Agricultural Livestock and Poultry Program QAD 511 Procedure 

Marketing 1400 Independence Avenue SW, Stop 0258 Effective March 11, 2024

Service Washington, DC 20250 Page 1 of 11

INSTRUMENT APPROVAL PROCESS: 

INSTRUMENT GRADING SYSTEMS FOR BEEF CARCASSES

1. Purpose

The U.S. Department of Agriculture (USDA), Agricultural Marketing Service (AMS), Livestock 

and Poultry (L&P) Program, has implemented a program to predict beef carcass quality and yield 

factors made by approved instrument systems. The purpose of this document is to provide the 

performance standard by which the instruments will be evaluated for the ability to accurately and 

precisely measure USDA grade factors used to predict final yield and quality grades. For a 

technology provider that wants their instrument approved, two like instruments must be tested, and 

each instrument must meet the following requirements to gain approval from the L&P Program. 

The approval process consists of three phases:

Phase I Submission of approval request and determination of designation as either a “New” 

instrument, which includes both previously unapproved instruments or currently 

approved instruments that have undergone significant upgrades. A currently 

approved instrument that has undergone non-significant upgrades will be considered 

an “Upgrade”, see Section 10. 

Phase II Demonstration of the accuracy and repeatability of marbling score, ribeye area, fat 

thickness and final yield grade predictions on stationary beef carcasses. 

Phase III Demonstration of the accuracy and precision of marbling score, ribeye area, fat 

thickness and final yield grade predictions at line speeds consistent with intended 

use.

Note: The intent of this document is for an instrument to pass both quality grade and yield grade 

requirements before being officially utilized by the industry. L&P reserves the right to approve an 

instrument for either quality grade or yield grade only, if a manufacturer can show that meeting the 

requirements for both grade measures either:

1. Creates a hardship that would significantly limit an applicant’s ability to reach the 

marketplace.

2. The manufacturer can show that the technology can provide output that can accurately and 

consistently predict grade factors for quality grade or yield grade, but not both. 

2. Scope

The intent of utilizing beef carcass instrument grading augmentation is to improve the accuracy and 

uniformity of grade application nationwide. 
Agricultural Livestock and Poultry Program QAD 511 Procedure 

Marketing 1400 Independence Avenue SW, Stop 0258 Effective March 11, 2024

Service Washington, DC 20250 Page 2 of 11

3. References

QAD 500 Procedure: Beef, Bullock, and Bull Grading Methods and Procedures

QAD 512 Procedure: Instrument Installation Process, Instrument Grading Systems for Beef 

Carcasses

QAD 513 Procedure: Data Requirements, Instrument Grading Systems for Beef Carcasses

QAD 515 Procedure: Beef Carcass Instrument Grading Procedures

United States Standards for Grades of Carcass Beef

Standard Practice for User Requirements for Livestock, Meat, and Poultry Evaluation Devices or 

Systems. American Society of Testing Materials (ASTM) International Standard F 2341-05. 

Standard Specification for Developing and Validating Prediction Equation(s) or Model(s) Used in 

Connection with Livestock, Meat, and Poultry Evaluation Device(s) or System(s) to Determine 

Value. ASTM International Standard F 2340-05 

4. Definitions

A “New” instrument is described as either a new instrument or a currently approved instrument that 

has undergone significant upgrades. 

• Significant upgrades can be described to include, but are not limited to: 

 Any change that can affect at least one of the predicted independent variables, to 

include both hardware and/or software changes. 

5. Approval Process

5.1 Phase I

Submission of approval request and determination of designation as either a “New” instrument or 

an “Upgrade” to an already approved instrument. 

5.1.1 The applying technology provider will send a request for approval to the Standards and 

Specifications Division (SSD). Details in this request should include:

i. Both a basic and an in-depth summary of how their instrument works.

ii. Whether this is a: 

• “New” instrument

 An instrument not previously approved for use.

 A significant “Upgrade” to a currently approved instrument. 
Agricultural Livestock and Poultry Program QAD 511 Procedure

Marketing 1400 Independence Avenue SW, Stop 0258 Effective March 11, 2024

Service Washington, DC 20250 Page 3 of 11

 List all upgrades to the instrument if utilizing a previously approved device 

as the base for the upgrade.

• A Non-significant “Upgrade” to a currently approved instrument.

iii. A signed affidavit acknowledging that L&P requires that no additional changes to an 

instrument will be made without prior knowledge of L&P after the initial sample 

collection has begun.

5.1.2 L&P will make the final determination if the instrument is a new instrument or a nonsignificant upgrade not needing approval or a significant upgrade needing approval. The instrument 

manufacturer may provide data and any other relevant information needed for L&P to make the 

decision to identify the instrument as a new instrument, a significant upgrade to an instrument, or a 

non-significant upgrade.

5.2 Phase II

Demonstration of the repeatability of marbling score, ribeye area, fat thickness and final yield grade 

predictions on stationary beef carcasses.

5.2.1 Marbling - A minimum of 200 stationary carcass sides will be measured three times1 to 

provide a minimum of 600 observations for the evaluation of repeatability. Instruments can be 

evaluated for repeatability at any of the two approval trials or prior to the approval trials (see 5.3 

Phase III). Repeatability will be estimated from carcasses that represent the U.S. fed beef 

population2 and are distributed over the full range of degrees of marbling, see Table 1.

Table 1

Marbling Scores Minimum required for 

Phase II

Minimum required 

for Phase III

Traces or less 5 50

Slight 0-49 20 200

Slight 50-99 30 200

Small 60 600

Modest 40 400

Moderate 25 300

Slightly Abundant 15 200

1 The Triple Placement method will be used for collecting images (data) from stationary carcasses for marbling, REA, 

and fat thickness. L&P may determine, for certain conditions, that a Triple Trigger method may be used as an 

alternative.

2 Estimates of the distribution of the U.S. fed beef population was provided by the AMSA Beef Grading Committee.
Agricultural Livestock and Poultry Program QAD 511 Procedure

Marketing 1400 Independence Avenue SW, Stop 0258 Effective March 11, 2024

Service Washington, DC 20250 Page 4 of 11

Moderately Abundant or higher 5 50

Totals per location 200 2000

5.3 Phase III

Demonstration of the accuracy and precision of marbling score, ribeye area, fat thickness and final 

yield grade predictions at typical line speeds.

5.3.1 Marbling - A “New” instrument must be tested in at least two separate trial locations to 

ensure that any carcass variation due to environmental conditions (e.g., bloom time/chain speed, 

carcass temperature, sawing, and ribbing) is considered. The locations will be regionally diverse 

and approved by L&P. The technology provider will be responsible for contacting the location and 

obtaining approval to collect samples.

At each trial location, a minimum of 1,000 carcasses will be collected for testing purposes 

and 1,000 carcasses for algorithm development purposes. These carcasses will adequately 

represent the U.S. fed beef population and the full range of marbling scores. Of the 2,000 

carcasses coming from each of the trial locations, no more than two-thirds (2/3) of the 

samples in any one of the marbling categories can come from a single location. The official 

marbling calls for the carcasses used in algorithm development will be provided to the 

applicant.

Sample collection will occur in each of the two locations for a minimum of 1 full production shift 

per location. If the total number of carcasses per marbling category cannot be collected during 

these two shifts, collection over a third shift may be necessary. At least 90% of samples within 

each marbling score must be fulfilled before the sample collection for an approval trial may be 

completed. It will be at the discretion of L&P as to whether additional production shifts and/or 

additional plant/facility will be utilized.

Carcasses must be presented for image capture and analysis at a continuous normal 

operating chain speed. The degree of marbling matrix shown in Table 1 will serve as the 

template in guiding carcass selection. L&P personnel will select the side of the carcass to be 

evaluated by the instrument and the Gold Standard Committee as outlined in 5.3.1.1. No 

more than 65 percent of the images will derive from one side (left or right).

5.3.1.1 The Gold Standard Committee (Committee) is responsible for establishing the Official 

Marbling Score. The Official Marbling Score will be determined by a panel of three L&P experts. 

The Committee will include three L&P experts from the following: National Meat Supervisor, 

Assistant National Meat Supervisor, and Meat Supervisors from the Gold Standard Team. Once the 

instrument trial has started, the Committee will remain unchanged until the completion of the 

instrument trial.

5.3.1.2 Procedure for Determining the Official Marbling Score
Agricultural Livestock and Poultry Program QAD 511 Procedure 

Marketing 1400 Independence Avenue SW, Stop 0258 Effective March 11, 2024

Service Washington, DC 20250 Page 5 of 11

Prior to or immediately following the on-line evaluation by the instrument, carcasses will be 

evaluated by the Committee in an adequately illuminated (a minimum of 100-foot candle power) 

area. The Committee will independently evaluate the marbling score in accordance with the 

United States Standards for Grades of Carcass Beef and will be recorded to the nearest 10 degrees 

of marbling. The marbling score will be numerically coded as follows:

Table 2

A final Official Marbling Score will be computed from the independent observations for each 

carcass. Carcass data will be excluded from the test if the standard deviation of the samples for that 

carcass is greater than 50.

Standard deviation of the carcass is calculated by finding the mean of the observations of the team 

members for that carcass. Once that is calculated, find the deviation from the mean of each score. 

Square each of these deviations from the mean. Add up all these squared deviations to get a sum of 

squares. Divide the sum of squares by (n-1). Take the square root of the resulting variance to get 

the resulting standard deviation. 

The Committee will compute the final Official Marbling Score and the standard deviation (SD) for 

each carcass. Carcasses will be ranked by the final Official Marbling Score in ascending order by 

trial location and then every other carcass will be assigned to a calibration data set. The other half 

of the data will become the test data set. Data entry, data checking, and assignment to calibration 

and test data sets will be conducted in duplicate by the Committee and the results reconciled before 

release of information to the technology provider. The Committee will provide all data parameters 

to the technology provider for the carcasses assigned to the calibration data set. The technology 

provider may use the calibration data to develop or refine their marbling score prediction equation. 

Degree of Marbling Marbling Score

Practically Devoid 100 

Traces 200

Slight 300

Small 400

Modest 500

Moderate 600

Slightly Abundant 700

Moderately Abundant 800

Abundant 900
Agricultural Livestock and Poultry Program QAD 511 Procedure

Marketing 1400 Independence Avenue SW, Stop 0258 Effective March 11, 2024

Service Washington, DC 20250 Page 6 of 11

5.3.1.3 Submission of Predicted Marbling Score and Prediction Equation

At the completion of the two trials, the technology provider will submit the instrument predicted 

marbling scores to L&P for comparison with the Official Marbling Scores of the test data set. L&P 

experts will then compute the necessary statistics to determine if the instrument system meets the 

requirements outline in Section 5. The technology provider will provide L&P with a copy of the 

prediction equation used as well as the values for each variable in the prediction equation for each 

of the predicted marbling scores, which L&P will use to validate the information collected and 

outputs computed.

5.3.1.4 Quality Control of Images and Data Collected

For the test to provide the best results possible, L&P will review all data and images to determine if 

the data and images from the selected carcasses were accurately obtained. Technology providers 

must submit to L&P the original image and the processed display screen image or thumbnail for 

each carcass in the test so that images can be reviewed to determine if proper image capture 

occurred.

6. Performance Requirements for Marbling

Approval will be granted in accordance with the following performance requirements. Approval of 

the instrument will be determined by L&P and will be based on the intended use of the instrument

system. The scope will be set forth in the final approval.

6.1 Phase II: Stationary Measurement of Repeatability

• 95% of the carcasses will have all three of the predicted marbling score observations within 

20 marbling score units of the mean predicted marbling score for that carcass.

6.2 Phase III: On-line Measurement of Accuracy and Precision

• Average residual = 0 ± 10 marbling score units where the residual is the difference between 

the predicted instrument marbling score and mean expert panel marbling score;

• The standard deviation of the residuals (rSD) from the mean expert panel marbling score ≤ 

35 marbling score units; and,

• Slope of 0 ± 0.075, using the residual from the MEPMS as the dependent variable (y-axis) 

and the average of the instrument marbling score and mean expert panel marbling score as 

the independent variable (x-axis).

7. Performance Requirements for Ribeye Area

Approval will be granted in accordance with the requirements listed below. Accuracy of ribeye 

measurement will be evaluated by comparing (correlation and regression) the ribeye area 

observation to the actual ribeye area.
Agricultural Livestock and Poultry Program QAD 511 Procedure

Marketing 1400 Independence Avenue SW, Stop 0258 Effective March 11, 2024

Service Washington, DC 20250 Page 7 of 11

To validate the accuracy of an instrument, a mean expert ribeye area measurement must be 

established for each carcass that is measured. Three experts experienced in measuring ribeye will 

individually grid the ribeye of each carcass with the official USDA ribeye grid. The three 

individual measurements will be averaged to establish the mean expert ribeye area measurement.

7.1 Phase II: Stationary Measurement of Repeatability

• Stationary Accuracy

 95% of predicted ribeye area measurements within 1.5 square inches of the mean expert 

ribeye area measurement.

• Stationary Repeatability

 95% of predicted ribeye area observations within 0.5 square inches of the mean of the 

three ribeye area observations for that carcass.

7.2 Phase III: On-line Measurement of Accuracy and Precision

A minimum of 200 carcasses (representing a range of at least 3.0 square inches from the smallest 

ribeye area to the largest ribeye area sampled) must be presented for image analysis under normal 

beef carcass grading conditions. Freshly ribbed carcasses must be presented to the instrument on a 

line where chain speeds are more than 300 head per hour. Following collection of images, 

carcasses should be placed on a stationary rail for determination of mean expert ribeye area as 

described in the Phase II.

• R2 = 0.85 or greater.

• 95% of ribeye area observations within 2.0 square inches of the mean expert ribeye area.

• Residual standard deviation (RSD) will not exceed 1.0 square inches.

8. Performance Requirements for Fat Thickness

Approval will be granted in accordance with the requirements listed below. Accuracy of fat 

measurement will be evaluated by comparing (correlation and regression) the fat thickness 

observation to the mean expert fat thickness of that carcass side. The mean expert fat thickness will 

be established by calculating the mean of the values derived from three experts, independently, 

measuring the fat thickness of the 12th – 13th rib interface of the carcass. The fat thickness is a 

single measurement of fat over the ribeye three-fourths of the length of a ribeye from the chine bone 

end.

A minimum of 200 carcasses must be presented for image capture and analysis under normal beef 

carcass grading conditions. Freshly ribbed carcasses must be moving at a minimum chain speed of 
Agricultural Livestock and Poultry Program QAD 511 Procedure

Marketing 1400 Independence Avenue SW, Stop 0258 Effective March 11, 2024

Service Washington, DC 20250 Page 8 of 11

300 head per hour. Following collection of images, carcasses should be placed on a stationary rail 

for determination of actual fat thickness as described above (Phase II).

8.1 Phase II: Stationary Measurement of Repeatability

• Stationary Accuracy

 95% of the predicted fat thickness observations within 0.1 inches of the mean expert fat 

thickness.

8.2 Phase III: On-line Measurement of Accuracy and Precision

• 95% of predicted fat thickness observations within 0.1 inches of the mean expert fat 

thickness.

• The average of the residuals (between instrument fat thickness and mean expert fat 

thickness) will equal 0 ± 0.05 inches.

9. Performance Requirements for Yield Grade

To validate the accuracy of an instrument, an actual expert calculated yield grade must be 

established for each carcass that is measured. The yield grade factors will be evaluated by three 

experts from the Gold Standard Team. The panel of three officials will evaluate:

• Adjusted Preliminary Yield Grade based on fat cover over the entire carcass.

• Percent Kidney, Pelvic, and Heart Fat estimated as a percent of the hot carcass weight 

(pounds) or, if removed, the actual percent.

• Ribeye Area, (square inches), determined by grid measurement. The panel will evaluate the 

same side evaluated by the instrument being tested.

• The actual expert calculated yield grade will be a consensus yield grade calculated to the 

nearest 0.1 yield grade unit using the USDA “short-cut method” or the official Yield Grade 

equation:

Yield Grade = 2.50 + (2.0 x adjusted fat thickness, inches)

+ (0.20 x percent kidney, pelvic, and heart fat)

+ (0.0038 x hot carcass weight, pounds)

- (0.32 x area of ribeye, square inches)

Approval will be granted in accordance with the requirements listed below. Accuracy of yield 

grade will be evaluated by comparing (correlation and regression) the yield grade observation to the 
Agricultural Livestock and Poultry Program QAD 511 Procedure

Marketing 1400 Independence Avenue SW, Stop 0258 Effective March 11, 2024

Service Washington, DC 20250 Page 9 of 11

mean expert calculated yield grade. A mean expert calculated yield grade is determined by 

calculating the mean of the three independent actual calculated yield grades.

9.1 Phase II: Stationary Measurement of Accuracy

• 95% of predicted yield grade observations within 0.5 yield grade units of the mean expert 

calculated yield grade.

9.2 Phase III: On-line Measurement of Accuracy and Precision

• 95% of predicted yield grade observations within 0.5 yield grade units of the mean expert

calculated yield grade.

• The residual standard deviation (RSD) will not exceed 0.25 yield grade units.

Note: Preliminary yield grade observations and adjusted fat thickness predictions will not be 

evaluated separately from the final yield grade.

10. Performance Requirements for a Non-Significant Upgrade of an already approved 

instrument.

Approval will be granted in accordance with the requirements listed below. A non-significant 

upgrade is an upgrade that does not potentially influence or practically affect at least one 

independent variable being measured.

10.1 The manufacturer will have the option to choose between expert panel to prototype testing 

or approved device to prototype testing. The manufacturer will notify L&P of its choice before 

initiation of testing.

10.2 The prototype instruments will be required to fulfill all requirements listed for Phase III 

measurements. If the manufacturer choses the approved device to prototype testing method, the 

measurement values given for the duplicate assessment will be used in place of the mean expert 

scores.

11. Supersedes

This document supersedes:

• USDA AMS Prime I: Instrument Grading Systems for Beef Carcasses, June 2006

• USDA AMS Addendum A: Instrument Grading Systems for Beef Carcasses, December 

2012

• USDA AMS: Procedures for Approval and Use of Instrument Systems for Beef Carcass 

Ribeye Measurement, February 2003
Agricultural Livestock and Poultry Program QAD 511 Procedure

Marketing 1400 Independence Avenue SW, Stop 0258 Effective March 11, 2024

Service Washington, DC 20250 Page 10 of 11

• USDA AMS: Procedures for Approval and Use of Vision-Based Instrument Systems for 

Beef Carcass Yield Grade Measurement, March 2005

• USDA AMS Addendum A: Procedures for Use of Approved Vision-Based Beef Carcass 

Yield Grade Instrument Systems for Fat Thickness Measurements, March 2007

12. Additional Requirements

12.1 Verification of Approved Equations

The L&P Program must be notified prior to implementing any modifications of the device, system, 

or process that would affect one or more independent variables of a L&P Program approved 

equation (see Section 5.2 ASTM International Standard F 2340-05; and Section 5.3.3. ASTM 

International Standard F 2341-05). The instrument manufacturer must allow L&P personnel access 

to the unencrypted software coding for an instrument immediately upon request by L&P. 

Immediate view of a specific instrument’s coding will be required (no generalized coding will be 

accepted.)

The manufacturer will keep an electronic log within the instrument system that tracks any updates 

or modifications to the software coding. This electronic log will be provided to L&P personnel

immediately upon request.

L&P reserves the right to confirm if changes have been made to the software by utilizing existing 

monitoring and quality control tools.

12.2 Conditional Approval

Approval of an instrument will be granted on a conditional basis of performance in the field setting 

for a minimum of one year from date of approval. During this conditional period, L&P may require 

additional testing of the instruments if field data suggests that the instruments are not performing to 

the levels achieved in the approval process.

12.3 Use of a Reference Device

The device manufacturer will provide L&P with an instrument to be used as a reference device. 

L&P will possess this device and will utilize it as a reference device when testing instruments in the 

production setting.

13. Subject to Change

The provisions of this procedure are subject to revision by the L&P Program at any time.
Agricultural Livestock and Poultry Program QAD 511 Procedure 

Marketing 1400 Independence Avenue SW, Stop 0258 Effective March 11, 2024

Service Washington, DC 20250 Page 11 of 11

14. Approval Request

Requests for approval will be submitted to: 

Chad Nelson Willy Horne

chad.nelson2@usda.gov willy.horne@usda.gov 

(402) 281-8704 (202) 841-2974 

__________________________________

Mark Perigen, Director

Quality Assessment Division

Livestock and Poultry Program

In accordance with Federal civil rights law and U.S. Department of Agriculture (USDA) civil rights regulations and policies, the USDA, its Agencies, offices, and 

employees, and institutions participating in or administering USDA programs are prohibited from discriminating based on race, color, national origin, religion, sex, 

gender identity (including gender expression), sexual orientation, disability, age, marital status, family/parental status, income derived from a public assistance 

program, political beliefs, or reprisal or retaliation for prior civil rights activity, in any program or activity conducted or funded by USDA (not all bases apply to all 

programs). Remedies and complaint filing deadlines vary by program or incident.

Persons with disabilities who require alternative means of communication for program information (e.g., Braille, large print, audiotape, American Sign Language, 

etc.) should contact the responsible Agency or USDA's TARGET Center at (202) 720-2600 (voice and TTY) or contact USDA through the Federal Relay Service at 

(800) 877-8339. Additionally, program information may be made available in languages other than English.

To file a program discrimination complaint, complete the USDA Program Discrimination Complaint Form, AD-3027, found online at How to File a Program 

Discrimination Complaint and at any USDA office or write a letter addressed to USDA and provide in the letter all of the information requested in the form. To 

request a copy of the complaint form, call (866) 632-9992. Submit your completed form or letter to USDA by: (1) mail: U.S. Department of Agriculture, Office of the 

Assistant Secretary for Civil Rights, 1400 Independence Avenue, SW, Washington, D.C. 20250-9410; (2) fax: (202) 690-7442; or (3) email: 

program.intake@usda.gov. 

USDA is an equal opportunity provider, employer, and lender.