580. ng/mL Acq. Date: 4/14/2012 Acq. Time: 11:57:29 AM Modified: Yes RT Window: 30.0 secExpected RT: 14.6 minUse Relative RT: No Int. Type: Manual Retention Time: 14.8 minArea: 12168421 countsHeight: 3010000. cpsStart Time: 14.5 minEnd Time: 15.1 min

12.5 13.0 13.5 14.0 14

1947 2025 2103 2181 22

Tim

0.0

2.0e54.0e56.0e58.0e51.0e61.2e61.4e61.6e61.8e62.0e62.2e62.4e62.6e62.8e6

Intensity, cps

14

Sample Name: "g120413a081" Sample ID: "GLP10-01-02-31-005" File: "g120413a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-TW2R-0 (dil 1:10)" Annotation: ""

Sample Index: 81 Sample Type: Unknown Concentration: N/A Calculated Conc: No Intercept Acq. Date: 4/14/2012 Acq. Time: 10:53:53 PM Modified: Yes RT Window: 30.0 secExpected RT: 14.6 minUse Relative RT: No Int. Type: Manual Retention Time: 14.6 minArea: 43613173 countsHeight: 8020000. cpsStart Time: 14.2 minEnd Time: 15.0 min

12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0

1947 2025 2103 2181 2259 2337 2415 2492 2570 2648

Time, min

0.0

5.0e51.0e61.5e62.0e62.5e63.0e63.5e64.0e64.5e65.0e65.5e66.0e66.5e67.0e67.5e68.0e6

Intensity, cps

14.56

14.42

Sample Name: "g120413a055" Sample ID: "GLP10-01-02-31-009" File: "g120413a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-TW3R-0 (dil 1:10)" Annotation: ""

Sample Index: 55 Sample Type: Unknown Concentration: N/A Calculated Conc: No Intercept Acq. Date: 4/14/2012 Acq. Time: 1:42:48 PM Modified: Yes RT Window: 30.0 secExpected RT: 14.6 minUse Relative RT: No Int. Type: Manual Retention Time: 14.8 minArea: 24528502 countsHeight: 5490000. cpsStart Time: 14.5 minEnd Time: 15.1 min

12.5 13.0 13.5 14.0 

1947 2025 2103 2181 

0.0

5.0e51.0e61.5e62.0e62.5e63.0e63.5e64.0e64.5e65.0e6

Intensity, cps

*** Ginger AG01330509 Results Name: g120413a.rdbData printed by STWPrinting Time: 11:15:44 AMPrinting Date: Wednesday, May 09, 2012

Page 7 of 9

G

Analysis of PFBS, PFH

New On-
Sample Name: "g120413a062" Sample ID: "GLP10-01-02-31-013" File: "g120413a.wiff"

Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-TW4R-0 (dil 1:10)" Annotation: ""

Sample Index: 62 Sample Type: Unknown Concentration: N/A Calculated Conc: 19.0 ng/mL Acq. Date: 4/14/2012 Acq. Time: 4:10:50 PM Modified: Yes RT Window: 30.0 secExpected RT: 14.6 minUse Relative RT: No Int. Type: Manual Retention Time: 14.6 minArea: 2294370 countsHeight: 618000. cpsStart Time: 14.3 minEnd Time: 14.8 min

12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0

1947 2025 2103 2181 2259 2337 2415 2492 2570 2648

Time, min

0.0

5.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e5

Intensity, cps

14.58

Sample Name: "g120413a064" Sample ID: "GLP10-01-02-31-015" File: "g120413a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-TW4R-LS (dil 1:10)" Annotation: ""

Sample Index: 64 Sample Type: Unknown Concentration: N/A Calculated Conc: 28.2 ng/mL Acq. Date: 4/14/2012 Acq. Time: 4:53:12 PM Modified: Yes RT Window: 30.0 secExpected RT: 14.6 minUse Relative RT: No Int. Type: Manual Retention Time: 14.6 minArea: 3365504 countsHeight: 935000. cpsStart Time: 14.3 minEnd Time: 14.8 min

12.5 13.0 13.5 14.0 14

1947 2025 2103 2181 22

Tim

0.0

5.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e58.5e59.0e5

Intensity, cps

14.46

Sample Name: "g120413a067" Sample ID: "GLP10-01-02-31-017" File: "g120413a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-TW5R-0 (dil 1:10)" Annotation: ""

Sample Index: 67 Sample Type: Unknown Concentration: N/A Calculated Conc: 43.3 ng/mL Acq. Date: 4/14/2012 Acq. Time: 5:57:06 PM Modified: Yes RT Window: 30.0 secExpected RT: 14.6 minUse Relative RT: No Int. Type: Manual Retention Time: 14.6 minArea: 5057137 countsHeight: 1400000. cpsStart Time: 14.3 minEnd Time: 14.9 min

12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0

1947 2025 2103 2181 2259 2337 2415 2492 2570 2648

Time, min

0.0

1.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e6

Intensity, cps

14.62

14.48

Sample Name: "g120413a074" Sample ID: "GLP10-01-02-31-021" File: "g120413a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-TW6R-0 (dil 1:10)" Annotation: ""

Sample Index: 74 Sample Type: Unknown Concentration: N/A Calculated Conc: 48.0 ng/mL Acq. Date: 4/14/2012 Acq. Time: 8:25:30 PM Modified: Yes RT Window: 30.0 secExpected RT: 14.6 minUse Relative RT: No Int. Type: Manual Retention Time: 14.8 minArea: 5564441 countsHeight: 1470000. cpsStart Time: 14.5 minEnd Time: 15.1 min

12.5 13.0 13.5 14.0 14

1947 2025 2103 2181 22

Tim

0.0

1.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e6

Intensity, cps

14

*** Ginger AG01330509 Results Name: g120413a.rdbData printed by STWPrinting Time: 11:15:44 AMPrinting Date: Wednesday, May 09, 2012

Page 8 of 9

G

Analysis of PFBS, PFH

New On-
Sample Name: "g120413a038" Sample ID: "GLP10-01-02-31-025" File: "g120413a.wiff"

Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-RW3R-RB (dil 1:10)" Annotation: ""

Sample Index: 38 Sample Type: Unknown Concentration: N/A Calculated Conc: < 0 Acq. Date: 4/14/2012 Acq. Time: 7:43:05 AM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 50 Base. Sub. Window: 1.00 minPeak-Split. Factor: 1 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 14.6 minUse Relative RT: No Int. Type: Valley Retention Time: 14.8 minArea: 2254 countsHeight: 697. cpsStart Time: 14.8 minEnd Time: 14.9 min

12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0

1947 2025 2103 2181 2259 2337 2415 2492 2570 2648

Time, min

0

500

10001500200025003000350040004500500055006000

Intensity, cps

Sample Name: "g120413a039" Sample ID: "GLP10-01-02-31-026" File: "g120413a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-TRIP01-0 (dil 1:10)" Annotation: ""

Sample Index: 39 Sample Type: Unknown Concentration: N/A Calculated Conc: < 0 Acq. Date: 4/14/2012 Acq. Time: 8:04:09 AM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 50 Base. Sub. Window: 1.00 minPeak-Split. Factor: 1 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 14.6 minUse Relative RT: No Int. Type: Valley Retention Time: 14.6 minArea: 1832 countsHeight: 542. cpsStart Time: 14.5 minEnd Time: 14.7 min

12.5 13.0 13.5 14.0 1

1947 2025 2103 2181 2

T

0

500

1000150020002500300035004000450050005500

Intensity, cps

Sample Name: "g120413a040" Sample ID: "GLP10-01-02-31-027" File: "g120413a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-TRIP01-LS (dil 1:10)" Annotation: ""

Sample Index: 40 Sample Type: Unknown Concentration: N/A Calculated Conc: 8.60 ng/mL Acq. Date: 4/14/2012 Acq. Time: 8:25:25 AM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 50 Base. Sub. Window: 1.00 minPeak-Split. Factor: 1 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 14.6 minUse Relative RT: No Int. Type: Valley Retention Time: 14.6 minArea: 1057216 countsHeight: 333000. cpsStart Time: 14.5 minEnd Time: 14.8 min

12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0

1947 2025 2103 2181 2259 2337 2415 2492 2570 2648

Time, min

0.0

2.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e5

Intensity, cps

14.58

15.11

Sample Name: "g120413a041" Sample ID: "GLP10-01-02-31-028" File: "g120413a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-TRIP01-HS (dil 1:10)" Annotation: ""

Sample Index: 41 Sample Type: Unknown Concentration: N/A Calculated Conc: 88.4 ng/mL Acq. Date: 4/14/2012 Acq. Time: 8:46:37 AM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 50 Base. Sub. Window: 1.00 minPeak-Split. Factor: 1 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 14.6 minUse Relative RT: No Int. Type: Valley Retention Time: 14.6 minArea: 9665200 countsHeight: 2890000. cpsStart Time: 14.5 minEnd Time: 14.9 min

12.5 13.0 13.5 14.0 1

1947 2025 2103 2181 2

T

0.0

2.0e54.0e56.0e58.0e51.0e61.2e61.4e61.6e61.8e62.0e62.2e62.4e62.6e62.8e6

Intensity, cps

*** Ginger AG01330509 Results Name: g120413a.rdbData printed by STWPrinting Time: 11:15:44 AMPrinting Date: Wednesday, May 09, 2012

Page 9 of 9

G

Analysis of PFBS, PFH

New On-
b120416a.rdb (PFOS): "Quadratic" Regression ("1 / x" weighting): y = -3.78e+003 x^2 + 6.38e+005 x + 1.29e+004 (r = 0.9995)

0 2 4 6 8 10 12 14 16 18 20 22 24 26 28 30 32 34 36 38 

Concentration, ng/mL

0.0

1.0e62.0e63.0e64.0e65.0e66.0e67.0e68.0e69.0e61.0e71.1e71.2e71.3e71.4e71.5e71.6e71.7e71.8e71.9e72.0e72.1e72.2e7

Area, counts

Workstation: ETSBUSTER Batch Name: b120416a.dab*Data worked up by KJU Page 1 of 1Printing Time: 2:17:59 PMPrinting Date: Wednesday, April 18, 2012

G

Analysis of PFBS, PFH

New On-
b120416a.rdb (PFBS): "Quadratic" Regression ("1 / x" weighting): y = -3.75e+003 x^2 + 5.22e+005 x + 5.55e+003 (r = 0.9993)

0 2 4 6 8 10 12 14 16 18 20 22 24 26 28 30 32 34 36 38 40 

Concentration, ng/mL

0.0

1.0e62.0e63.0e64.0e65.0e66.0e67.0e68.0e69.0e61.0e71.1e71.2e71.3e71.4e71.5e71.6e71.7e7

Area, counts

Workstation: ETSBUSTER Batch Name: b120416a.dab*Data worked up by KJU Page 1 of 1Printing Time: 2:17:43 PMPrinting Date: Wednesday, April 18, 2012

G

Analysis of PFBS, PFH

New On-
Sample Name: "b120416a020" Sample ID: "11012-10-11" File: "b120416a.wiff"

Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "10 ng/mL FC std in Milli Q Water" Annotation: ""

Sample Index: 20 Sample Type: Standard Concentration: 9.28 ng/mL Calculated Conc: 9.80 ng/mL Acq. Date: 4/16/2012 Acq. Time: 10:15:48 PM Modified: Yes RT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: No Int. Type: Manual Retention Time: 16.0 minArea: 5899151 countsHeight: 1.34e+006 cpsStart Time: 15.6 minEnd Time: 16.2 min

14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5

Time, min

0.0

5.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e58.5e59.0e59.5e51.0e61.1e61.1e61.2e61.2e61.3e61.3e6

Intensity, cps

16.03

15.89

15.71

Sample Name: "b120416a024" Sample ID: "Solvent Blank" File: "b120416a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "Milli Q Water TN12-0142" Annotation: ""

Sample Index: 24 Sample Type: Unknown Concentration: N/A Calculated Conc: < 0 Acq. Date: 4/16/2012 Acq. Time: 11:40:12 PM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 95 Base. Sub. Window: 1.00 minPeak-Split. Factor: 8 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: No Int. Type: Base To Base Retention Time: 16.0 minArea: 1091 countsHeight: 4.17e+002 cpsStart Time: 16.0 minEnd Time: 16.1 min

14.0 14.5 15.0 

0

200400600800

1000120014001600180020002200240026002800300032003400360038004000

Intensity, cps

Sample Name: "b120416a027" Sample ID: "LCS-120411-1" File: "b120416a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "0.5ppb LCS" Annotation: ""

Sample Index: 27 Sample Type: QC Concentration: 0.462 ng/mL Calculated Conc: 0.348 ng/mL Acq. Date: 4/17/2012 Acq. Time: 12:43:31 AM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 95 Base. Sub. Window: 1.00 minPeak-Split. Factor: 8 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: No Int. Type: Base To Base Retention Time: 16.0 minArea: 234201 countsHeight: 5.09e+004 cpsStart Time: 15.8 minEnd Time: 16.1 min

14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5

Time, min

0.0

2000.04000.06000.08000.0

1.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e43.4e43.6e43.8e44.0e44.2e44.4e44.6e44.8e45.0e4

Intensity, cps

16.02

15.88

16.42

17.72 14.36 16.59 14.59 15.71 17.17 17.33 15.44

Sample Name: "b120416a030" Sample ID: "LCS-120411-4" File: "b120416a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "5ppb LCS" Annotation: ""

Sample Index: 30 Sample Type: QC Concentration: 4.62 ng/mL Calculated Conc: 4.20 ng/mL Acq. Date: 4/17/2012 Acq. Time: 1:46:54 AM Modified: Yes RT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: No Int. Type: Manual Retention Time: 16.0 minArea: 2626788 countsHeight: 6.19e+005 cpsStart Time: 15.6 minEnd Time: 16.2 min

14.0 14.5 15.0 15

0.0

5.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e5

Intensity, cps

15

*** Buster J2930203Printing Time: 11:27:17 AMPrinting Date: Wednesday, May 09, 2012

Page 1 of 3

G

Analysis of PFBS, PFH

New On-
Sample Name: "b120416a033" Sample ID: "LCS-120411-7" File: "b120416a.wiff"

Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "30ppb LCS" Annotation: ""

Sample Index: 33 Sample Type: QC Concentration: 27.8 ng/mL Calculated Conc: 26.5 ng/mL Acq. Date: 4/17/2012 Acq. Time: 2:50:10 AM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 95 Base. Sub. Window: 1.00 minPeak-Split. Factor: 8 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: No Int. Type: Valley Retention Time: 16.0 minArea: 14250414 countsHeight: 3.03e+006 cpsStart Time: 15.6 minEnd Time: 16.3 min

14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5

Time, min

0.0

2.0e54.0e56.0e58.0e51.0e61.2e61.4e61.6e61.8e62.0e62.2e62.4e62.6e62.8e63.0e6

Intensity, cps

16.02

15.87

15.70

Sample Name: "b120416a044" Sample ID: "GLP10-01-02-31-001" File: "b120416a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-TW1R-0" Annotation: ""

Sample Index: 44 Sample Type: Unknown Concentration: N/A Calculated Conc: 27.7 ng/mL Acq. Date: 4/17/2012 Acq. Time: 6:42:17 AM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 95 Base. Sub. Window: 1.00 minPeak-Split. Factor: 8 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: No Int. Type: Base To Base Retention Time: 15.9 minArea: 1747791 countsHeight: 1.96e+005 cpsStart Time: 15.5 minEnd Time: 16.2 min

14.0 14.5 15.0 

0.0

1.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e51.9e5

Intensity, cps

Sample Name: "b120416a047" Sample ID: "GLP10-01-02-31-004" File: "b120416a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-TW1R-HS" Annotation: ""

Sample Index: 47 Sample Type: Unknown Concentration: N/A Calculated Conc: 94.7 ng/mL Acq. Date: 4/17/2012 Acq. Time: 7:45:24 AM Modified: Yes RT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: No Int. Type: Manual Retention Time: 16.0 minArea: 5713870 countsHeight: 9.27e+005 cpsStart Time: 15.5 minEnd Time: 16.2 min

14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5

Time, min

0.0

5.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e58.5e59.0e5

Intensity, cps

16.01

15.86

15.70

Sample Name: "b120416a040" Sample ID: "GLP10-01-02-31-025" File: "b120416a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-RW3R-RB" Annotation: ""

Sample Index: 40 Sample Type: Unknown Concentration: N/A Calculated Conc: < 0 Acq. Date: 4/17/2012 Acq. Time: 5:17:54 AM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 95 Base. Sub. Window: 1.00 minPeak-Split. Factor: 8 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: No Int. Type: Base To Base Retention Time: 16.0 minArea: 1220 countsHeight: 3.94e+002 cpsStart Time: 15.9 minEnd Time: 16.1 min

14.0 14.5 15.0 

0

200400600800

10001200140016001800200022002400260028003000320034003600

Intensity, cps

*** Buster J2930203Printing Time: 11:27:17 AMPrinting Date: Wednesday, May 09, 2012

Page 2 of 3

G

Analysis of PFBS, PFH

New On-
Sample Name: "b120416a041" Sample ID: "GLP10-01-02-31-026" File: "b120416a.wiff"

Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-TRIP01-0" Annotation: ""

Sample Index: 41 Sample Type: Unknown Concentration: N/A Calculated Conc: < 0 Acq. Date: 4/17/2012 Acq. Time: 5:39:01 AM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 95 Base. Sub. Window: 1.00 minPeak-Split. Factor: 8 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: No Int. Type: Base To Base Retention Time: 15.9 minArea: 271 countsHeight: 2.02e+002 cpsStart Time: 15.9 minEnd Time: 15.9 min

14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5

Time, min

0

100200300400500600700800900

1000110012001300140015001600170018001900200021002200230024002500

Intensity, cps

16.41

17.72

16.29

17.39

17.90 15.54 16.59 17.17

14.32 18.26 15.47 15.62 15.91

14.41

14.79

14.28 14.03

Sample Name: "b120416a042" Sample ID: "GLP10-01-02-31-027" File: "b120416a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-TRIP01-LS" Annotation: ""

Sample Index: 42 Sample Type: Unknown Concentration: N/A Calculated Conc: 6.96 ng/mL Acq. Date: 4/17/2012 Acq. Time: 6:00:04 AM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 95 Base. Sub. Window: 1.00 minPeak-Split. Factor: 8 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: No Int. Type: Base To Base Retention Time: 16.0 minArea: 454899 countsHeight: 8.94e+004 cpsStart Time: 15.6 minEnd Time: 16.2 min

14.0 14.5 15.0 

0.0

5000.0

1.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e48.0e48.5e4

Intensity, cps

14.34 

15.46 

Sample Name: "b120416a043" Sample ID: "GLP10-01-02-31-028" File: "b120416a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-TRIP01-HS" Annotation: ""

Sample Index: 43 Sample Type: Unknown Concentration: N/A Calculated Conc: 70.1 ng/mL Acq. Date: 4/17/2012 Acq. Time: 6:21:07 AM Modified: Yes RT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: No Int. Type: Manual Retention Time: 16.0 minArea: 4299439 countsHeight: 8.42e+005 cpsStart Time: 15.6 minEnd Time: 16.2 min

14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5

Time, min

0.0

5.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e5

Intensity, cps

16.02

15.87

15.70

*** Buster J2930203Printing Time: 11:27:17 AMPrinting Date: Wednesday, May 09, 2012

Page 3 of 3

G

Analysis of PFBS, PFH

New On-
Sample Name: "b120416a020" Sample ID: "11012-10-11" File: "b120416a.wiff"

Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "10 ng/mL FC std in Milli Q Water" Annotation: ""

Sample Index: 20 Sample Type: Standard Concentration: 10.0 ng/mL Calculated Conc: 10.2 ng/mL Acq. Date: 4/16/2012 Acq. Time: 10:15:48 PM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 50 Base. Sub. Window: 1.00 minPeak-Split. Factor: 1 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: No Int. Type: Valley Retention Time: 13.6 minArea: 4930110 countsHeight: 1.34e+006 cpsStart Time: 13.5 minEnd Time: 13.8 min

11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0

Time, min

0.0

5.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e58.5e59.0e59.5e51.0e61.1e61.1e61.2e61.2e61.3e61.3e6

Intensity, cps

13.56

Sample Name: "b120416a024" Sample ID: "Solvent Blank" File: "b120416a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "Milli Q Water TN12-0142" Annotation: ""

Sample Index: 24 Sample Type: Unknown Concentration: N/A Calculated Conc: < 0 Acq. Date: 4/16/2012 Acq. Time: 11:40:12 PM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 50 Base. Sub. Window: 1.00 minPeak-Split. Factor: 1 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: No Int. Type: Valley Retention Time: 13.5 minArea: 1362 countsHeight: 3.91e+002 cpsStart Time: 13.5 minEnd Time: 13.6 min

11.5 12.0 12.5 1

0

100200300400500600700800900

10001100120013001400150016001700180019002000

Intensity, cps

Sample Name: "b120416a027" Sample ID: "LCS-120411-1" File: "b120416a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "0.5ppb LCS" Annotation: ""

Sample Index: 27 Sample Type: QC Concentration: 0.498 ng/mL Calculated Conc: 0.505 ng/mL Acq. Date: 4/17/2012 Acq. Time: 12:43:31 AM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 50 Base. Sub. Window: 1.00 minPeak-Split. Factor: 1 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: No Int. Type: Valley Retention Time: 13.6 minArea: 268240 countsHeight: 7.52e+004 cpsStart Time: 13.5 minEnd Time: 13.7 min

11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0

Time, min

0.0

5000.0

1.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e4

Intensity, cps

13.55

11.71 11.84 13.43 14.21 14.63 14.77 15.12 15.78

Sample Name: "b120416a030" Sample ID: "LCS-120411-4" File: "b120416a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "5ppb LCS" Annotation: ""

Sample Index: 30 Sample Type: QC Concentration: 4.98 ng/mL Calculated Conc: 5.12 ng/mL Acq. Date: 4/17/2012 Acq. Time: 1:46:54 AM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 50 Base. Sub. Window: 1.00 minPeak-Split. Factor: 1 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: No Int. Type: Valley Retention Time: 13.5 minArea: 2576292 countsHeight: 7.20e+005 cpsStart Time: 13.4 minEnd Time: 13.7 min

11.5 12.0 12.5 1

0.0

5.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e5

Intensity, cps

*** Buster J2930203Printing Time: 11:28:38 AMPrinting Date: Wednesday, May 09, 2012

Page 1 of 3

G

Analysis of PFBS, PFH

New On-
Sample Name: "b120416a033" Sample ID: "LCS-120411-7" File: "b120416a.wiff"

Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "30ppb LCS" Annotation: ""

Sample Index: 33 Sample Type: QC Concentration: 29.9 ng/mL Calculated Conc: 29.3 ng/mL Acq. Date: 4/17/2012 Acq. Time: 2:50:10 AM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 50 Base. Sub. Window: 1.00 minPeak-Split. Factor: 1 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: No Int. Type: Valley Retention Time: 13.6 minArea: 12068689 countsHeight: 3.04e+006 cpsStart Time: 13.4 minEnd Time: 13.8 min

11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0

Time, min

0.0

2.0e54.0e56.0e58.0e51.0e61.2e61.4e61.6e61.8e62.0e62.2e62.4e62.6e62.8e63.0e6

Intensity, cps

13.56

Sample Name: "b120416a057" Sample ID: "GLP10-01-02-31-013" File: "b120416a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-TW4R-0" Annotation: ""

Sample Index: 57 Sample Type: Unknown Concentration: N/A Calculated Conc: 179. ng/mL Acq. Date: 4/17/2012 Acq. Time: 11:15:54 AM Modified: Yes RT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: No Int. Type: Manual Retention Time: 13.5 minArea: 1824040 countsHeight: 4.95e+005 cpsStart Time: 13.3 minEnd Time: 13.7 min

11.5 12.0 12.5 13

0.0

2.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e54.4e54.6e54.8e5

Intensity, cps

Sample Name: "b120416a059" Sample ID: "GLP10-01-02-31-016" File: "b120416a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-TW4R-HS" Annotation: ""

Sample Index: 59 Sample Type: Unknown Concentration: N/A Calculated Conc: 278. ng/mL Acq. Date: 4/17/2012 Acq. Time: 11:58:03 AM Modified: Yes RT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: No Int. Type: Manual Retention Time: 13.5 minArea: 2793103 countsHeight: 7.74e+005 cpsStart Time: 13.3 minEnd Time: 13.7 min

11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0

Time, min

0.0

5.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e5

Intensity, cps

13.53

Sample Name: "b120416a066" Sample ID: "GLP10-01-02-31-021" File: "b120416a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-TW6R-0" Annotation: ""

Sample Index: 66 Sample Type: Unknown Concentration: N/A Calculated Conc: 156. ng/mL Acq. Date: 4/17/2012 Acq. Time: 2:25:37 PM Modified: Yes RT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: No Int. Type: Manual Retention Time: 13.5 minArea: 1599713 countsHeight: 4.53e+005 cpsStart Time: 13.3 minEnd Time: 13.7 min

11.5 12.0 12.5 13

0.0

2.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e54.4e5

Intensity, cps

*** Buster J2930203Printing Time: 11:28:38 AMPrinting Date: Wednesday, May 09, 2012

Page 2 of 3

G

Analysis of PFBS, PFH

New On-
Sample Name: "b120416a068" Sample ID: "GLP10-01-02-31-024" File: "b120416a.wiff"

Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-TW6R-HS" Annotation: ""

Sample Index: 68 Sample Type: Unknown Concentration: N/A Calculated Conc: 238. ng/mL Acq. Date: 4/17/2012 Acq. Time: 3:07:39 PM Modified: Yes RT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: No Int. Type: Manual Retention Time: 13.5 minArea: 2399904 countsHeight: 6.76e+005 cpsStart Time: 13.3 minEnd Time: 13.7 min

11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0

Time, min

0.0

5.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e5

Intensity, cps

13.51

*** Buster J2930203Printing Time: 11:28:38 AMPrinting Date: Wednesday, May 09, 2012

Page 3 of 3

G

Analysis of PFBS, PFH

New On-
b120427a.rdb (PFOS): "Quadratic" Regression ("1 / x" weighting): y = -348 x^2 + 2.79e+005 x + 9.24e+003 (r = 0.9998)

5 10 15 20 25 30 35 40 45 50 55 60 65 70 75 

Concentration, ng/mL

0.0

1.0e62.0e63.0e64.0e65.0e66.0e67.0e68.0e69.0e61.0e71.1e71.2e71.3e71.4e71.5e71.6e71.7e71.8e71.9e72.0e72.1e72.2e72.3e7

Area, counts

Workstation: ETSBUSTER Batch Name: b120427a.dab*Data worked up by KJU Page 1 of 1Printing Time: 2:04:20 PMPrinting Date: Monday, April 30, 2012

G

Analysis of PFBS, PFH

New On-
b120427a.rdb (PFHS): "Quadratic" Regression ("1 / x" weighting): y = -642 x^2 + 2.39e+005 x + 6.09e+004 (r = 0.9996)

5 10 15 20 25 30 35 40 45 50 55 60 65 70 75 80 

Concentration, ng/mL

0.0

1.0e62.0e63.0e64.0e65.0e66.0e67.0e68.0e69.0e61.0e71.1e71.2e71.3e71.4e71.5e71.6e71.7e71.8e7

Area, counts

Workstation: ETSBUSTER Batch Name: b120427a.dab*Data worked up by KJU Page 1 of 1Printing Time: 2:03:53 PMPrinting Date: Monday, April 30, 2012

G

Analysis of PFBS, PFH

New On-
b120427a.rdb (PFBS): "Quadratic" Regression ("1 / x" weighting): y = -525 x^2 + 2.3e+005 x + 5.54e+004 (r = 0.9996)

5 10 15 20 25 30 35 40 45 50 55 60 65 70 75 80 

Concentration, ng/mL

0.0

1.0e62.0e63.0e64.0e65.0e66.0e67.0e68.0e69.0e61.0e71.1e71.2e71.3e71.4e71.5e71.6e71.7e71.8e7

Area, counts

Workstation: ETSBUSTER Batch Name: b120427a.dab*Data worked up by KJU Page 1 of 1Printing Time: 2:04:06 PMPrinting Date: Monday, April 30, 2012

G

Analysis of PFBS, PFH

New On-
Sample Name: "b120427a012" Sample ID: "11012-10-11" File: "b120427a.wiff"

Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "10 ng/mL FC std in Milli Q Water" Annotation: ""

Sample Index: 12 Sample Type: Standard Concentration: 10.0 ng/mL Calculated Conc: 10.6 ng/mL Acq. Date: 4/27/2012 Acq. Time: 5:30:22 PM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 50 Base. Sub. Window: 1.00 minPeak-Split. Factor: 1 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: No Int. Type: Valley Retention Time: 13.5 minArea: 2433472 countsHeight: 5.89e+005 cpsStart Time: 13.4 minEnd Time: 13.7 min

11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0

Time, min

0.0

5.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e5

Intensity, cps

13.53

13.70

Sample Name: "b120427a020" Sample ID: "Solvent Blank" File: "b120427a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "Milli Q Water TN12-0158" Annotation: ""

Sample Index: 20 Sample Type: Unknown Concentration: N/A Calculated Conc: < 0 Acq. Date: 4/27/2012 Acq. Time: 8:18:29 PM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 50 Base. Sub. Window: 1.00 minPeak-Split. Factor: 1 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: No Int. Type: Valley Retention Time: 13.4 minArea: 611 countsHeight: 1.77e+002 cpsStart Time: 13.3 minEnd Time: 13.4 min

11.5 12.0 12.5 1

0

100200300400500600700800900

10001100120013001400

Intensity, cps

Sample Name: "b120427a021" Sample ID: "LCS-120426-1" File: "b120427a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "5ppb LCS" Annotation: ""

Sample Index: 21 Sample Type: QC Concentration: 4.98 ng/mL Calculated Conc: 4.95 ng/mL Acq. Date: 4/27/2012 Acq. Time: 8:39:30 PM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 50 Base. Sub. Window: 1.00 minPeak-Split. Factor: 1 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: No Int. Type: Valley Retention Time: 13.6 minArea: 1183526 countsHeight: 2.96e+005 cpsStart Time: 13.4 minEnd Time: 13.7 min

11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0

Time, min

0.0

2.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e5

Intensity, cps

13.56

14.21

Sample Name: "b120427a024" Sample ID: "LCS-120426-4" File: "b120427a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "20ppb LCS" Annotation: ""

Sample Index: 24 Sample Type: QC Concentration: 20.0 ng/mL Calculated Conc: 18.6 ng/mL Acq. Date: 4/27/2012 Acq. Time: 9:42:39 PM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 50 Base. Sub. Window: 1.00 minPeak-Split. Factor: 1 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: No Int. Type: Valley Retention Time: 13.6 minArea: 4157125 countsHeight: 1.04e+006 cpsStart Time: 13.4 minEnd Time: 13.7 min

11.5 12.0 12.5 1

0.00

5.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e6

Intensity, cps

*** Buster J2930203Printing Time: 11:31:04 AMPrinting Date: Wednesday, May 09, 2012

Page 1 of 3

G

Analysis of PFBS, PFH

New On-
Sample Name: "b120427a027" Sample ID: "LCS-120426-7" File: "b120427a.wiff"

Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "50ppb LCS" Annotation: ""

Sample Index: 27 Sample Type: QC Concentration: 49.8 ng/mL Calculated Conc: 50.4 ng/mL Acq. Date: 4/27/2012 Acq. Time: 10:45:36 PM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 50 Base. Sub. Window: 1.00 minPeak-Split. Factor: 1 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: No Int. Type: Valley Retention Time: 13.6 minArea: 10333827 countsHeight: 2.40e+006 cpsStart Time: 13.5 minEnd Time: 13.8 min

11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0

Time, min

0.0

1.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e61.6e61.7e61.8e61.9e62.0e62.1e62.2e62.3e6

Intensity, cps

13.57

Sample Name: "b120427a038" Sample ID: "GLP10-01-02-31-009" File: "b120427a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-TW3R-0 (dil 1:100)" Annotation: ""

Sample Index: 38 Sample Type: Unknown Concentration: N/A Calculated Conc: 292. ng/mL Acq. Date: 4/28/2012 Acq. Time: 2:36:55 AM Modified: Yes RT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: No Int. Type: Manual Retention Time: 13.6 minArea: 723278 countsHeight: 1.81e+005 cpsStart Time: 13.3 minEnd Time: 13.7 min

11.5 12.0 12.5 13

0.0

1.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e5

Intensity, cps

Sample Name: "b120427a040" Sample ID: "GLP10-01-02-31-009; LMS" File: "b120427a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-TW3R-0 2000ppb LMS (dil 1:100)" Annotation: ""

Sample Index: 40 Sample Type: Unknown Concentration: N/A Calculated Conc: 2280. ng/mL Acq. Date: 4/28/2012 Acq. Time: 3:18:57 AM Modified: Yes RT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: No Int. Type: Manual Retention Time: 13.5 minArea: 5024434 countsHeight: 1.23e+006 cpsStart Time: 13.3 minEnd Time: 13.7 min

11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0

Time, min

0.00

5.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e61.10e61.15e61.20e6

Intensity, cps

13.55

Sample Name: "b120427a046" Sample ID: "GLP10-01-02-31-017" File: "b120427a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-TW5R-0 (dil 1:50)" Annotation: ""

Sample Index: 46 Sample Type: Unknown Concentration: N/A Calculated Conc: 533. ng/mL Acq. Date: 4/28/2012 Acq. Time: 5:25:16 AM Modified: Yes RT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: No Int. Type: Manual Retention Time: 13.5 minArea: 2453229 countsHeight: 6.10e+005 cpsStart Time: 13.3 minEnd Time: 13.7 min

11.5 12.0 12.5 13

0.0

5.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e5

Intensity, cps

*** Buster J2930203Printing Time: 11:31:04 AMPrinting Date: Wednesday, May 09, 2012

Page 2 of 3

G

Analysis of PFBS, PFH

New On-
Sample Name: "b120427a048" Sample ID: "GLP10-01-02-31-017; LMS" File: "b120427a.wiff"

Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "DAL-GW-TW5R-0 LMS (dil 1:50)" Annotation: ""

Sample Index: 48 Sample Type: Unknown Concentration: N/A Calculated Conc: 988. ng/mL Acq. Date: 4/28/2012 Acq. Time: 6:07:16 AM Modified: Yes RT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: No Int. Type: Manual Retention Time: 13.5 minArea: 4403931 countsHeight: 1.06e+006 cpsStart Time: 13.3 minEnd Time: 13.7 min

11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0

Time, min

0.00

5.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e6

Intensity, cps

13.52

*** Buster J2930203Printing Time: 11:31:04 AMPrinting Date: Wednesday, May 09, 2012

Page 3 of 3

G

Analysis of PFBS, PFH

New On-
Sample Name: "b120427a012" Sample ID: "11012-10-11" File: "b120427a.wiff"

Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "10 ng/mL FC std in Milli Q Water" Annotation: ""

Sample Index: 12 Sample Type: Standard Concentration: 9.28 ng/mL Calculated Conc: 9.65 ng/mL Acq. Date: 4/27/2012 Acq. Time: 5:30:22 PM Modified: No Proc. Algorithm: IntelliQuan - MQII Noise Percentage: 95 Base. Sub. Window: 1.00 minPeak-Split. Factor: 8 Report Largest Peak: Yes Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: No Int. Type: Valley Retention Time: 16.0 minArea: 2668335 countsHeight: 5.03e+005 cpsStart Time: 15.6 minEnd Time: 16.1 min

14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5

Time, min

0.0

2.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e54.4e54.6e54.8e55.0e5

Intensity, cps

15.99

15.83

Sample Name: "b120427a020" Sample ID: "Solvent Blank" File: "b120427a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "Milli Q Water TN12-0158" Annotation: ""

Sample Index: 20 Sample Type: Unknown Concentration: N/A Calculated Conc: < 0 Acq. Date: 4/27/2012 Acq. Time: 8:18:29 PM Modified: Yes RT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: No Int. Type: Manual Retention Time: 16.0 minArea: 4791 countsHeight: 7.78e+002 cpsStart Time: 15.8 minEnd Time: 16.1 min

14.0 14.5 15.0 15

0

200400600800

10001200140016001800200022002400260028003000320034003600380040004200440046004800500052005400

Intensity, cps

Sample Name: "b120427a021" Sample ID: "LCS-120426-1" File: "b120427a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "5ppb LCS" Annotation: ""

Sample Index: 21 Sample Type: QC Concentration: 4.62 ng/mL Calculated Conc: 4.26 ng/mL Acq. Date: 4/27/2012 Acq. Time: 8:39:30 PM Modified: Yes RT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: No Int. Type: Manual Retention Time: 16.0 minArea: 1190720 countsHeight: 2.30e+005 cpsStart Time: 15.6 minEnd Time: 16.1 min

14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5

Time, min

0.0

1.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e51.9e52.0e52.1e52.2e5

Intensity, cps

16.00

15.86

16.37 15.69

Sample Name: "b120427a024" Sample ID: "LCS-120426-4" File: "b120427a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "20ppb LCS" Annotation: ""

Sample Index: 24 Sample Type: QC Concentration: 18.5 ng/mL Calculated Conc: 15.7 ng/mL Acq. Date: 4/27/2012 Acq. Time: 9:42:39 PM Modified: Yes RT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: No Int. Type: Manual Retention Time: 16.0 minArea: 4313013 countsHeight: 8.07e+005 cpsStart Time: 15.6 minEnd Time: 16.2 min

14.0 14.5 15.0 15

0.0

5.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e5

Intensity, cps

*** Buster J2930203Printing Time: 11:33:49 AMPrinting Date: Wednesday, May 09, 2012

Page 1 of 4

G

Analysis of PFBS, PFH

New On-
Sample Name: "b120427a027" Sample ID: "LCS-120426-7" File: "b120427a.wiff"

Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "50ppb LCS" Annotation: ""

Sample Index: 27 Sample Type: QC Concentration: 46.2 ng/mL Calculated Conc: 46.3 ng/mL Acq. Date: 4/27/2012 Acq. Time: 10:45:36 PM Modified: Yes RT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: No Int. Type: Manual Retention Time: 16.0 minArea: 12177430 countsHeight: 2.20e+006 cpsStart Time: 15.6 minEnd Time: 16.2 min

14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5

Time, min

0.0

1.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e61.6e61.7e61.8e61.9e62.0e62.1e6

Intensity, cps

16.00

15.86

Sample Name: "b120427a034" Sample ID: "GLP10-01-02-31-005" File: "b120427a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-TW2R-0 (dil 1:50)" Annotation: ""

Sample Index: 34 Sample Type: Unknown Concentration: N/A Calculated Conc: 995. ng/mL Acq. Date: 4/28/2012 Acq. Time: 1:12:43 AM Modified: Yes RT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: No Int. Type: Manual Retention Time: 16.0 minArea: 5421689 countsHeight: 6.29e+005 cpsStart Time: 15.5 minEnd Time: 16.2 min

14.0 14.5 15.0 15

0.0

5.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e5

Intensity, cps

Sample Name: "b120427a036" Sample ID: "GLP10-01-02-31-005; LMS" File: "b120427a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-TW2R-0 LMS (dil 1:50)" Annotation: ""

Sample Index: 36 Sample Type: Unknown Concentration: N/A Calculated Conc: 2020. ng/mL Acq. Date: 4/28/2012 Acq. Time: 1:54:50 AM Modified: Yes RT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: No Int. Type: Manual Retention Time: 16.0 minArea: 10706006 countsHeight: 1.53e+006 cpsStart Time: 15.5 minEnd Time: 16.1 min

14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5

Time, min

0.0

1.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e6

Intensity, cps

15.99

15.84

15.69

Sample Name: "b120427a038" Sample ID: "GLP10-01-02-31-009" File: "b120427a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-TW3R-0 (dil 1:100)" Annotation: ""

Sample Index: 38 Sample Type: Unknown Concentration: N/A Calculated Conc: 3340. ng/mL Acq. Date: 4/28/2012 Acq. Time: 2:36:55 AM Modified: Yes RT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: No Int. Type: Manual Retention Time: 16.0 minArea: 8941101 countsHeight: 1.12e+006 cpsStart Time: 15.5 minEnd Time: 16.1 min

14.0 14.5 15.0 15

0.00

5.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e61.10e6

Intensity, cps

15

*** Buster J2930203Printing Time: 11:33:49 AMPrinting Date: Wednesday, May 09, 2012

Page 2 of 4

G

Analysis of PFBS, PFH

New On-
Sample Name: "b120427a041" Sample ID: "GLP10-01-02-31-009; LMS" File: "b120427a.wiff"

Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-TW3R-0 5000ppb LMS (dil 1:100)" Annotation: ""

Sample Index: 41 Sample Type: Unknown Concentration: N/A Calculated Conc: 8480. ng/mL Acq. Date: 4/28/2012 Acq. Time: 3:39:57 AM Modified: Yes RT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: No Int. Type: Manual Retention Time: 16.0 minArea: 9070510 countsHeight: 1.45e+006 cpsStart Time: 15.5 minEnd Time: 16.2 min

14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5

Time, min

0.0

1.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e6

Intensity, cps

15.97

15.82

Sample Name: "b120427a056" Sample ID: "GLP10-01-02-31-013" File: "b120427a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-TW4R-0 (dil 1:10)" Annotation: ""

Sample Index: 56 Sample Type: Unknown Concentration: N/A Calculated Conc: 244. ng/mL Acq. Date: 4/28/2012 Acq. Time: 8:55:36 AM Modified: Yes RT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: No Int. Type: Manual Retention Time: 16.0 minArea: 6604817 countsHeight: 1.09e+006 cpsStart Time: 15.6 minEnd Time: 16.1 min

14.0 14.5 15.0 15

0.00

5.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e6

Intensity, cps

Sample Name: "b120427a058" Sample ID: "GLP10-01-02-31-013;LMS" File: "b120427a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-TW4R-0 LMS (dil 1:10)" Annotation: ""

Sample Index: 58 Sample Type: Unknown Concentration: N/A Calculated Conc: 450. ng/mL Acq. Date: 4/28/2012 Acq. Time: 9:37:38 AM Modified: Yes RT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: No Int. Type: Manual Retention Time: 16.0 minArea: 11864095 countsHeight: 1.98e+006 cpsStart Time: 15.6 minEnd Time: 16.2 min

14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5

Time, min

0.0

1.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e61.6e61.7e61.8e61.9e6

Intensity, cps

15.96

15.81

Sample Name: "b120427a046" Sample ID: "GLP10-01-02-31-017" File: "b120427a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-TW5R-0 (dil 1:50)" Annotation: ""

Sample Index: 46 Sample Type: Unknown Concentration: N/A Calculated Conc: 704. ng/mL Acq. Date: 4/28/2012 Acq. Time: 5:25:16 AM Modified: Yes RT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: No Int. Type: Manual Retention Time: 16.0 minArea: 3870595 countsHeight: 5.90e+005 cpsStart Time: 15.6 minEnd Time: 16.1 min

14.0 14.5 15.0 15

0.0

5.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e5

Intensity, cps

*** Buster J2930203Printing Time: 11:33:49 AMPrinting Date: Wednesday, May 09, 2012

Page 3 of 4

G

Analysis of PFBS, PFH

New On-
Sample Name: "b120427a048" Sample ID: "GLP10-01-02-31-017; LMS" File: "b120427a.wiff"

Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-TW5R-0 LMS (dil 1:50)" Annotation: ""

Sample Index: 48 Sample Type: Unknown Concentration: N/A Calculated Conc: 1120. ng/mL Acq. Date: 4/28/2012 Acq. Time: 6:07:16 AM Modified: Yes RT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: No Int. Type: Manual Retention Time: 16.0 minArea: 6078162 countsHeight: 9.88e+005 cpsStart Time: 15.5 minEnd Time: 16.1 min

14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5

Time, min

0.0

5.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e58.5e59.0e59.5e5

Intensity, cps

15.97

15.82

Sample Name: "b120427a050" Sample ID: "GLP10-01-02-31-021" File: "b120427a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-TW6R-0 (dil 1:10)" Annotation: ""

Sample Index: 50 Sample Type: Unknown Concentration: N/A Calculated Conc: 287. ng/mL Acq. Date: 4/28/2012 Acq. Time: 6:49:24 AM Modified: Yes RT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: No Int. Type: Manual Retention Time: 16.0 minArea: 7742598 countsHeight: 1.25e+006 cpsStart Time: 15.6 minEnd Time: 16.1 min

14.0 14.5 15.0 15

0.00

5.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e61.10e61.15e61.20e6

Intensity, cps

Sample Name: "b120427a052" Sample ID: "GLP10-01-02-31-021; LMS" File: "b120427a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "DAL-GW-TW6R-0 LMS (dil 1:10)" Annotation: ""

Sample Index: 52 Sample Type: Unknown Concentration: N/A Calculated Conc: 456. ng/mL Acq. Date: 4/28/2012 Acq. Time: 7:31:25 AM Modified: Yes RT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: No Int. Type: Manual Retention Time: 16.0 minArea: 12003443 countsHeight: 1.93e+006 cpsStart Time: 15.5 minEnd Time: 16.1 min

14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5

Time, min

0.0

1.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e61.6e61.7e61.8e61.9e6

Intensity, cps

15.97

15.82

*** Buster J2930203Printing Time: 11:33:49 AMPrinting Date: Wednesday, May 09, 2012

Page 4 of 4

G

Analysis of PFBS, PFH

New On-
Sample Name: "b120427a012" Sample ID: "11012-10-11" File: "b120427a.wiff"

Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "10 ng/mL FC std in Milli Q Water" Annotation: ""

Sample Index: 12 Sample Type: Standard Concentration: 9.98 ng/mL Calculated Conc: 10.7 ng/mL Acq. Date: 4/27/2012 Acq. Time: 5:30:22 PM Modified: No Proc. Algorithm: IntelliQuan - IQA Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: No Int. Type: Valley Retention Time: 15.0 minArea: 2553392 countsHeight: 6.63e+005 cpsStart Time: 14.9 minEnd Time: 15.3 min

13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5

Time, min

0.0

5.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e5

Intensity, cps

15.04

Sample Name: "b120427a020" Sample ID: "Solvent Blank" File: "b120427a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "Milli Q Water TN12-0158" Annotation: ""

Sample Index: 20 Sample Type: Unknown Concentration: N/A Calculated Conc: < 0 Acq. Date: 4/27/2012 Acq. Time: 8:18:29 PM Modified: No Proc. Algorithm: IntelliQuan - IQA Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: No Int. Type: Valley Retention Time: 15.1 minArea: 1272 countsHeight: 2.87e+002 cpsStart Time: 15.0 minEnd Time: 15.2 min

13.0 13.5 14.0 14

0

200400600800

10001200140016001800200022002400260028003000320034003600380040004200440046004800

Intensity, cps

Sample Name: "b120427a021" Sample ID: "LCS-120426-1" File: "b120427a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "5ppb LCS" Annotation: ""

Sample Index: 21 Sample Type: QC Concentration: 4.97 ng/mL Calculated Conc: 4.85 ng/mL Acq. Date: 4/27/2012 Acq. Time: 8:39:30 PM Modified: No Proc. Algorithm: IntelliQuan - IQA Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: No Int. Type: Valley Retention Time: 15.1 minArea: 1207355 countsHeight: 3.12e+005 cpsStart Time: 14.9 minEnd Time: 15.2 min

13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5

Time, min

0.0

2.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e5

Intensity, cps

15.06

17.46 15.52 17.10

Sample Name: "b120427a024" Sample ID: "LCS-120426-4" File: "b120427a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "20ppb LCS" Annotation: ""

Sample Index: 24 Sample Type: QC Concentration: 19.9 ng/mL Calculated Conc: 17.7 ng/mL Acq. Date: 4/27/2012 Acq. Time: 9:42:39 PM Modified: No Proc. Algorithm: IntelliQuan - IQA Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: No Int. Type: Valley Retention Time: 15.1 minArea: 4091996 countsHeight: 1.04e+006 cpsStart Time: 14.9 minEnd Time: 15.3 min

13.0 13.5 14.0 14

0.00

5.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e6

Intensity, cps

*** Buster J2930203Printing Time: 11:32:19 AMPrinting Date: Wednesday, May 09, 2012

Page 1 of 3

G

Analysis of PFBS, PFH

New On-
Sample Name: "b120427a027" Sample ID: "LCS-120426-7" File: "b120427a.wiff"

Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "50ppb LCS" Annotation: ""

Sample Index: 27 Sample Type: QC Concentration: 49.7 ng/mL Calculated Conc: 49.1 ng/mL Acq. Date: 4/27/2012 Acq. Time: 10:45:36 PM Modified: No Proc. Algorithm: IntelliQuan - IQA Min. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: No Int. Type: Valley Retention Time: 15.1 minArea: 10271750 countsHeight: 2.53e+006 cpsStart Time: 14.9 minEnd Time: 15.3 min

13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5

Time, min

0.0

1.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e61.6e61.7e61.8e61.9e62.0e62.1e62.2e62.3e62.4e62.5e6

Intensity, cps

15.07

Sample Name: "b120427a034" Sample ID: "GLP10-01-02-31-005" File: "b120427a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-TW2R-0 (dil 1:50)" Annotation: ""

Sample Index: 34 Sample Type: Unknown Concentration: N/A Calculated Conc: 563. ng/mL Acq. Date: 4/28/2012 Acq. Time: 1:12:43 AM Modified: Yes RT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: No Int. Type: Manual Retention Time: 15.1 minArea: 2678296 countsHeight: 5.69e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min

13.0 13.5 14.0 14.

0.0

5.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e5

Intensity, cps

Sample Name: "b120427a036" Sample ID: "GLP10-01-02-31-005; LMS" File: "b120427a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-TW2R-0 LMS (dil 1:50)" Annotation: ""

Sample Index: 36 Sample Type: Unknown Concentration: N/A Calculated Conc: 1520. ng/mL Acq. Date: 4/28/2012 Acq. Time: 1:54:50 AM Modified: Yes RT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: No Int. Type: Manual Retention Time: 15.0 minArea: 6734170 countsHeight: 1.60e+006 cpsStart Time: 14.8 minEnd Time: 15.2 min

13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5

Time, min

0.0

1.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e6

Intensity, cps

15.05

14.91

Sample Name: "b120427a038" Sample ID: "GLP10-01-02-31-009" File: "b120427a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-TW3R-0 (dil 1:100)" Annotation: ""

Sample Index: 38 Sample Type: Unknown Concentration: N/A Calculated Conc: 1210. ng/mL Acq. Date: 4/28/2012 Acq. Time: 2:36:55 AM Modified: Yes RT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: No Int. Type: Manual Retention Time: 15.1 minArea: 2858749 countsHeight: 6.16e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min

13.0 13.5 14.0 14.

0.0

5.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e5

Intensity, cps

*** Buster J2930203Printing Time: 11:32:19 AMPrinting Date: Wednesday, May 09, 2012

Page 2 of 3

G

Analysis of PFBS, PFH

New On-
Sample Name: "b120427a040" Sample ID: "GLP10-01-02-31-009; LMS" File: "b120427a.wiff"

Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "DAL-GW-TW3R-0 2000ppb LMS (dil 1:100)" Annotation: ""

Sample Index: 40 Sample Type: Unknown Concentration: N/A Calculated Conc: 3160. ng/mL Acq. Date: 4/28/2012 Acq. Time: 3:18:57 AM Modified: Yes RT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: No Int. Type: Manual Retention Time: 15.0 minArea: 6984904 countsHeight: 1.66e+006 cpsStart Time: 14.7 minEnd Time: 15.2 min

13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5

Time, min

0.0

1.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e61.6e6

Intensity, cps

15.03

14.90

*** Buster J2930203Printing Time: 11:32:19 AMPrinting Date: Wednesday, May 09, 2012

Page 3 of 3

G

Analysis of PFBS, PFH

New On-
ATTACHMENT C: ANALYTICAL METHOD(S) 

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 65 of 89
Page 66 of 89
ETS-8-044.1 Page 2 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Water by LC/MS/MS; 

Direct Injection Analysis

1 Scope and Application

This method describes the direct injection analysis of perfluorinated compounds (PFCs) from 

water matrices using high-performance liquid chromatography tandem mass spectrometry 

(HPLC/MS/MS). The method is generally applicable but not limited to the measurement of 

perfluoroalkyl sulfonamides and perfluorinated alkyl acids (PFAAs) such as perfluorosulfonic 

acids (PFSAs) and perfluorocarboxylic acids (PFCAs) (Table 1). Water samples containing 

heavy particulate may require preparation by an alternate method such as ETS-8-154 

“Determination of Perfluorinated Acids, Alcohols, Amides, and Sulfonates In Water By Solid 

Phase Extraction and High Performance Liquid Chromatography/Mass Spectrometry”. The 

method is applicable to both external standard and internal standard calibration1

. 

Table 1. Representative Target Analytes

Acronym Analyte

Chemical Abstract Services 

Registry Number (CASRN)

PFBA (C4 Acid) Perfluorobutanoic acid 375-22-4

PFPeA (C5 Acid) Perfluoropentanoic acid 2706-90-3

PFHxA (C6 Acid) Perfluorohexanoic acid 307-24-4

PFHpA (C7 Acid) Perfluoroheptanoic acid 375-85-9

PFOA (C8 Acid) Perfluorooctanoic acid 335-67-1

PFNA (C9 Acid) Perfluorononanoic acid 375-95-1

PFDA (C10 Acid) Perfluorodecanoic acid 335-76-2

PFUnA (C11 Acid) Perfluoroundecanoic acid 2058-94-8

PFDoA (C12 Acid) Perfluorododecanoic acid 307-55-1

PFTrDA (C13 Acid) Perfluorotridecanoic acid 72629-94-8

PFBS (C4 Sulfonate) Perfluorobutanesulfonic acid 375-73-5

PFHS (C6 Sulfonate) Perfluorohexanesulfonic acid 355-46-4

PFOS (C8 Sulfonate) Perfluorooctanesulfonic acid 1763-23-1

FBSA (C4 Sulfonamide Perfluorobutanesulfonamide 30334-69-1

FOSA (C8 Sulfonamide) Pefluorooctanesulfonamide 754-91-6

The Minimum Reporting Level (MRL) is the Limit of Quantitation (LOQ) that meets Data 

Quality Objectives (DQOs) that are developed based on the intended use of this method. 

Method Flexibility – This is a performance-based method and may be generally applied to the 

determination of perfluorinated compounds in water matrices when analysis batch quality 

control (QC) criteria are met2

. Each set of samples are prepared in an analysis batch with 

calibration standards, LCSs, blanks, and continuing calibration check standards analyzed on 

the same instrument during a time period that begins and ends with the analysis of the 

appropriate continuing calibration check standards. The laboratory is permitted to modify the 

LC column, mobile phase composition, LC conditions, and MS/MS conditions. Method 

modifications should be considered to improve method performance or to meet data quality 

objectives for the study. In all cases where method modifications are implemented, the batch 

 

1 The method is supported by validation with internal standard calibration for C4-C13 PFCAs, C4, C6, and C8 PFSAs, 

and C8 perfluoroalkane sulfonamide in laboratory control samples under 3M method validation E11-0667.

2 Guidance for establishing method QC Criteria based on a.) FDA May 2001, “Guidance for Industry, Bioanalytical 

Method Validation”, b.) EPA Method 537, and c.) European Commission: Guidance for Generating and Reporting 

Methods of Analysis in Support of Pre-registration Data Requirements for Annex II (Part A, section 4) and Annex III 

(Part A,section 5) of Directive 91/414, SANCO/3029/99 rev. 4 (11/07/00).

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 67 of 89
ETS-8-044.1 Page 3 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Water by LC/MS/MS; 

Direct Injection Analysis

analytical QCs (section 9) must be completed and pass QC acceptance criteria (section 13) if 

the data from the analytical batch are to be reported. 

2 Method Summary

Water samples are analyzed as neat aqueous sample or as solvent diluted aqueous samples 

by direct injection using LC/MS/MS. Samples containing heavy particulate may not be 

suitable for analysis by this method. Samples containing suspended particulate should be 

centrifuged or filtered prior to removing a sample aliquot or diluting with solvent. The water 

sample is mixed well prior to removing an aliquot or diluting, if necessary, with ASTM Type I 

water, HPLC water, other suitable water, or solvent (methanol). 

Quantitation is by stable isotope internal standard calibration in laboratory reagent water. All 

perfluorinated compounds (PFCs) target analyte concentrations of perfluorosulfonic acids 

(PFSAs) and perfluorocarboxylic acids (PFCAs) are reported as anions and corrected for 

their salt or free acid forms. Alternatively, quantitation may be performed by external 

standard calibration. 

This is a performance-based method. Method uncertainty for each target analyte is 

determined for each analytical batch using multiple laboratory control spikes at multiple 

concentrations. This method also requires that the precision and accuracy for each sample 

be determined using field matrix spikes to verify that the method is applicable to each sample 

matrix. 

Calibration standards for PFUnA, PFDoA, PFTrDA, and FOSA have been found to be 

unstable for more than 2 days in 100% water. Samples requiring analysis for these 

compounds by this method should be diluted 1:1 with methanol and analyzed against a 

calibration curve prepared in 1:1 synthetic groundwater:MeOH.

3 Definitions 

3.1 Analysis Batch 

A set of study samples that are prepared with calibration standards, laboratory control 

samples, and procedural blanks, and analyzed on the same instrument during a time period 

that begins and ends with the analysis of the appropriate continuing calibration check 

standards.

3.2 Analytical Sample 

A portion of a laboratory sample prepared for analysis.

3.3 Calibration Standard

A solution prepared by spiking a known volume of the Working Standard (WS) into a 

predetermined amount of ASTM Type I, HPLC grade water, or other suitable water (i.e. 

matrix water), and analyzed according to this method. Calibration standards are used to 

calibrate the instrument response with respect to analyte concentration.

3.4 Laboratory Duplicate Sample (LDS, or Lab Dup)

A laboratory duplicate sample is a separate aliquot of a sample taken in the analytical 

laboratory that is analyzed separately with identical procedures. Analysis of LDSs compared 

to that of the first aliquot give a measure of the precision associated with laboratory 

procedures, but not with sample collection, preservation, or storage procedures.

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 68 of 89
ETS-8-044.1 Page 4 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Water by LC/MS/MS; 

Direct Injection Analysis

3.5 Field Blank (FB)/Trip Blank (TB)

ASTM Type I, HPLC grade water, or other suitable water, placed in a sample container in the 

laboratory and treated as a sample in all respects, including exposure to sampling site 

conditions, storage, preservation and all analytical procedures. The purpose of the TB is to 

determine if test substances or other interferences are present in the field environment. This 

sample is also referred to as a Trip Blank. 

3.6 Field Duplicate Sample (FDS, Field Dup)

A sample collected in duplicate at the same time from the same location as the sample. The 

FDS is handled under identical circumstances and treated exactly the same throughout field 

and laboratory procedures. Analysis of the FDS compared to that of the first sample gives a 

measure of the precision associated with sample collection, preservation and storage, as well 

as with laboratory procedures.

3.7 Field Matrix Spike (FMS)

A sample to which known quantities of the target analytes, ISs and SRSs are added to the 

sample bottle in the laboratory before the bottles are sent to the field for collection of aqueous 

samples. A known, specific volume of sample must be added to the sample container 

without rinsing. This may be accomplished by making a “fill to this level” line on the outside of 

the sample container. The FMS is analyzed to ascertain if any matrix effects, interferences, 

or stability issues may complicate the interpretation of the sample analysis.

3.8 Trip Blank Matrix Spike (TBMS) 

An aliquot of ASTM Type I, HPLC grade water, or other suitable water, to which known 

quantities of the target analytes, ISs and SRSs are added in the laboratory prior to the 

shipment of the collection bottles. The TBMS is analyzed exactly like a study sample to help 

determine if the method is in control and whether a loss of analyte or analytical bias could be 

attributed to sample holding time, sample storage and/or shipment issues. A low and high 

TBMS are appropriate when expected sample concentrations are not known or may vary.

3.9 Internal Standard (IS)

A compound added to each study sample, calibration standard, laboratory control samples, 

and procedural blanks at a consistent level (typically around 1 ng/mL). The internal 

standard(s) are stable isotope labeled versions of the target analytes. The area count ratio of 

the target analyte to the internal standard is used for calibration. Surrogate ISs are applied 

when stable isotope ISs of target analytes are unavailable. A surrogate IS is not necessarily 

a stable isotope labeled version of the target analyte, but is treated as an internal standard for 

quantitation.

3.10 Laboratory Control Sample (LCS)

An aliquot of control matrix to which known quantities of the target analytes, ISs and SRSs

(when applicable) are added in the laboratory at the time when samples are aliquotted. At 

least three levels (two levels for SRSs) in triplicate are included, one generally at the low end 

of the calibration curve and one near the mid range and the upper end of the curve. The 

LCSs are analyzed exactly like a laboratory sample to determine whether the stability of the 

standards. LCSs should be prepared each day samples are aliquoted. 

3.11 Laboratory Matrix Spike (LMS)

A laboratory matrix spike is an aliquot of a sample to which known quantities of target 

analytes, ISs and SRSs (when applicable) are added in the laboratory. The LMS is analyzed 

exactly like a laboratory sample to determine whether the sample matrix contributes bias to 

the analytical results. The endogenous concentrations of the analytes in the sample matrix 

must be determined in a separate aliquot and the measured values in the LMS corrected for 

these concentrations. LMSs are optional for analysis of aqueous samples.

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 69 of 89
ETS-8-044.1 Page 5 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Water by LC/MS/MS; 

Direct Injection Analysis

3.12 Laboratory Sample

A portion or aliquot of a sample received from the field for testing.

3.13 Limit of Quantitation (LOQ)

The lower limit of quantitation (LLOQ) for an analytical batch is the lowest concentration that 

can be reliably quantitated within the specified limits of precision and accuracy. The LLOQ is 

generally selected as the lowest non-zero standard in the calibration curve that meets method 

acceptance criteria. The LLOQ for each target analyte is established for each analysis batch 

as the lowest calibration standard with area counts at least twice that of the average area 

counts of the procedural blanks.

The upper limit of quantitation (ULOQ) for an analytical batch is the highest concentration that 

can be reliably quantitated within the specified limits of precision and accuracy. The highest 

standard in the calibration curve that meets method acceptance criteria is defined as the 

ULOQ.

3.14 Method/Procedural Blank

An aliquot of control matrix that is treated exactly like a laboratory sample including exposure 

to all glassware, equipment, solvents, and reagents that are used with other laboratory 

samples. The method blank is used to determine if test substances or other interferences are 

present in the laboratory environment, the reagents, or the apparatus.

3.15 Sample

A sample is an aliquot removed from a larger quantity of material intended to represent the 

original source material.

3.16 Stock Standard Solution (SSS)

A concentrated solution of a single-analyte prepared in the laboratory with an assayed 

reference compound.

3.17 Surrogate Internal Standard

An IS that is not necessarily a stable isotopically labeled target analyte, but is treated as an 

internal standard for quantitation. Surrogate ISs are used when isotopically labeled 

counterparts of the target analyte are not commercially or readily available.

3.18 Surrogate Recovery Standard (SRS)

An isotopically labeled standard, not used as an internal standard, that is added to each 

sample and appropriate QC sample as a means to evaluate the method performance for a 

chemical class of compounds (e.g., PFSAs, PFCAs). 

3.19 Working Standard (WS)

A solution of several analytes prepared in the laboratory from SSSs and diluted as needed to 

prepare calibration standards and other required analyte solutions.

4 Warnings and Cautions

4.1 Health and Safety

The acute and chronic toxicity of the standards for this method have not been precisely 

determined; however, each should be treated as a potential health hazard. The analyst 

should wear gloves, a lab coat, and safety glasses to prevent exposure to chemicals that 

might be present.

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 70 of 89
ETS-8-044.1 Page 6 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Water by LC/MS/MS; 

Direct Injection Analysis

The laboratory is responsible for maintaining a safe work environment and a current 

awareness of local regulations regarding the handling of the chemicals used in this method. 

A reference file of material safety data sheets (MSDS) should be available to all personnel 

involved in these analyses.

4.2 Cautions

The analyst must be familiar with the laboratory equipment and potential hazards including, 

but not limited to, the use of solvents, pressurized gas and solvent lines, high voltage, and 

vacuum systems. Refer to the appropriate equipment procedure or operator manual for 

additional information and cautions.

5 Interferences

During sample preparation and analysis, major potential contaminant sources are reagents 

and glassware. All materials used in the analyses shall be demonstrated to be free from 

interferences under conditions of analysis by running method blanks.

Parts and supplies that contain Teflon® should be avoided or minimized due to the possibility 

of interference and/or contamination. These may include, but are not limited to: wash bottles, 

Teflon® lined caps, autovial caps, HPLC parts, etc.

The use of disposable micropipettes or pipettes to aliquot standard solutions is recommended 

to make calibration standards and matrix spikes.

6 Instrumentation, Supplies, and Materials

6.1 Instrumentation

Analytical balance capable of reading to 0.0001g 

HPLC/MS/MS or HPLC/MS system, as described in Section 10. 

6.2 Supplies and Materials

Sample collection bottles—HDPE (e.g., NalgeneTM) wide-mouth bottles with screw cap. 

Note: Do not use fluorinated or Teflon® bottles or lined caps.

Coolers or boxes for sample shipment.

15-mL and 50-mL disposable polypropylene centrifuge tubes.

Class A pipettes and volumetric flasks, various.

2 mL HPLC autovials 

Disposable pipettes, polypropylene or glass as appropriate

Centrifuge capable of spinning 15-mL and 50-mL polypropylene tubes at 3000 rpm.

7 Reagents and Standards

Note: Suppliers and catalog numbers are for illustrative purposes only. Equivalent 

performance may be achieved using chemicals obtained from other suppliers. Do not use a 

lesser grade of chemical than those listed. 

7.1 Chemicals

Water – Milli-Q, HPLC grade, or other suitably appropriate sources

Calcium Acetate - A.C.S. Reagent Grade

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 71 of 89
ETS-8-044.1 Page 7 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Water by LC/MS/MS; 

Direct Injection Analysis

Magnesium Acetate - A.C.S. Reagent Grade

Methanol – HPLC grade

Ammonium Acetate – A.C.S. Reagent Grade

7.2 Representative Target Analytes, ISs, and SRSs

PFBA, Heptafluorobutyric Acid, (C4 Perfluorinated Acid)

PFPeA, Nonafluoropentanoic Acid (C5 Perfluorinated Acid)

PFHxA, Perfluorohexanoic Acid (C6 Perfluorinated Acid)

PFHpA, Tridecafluoroheptanoic Acid, (C7 Perfluorinated Acid)

PFOA, Ammonium perfluorooctanoate, (C8 Perfluorinated Acid)

PFNA, Heptadecafluorononanoic Acid, (C9 Perfluorinated Acid)

PFDA, Nonadecafluorodecanoic Acid (C10 Perfluorinated Acid)

PFUnA, Perfluoroundecanoic Acid, (C11 Perfluorinated Acid)

PFDoA, Perfluorododecanoic Acid, (C12 Perfluorinated Acid)

PFTrDA, Perfluorotridecanoic Acid, (C13 Perfluorinated Acid)

FBSA, Perfluorobutanesulfonamide

FOSA, Perfluorooctanesulfonylamide 

PFBS, Potassium Perfluorobutanesulfonate

PFHS, Perfluorohexanesulfonate

PFOS, Potassium perfluorooctanesulfonate

PFOA [1,2, 3, 4-

13C], 13C4-isotopically labeled perfluorooctanoic acid (SRS)

PFOS [1,2, 3, 4-

13C], 13C4-isotopically labeled Perfluorooctanesulfonate (SRS)

PFUnA [1,2-

13C], 13C2-isotopically labeled Perfluoroundecanoic acid (SRS)

A custom mix of ISs in a methanolic solution containing ([1,2,3,4-

13C4]PFBA, [1,2 - 13C2]PFHxA, [1,2,3,4,5,6,7,8-

13C8]PFOA, [1,2,3,4,5,6,7,8,9-13C9]PFNA, [1,2 -13C2]PFDA, 

[1,2,3,4,5,6,7 -13C7]PFUnA, [1,2 -

13C2]PFDoA, [1,2,3 -13C3]PFHS, [1,2,3,4,5,6,7,8-13C8]PFOS, 

and [1,2,3,4,5,6,7,8-

13C8]PFOSA (Wellington Laboratories, Guelph, ON) in combination with 

added ([1,2,3,4,5-

13C5]PFPeA, ([1,2,3,4-13C4]PFHpA, and [18O2]PFBS can be used to prepare 

a stock IS solution. Alternatively, individual stable isotope ISs can be used to prepare a stock 

IS mixture.

Other ISs can be applied.

7.3 Reagent Preparation

2 mM Ammonium acetate solution (Analysis)—Weigh 0.3 g of Ammonium acetate and 

dissolve in 2.0 L of reagent water. 

Synthetic Groundwater (containing 25 ppm Ca and Mg) – Weigh 0.61 g of Calcium Acetate 

and 0.92 g of Magnesium Acetate and dissolve in 6.0 L of reagent water.

Note: Alternative volumes may be prepared as long as the ratios of the solvent to solute ratios 

are maintained.

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 72 of 89
ETS-8-044.1 Page 8 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Water by LC/MS/MS; 

Direct Injection Analysis

7.4 Stock Standard Solution (SSS) and Working Standard Solution 

Preparation

The following standard preparation procedure serves as an example. Weighed amounts and 

final volumes may be changed to suit the needs of a particular study. For example, μL 

volumes may be spiked into volumetric flasks when diluting stock solutions to appropriate 

levels.

100 μg/mL target analyte SSSs—Weigh out 10 mg of analytical standard (corrected for 

percent salt, acid [ETS-4-031] and purity) and dilute to 100 mL with methanol or other 

suitable solvent, in a 100 mL volumetric flask. Transfer to a 125 mL LDPE bottle or other 

suitable container. Prepare a separate solution for each analyte. Expiration dates and 

storage conditions of stock solutions should be assigned in accordance with laboratory 

standard operating procedure. An example of purity and salt correction is given below for 

PFOS.

moclecular salt of weight

molecular anion of weight factor correction salt =

9275.0

538

499 PFOS (K factor correction )salt == +

10 mg C8F17S03

-

K+ with purity 90% = 8.35 mg C8F17S03

– (10 mg*0.90*0.9275=8.35 mg)

10 μg/mL (10,000 ng/mL) mixed working standard—Add 5.0 mL each of the 100 μg/mL 

SSSs to a 50 mL volumetric flask and bring up to volume with solvent. 

1 μg/mL (1,000 ng/mL) mixed working standard—Add 0.5 mL of the 100 μg/mL SSSs to a 

50 mL volumetric flask and bring up to volume with solvent. 

0.1 μg/mL (100 ng/mL) mixed standard—Add 0.05 mL of the 100 μg/mL SSSs to a 50 mL 

volumetric flask and bring up to volume with solvent. 

Storage Conditions—Store all SSSs and working standards in accordance with laboratory 

standard operating procedure or in a refrigerator at 4°±2°C for a maximum period of 6 months 

from the date of preparation.

7.5 Calibration Standards

Calibration can be performed by IS or external calibration. Using the working standards 

described above, prepare calibration solutions in ASTM Type I water, HPLC water, other 

suitable water, or a mixture of solvent and water using the information in Table 2 as a 

guideline. Note: Volumes of water or water/solvent mixtures and working standards may be 

adjusted to meet the data quality objectives addressed in the general project outline. 

Calibration levels other than those listed below can be prepared as needed.

For the quantitation of PFOA and PFOS, reference materials of certified mixed linear and 

branched isomer are preferred. Alternately, reference materials of primarily linear isomers of 

PFOA and/or PFOS may be used, however, when quantitating with predominantly linear 

reference standards, additional LCS samples containing both linear and branched isomers of 

PFOA and PFOS are required3

. 

7.5.1 Internal Standard (IS) and Surrogate Recovery Standard (SRS) 

For IS calibration, stable isotope internal standards of each target analyte or appropriate 

surrogate ISs should be spiked at the same level in all calibration standards. Once the 

calibration standards have been prepared as stated above in Section 7.5, all calibration 

standards are spiked with a separate internal standard spiking solution. Typically the 

 

3 A report summarizing an assessment of the use of reference standards containing certified linear and branched isomers of 

PFOA/PFOS can be found in 3M report E11-0560.

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 73 of 89
ETS-8-044.1 Page 9 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Water by LC/MS/MS; 

Direct Injection Analysis

concentration of the internal standard is consistent with the internal standard concentration 

expected in the samples being prepared, usually 1 ng/mL. The concentration of the internal 

standard spiking solution is typically 2 μg/mL. A separate zero point or method blank is 

typically prepared at the same time as the calibration standards, using the same solution 

used to prepare the standards (ASTM Type I water, HPLC water, other suitable water, or a 

solvent/water mixture), and is spiked with the internal standard at the same concentration as 

the calibration curve, typically at 1 ng/mL. 

If the samples being analzyed were pre-spiked with SRSs, the calibration curve prepared in 

Section 7.5 is spiked with a separate SRS spiking solution. Typically, the sample bottles are 

spiked with a SRS at 0.1 ng/mL. The final calibration curve must consist of at least six 

calibration points after analysis. The following table provides an example of spike 

concentrations and volumes used to achieve a multi-point extracted calibration curve with 

internal standard and surrogate standard.

Table 1 lists recommended stable isotope internal standards for several PFSA and PFCA 

target compounds. A custom mix of isotopically labeled target analytes in a methanolic 

solution containing ([1,2,3,4-

13C4]PFBA, [1,2 -13C2]PFHxA, [1,2,3,4,5,6,7,8-13C8]PFOA, 

[1,2,3,4,5,6,7,8,9-

13C9]PFNA, [1,2,3,4,5,6 -13C6]PFDA, [1,2,3,4,5,6,7 -13C7]PFUnA, [1,2 - 13C2]PFDoA, [1,2,3-

13C3]PFHS, [1,2,3,4,5,6,7,8-13C8]PFOS, and [1,2,3,4,5,6,7,8-13C8]FOSA 

(Wellington Laboratories, Guelph, ON) in combination with added ([1,2,3,4,5-

13C5]PFPeA, 

([1,2,3,4-

13C4]PFHpA, and [18O2]PFBS can be used to prepare a stock IS solution. 

Alternative sources of certified stable isotope labeled target analytes are applicable. 

Alternatively, individual stable isotope ISs can be used to prepare a stock IS mixture. The 

table below lists the recommended stable isotope ISs and SRSs applied in the method. 

Other stable isotope ISs and SRSs of target analytes not listed in the table may be used if 

supported by validation and/or analysis batch QCs meeting method acceptance criteria (e.g., 

[

13C2]-PFOA). The same internal standard should be used for a given analyte throughout the 

entire project/study. Note: some of the compounds listed below are appropriate to use as 

surrogate ISs when a stable isotope IS of a target analyte is not available. Generally, 

surrogate isotopically labeled PFCAs are used for PFCAs, and surrogate isotopically labeled 

PFSAs are used for PFSAs.

Table 2 provides examples of spike concentrations and volumes used to achieve a multi-point 

calibration curve with ISs and SRSs. 

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 74 of 89
ETS-8-044.1 Page 10 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Water by LC/MS/MS; 

Direct Injection Analysis

Table 1. Stable Isotope PFCAs and PFSAs used for ISs and SRSs

Compound Name Synonym or Acronym Analytical Purpose Reference Standard 

Source

13C4-Perfluorobutanoic acid [1,2,3,4-13C4]PFBA IS for PFBA Wellington Labs 

(Mix or Individual)

13C4-Perfluoropentanoic acid [1,2,3,4,5-13C5]PFPeA IS for PFPeA Wellington Labs 

(Mix or Individual)

13C2-Perfluorohexanoic acid [1,2 -13C2]PFHxA IS for PFHxA Wellington Labs 

(Mix or Individual) 

13C4-Perfluoroheptanoic acid [1,2,3,4-13C4]PFHpA IS for PFHpA Wellington Labs 

(Mix or Individual)

13C8-Perfluorooctanoic acid [1,2,3,4,5,6,7,8-13C8]PFOA IS for PFOA and [1,2,3,4 13C4]PFOA

Wellington Labs 

(Mix or Individual)

13C9-Perfluorononanoic acid [1,2,3,4,5,6,7,8,9-13C9]PFNA IS for PFNA Wellington Labs 

(Mix or Individual)

13C6-Perfluorodecanoic acid [1,2,3,4,5,6 -13C6]PFDA IS for PFDA Wellington Labs 

(Mix or Individual)

13C7-Perfluoroundecanoic acid [1,2,3,4,5,6,7 -13C7]PFUnA IS for PFUnA Wellington Labs 

(Mix or Individual)

13C2-Perfluorododecanoic acid [1,2 -13C2]PFDoA IS for PFDoA, *PFTA Wellington Labs 

(Mix or Individual)

18O2-Ammonium Perfluorobutane sulfonate [18O2]PFBS IS for PFBS RTI International 

(Individual)

13C3-Ammonium Perfluorohexane sulfonate [1,2,3-13C3]PFHS IS for PFHS Wellington Labs 

(Mix or Individual)

13C8-Sodium Perfluorooctane sulfonate [1,2,3,4,5,6,7,8-13C8]PFOS IS for PFOS and 

PFOS[1,2,3,4 13C4], 

Wellington Labs 

(Mix or Individual)

13C8-Perfluorooctanesulfonamide [1,2,3,4,5,6,7,8-13C8]FOSA IS for FOSA

Wellington Labs (mix)

RTI International 

(Individual)

13C4-Perfluorooctanoic acid [1,2,3,4-13C4]PFOA SRS for all PFCAs: C4-C8 Wellington

13C2-Perfluoroundecanoic acid [1,2 -13C2]PFUnA SRS for all PFCAs C9-C13 Wellington

13C8-Perfluorooctane sulfonate [1,2,3,4-13C4]PFOS SRS for all PFSAs: C4, 

C6, and C8 Wellington 

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 75 of 89
ETS-8-044.1 Page 11 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Water by LC/MS/MS; Direct Injection Analysis

Table 2. Example Preparation of Calibration Curve with ISs and SRSs

Sample Description Concentration 

of WS, μg/mL

Volume of 

WS, μL 

Volume of IS 

(2 μg/mL), μL 

Concentration of 

Surrogate, μg/mL

Volume of 

Surrogate, μL 

Volume of ASTM Type I Water, or 

other suitable solvent (1), mL

0.025 ng/mL curve point 0.10 25 50 0.20 12.5 100

0.030 ng/mL curve point 0.10 30 50 0.20 15 100

0.04 ng/mL curve point 0.10 40 50 0.20 20 100

0.05 ng/mL curve point 0.10 50 50 0.20 25 100

0.1 ng/mL curve point 0.10 100 50 0.20 50 100

0.25 ng/mL curve point 0.10 250 50 0.20 125 100

0.5 ng/mL curve point 1.0 50 50 0.20 250 100

1 ng/mL curve point 1.0 100 50 0.20 500 100

2.5 ng/mL curve point 10.0 25 50 10.0 25 100

5.0 ng/mL curve point 10.0 50 50 10.0 50 100

10.0 ng/mL curve point 10.0 100 50 10.0 100 100

25.0 ng/mL curve point 10.0 250 50 10.0 NA 100

50.0 ng/mL curve point 10.0 500 50 10.0 NA 100

75.0 ng/mL curve point 10.0 750 50 10.0 NA 100

100 ng/mL curve point 10.0 1000 50 10.0 NA 100

 

N/A - Not Applicable

(1) Samples requiring analysis for PFUnA, PFDoA, PFTrDA, and FOSA should be analyzed against a calibration curve prepared in 1:1 synthetic groundwater:MeOH.

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 76 of 89
ETS-8-044.1 Page 12 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Water by LC/MS/MS; Direct Injection 

Analysis

8 Sample Collection and Bottle Preparation

Sample collection bottles are prepared by 3M Environmental Laboratory (or subcontract supplier) personnel for 

shipment at ambient temperature to the collection site. Typically, four separate collection bottles are 

associated with a single collection site: sample, field duplicate sample, low field matrix spike, and high field 

matrix spike. Alternatively, the sample and field duplicate sample may contain SRSs in lieu of additional target 

analyte low field matrix spike and target analyte high field matrix spike samples. Depending on the scope of 

the project, additional replicates of the field sample and field matrix spikes may be added. Also, it is not 

uncommon for additional mid-level field matrix spikes to be collected if the expected sample concentrations are 

truly unknown or could span a large concentration range. 

High-density polyethylene (HDPE) wide-mouth Nalgene bottles are used for the sample collection containers. 

(Volumes of the bottles may vary depending on how much sample is required to meet data quality objectives.) 

Sample collection volumes are project specific and based on data quality objectives. The Nalgene bottles do 

not require any pretreatment prior to use. Typically, placement of a sample bottle volumetric “fill to here” line is 

done by using a sample bottle marker template. Alternatively, bottles may be weighed prior to bottle 

preparation and weighed again after samples have been collected.

All bottles should be clearly labeled to indicate its intended use as a sample, field sample duplicate, low field 

matrix spike, high field matrix spike, sample/SRS field matrix spike, field duplicate sample/SRS field matrix 

spike, trip blank, or trip blank matrix spike. If each location has different designated spike levels, the label 

should also clearly indicate the sample location designation. Generally, a set of bottles for a given collection 

site are prepared then grouped together in plastic bags for organizational purposes. For each sample 

collection event, at least one set of trip blank and trip blank matrix spikes are prepared. 

Bottle preparation should be documented in a Note to File or on a sample preparation worksheet and should 

include the following information: date prepared, total number of bottles prepared, number of sample sites, the 

standard identification numbers and spike volumes used to prepare spiked bottles, the “fill to here” volume, and 

any other pertinent information needed for reconstructibility of the data. The Note to File will be included in the 

final data package for the project. 

Samples are collected in the field and shipped to the laboratory at ambient temperature.

8.1 Field Matrix Spike Sample (FMS)

Field matrix spike samples are a requirement of the method. A FMS sample is defined as a QC sample to 

which known quantities of appropriate target analytes are added to the sample bottle in the field or in the 

laboratory before the bottles are sent to the field. The sample and field duplicate sample may contain 

appropriate SRSs in lieu of target analyte FMS samples. Sample quantities are determined volumetrically or 

gravimetrically. A known, specific volume or weight of sample is added to the sample container without

rinsing. Volumetric sample measurements may be acquired by a laboratory applied “fill to this level” line on the 

outside of the sample container. Target analyte FMS samples should be spiked at approximately 0.5-10 times 

the expected analyte concentration in the sample. If the expected range of analyte concentrations is unknown, 

multiple spikes at varying levels may be prepared to increase the likelihood that a spike at an appropriate level 

is made. Typically a low and a high target analyte spike are prepared for each sampling location. In those 

instances where SRSs are to be used in lieu of target analyte FMS samples, the sample and field duplicate 

sample are spiked at approximately 2-5 times the target LOQ. The FMS is analyzed to ascertain if matrix 

effects or sample holding time contributes bias to the analytical results. For the sample bottles designated for 

matrix spikes, an appropriate volume of matrix spiking solution is added to the empty bottle prior to sampling. 

The volume of spike solution added should produce the desired final concentration of target analytes once the 

bottle is filled with sample to the “fill to here line”. The matrix spiking solution(s) should be prepared in a 

suitable solvent and contain all of the appropriate target analytes, ISs, and SRSs. The target analyte matrix 

spiking solution is often the same as the working standards used to create the calibration standards. An 

example of a bottle spike is given below.

“Fill to here” volume = 200 mL (A 250 mL Nalgene bottle is used)

Desired Field Spike Concentration = 0.25 ng/mL

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 77 of 89
ETS-8-044.1 Page 13 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Water by LC/MS/MS; Direct Injection 

Analysis

500 μL of a 0.1 μg/mL spiking solution (containing the target analytes) is added to the bottle and the bottle cap 

promptly sealed.

8.2 Internal Standard and Surrogate Recovery Standard 

If analysis of a surrogate recovery standard (SRS) is included in the project objectives, an appropriate volume 

of a surrogate standard solution is added to all the bottles prior to sampling and SPE. Typically sample bottles 

are spiked with surrogate recovery standards at a final desired spike concentration of 0.1 ng/mL.

If quantitation by internal standard (IS) is included in the project objective, an appropriate volume of internal 

standard solution is added to all the bottles prior to sampling and SPE. Typically sample bottles are spiked 

with internal standard at a final desired spike concentration of 1 ng/mL.

For the trip blank, the SRS spike and IS spike is added to the bottle and then ASTM Type I water (HPLC grade 

reagent water or other suitable water may used) is added to the “fill to here” line. The bottle is capped and 

sealing tape may be placed around the outer edge of the cap. Trip blank matrix spikes are prepared by adding 

the appropriate volume of target analyte spiking solution, IS, and SRS spiking solutions and filling the bottle to 

the desired volume with the appropriate water and capping and sealing the cap.

9 Quality Control and Data Quality Objectives

9.1 Data Quality Objectives

This method and required quality control samples is designed to generate data accurate to ±30% with a 

targeted LOQ of 0.025 ng/mL. Any deviations from the quality control measures spelled out below will be 

documented in the raw data and footnoted in the final report.

9.2 Method/Procedural Blanks

The method/procedural blank is zero point calibration standard (which includes ISs) analyzed in a regular basis 

with each analysis batch. At a minimum, method blanks are analyzed prior to instrument calibration, prior to 

the analysis of CCV samples, after every 10 sample injections, and at the end of the analytical run.

The mean area count or area ratios when using internal standard calibration, for each analyte in the method 

blanks must be less than 50% of the area count counts or area ratios when using internal standard calibration, 

of the LOQ standard. The standard deviation of the area counts, or area ratios when using internal standard 

calibration, of these method blanks should be calculated. A specific %RSD acceptance criteria is not specified 

but is assessed on an analytical batch basis. If the mean area counts or area ratios when using internal 

standard calibration, of the method blanks exceed 50% of the LOQ standard, then the LOQ must be raised to 

the first standard level in the curve that meets criteria. Method blanks may be eliminated if technical 

justification can be provided (e.g. the procedural blank was analyzed after an unexpectedly high level sample). 

If any procedural blanks are removed from the LOQ determination, document in the raw data and report as 

appropriate. Laboratory Sample Replicates / Field Duplicate Sample 

Typically, samples are collected in duplicates in the field. The relative percent difference (RPD) of duplicate 

samples should be ≤20% for the precision of sample preparation and analysis to be considered in control. 

Replicate samples not meeting the ≤20% RPD criteria are flagged and reported as outside of QC acceptance 

criteria.

9.3 Laboratory Matrix Spikes (LMSs)

LMSs may be performed in lieu of FMSs if FMSs have previously been performed for the sample matrix. 

Additionally, LMSs may be performed in lieu of FMSs for a sample matrix if the FMS levels were not 

appropriate for determining spike recoveries relative to endogenous levels of target analytes and appropriate 

SRSs. Generally, each sample location represents a different sample and sample matrix. LMSs are prepared 

for each sample and analyzed to determine the matrix effect on spike recovery efficiency of each target analyte 

and appropriate SRSs. LMSs should be prepared at a minimum of one level and in duplicate. LMS 

concentrations should be prepared at approximately 0.5-10 times the endogenous concentration or 

approximately 4-10 times the LOQ concentration of each target analyte. 

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 78 of 89
ETS-8-044.1 Page 14 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Water by LC/MS/MS; Direct Injection 

Analysis

Lab matrix spike recoveries should fall within ± 30% of expected values. Sample data with LMS recovery 

outside of ±30% but within ±50% of the expected value are flagged and reported as outside of QC acceptance 

criteria. Data with LMS recovery outside of ±50% of the expected value are reported as NR, where NR is 

defined as “Not Reportable” data outside of QC acceptance criteria.

9.4 Lab Control Sample

Lab control spikes are prepared for each analysis batch to determine method accuracy and precision. LCSs 

should be prepared at three levels in triplicate for each target analyte and at a minimum of two levels in 

triplicate for appropriate SRSs. Low lab control spikes should be prepared at a concentration in the range of 

approximately four to ten times higher than the targeted lower LOQ, the mid lab control spikes should be 

prepared at a concentration near the mid-point of the calibration curve and the high lab control spikes at 

approximately 80% of the upper LOQ. For each target analyte and SRSs, the percent relative standard 

deviation (method precision) for each control spike level must be less than or equal to 20% and the average 

recovery (method accuracy) for each control spike level must be 80-120%. Sample data for target analytes 

outside of the laboratory control spike acceptance criteria will be handled as follows:

If the average recovery of a spiking level falls outside method acceptance, but at least 67% (6 out of 9) of LCS 

samples are within 20% of their respective nominal value (33% of the QC samples, not all replicates at the 

same concentration, may be outside 20% of nominal value), the average recovery will be flagged as outside 

method acceptance criteria. All LCS samples will be control charted as per ETS-4-026. If the average 

recovery of one of the spiking levels exceeded the analytical method uncertainty as determined by ETS-12-

012, that analytical batch uncertainty will be expanded for that particular study.

If more than 67% of the LCS samples fail to meet method acceptance criteria, the data will not be reported. 

Calibration standards consisting of mixed branched and linear isomer PFOS/PFOA are preferred. However, 

for PFOS/PFOA target analytes, if the calibration standards are comprised of predominantly linear isomers 

only, at least one level of triplicate LCSs should be prepared using PFOS/PFOA which contains a mix of linear 

and branched isomers. These LCSs will be used to demonstrate quantitative equivalency (or quantitative bias) 

of the isomeric mix when using a predominantly linear standard for calibration. The mixed linear and branched 

isomer PFOS/PFOA LCSs recoveries should fall within ± 30% of expected values. Alternatively, in lieu of 

mixed branched and linear isomer PFOS/PFOA LCSs, mixed branched and linear isomer PFOS/PFOA 

TBMSs may be applied to demonstrate method accuracy and precision.

9.5 Field Matrix Spikes (FMSs) / Surrogate Recovery Standards (SRSs)

FMSs are prepared for each sampling location and analyzed to determine the matrix effect and sample holding 

time on the spike recovery of each target analyte and/or appropriate SRS. Generally, each sample location 

represents a different sample and sample matrix. 

FMSs are QC samples to which known quantities of appropriate target analytes are added to the sample bottle 

in the laboratory before the bottles are sent to the field. Typically a low and a high target analyte FMS are 

prepared for each sampling location. The sample and field duplicate sample may contain appropriate SRSs in 

lieu of target analyte low field matrix spike and target analyte high field matrix spike samples. 

Field matrix spike method acceptance criteria are recoveries within ±30% of the expected value. If FMS 

recovery (target analyte or SRS spike) is outside of ±30% of the expected value or could not be assessed 

because the FMS (target analyte) was spiked at an inappropriate level, the sample result is reported as follows: 

1.) If target analyte FMS recovery could not be assessed because the FMS's were at an inappropriate level, 

then Laboratory Matrix Spikes (LMS) may be substituted. If LMS recoveries are within ±30% the data are 

reportable and flagged to indicate that the FMS spikes levels were inappropriate.

2.) If multiple target analyte FMS's were prepared on a sample and the closest FMS level to the reported 

sample meets the ±30% acceptance criteria but additional FMS's are outside the ±30% acceptance range, the 

data are reportable and flagged to indicate that while there were failing FMS’s, the uncertainty will not be 

expanded since the most appropriate spike level passed. 

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 79 of 89
ETS-8-044.1 Page 15 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Water by LC/MS/MS; Direct Injection 

Analysis

3.) If the target analyte FMS recoveries are outside of the ±30% acceptance range but at least 30 acceptable 

historical reportable FMS sample results are available, the data may be reported but flagged with an expanded 

uncertainty and as not meeting FMS criteria.

4.) Sample data with FMS recovery outside of ±30% but within ±50% of the expected value are flagged and 

reported as outside of QC acceptance criteria with an expanded uncertainty.

5.) If FMS recovery is outside of ±50%, the sample result is reported as NR, where NR is defined as “Not 

Reportable” due to noncompliant QC results. 

The targeted fortification levels should be at least 50% of the endogenous level and less than 10 times the 

endogenous level to be used without justification to determine the statement of accuracy for analytical results.

Note: It is possible for bottles utilized for Field Matrix Spike samples to be under-filled or over-filled during 

sample collection. Since this scenario will effect the actual concentration of the FMS sample (surrogate and 

internal standard concentrations will also be effected, if used), it is important that any obvious under-filling or 

over-filling of sample bottles be documented in the data package and taken into account in the FMS, ISs, or 

SRSs recovery calculations. Samples over-filled or under-filled by more than 10% will be require recalculation 

of the FMS, ISs, and SRS true values.

The average of the sample and the field duplicate should be used to calculate the recovery.

10 Procedures

10.1 Water Sample Preparation

This method is applicable to water samples. Samples containing heavy particulate may not be suitable for 

analysis by this method. Samples containing suspended particulate should be centrifuge prior to removing a 

sample aliquot, or filtered. 

• Thoroughly mix sample before removing an aliquot and placing in a labeled autovial. 

• Dilute sample, if necessary, with ASTM Type I water, HPLC water, other suitable water, or solvent 

(methanol). 

• Lab control spikes are prepared for each analysis batch to determine method accuracy and precision. 

LCSs should be prepared at three levels in triplicate for each target analyte and at a minimum of two 

levels in triplicate for appropriate SRSs. Low lab control spikes should be prepared at a concentration 

in the range of approximately four to ten times higher than the targeted lower LOQ, the mid lab control 

spikes should be prepared at a concentration near the mid-point of the calibration curve and the high

lab control spikes at approximately 80% of the upper LOQ. For IS quantitation, stable isotope internal 

standards of each target analyte or appropriate surrogate ISs should be spiked at the same level as the 

samples being analyzed, in all LCSs. 

• If LCSs are being prepared using synthetic groundwater, allow the LCSs samples to equilibrate for a 

minimum of 4 hours before aliquoting for analysis or diluting with solvent (methanol).

11 Sample Analysis - LC/MS/MS

11.1 Instrument Setup

Note: In this example, an Applied Biosystems Sciex API 4000 (API 5000 or API 5500) Tandem Mass 

Spectrometer (LC/MS/MS) is used. Other brands/models of LC/MS/MS instruments as well as single 

quadrupole mass spectrometers (LC/MS) may be used as long as the method acceptance criteria are met. 

Brand names, suppliers, part numbers, and models are for illustrative purposes only. Equivalent performance 

may be achieved using apparatus and materials other than those specified here, but demonstration of 

equivalent performance that meets the requirements of this method is the responsibility of the laboratory. The 

operator must optimize and document the equipment and settings used.

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 80 of 89
ETS-8-044.1 Page 16 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Water by LC/MS/MS; Direct Injection 

Analysis

Establish the LC/MS/MS system and operating conditions equivalent to the following:

Mass Spec: Applied Biosystems API 4000, API 5000, or API 5500

Ion Source: Turbo Ion Spray (ABS)

Mode: Electrospray Negative

Scan Type: MRM (Multiple Reaction Monitoring) 

Computer: Dell DHM

Software: Windows 2000 or Windows XP, Analyst 1.4.2 or higher versions

HPLC: Agilent Series 1100,1200, or 1290

Agilent Quaternary Pump

Agilent Vacuum Degasser

Agilent Autosampler

Agilent Column Oven

Note: One or more C18 HPLC analytical columns (2.1 mm x 100 mm, 5μm or 2.1 mm x 50 mm, 5μm) may be 

attached on-line after the purge valve and before the sample injection port to retard and separate any residue 

contaminants that may be in the mobile phase and/or HPLC system.

HPLC Column: Betasil C18, 4.6mm x 100mm, 5μm (ThermoElectron Corporation)

Column Temperature: 35°C 

Injection Volume: 5μL

Mobile Phase (A): 2mM Ammonium Acetate in ASTM Type I water (See 7.3)

Mobile Phase (B): Methanol

Table 3. Liquid Chromatography Gradient Program. 

Step 

Number

Total Time 

(min)

Flow Rate 

(μL/min)

Percent A

(2 mM ammonium 

acetate)

Percent B

(Methanol)

0 0 750 97.0 3.0

1 2.0 750 97.0 3.0

2 14.5 750 5.0 95.0

3 15.5 750 5.0 95.0

4 16.5 750 97.0 3.0

5 20.0 750 97.0 3.0

Note: Other HPLC gradients may be used as long as the method criteria and project data quality objectives are 

met.

It may be necessary to adjust the HPLC gradient in order to optimize instrument performance. Columns with 

different dimensions (e.g. 2.1mm x 30mm) and columns from different manufacturers (Keystone Betasil C18 

etc.) may be used.

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 81 of 89
ETS-8-044.1 Page 17 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Water by LC/MS/MS; Direct Injection 

Analysis

Table 4 Suggested MRM Transitions for Target Analytes, Surrogates, and Internal Standards

Analyte Analyte Description Mass Transition

Q1 (amu)

Mass Transition

Q3 (amu)

PFBA (C4 Acid) Target 213 169

PFPeA (C5 Acid) Target 263 219

PFHxA (C6 Acid) Target 313 269, 119

PFHpA (C7 Acid) Target 363 319, 169

PFOA (C8 Acid) Target 413 369, 219, 169

PFNA (C9 Acid) Target

463 419, 169, 219

PFDA (C10 Acid) Target 513 469, 269, 219

PFUnA (C11 Acid) Target 563 519, 269, 219

PFDoA (C12 Acid) Target 613 569, 169, 319

PFTA (C13 Acid) Target 663 619, 369, 319

FBSA (C4 Sulfonamide) Target 298 78

FOSA (C8 Sulfonamide) Target 498 78

PFBS (C4 Sulfonate) Target 299 99, 80

PFHS (C6 Sulfonate) Target 399 99, 80

PFOS (C8 Sulfonate) Target 499 80, 99, 130

[1,2,3,4 -

13C4]PFBA IS for PFBA 217 172

[1,2,3,4,5 -

13C5]PFPeA IS for PFPeA 268 223

[1,2 -

13C2]PFHxA IS for PFHxA 315 270

[1,2,3,4- 13C4]PFHpA IS for PFHpA 367 322

[1,2,3,4,5,6,7,8-

13C8]PFOA IS for PFOA 421 376

[1,2,3,4,5,6,7,8,9-

13C9]PFNA IS for PFNA 472 427

[1,2,3,4,5,6 -

13C6]PFDA IS for PFDA 519 474

[1,2,3,4,5,6,7 -

13C7]PFUnA IS for PFUnA 570 525

[1,2 -

13C2]PFDoA IS for PFDoA and PFTA 615 570

[

18O2]PFBS IS for PFBS 303 84

[1,2,3-

13C3]PFHS IS for PFHS 402 80

[1,2,3,4- 13C4]PFOS IS for PFOS 503 80

[1,2,3,4,5,6,7,8-

13C8]FOSA IS for FOSA 507 80

[1,2,3,4-

13C4]PFOA Surrogate (C4-C8 Acids) 417 372

[1,2,3,4- 13C4]PFOS Surrogate(Sulfonates, FOSA) 503 80

[1,2 -

13C2]PFUnA Surrogate (C9-C13 Acids) 565 520

Multiple transitions for monitoring the analytes is an option. The use of one daughter ion is acceptable if data 

sensitivity and selectivity is achieved and provided that retention time criteria are met to assure adequate 

specificity. While the daughter ions may be chosen at the discretion of the analyst, mass transition 99 is 

suggested for PFOS. Quantitation may be performed using the total ion chromatogram (TIC, or summed 

MRMs) for a given analyte. For example, the PFOA TIC would sum all three of the monitored transitions. Use 

of the suggested primary ion is recommended. Retention times may vary slightly, on a day-to-day basis, 

depending on the batch of mobile phase and the gradient, column, guard column(s) used etc. Drift in retention 

times is acceptable within an analytical run, as long as the drift continues through the entire analysis and the 

standards are interspersed throughout the analytical run.

11.2 Calibration Curve

Quantitation is by internal standard or external standard calibration. Calibration standards may be prepared in 

ASTM Type I, HPLC water, other suitable water, or a solvent/water mixture. If internal standard calibration 

does not meet calibration acceptance criteria, external calibration can be applied. See Table 1 for 

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 82 of 89
ETS-8-044.1 Page 18 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Water by LC/MS/MS; Direct Injection 

Analysis

recommended application of available internal standards. Quantitation of PFOA and PFOS is by summed 

analyte-specific mass transitions. 

Analyze the standard curve prior to each set of samples. If internal standards were added to the calibration 

standards area ratios are used to generate the calibration curve. The standard curve may be plotted using a 

linear regression (y = mx + b), weighted 1/x or unweighted, or by quadratic fit (y = ax2 + bx + c), weighted 1/x or 

unweighted, using suitable software. The mathematical method used to calculate the calibration curve should 

be applied consistently throughout a study. Any change should be thoroughly documented in the raw data.

High and/or low points may be excluded from the calibration curves to provide a better fit over the range 

appropriate to the data or because they did not meet the pre-determined acceptance criteria. Low-level curve 

points should also be excluded if their area counts (or area ratio if quantitating by IS) are not at least twice that 

of the average area counts (or area ratio if quantitating by IS) of method and/or solvent blanks. The coefficient 

of determination (r2

) value for the calibration curve must be greater than or equal to 0.990 (or a correlation 

coefficient (r) of 0.995). Each point in the curve must be within ±25% of the theoretical concentration with the 

exception of the LLOQ, which may be within ±30%. Justification for exclusion of calibration curve points will be 

noted in the raw data. A minimum of 6 points will be used to construct the calibration curve. 

If the calibration curve does not meet acceptance criteria, perform routine maintenance or prepare a new 

standard curve (if necessary) and reanalyze. 

11.3 Continuing Calibration Verification (CCV)

Continuing calibration verifications (CCV) are analyzed to verify the accuracy of the calibration curve. Analyze 

a mid-range calibration standard, one of the same standards used to construct the calibration curve, at a 

minimum after every tenth sample, not including solvent blanks, with a minimum of one per sample set. 

Calibration verification injections must be within ±25% to be considered acceptable. The calibration curve and 

the last passing CCV will then bracket acceptable samples. Multiple CCV levels may be used. Samples must 

be bracketed by passing CCVs or the calibration curve and a passing CCV to be reportable.

11.4 System Suitability

A minimum of three system suitability samples should be injected at the beginning of each analytical run, prior 

to the analysis of the calibration curve. Typically these samples are at a concentration near the mid-level of the 

calibration curve and are repeated injections from one autosampler vial. It is suggested that the system 

suitability injections have area counts or area ratios when using internal standard calibration, with a target RSD 

of ≤5% and a target retention time RSD of ≤2%. There is no defined acceptability limit on these results as the 

%RSD value is dependent on the number of MRM transitions being monitored in the LC/MS/MS run or time 

period. Ultimately, any effects on these parameters for the System Suitability samples will also be evident on 

all standards and QC samples analyzed as part of the analysis batch. Any effect of system suitability is 

incorporated within QC acceptance criteria.4

11.5 Sample Analysis and QCs

For each analysis batch, the instrument analysis run sequence should include an initial calibration curve, 

samples, FDSs, interspersed blanks, interspersed CCVs, appropriate QCs (i.e., LCSs, LMSs, FMSs, TBMSs, 

and TBs), and a final CCV or calibration curve bracketing samples and appropriate QCs

Inject the same volume (between 5 - 100μL) of each standard, analytical sample and blank into the instrument 

(unless an on-instrument sample dilution is desired).

Samples containing analytes that are quantitated above the concentration of the highest standard in the curve 

should be further diluted and reanalyzed. 

 

4 3M Environmental Laboratory study E08-0096 evaluated the effect on these results as a function of the number of MRMs being monitored. 

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 83 of 89
ETS-8-044.1 Page 19 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Water by LC/MS/MS; Direct Injection 

Analysis

12 Data Analysis and Calculations

The chromatography analysis software will typically calculate the amount of target analyte in the sample 

extracts using the established calibration curve.

Calculate the percent recovery of the LCS using the following equation:

100%*

) mL

ng Concentrat Spike (ion 

) mL

ng LCSConcentrat (ion 

LCS% recovery =

 

Calculate the percent recovery of the LMS using the following equation:

100%*

) mL

ng Spike Concentrat ( ion

) mL

ng -) Concentrat ( Sample of ion mLng LMS Concentrat ( ion

LMS recovery % =

For samples fortified with known amounts of analyte prior to extraction, use the following equation to calculate 

the percent recovery.

100 

(ng/mL) added Analyte

(ng/mL) sample in found analyte Average- (ng/mL) found analyte Total Recovery = ×

13 Analysis Batch Method Performance Criteria

Any method performance parameters that are not achieved must be considered in the evaluation of the data. 

Nonconformance to any specified parameters must be described and discussed in the final report if the 

Technical Manager (non-GLP study) or Study Director (GLP study) chooses to report the data.

If criteria listed in this method performance section are not met, maintenance may be performed on the system 

and samples reanalyzed, or other actions taken as appropriate. Document all actions in the raw data.

If data are to be reported when performance criteria have not been met, the data must be footnoted on tables 

and discussed in the text of the report.

13.1 System Suitability – Analysis Batch

A minimum of three system suitability samples should be injected at the beginning of each analytical run. 

These samples are run prior to the calibration curve. It is suggested that the system suitability injections have 

area counts with a target RSD of ≤5% and a target retention time RSD of ≤2%. There is no defined 

acceptability limit on these results as the %RSDs are dependent on the number of MRM transitions being 

monitored in the LC/MS/MS run or time period. Any effect of system suitability is incorporated in the QC 

acceptance criteria.

13.2 Calibration and Limit of Quantitation (LOQ) – Analysis Batch

Calibration Curve: The coefficient of determination (r2) value for the calibration curve must be greater than or 

equal to 0.990 corresponding to a correlation coefficient (r) = 0.995. Each point in the curve must be within 

±25% of the theoretical concentration with the exception of the LLOQ, which may be within ±30%. 

CCV Performance: The calibration standards that are interspersed throughout the analytical sequence are 

evaluated as continuing calibration verifications in addition to being part of the calibration curve. The accuracy 

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 84 of 89
ETS-8-044.1 Page 20 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Water by LC/MS/MS; Direct Injection 

Analysis

of each curve point must be within 25% of the theoretical value (within 30% for lowest curve point). Samples 

that are bracketed by CCVs not meeting these criteria must be reanalyzed.

Limits of Quantitation (LOQ): The lower LOQ (LLOQ) is the lowest non-zero active standard in the 

calibration curve; the peak area of the LLOQ must be at least 2X that of the average area counts for all 

prepared procedural blank(s). By definition, the measured value of the LLOQ must be within 30% of the 

theoretical value.

Demonstration of Specificity: Specificity is demonstrated by chromatographic retention time (within 4% of 

standard) and the mass spectral response of unique ions.

13.3 Blanks – Method/Procedural Blanks and Trip

Method/Procedural Blanks: Multiple procedural blanks should be interspersed throughout the analysis batch 

and the analytical sequence. At a minimum, method blanks are analyzed prior to instrument calibration, prior 

to the analysis of CCV samples, after every 10 sample injections, and at the end of the analytical run. 

The mean area counts (or area ratios when using IS calibration) for each analyte must be less than 50% of the 

area count of the LOQ standard. If the area counts of the procedural blanks exceed 50% of the LOQ standard, 

then the LOQ must be raised to the first standard level that meets criteria.

Trip Blank: A trip blank of ASTM Type I water (or lab equivalent) is prepared in a sample container in the 

laboratory and treated as a sample, including exposure to shipping, sampling site conditions, storage, 

preservation and all analytical procedures. The trip blanks results for each analyte are included with the 

reported sample results. 

13.4 Data Accuracy and Precision – Analysis Batch

Lab Control Spikes: The average recovery at each LCS level for each target analyte and appropriate SRS 

should be within 80-120% and the percent relative standard deviation of the recoveries must be less than or 

equal to 20%. If the average recovery of a spiking level falls outside method acceptance, but at least 67% (6

out of 9) of LCS samples are within 20% of their respective nominal value (33% of the QC samples, not all 

replicates at the same concentration, may be outside 20% of nominal value), the average recovery will be 

flagged as outside method acceptance criteria. All LCS samples will be control charted as per ETS-12-012. If 

the average recovery of one of the spiking levels exceeded the analytical method uncertainty as determined by 

ETS-12-012, that analytical batch uncertainty will be expanded for that particular study. The average recovery 

at each LCS level for mixed branched/linear isomer PFOA and PFOS should be within 70-130% and the 

percent relative standard deviation of the recoveries must be less than or equal to 20%. 

Field Duplicates: The relative percent difference (RPD) of duplicate samples should be less than 20% for the 

precision of sample preparation and analysis to be considered in control. Replicate samples not meeting the 

20% RPD criteria are flagged and reported as outside of QC acceptance criteria.

Field Matrix Spikes: FMS acceptance criteria are recoveries within ±30% of the expected value for each 

target analyte and appropriate SRS. Sample data with FMS recovery outside of ±30% but within ±50% of the 

expected value are flagged and reported as outside of QC acceptance criteria. Data with FMS recovery 

outside of ±50% of the expected value are reported as NR, where NR is defined as “Not Reportable” data 

outside of QC acceptance criteria. If FMS recovery could not be assessed because FMSs were at an 

inappropriate level, then Laboratory Matrix Spikes (LMSs) may be substituted. If LMS recoveries are within 

±30% for each target analyte and SRSs the data are reportable but flagged as not meeting the FMS method 

acceptance criteria.

13.5 Analytical Method Uncertainty

Analytical method uncertainty for each target analyte and SRS is determined with control charted historical 

analysis batch LCS data for the method and reported with each analysis batch.5

 Uncertainty determinations 

 

5 Method uncertainty based on INTERNATIONAL ANS/ISO/IED STANDARD 17025 reference (GUM, Guide to the Expression of 

Uncertainty in Measurement). Method application demonstrated in ETS-12-012, citing references: a.) EURACHEM/CITAC Guide, 

“Quantifying Uncertainty in Analytical Measurement,” Second Edition; Editors: S.L.R. Ellison, M. Rosslein, and A. Williams. 

b.)Georgian, Thomas, “Estimation of Laboratory Analytical Uncertainty Using Laboratory Control Samples,” Environmental Testing & 

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 85 of 89
ETS-8-044.1 Page 21 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Water by LC/MS/MS; Direct Injection 

Analysis

are based on INTERNATIONAL ANS/ISO/IED STANDARD 17025 reference (GUM, Guide to the Expression 

of Uncertainty in Measurement) and described in ETS-12-012. At least thirty data points are required for 

determining analytical method uncertainty. The method uncertainty is defined as 2x the standard deviation of 

the percent recoveries of the pooled lab control spikes. While all LCS data points are control charted, only the 

most recent fifty data points are used for determining the method uncertainty.

When less than thirty LCS data points have been generated for a given analyte, the analysis batch LCSs are 

used to determine the data uncertainty. If FMSs meet the ±30% recovery criteria at a level appropriate to the 

endogenous level, and the LCS meet the ±20% recovery criteria, then the uncertainty of the data is determined 

as within 100±20%. 

Analysis batch sample data with FMS recovery outside of ±30% but within ±50% of the expected value are 

flagged and reported as outside of QC acceptance criteria with expanded uncertainties. Data with FMS 

recovery outside of ±50% of the expected value are reported as NR, where NR is defined as “Not Reportable” 

data outside of QC acceptance criteria. If FMS recovery could not be assessed because FMSs were at an 

inappropriate level, then Laboratory Matrix Spikes (LMSs) may be substituted. If LMS recoveries are within 

±30% for each target analyte and appropriate SRSs the data are reportable but flagged as not meeting the 

FMS method acceptance criteria with uncertainties of ±30%. If FMS do not meet the ±30% recovery criteria, 

and historical FMS data does not exist, the analytical uncertainty is evaluated on a sample-by-sample basis, 

the data may be reported with expanded uncertainty and are flagged. 

13.6 Quantitation of PFOA/PFOS – Analysis Batch

Calibration standards consisting of mixed branched and linear isomer PFOS/PFOA are preferred. 

Quantitation is performed by integrating the linear and branched isomers together. Alternately, the linear and 

branched isomers can be integrated separately, applying the appropriate true value to each calibration curve 

point for each isomer. The LCS and samples are then quantitated by integrating the linear and branched 

isomers separately (requires separate analytical results files) and quantitating the resulting peak against the 

linear or branched calibration curve. The results from both integrations are then summed to produce the final 

result. Integrating the linear and branched isomers separately may be helpful for those samples where the 

linear/branched ratios do not closely match those of the reference standards.

However, for PFOS/PFOA target analytes, if the calibration standards are comprised of predominantly linear 

isomers only the method requires the addition of LCSs of mixed branched/linear isomer PFOS/PFOA. The 

purpose of including these LCSs is to demonstrate quantitative equivalency (or quantitative bias) of the 

isomeric mix when using a predominantly linear PFOS or PFOA standard for calibration. Alternatively, in lieu 

of mixed branched and linear isomer PFOS/PFOA LCSs, mixed branched and linear isomer PFOS/PFOA 

TBMSs may be applied to demonstrate method accuracy and precision.

An alternate method of quantitation can be performed whereby only the linear isomer of PFOS/PFOA is 

integrated and used for generating the calibration curve. The LCS and samples are then quantitated by 

integrating the linear and branched isomers separately (requires separate analytical results files) and 

quantitating the resulting peak against the linear calibration curve. The results from both integrations are then 

summed to produce the final result. Integrating the linear and branched isomers separately reduces the oncolumn concentration for those samples that contain both linear and branched isomers of PFOA/PFOS. This 

ensures that the concentration detected is within the a range of the calibration curve that is comparable 

regardless of whether the calibration curve was generated using predominantly linear isomers of PFOS/PFOA 

or linear plus branched isomers of PFOS/PFOA.

14 Pollution Prevention and Waste Management

Waste generated when performing this method will be disposed of appropriately. The original samples will be 

archived at the 3M Environmental Laboratory in accordance with internal procedures.

 

Analysis, November/December 2000. c.)Taylor, B.N. and CE. Kuyatt, NIST Technical Note 1297, 1994 Edition: “Guidelines for 

Evaluating and Expressing the Uncertainty of NIST Measurement Results.”d.)Adams, T.M., “A2LA Guide for the Estimation of 

Measurement Uncertainty in Testing”, July 2002.

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 86 of 89
ETS-8-044.1 Page 22 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Water by LC/MS/MS; Direct Injection 

Analysis

15 Records

Each data package generated for a study must include all supporting information for reconstruction of the data. 

Information for the data package must include, but is not limited to the following items: study or project 

number, sample and standard prep sheets/records, instrument run log (instrument batch records, instrument 

acquisition method, summary pages), instrument results files, chromatograms, calibration curves, and data 

calculations.

16 Affected Documents

None.

17 Revisions

Revision 

Number Summary of Changes 

1 Section 1. Included the use of internal standard calibration by this method. 

Section 2. Included the use of internal standard calibration by this method. Included the 

use of a solvent/water mixture when analyzing for PFUnA, PFDoA, PFTrDA, and FOSA.

Section 3. Added definitions for internal standard, surrogate internal standard, and 

surrogate recovery standard.

Section 6.Removed the details regarding the instrument parameters to section 10 of the 

method. 

Section 7. Updated reference standards to include internal standards and surrogates. 

Changed concentration levels for working standards and included the use of internal 

standards and surrogates.

Section 8. Inserted a new section on sample bottle preparation.

Section 9 Quality Control. This section was previously section 10 in ETS-8-044.0. Updated 

QC criteria to be consistent with method ETS-8-154.4.

Section 10 Procedures. This section was previously section 8 (Sample Handling) in ETS-8-

044.0. Added detail regarding the preparation of LCSs. Included the use of methanol as a 

dilution solvent. 

Section 11 Sample Analysis. This section was previously section 10 in ETS-8-044.0. 

Included the details regarding the instrument parameters. 

Section 12 Data Analysis and Calculations. This section was previously section 11 in ETS8-044.0. Removed the equation for calculating the analytes concentration, indicating that 

this is done by the instrument software.

Section 13 Method Performance. This section was previously section 12 in ETS-8-044.0. 

Updated QC criteria to be consistent with ETS-8-154.4. Added information on the 

determination of analytical method uncertainty and quantitation of PFOA/PFOS.

Section 14 Pollution Prevention. This section was previously section 13 in ETS-8-044.0.

Section 15 Records. This section was previously section 14 in ETS-8-044.0. 

Section 16 Affected Documents. This section was previously section 15 in ETS-8-044.0.

Section 17 Revisions. This section was previously section 16 in ETS-8-044.0. 

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 87 of 89
ATTACHMENT D: DEVIATION(S) 

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 88 of 89
RECORD OF DEVIATION/NONCONFORMANCE

ETS-4-008.7

Documentation of Deviations and Control of Nonconforming Testing

Page 1 of 1

I. Identification

Study / Project No.

GLP10-01-02-31

Date(s) of Occurrence:

4/16/12

Document Number:

ETS-8-044.1

Deviation type  SOP  Equipment Procedure  Method

(Check one)  Protocol  GPO  Other:

II. Description (attach extra pages as needed)

Method Requirements:

1. LCS recovery within ±20% (section 13.4).

2. FMS recovery within ±30% (section 13.4). 

Actual procedure/process:

1. The low set of LCSs had an average recovery of 75.2% for PFOS.

2. The sampling location DAL GW TW1R HS had a PFOS FMS recovery of 65.8%.

III. Actions Taken

(such as amendment issued, SOP revision, etc.)

Corrective Action ( Yes  No) Reference:

Acceptability of the nonconforming work:

1. The non-compliant LCSs will be flagged in the final report. All LCSs were used in the determination of the 

analytical method uncertainty. Since the %bias was less than the calculated method uncertainty for 

PFOS of ±35%, the method uncertainty does not need to be expanded further.

 

2. The sampling location with a FMS recovery outside method acceptance criteria will be flagged in the 

report. Since the analytical method uncertainty was calculated at 35%, the uncertainty for DAL GW 

TW1R does not need to be expanded further. 

Actions:  Halting of Work  Client Notification  Work Recall  Withholding of Report

 Other: Deviations will be noted in final report.

Project Lead/PAI Approval: Date:

Study Director (if GLP): Date:

Sponsor Approval (for GLP protocol deviations): NA Date: NA

Technical Reviewer (optional): NA Date: NA

Laboratory Department Manager Approval: Date:

IV. Authorization to Resume Work

Where halting of work occurred, resumption of work must first be approved by Laboratory Management

Laboratory Department Manager Approval: NA Date: NA

Deviation No. ______________

(assigned by Study Director or Team Leader at the end of study or project)

GLP10-01-02; Interim Report 31 

Analysis of PFBS, PFHS, and PFOS in Groundwater 

New On-Site WWTP Wells - April 2012

Page 89 of 89