# 

GLP10

- 01

- 02; Interim Report 30

- Analysis of PFBS, PFHS, and PFOS in Ground w ater

Samples Collected at Off

- Site Wells on Bert Jeffries Property in Decatur, AL in April 2012

Study Title

Analysis of Perfluorooctane Sulfonate (PFOS), Perfluorohexane Sulfon ate (PFHS) and

Perfluorobutane Sulfonate (PFBS) in Groundwater, Soil and Sediment for the 3M Decatur Phase 3

Site

- Related Monitoring Program

Data Requirement

EPA TSCA Good Laboratory Practice Standards 40 CFR Part 792

Study Director

Jaisimha Kesari P.E., D EE

Weston Solutions, Inc.

1400 Weston Way

West Chester, PA 19380

Phone: 610

- 701

- 3761

Author

Susan Wolf

3M Environmental Laboratory

Interim Report Completion Date

Date of signing

Performing Laboratory

3M Environmental Health and Safety Operations

Enviro nmental Laboratory

3M Center, Bldg 260

- 05 -N- 17

St. Paul , MN 55144

Project Identification

GLP10

- 01

- 02

- 30

Total Number of Pages

112

Testing Cert #2052.01 The testing reported herein meet the requirements of ANSI/ISO/IEC 17025:2005 “General Requirements for

the Competence of Testing and Calibration Laboratories”, in accordance with the A2LA Testing Certificate #

205 2.01. Testing that complies with this International Standard also meets principles of ISO 9001:2000.

# This page has been reserved for specific country requirements.

### GLP10-01-02 ; Interi m Repor t 3 0

Analysi s o f PFBS

, PFHS

, an d PFO S I n Groundwate r Sample s

Ber t Jeffrie s Propert y - Apri l 201 2



# GL P COMPLIANC E STATEMEN T

Repor t Title : Interi m Repor

t 3 0

- Analysi s o

f PFBS

, PFHS

, an d PFO S i n Groundwate r Sample s

Collecte d a

t Off-Sit e Well s o n Ber

t Jeffrie s Propert

y i n Decatur

, A L i n Apri l 2012

.

Study : Analysi s o f Perfluorooctan e Sulfonat e (PFOS)

, Perfluorohexan e Sulfonat e (PFHS ) an d

Perfluorobutan e Sulfonat e (PFBS ) i n Groundwater

, Soi l an d Sedimen t fo r th e 3 M Decatu r Phas e 3

Site-Relate d Monitorin g Program

.

Thi s analytica l phas e wa s conducte d i n complianc e wit h Toxi c Substance s Contro l Ac t (TSCA ) Goo d

Laborator y Practic e (GLP ) Standards

, 4 0 CF R 792

, wit h th e exception s liste d below :

• Thes e ar e environmenta l sample s wher e ther e i s n o specifi c tes t substance

, n o specifi c tes t

syste m an d n o dosin g o

f a tes t system

.

• Th e referenc e substance s hav e no t bee n characterize d unde r th e GLP s an d th e stabilit y unde r

storag e condition s a

t th e tes t sit e hav

e no t bee n detemnine d unde r GLPs

.

Gar y Hoh^nsteiri n Sponso r Representativ e Dat e

Jaisimh a Kesari

, P.E.

, DEE

, Stud y Directo r ' Dat e

Pag e 3 o f 11 2

### GLP10-01-02 ; Interi m Repor t 3 0

Analysi s o f PFBS

, PFHS

, an d PFO S i n Groundwate r Sample s

Ber t Jeffrie s Propert y - Apri l 201 2



# QUALIT Y ASSURANC E STATEMEN T

Repor t Title : Interi m Repor

t 3 0

- Analysi s o

f PFBS

, PFHS

, an d PFO S i n Groundwate r Sample s

Collecte d a

t Off-Sit e Well s o n Ber

t Jeffrie s Propert

y i n Decatur

, A L i n Apri l 2012

.

Study : Analysi s o

f Perfluorooctan e Sulfonat e (PFOS)

, Perfluorohexan e Sulfonat e (PFHS ) an d

Perfluorobutan e Sulfonat e (PFBS ) i n Groundwater

, Soi l an d Sedimen t fo r th e 3 M Decatu r Phas e 3

Site-Relate d Monitorin g Program

.

Thi s analytica l phas e wa s audite d b y th e 3 M Environmenta l Laborator y Qualit y Assuranc e Uni t (QAU)

,

a s indicate d i n th e followin g table

. Th e finding s wer e reporte d t o th e principa l investigato r (P.I.)

, laborator y managemen t an d stud y director

.

Inspectio n Date s Phas e Dat e R e porte d t o

Inspectio n Date s Phas e Testin g Facilit y IVIanagemen t Stud y Directo r

4/27/12

, 4/30/1 2 Dat a an d Repor

t 5/1/1 2 5/1/1 2



# 5 I I

Date/ ' QA U

Pag e 4 o f 11 2

# T ABLE OF 

# C ONTENTS

GLP Compliance Statement

...................................................................................................................... 3

Quality Assurance Statement

.................................................................................................................... 4

Table of Contents

................................................................................................ ....................................... 5

List of Tables

................................................................................................ .............................................. 6

1 Study Information

................................................................................................ ................................ 8

2 Summary

................................................................................................ ............................................. 9

3 Introduction

................................................................................................ ........................................ 10

4 Test & Control Substances

.............................................................................................................. 11

5 Reference Substances

..................................................................................................................... 11

6 Test System

................................................................................................ ...................................... 13

7 Method Summary

............................................................................................................................. 13

7.1 Methods

............................................................................................................................ 13

7.2 Sample Collection

............................................................................................................. 13

7.3 Sam ple Preparation

.......................................................................................................... 13

7.4 Analysis

............................................................................................................................. 13

8 Analytical Results

.............................................................................................................................. 15

8.1 Calibration

......................................................................................................................... 15

8.2 System Suitability

............................................................................................................. 15

8.3 Limit of Quantitation (LOQ)

.............................................................................................. 15

8 .4 Continuing Calibration

...................................................................................................... 15

8.5 Blanks

................................................................................................ ................................ 15

8.6 Lab Control Spikes (LCSs)

.............................................................................................. 16

8.7 Analytical Method Uncertainty

......................................................................................... 18

8.9 Field Matrix Spikes (FMS)

................................................................................................ 18

# 9 Data Summary and Discussion

....................................................................................................... 18

10 Conclusion

................................................................................................ ......................................... 28

11 Data/Sample Retention

.................................................................................................................... 28

12 Attachments

................................................................................................ ...................................... 28

13 Signatures

................................................................................................ ......................................... 29



# L IST OF 

# T ABLES

Table 1. S ummarized PFBS, PFHS, and PFOS Results (Bert Jeffries Property, April 2012).

.. 10

Table 2. Sample Description Key Code.

....................................................................................... 13

Table 3. Instrument Parameters.

................................................................................................... 14

Table 4. Liquid Chromatography Conditions.

............................................................................... 14

Table 5. Mass Transitions.

............................................................................................................. 14

Table 6. Limit of Quantitation (LOQ).

............................................................................................ 15

Table 7. Laboratory Control Spike Recovery.

............................................................................... 17

Table 8. Analytical Uncertainty.

..................................................................................................... 18

Table 9. Field Matrix Spike Levels.

............................................................................................... 18

Table 10. JPAL GW MW1R 120405

.............................................................................................. 20

Table 11. JPAL GW MW3R 120405

.............................................................................................. 20

Table 12. JPAL GW MW4R 120405

.............................................................................................. 21

Table 13. JPAL GW MW4L 120405

............................................................................................... 21

Table 14. JPAL GW MW5R 120405

.............................................................................................. 22

Table 15. JPAL GW MW5L 120405

............................................................................................... 22

Table 16. JPAL GW MW6R 120405

.............................................................................................. 23

Table 17. JPAL GW MW6L 120405

............................................................................................... 23

Table 18. JPAL GW MW7R 120405

.............................................................................................. 24

Table 19. JPAL GW MW8R 120 405

.............................................................................................. 24

Table 20. JPAL GW MW9R 120405

.............................................................................................. 25

# Table 21. JPAL GW MW9L 120405

............................................................................................... 25

Table 22. JPAL GW MW10R 120405

............................................................................................ 26

Table 23. JPAL GW MW11L 120405

............................................................................................ 26

Table 24. Rinseate Blank

................................................................................................................ 27

Table 25. Trip Blank

........................................................................................................................ 27

# 1 **Study Information**

Sponsor

3M Company

Sponsor Representative

Gary Hohenstein

3M EHS Operations

3M Building 224

- 5W

- 03

Saint Paul, MN 55144

- 1000

Phone: (651) 737

- 3570

Study Director

Jaisimha Kesari, P.E., DEE

Weston Solutions, Inc.

West Chester, PA 19380

Phone: (610) 701

- 3761

Fax: (610) 701

- 7401

j.kesari@westonsolutions.com

Study Location

Testing Facility

3M EHS Operations

3M Environmental Laboratory

Building 260

- 5N

- 17

St. Paul, MN 55144

Study Personnel

William K. Reagen, Ph.D., 3M Laboratory Manager

Cleston Lange, Ph.D., Principal Analytical Investigator, ( clange@mmm.com ); phone (651)

- 733

- 9860

Susan Wolf, 3M Analyst

Chelsi e Grochow, Analyst

Kevin Eich, Analyst

Kelly Ukes, Analyst

Study Dates

Study Initiation: March 8, 2010

Interim 30 Experimental Termination: April 26, 2012

Interim Report Completion: Date of Interim Report Signing

Location of Archives

All original raw data and the analytical report have been archived at the 3M Environmental

Laboratory according to 40 CFR Part 792. The test substance and analytical reference standard

reserve samples are archived at the 3M Environmental Laboratory according to 40 CFR Part 792

.

# 2 **Summary**

The 3M Environmental Laboratory prepared sample containers for fourteen locations on the Bert

Jeffries property. A total of forty

- five sample bottles were received at the 3M Environmental Laboratory

for pe rfluorobutane sulfonate (PFBS)

, pe rfluo robutane sulfonate (PFBS)

, perfluorooctane sulfonate

(PFOS ) analysis from Weston personnel on April 9, 2012. All samples were logged into the laboratory

information management system (LIMS) under project GLP10

- 01

- 02

- 30.

For each sampling location, a field sample, field duplicate sample and field matrix spike (FMS) sample

were collected. Samples also included a trip blank containing Milli

- QTM water and an appropriate trip

blank spike. A single equipment rinseate blank was also collected

. The equipment rinseate blank did

not have FMS samples prepared for determination of PFOA recovery. All sample bottles were fortified

with internal standards and surrogate recovery standard 13 

# C 4 

# - PFOS prior to sample collection. All of

the samples were prepared and an alyzed for PFBS, PFHS, PFOS , and the surrogate recovery

standard 13 

# C 4 

# - PFOS following 3 M Environmental Laboratory Method ETS -8- 044.1.

The average measured PFBS, PFHS, and PFOS concentrations are summarized in Table 1. The trip

blank sample was below the lower limit of quantitation (LLOQ) for all analytes, indicating adequate

control of sample contamination during shipping and sample collections. The analytical method

uncertainties associated with the reported results are: PFBS + 13 %, PFHS + 14 % and PFOS + 18 %.

# Table 1

. S ummarized PFBS, PFHS, and PFOS Results (Bert Jeffries Property

, April 2012 ).

Sampl ing Location PFBS

Avg. Conc

.

(ng/mL)

± %RPD PFHS

Avg. Conc

.

(ng/mL)

± %RPD PFOS

Avg. Conc

.

(ng/mL)

± %RPD

JPAL GW MW1R 1 20405 0.113 ± 2.7% 0.861 ± 1.0% 1.61 ± 1.9%

JPAL GW MW3R 120405 1.79 ± 0.56% 8.73 ± 7.2% 9.70 ± 3.5%

JPAL GW MW4R 120405 0.189 ± 0.53% 4.88 ± 3.1% 5.81 ± 2.2%

JPAL GW MW4L 120405 0.0772 ± 3.2% 0.205 ± 4.4% 0.348 ± 0.57%

JPAL GW MW5R 120405 0.0402 ± 0.7 5% 0.128 ± 8.6% 0.128 ± 4.7%

JPAL GW MW5L 120405 0.263 ± 2.7% 0.400 ± 0.25% 2.07 ± 6.8%

JPAL GW MW6R 120405 0.334 ± 1.5% 1.06 ± 5.7% 7.91 ± 0.38%

JPAL GW MW6L 120405 0.0551 ± 0.91% 0.0952 ± 0.63% 0.372 ± 3.5%

JPAL GW MW7R 120405 0.536 ± 0.75% 1.87 ± 1. 1% 22.5 ± 2.2%

JPAL GW MW8R 120405 0.0508 ± 6.3% 0.128 ± 5.5% 0.410 ± 6.6%

JPAL GW MW9R 120405 <0.0250 0.0650 ± 1.1% 0.0620 ± 10%

JPAL GW MW9L 120405 <0.0250 <0.0250 0.0425 ± 0.71%

JPAL GW MW10R 120405 0.0579 ± 8.1% 0.264 ± 4.2% 0.238 ± 5.0%

JPAL GW M W11L 120405 <0.0250 0.0886 ± 7.6% 0.496 ± 6.5%

Trip Blanks (Milli

- QTM Water) <0.0250 <0.0250 <0.0232

GLP10

- 01

- 02

- 30 Equipment rinseate blank:

JPAL GW MW3R <0.0 250 <0.0250 <0.232

The analytical method uncertainties associated with the reported results are: PFBS ± 13%, PFHS ± 14 %, and PFOS ± 18%.



# 3 **Introduction**

This analytical study was conducted as part of the Phase 3 En vironmental M onitoring and A ssessment

P rogram for the 3M facility located in Decatur, Alabama. The objective of the overall program is to gain

information regarding concentrations of perfluoroo ctane sulfonate (PFOS), perfluorohexane sulfonate

(PFHS) and perfluorobutane sulfonate (PFBS)

, in various environmental media such as groundwater

, soils and sediments that are associated with and near the Decatur facility. This analytical study was

conduc ted to analyze ground water samples collected from fourteen locations from off

- site wells near

Jeffries Landfill in Decatur, AL for PFBS, PFHS, and PFOS in an effort to characterize regional

groundwater conditions.

The 3M Environmental Laboratory prepared sample containers (250 mL high

- density polyethylene

bottles) which were shipped to Decatur, AL Weston personnel prior to field sampling. Sample

containers for each sampling location included a field sample, field sample duplicate, and one field

matrix spike sample. Each empty container was marked with a “fill to here” line to produce a final

sample volume of 200 mL. Containers designated for field matrix samples were fortified with an

appropriate matrix spike solution containing PFBS (linear isomer ), PFHS (linear isomer ), and PFOS

(linear and branched isomers ) prior to being sent to the field for sample collection. All sample bottles

included the addition of 18 

# O 2 

# - PFBS, 13 

# C 3 

# - PFHS, and 13 

# C 8 

# - PFOS (internal standard) at a nominal

# concentration of 1 ng/mL

. All sample bottles also included the addition of 13 

# C 4 

# - PFOS (surrogate

recovery standard) at a nominal concentration of 0.1 ng/mL. See section 8.8 of the report for field

matrix spike levels.

Samples were prepared and analyzed according to the procedure defined in 3M Environmental

Laboratory method ETS -8- 044. 1 “ Method of Analysis for the Determination of Perfluorinated

Compounds In Water by LC/MS/MS; Direct Injection Analysis”. The use of internal standards was

used to aid in the data quality objectives.

Table 1 summarizes the average PFBS, PFHS, and PFOS concentrations for the duplicate surface

water samples collected and the trip blank sample

. Table s 10

- 25 summarize the individual sample

results and the associated fi eld matrix spike recoveries. All results for the quality control samples

prepared and analyzed with the samples are reported and discussed elsewhere in this report



# 4 **Test & Control Substances**

There was not a test substance or control substances in the classic sense of a GLP study. This study

was purely analytical in nature.



# 5 **Reference Substances**

Reference Substance PFBS

(predominantly linear) 18 

### O

2 

### - PFBS



## Chemical Name Potassium

Perfluorobutane sulfonate [ 18 

## O 2 

## ]

- Ammonium

Perfluorobutanesulfonate

Chemical Fo rmula 

# C 4 

# F 9 

# SO 3 - 

# K + 

# C 4 

# F 9 

# S[ 18 

# O 2 

# ]O- NH 4 +



## Identifier NA NA

Use Target Analyte Reference

Standard Internal Standard

Source 3M RTI International

Expiration Date 01/10/2017 03/09/2015

Storage Conditions Frozen Frozen

Chemical Lot Number 41

- 2600

- 8442 -5 11546 -1 07 -2

TCR Number TCR

- 121 TCR

- 1024 , TCR

- 1040

Physical Description White Powder Liquid

Purity 96.7% >99%

### Reference Substance PFHS

(Linear) 13 

### C

3 

### - PFHS



## Chemical Name Sodium Perfluorohexane

sulfonate Sodium

Perfluorohexane sulfonate

Chemical Formula 

# C 6 

# F 13 

# SO 3 - 

# Na + 13 

# C 3 12 

# C 3 

# F 13 

# SO 3 - 

# Na +



## Identifier L- PFHXS MPFC -C- 0112

Use Target Analyte Reference

Standard Internal Standard

Source Wellington Wellington

Expiration Date 03/25/2018 01/24/2015

Storage Conditions Frozen Frozen

Chemical Lot Number LPFHxSAM08 MPF C-C- 0112

TCR Number TCR08

- 0018 TCR12

- 0004

Physical Description Crystalline Liquid

Purity 100% 5 μ g/mL (1)



### Reference

Substance PFOS

(Linear + Branched) PFOS

(Linear + Branched) 13 

### C

4 

### - PFOS 13 

### C

8 

### - PFOS



## Chemical Name Potassium

Perfluorooctane

sulfonate Pota ssium

Perfluorooctane

sulfonate Sodium

Perfluorooctane

sulfonate Sodium

Perfluorooctane

sulfonate

Chemical Formula 

# C 8 

# F 17 

# SO 3 - 

# K + 

# C 8 

# F 17 

# SO 3 - 

# K + 13 

# C 4 12 

# C 4 

# F 17 

# SO 3 - 

# Na + 13 

# C 8 

# F 17 

# SO 3 - 

# Na +



## Identifier Br

- PFOSK CAS # 2795

- 39 -3 MPFOS MPFC -C- 0112

Use Target Analyte

Referen ce Standard FMS Reference

Standard Surrogate Recovery

Standard Internal Standard

Source Wellington Sigma Aldrich Wellington Wellington

Expiration Date 12/01/2014 0 2/ 0 4/2014 09/08/2013 01/24/2015

Storage Conditions Frozen Ambient Frozen Frozen

Chemical Lot Number brPFOSK1111 1424328V MPFOS0910 MPFC -C- 0112

TCR Number TCR11

- 0041 TCR11

- 0028 TCR10

- 0044 TCR12

- 0004

Physical Description Liquid White Powder Liquid Liquid

Purity 99.9% 99.7% >98% 5 μ g/mL (1)



### (1) Custom mixture of seven mass

- labeled ( 13 

### C) perfluoroalkylcarboxylic acids, two mass labeled ( 13 

### C)

p erfluoroalkylsulfonates and one mass

- labeled ( 13 

### C ) perfluoro -1- octanesulfonamide.

# 6 **Test System**

The test systems for this study are groundwater samples collected from wells located in Decatur, AL by

West on Solutions, Inc. personnel. Samples for this study are “real world” samples, not dosed with a

specific lot of test substance.

Table 2 . Sample Description Key Code.

String Number String Descriptor Example

**Example JPAL** 

**- GW** 

**- MW1R** 

**- 0** 

**- 120405** 

1 Sample Location JPAL= Jeffries Property, Alabama

2 Sample Type GW= Ground Water

3 Well Identifier Example: MW1R

4 Well Level R = Residuum shallow water

- bearing zone

L = Bedrock water

-bearing zone

S = Epikarst middle water

- bearing zone

5 Sam pling Date 120405

– April 5, 2012

6 Sample Type 0=primary sample

DB=duplicate sample

FMS = Field Matrix Spike



# 7 **Method Summary**

7.1 Method

Analysis for all analytes was completed following 3M Environmental Laboratory method ETS -8- 044. 1

“ Method of Analysis for the Determination of Perfluorinated Compounds In Water by High Performance

Liquid Chromatography/Mass Spectrometry Direct Injection Analysis”.



# 7.2 Sample Collection

Samples were collected in 250 mL Nalgene TM 

# (high

- density polyethylene) bottles prepared at the 3M

Environmental Laboratory. Sample bottles associated with GLP10

- 01

- 02

- 30 were returned to the

laboratory at ambient conditions on April 9 , 2012. Samples were stored refrigerated at the laboratory

after receipt. A set of laboratory prepared Trip Blank and Trip Blank field matrix spikes were sent with

the sample collection bottles.



# 7.3 Sample Preparation

Samples were prepared by removing an aliquot of the well mixed sample and placing it in an autovial for

analysis.

During the preparation of the laboratory control samples, an aliquot of a separate internal standard

spiking solution was added to the laboratory control samples (nominal concentration of 1 ng/mL). The

sample bottles were spiked with an internal standard mix at a nominal concentration of 1 ng/mL prior to

being sent to the field for sample collection

.



# 7.4 Analysis

All study samples and quality control samples were analyzed for PFBS, PFHS, and PFOS using high

performance liquid chromatography/ tandem mass spectrometry (HPLC/MS/MS). Detailed i nstrument

parameters, the liquid chromatography gradient program, and the specific mass transitions analyzed are

# described in the raw data hard copies placed in the final data packet, and are briefly described below in

Table 3

, Table 4 and Table 5

..

Table 3 . Instrument Parameters

.

Instrument Name ETS Buster



## Analytical Method Followed ETS

- 8

- 044.1



## Analysis Date 4/23/12



## Liquid Chromatograph Agilent 1100



## Guard column Betasil C18 (4.6 mm X 100 mm), 5 μ



## Analytical column Betasil C18 (4.6 mm X 100 mm), 5 μ



## Injection Volume 50 μ L



## Mass Spectrometer Applied Biosystems API 4000



## Ion Source Turbo Spray



## Electrode Turbo ion electrode



## Polarity Negative



## Software Analyst 1.4 .2



# Table 4 . Liquid Chromatography Conditi ons.

**Step**

**Number Total Time** 

**(min) Flow Rate** 

**(** μ

**L/min) Percent A** 

**(2 mM ammonium acetate) Percent B** 

**( m**  **ethanol)**

ETS

- 8

- 044. 1

0 0 .0 750 97.0 3 .0

1 2 .0 750 97 .0 3 .0

2 14.5 750 5. 0 9 5 .0

3 15.5 750 5. 0 95 .0

4 16.5 750 97 .0 3 .0

5 20 .0 750 97 .0 3 .0



# Table 5. Mass Transitions.

**Analyte Mass Transition** 

**Q1/Q3 Reference** 

**Material**

**Structure Internal Standard**  **Mass Transition**

**Q1/Q3**

**PFBS 299/80** 

**Linear [**  **18** 

### O **2** 

### ]-

**PFBS 303/84** 

**299/99**

**PFHS 399/80** 

**Linear [**  **13** 

### C **3** 

### ]-

**PFHS 402/80** 

**399/99**

**PFOS 499/**  **80**

**Linear +** 

**Branched [**  **13** 

### C **8** 

### ]-

**PFOS 507/80**  **499/99**

**499/130**

**[ 13**  

### C

**4** 

### ]

**- PFOS**  **503/80 Linear**  **[ 13**  

### C **8** 

### ]

**- PFOS**  **507/80**

Dwell time was 50 msec for each transition. The individual transitions were summed to produce a “total ion

chromatogram” (TIC), which was used for quantitation

.

# 8 **Analytical Results** 

8.1 Calibration

Samples were analyzed using a stable isotope internal standard calibration curve. Calibration standards

were prepared by spiking known amounts of the stock solution containing the target analytes into a

la boratory

- prepared synthetic groundwater containing calcium and magnesium. A separate internal

standard spiking solution was prepared and an aliquot was added at the same level to all calibration

standards and laboratory control samples at a nominal concentration of 1 ng/mL. A calibration curve

ranging from approximately 0.025 ng/mL to 25 ng/mL (0.025 ng/mL to 10 ng/mL for 13 

# C 4 

# - PFOS

surrogate) was analyzed

. The reference standard used for the calibration standards for PFOS

contained both linear and branched isomers. A quadratic, 1/x weighted, calibration curve of the

standard peak area/peak area ratios was used to fit the data for each analyte. The data were not forced

through zero during the fitting process. Calculating the standard concentrations using the peak

area/peak area ratios and the resultant calibration curve confirmed accuracy of each curve point.

Each curve point was quantitated using the overall calibration curve and reviewed for accuracy. Method

calibration accuracy requirements of 100±25% (100±30% for the lowest curve point) were met for all

analytes. The correlation coefficient (r) was greater than 0.995 for PFBS, PFHS, PFOS, and 13 

# C 4 

# - PFOS.



# 8.2 System Suitability

A calibration standard was analyzed four times at the beginning of each analytical sequence to

demonstrate overall system suitability. The acceptance criteria of less than or equal to 5% relative

standard deviation (RSD) for peak area and retention time criteria of less than or equal to 2% RSD w as

met for PFBS, PFHS, PFOS, and 13 

# C 4 

# - PFOS.



# 8.3 Limit of Quantitation (LOQ)

The LOQ for this analysis is the lowest non

- zero calibration standard in the curve that meets linearity

and accuracy requirements and for which the area counts or area ratio are at least twice those of the

appropriate blanks. The LOQ for all analytes can be found in Table 6.

Table 6 . Limit of Quantitation (LOQ).

**Analysis Date PFBS** 

**LOQ, ng/mL PFHS** 

**LOQ, ng/mL PFOS** 

**LOQ, ng/mL**

4/23/12 0.0250 0.0250 0.02 32



# 8.4 Continuing Calibration

During the course of each analytical sequence, continuing calibration verification samples (CCVs) were

analyzed to confirm that the instrument response and the initial calibration curve were still in control. All

CCVs met method criteria of 100% ± 25% for PFBS, PFHS, P FOS, and 13 

# C 4 

# - PFOS

.



# 8.5 Blanks

Three types of blanks were prepared and analyzed with the samples: procedural blanks, trip blanks, and

equipment rinseate blanks. Procedural blank results were reviewed and used to evaluate method

performance and to determine the LOQ. Trip blanks reflect the shipping and sample collection

conditons the sample bottles and samples experience. Equipment rinseate blanks are aqueous

samples that reflect the efficiency of equipment cleaning in the field between different sample col lections

to ensure no cross contamination of samples from the equipment.

# 8.6 Lab Control Spikes (LCSs)

Low, mid, and high lab control spikes were prepared for the target analytes and analyzed in triplicate,

while only low and high lab control spikes were prepared for the 13 

# C 4 

# - PFOS surrogate. LCSs were

prepared by spiking known amounts of the analyte into synthetic groundwater to produce the desired

concentration. The spiked water samples were then prepared and analyzed in the same manner as the

samples. The method acceptance criteria, average of LCS at each level should be within 100% ± 20%

with an RSD ≤20%

, were met for all analytes.

The following calculations were used to generate data in Table 7 for laboratory control spikes.

100% *

ion Concentrat Spike ion Concentrat Calculated Recovery Percent LC S

=

100% *

recovery LCS average replicates LCS dev iation standard RSD LCS%

=

# Table 7 . Laboratory Control Spike Recovery

.

ETS -8- 044.1

Analyzed 4/23 /12 

## PFBS **PFHS**

**Lab ID Spiked** 

**Concentration**

**(ng/mL) Calculated** 

**Concentration**

**(ng/mL) %Recovery**  **Spiked**

**Concentration**

**(ng/mL) Calcula**  **ted**

**Concentration**

**(ng/mL) %Recovery** 

LCS

- 120416

- 1 0.198 0.203 102 0.198 0.201 102

LCS

- 120416

- 2 0.198 0.216 109 0.198 0.210 106

LCS

- 120416

- 3 0.198 0.222 112 0.198 0.215 108

Average ± %RSD 108 % ± 4.8 % 105 % ± 2. 9 %

LCS

- 120416

- 4 1.98 2.13 108 1.98 2.10 106

LCS

- 120416

- 5 1.98 2.16 109 1.98 2.30 116

LCS

- 120416

- 6 1.98 2.10 106 1.98 2.21 112

Average ± %RSD 1 08 % ± 1.4 % 111 % ± 4. 5 %

LCS

- 120416

- 7 9.94 9.83 98.9 9.92 10.0 101

LCS

- 120416

- 8 9. 94 9.58 96.4 9.92 9.16 92.3

LCS

- 120416

- 9 9.94 9.41 94.7 9.92 9.39 94.7

Average ± %RSD 96.7 % ± 2.2 % 96.0 % ± 4.7 %

ETS -8- 044.1

Analyzed 4/23/12 

## PFOS (Linear + B **ranched) 13**  

## C **4** 

## - **PFOS surrogate**

**Lab ID Spiked** 

**Concentration**

**(ng/mL) Calculated** 

**Concentration**

**(ng/mL) %Recovery**  **Spiked**

**Concentration**

**(ng/mL) Calculated** 

**Concentration**

**(ng/mL) %Recovery** 

LCS

- 120416

- 1 0.184 0.179 97.3 0.189 0.202 107

LCS

- 120416

- 2 0.184 0.198 107 0.189 0.199 105

LCS

- 120416

- 3 0.184 0.199 108 0.189 0.193 102

Average ± %RSD 104 % ± 5

. 7 % 105 % ± 2.4 %

LCS

- 120416

- 4 1.84 1.84 99.8 1.89 1.91 101

LCS

- 120416

- 5 1.84 1.91 104 1.89 1.99 105

LCS

- 120416

- 6 1.84 1.95 106 1.89 2.01 106

Average ± %RSD 103 % ± 3.1 % 104 % ± 2.5 %

LCS

- 120416

- 7 9.22 9.16 99.3 NA NA NA

LCS

- 120416

- 8 9.22 8.78 95.3 NA NA NA

LCS

- 120416

- 9 9.22 8.59 93.2 NA NA NA

Average ± %RSD 95.9 % ± 3.2 % NA

NA = Not Applicable

# 8.7 Analytical Method Uncertainty

Analytical uncertainty is based on historical QC data that is control charted and used to evaluate

method accuracy and preci sion. The method uncertainty is calculated following ETS

- 12

- 012.2. The

standard deviation is calculated for the set of accuracy results (in %) obtained for the QC samples. The

expanded uncertainty is calculated by multiplying the standard deviation by a factor of 2, which

corresponds to a confidence level of 95%.

Table 8 . Analytical Uncertainty.

Analyte Standard Deviation Method Uncertainty

PFBS 6.42 ± 13 %

PFHS 7.00 ± 14 %

PFOS 9.10 ± 18 %



# 8.8 Field Matrix Spikes (FMS)

Low , mid, and high field matrix spikes (FMS) were collected at each sampling point to verify that the

analytical method is applicable to the collected matrix. Field matrix spikes were generated by adding a

measured volume of field sample to a container spike d by the laboratory with PFBS (linear), PFHS

(linear), and PFOS (linear + branched ) prior to shipping sample containers for sample collection. Field

matrix spike recoveries within method acceptance criteria of 100±30% confirm that “unknown”

components in the sample matrix do not significantly interfere with the extraction and analysis of the

analytes of interest. Field matrix spike concentrations must be 50% of the sample concentration to be

considered an appropriate field spike. Field matrix spikes are presented in section 9 of this report.

Table 9 . Field Matrix Spike Levels.

Sampling Location PFBS,

ng/mL PFHS,

ng/mL PFOS,

ng/mL



# All Locations and Trip Blank 2.00 2.00 2.00

100% *

on Concentrat Spike Dup.) Sample Field & Sample Field: ion Concentrat Average FMS of ion Concentrat ( Sample Recovery FMS −

=



# 9 **Data Summary and Discussion**

The tables below summarize the sample results and field matrix spike recoveries for the sampling

locations as well as the Trip Blank

. Results and average values are rounded to three significant figures

according to EPA rounding rules. Because of rounding, values may vary slightly from those listed in the

raw data. Field matrix spike recoveries meeting the method acceptance criteria of ± 30%, demonstrate

that the method was appropriate for the given matrix and their respective quantitative ranges

.

JPAL GW M W 3R 120405; The field matrix spike level was not appropriate as compared to the PFHS

and PFOS concentration s in the sample s . However, the 13 

# C 4 

# - PFOS surrogate recovery added to all

sample bottles met method acceptance criteria.

# JPAL GW MW4 R 120405; The field matrix spike level was not appropriate as compared to the PFHS

and PFOS concentrations in the sample s . However, the 13 

# C 4 

# - PFOS surrogate recovery added to all

sample bottles met method acceptance criteria.

JPAL GW MW6 R 120405; The field matrix spike level was not appropriate as compared to the PFOS

concentration in the sample s . However, the 13 

# C 4 

# - PFOS surrogate recovery added to all sample bottles

met method acceptance criteria.

JPAL GW MW7 R 120405; The field matrix spike level was not appropriate as compared to the PFOS

concentration in the samples. However, the 13 

# C 4 

# - PFOS surrogate recovery added to all sample bottles

met method acceptance criteria.

# Table 10

. JPAL GW MW1R 120405

**PF BS**  **PF HS**  **PF OS**  **13** 

## C **4** 

## - **PFOS**

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **%Recovery**

GLP10

- 01

- 02

- 30

- 001 JPAL

- GW

- MW1R

- 0

- 120405 0.114 NA 0.865 NA 1.62 NA 95.8

GLP10

- 01

- 02

- 30

- 002 JPAL

- GW

- MW1R

- DB

- 120405 0.111 NA 0.856 NA 1.59 NA 99.3

GLP10

- 01

- 02

- 30

- 003 JPA L

- GW

- MW1R

- FMS

- 120405 2.48 118 3.31 122 4.05 122 110

**Average Concentration (ng/mL) ±**  **%RPD /%RSD**  **0.113 ng/mL**  **± 2.7**  **% 0.861**  **ng/mL ±**  **1.0% 1.61**  **ng/mL ±**  **1.9 %**  **102 %**  **± 7.2**  **%**

NA = Not Applicable



# Table 11

. JPAL GW MW3R 120405

**PF BS**  **PF HS**  **PF OS**  **13** 

## C **4** 

## - **PFOS**

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **%Recovery**

GLP10

- 01

- 02

- 30

- 004 JPAL

- GW

- MW3R

- 0

- 120405 1.78 NA 9.04 NA 9.53 NA 97.8

GLP10

- 01

- 02

- 30

- 005 JPAL

- GW

- MW3R

- DB

- 120405 1.79 NA 8.41 NA 9.87 NA 104

GLP10

- 01

- 02

- 30

- 006 JPAL

- GW

- MW3R

- FMS

- 120405 3.63 92.3 10.2 NC 10.9 NC 88.1

**Average Concentration (ng/mL) ±**  **%RPD /%RSD**  **1.79 ng/mL**  **± 0.56%**  **8.73 ng/mL**  **± 7.2%**  **9.70 ng/mL**  **± 3.5**  **% 96.7**  **% ±**  **8.4 %** 

NA = Not Applicable

NC = Not Calculated; the sample concent ration is greater than 2x the spike level.

# Table 12

. JPAL GW MW4R 120405

**PF BS**  **PF HS**  **PF OS**  **13** 

## C **4** 

## - **PFOS**

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **%Recovery**

GLP10

- 01

- 02

- 30

- 0 07 JPAL

- GW

- MW4R

- 0

- 120405 0.188 NA 4.95 NA 5.87 NA 93.1

GLP10

- 01

- 02

- 30

- 008 JPAL

- GW

- MW4R

- DB

- 120405 0.189 NA 4.80 NA 5.74 NA 93.2

GLP10

- 01

- 02

- 30

- 009 JPAL

- GW

- MW4R

- FMS

- 120405 2.15 98.1 6.65 NC 7.85 NC 98.1

**Average Concentration (ng/mL) ±**  **%RPD /%RSD**  **0.189 ng/m**  **L ±**  **0.53% 4.88**  **ng/mL ±**  **3.1% 5.81**  **ng/mL ±**  **2.2 %**  **94.8 %**  **± 3.0**  **%**

NA = Not Applicable

NC = Not Calculated; the sample concentration is greater than 2x the spike level.



# Table 13

. JPAL GW MW4L 120405

**PF BS**  **PF HS**  **PF OS**  **13** 

## C **4** 

## - **PFOS**

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **%Recovery**

GLP10

- 01

- 02

- 30

- 010 JPAL

- GW

- MW4L

- 0

- 120405 0.0759 NA 0.200 NA 0.347 NA 97.1

GLP10

- 01

- 02

- 30

- 011 JPAL

- GW

- MW4L

- DB

- 120405 0.0784 NA 0.209 NA 0.349 NA 93.6

GL P10

- 01

- 02

- 30

- 012 JPAL

- GW

- MW4L

- FMS

- 120405 2.05 98.6 2.15 97.3 2.21 93.1 91.8

**Average Concentration (ng/mL) ±**  **%RPD /%RSD**  **0.0772 ng/mL**  **± 3.2%**  **0.205 ng/mL**  **± 4.4%**  **0.348 ng/mL**  **± 0.57**  **% 94.2**  **% ±**  **2.8 %** 

NA = Not Applicable

# Table 14

. JPAL GW MW5R 120405

**PF BS**  **P F** 

**HS PF**  **OS 13**  

## C **4** 

## - **PFOS**

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **%Recovery**

GLP10

- 01

- 02

- 30

- 013 JPAL

- GW

- MW5R

- 0

- 120405 0.0400 NA 0.133 NA 0.125 NA 96.9

GLP10

- 01

- 02

- 30

- 014 JPAL

- GW

- M W5R

- DB

- 120405 0.0403 NA 0.122 NA 0.131 NA 98.4

GLP10

- 01

- 02

- 30

- 015 JPAL

- GW

- MW5R

- FMS

- 120405 2.03 99.5 2.14 101 1.99 93.1 101

**Average Concentration (ng/mL) ±**  **%RPD /%RSD**  **0.0402 ng/mL**  **± 0.75%**  **0.128 ng/mL**  **± 8.6%**  **0.128 ng/mL**  **± 4.7%**  **98.9% ±**  **2.4%**

NA = Not Applic able



# Table 15

. JPAL GW MW5L 120405

**PF BS**  **PF HS**  **PF OS**  **13** 

## C **4** 

## - **PFOS**

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **%Recovery**

GLP10

- 01

- 02

- 30

- 016 JPAL

- GW

- MW5L

- 0

- 120405 0.259 NA 0.399 NA 2.14 NA 107

GLP10

- 01

- 02

- 30

- 017 JPAL

- GW

- MW5L

- DB

- 120405 0.266 NA 0.400 NA 2.00 NA 95.2

GLP10

- 01

- 02

- 30

- 018 JPAL

- GW

- MW5L

- FMS

- 120405 2.23 98.4 2.33 96.5 4.00 96.5 99.6

**Average Concentration (ng/mL) ±**  **%RPD /%RSD**  **0.263 ng/mL**  **± 2.7%**  **0.400 ng/mL**  **± 0.25%**  **2.07 ng/**  **mL ±**  **6.8% 100%**  **± 5.8%** 

NA = Not Applicable

# Table 16

. JPAL GW MW6R 120405

**PF BS**  **PF HS**  **PF OS**  **13** 

## C **4** 

## - **PFOS**

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **%Recovery**

GLP10

- 01

- 02

- 30

- 019 JPAL

- GW

- MW6R

- 0

- 120405 0.336 NA 1.09 NA 7.89 NA 101

GLP10

- 01

- 02

- 30

- 020 JPAL

- GW

- MW6R

- DB

- 120405 0.331 NA 1.03 NA 7.92 NA 97.1

GLP10

- 01

- 02

- 30

- 021 JPAL

- GW

- MW6R

- FMS

- 120405 2.31 98.8 3.02 98.0 9.47 NC 96.9

**Average Concentration (ng/mL) ±**  **%RPD /%RSD**  **0.334 ng**  **/mL ±**  **1.5% 1.06**  **ng/mL ±**  **5.7% 7.91**  **ng/mL ±**  **0.38% 98.3%**  **± 2.4%** 

NA = Not Applicable

NC = Not Calculated; the sample concentration is greater than 2x the spike level.



# Table 17

. JPAL GW MW6L 120405

**PF BS**  **PF HS**  **PF OS**  **13** 

## C **4** 

## - **PFOS**

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **%Recovery**

GLP10

- 01

- 02

- 30

- 022 JPAL

- GW

- MW6L

- 0

- 120405 0.0548 NA 0.0949 NA 0.365 NA 101

GLP10

- 01

- 02

- 30

- 023 JPAL

- GW

- MW6L

- DB

- 120405 0.0553 NA 0.0955 NA 0.378 NA 97.8

G LP10

- 01

- 02

- 30

- 024 JPAL

- GW

- MW6L

- FMS

- 120405 2.00 97.2 2.04 97.2 2.21 91.9 88.3

**Average Concentration (ng/mL) ±**  **%RPD /%RSD**  **0.0551 ng/mL**  **± 0.91%**  **0.0952 ng/mL**  **± 0.63%**  **0.372 ng/mL**  **± 3.5%**  **95.6% ±**  **6.8%**

NA = Not Applicable

# Table 18

. JPAL GW MW7R 120405

**PF B**  **S PF**  **HS PF**  **OS 13**  

## C **4** 

## - **PFOS**

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **%Recovery**

GLP10

- 01

- 02

- 30

- 025 JPAL

- GW

- MW7R

- 0

- 120405 0.534 NA 1.88 NA 22.7 NA 94.7

GLP10

- 01

- 02

- 30

- 026 JPAL

- GW

- M W7R

- DB

- 120405 0.538 NA 1.86 NA 22.2 NA 91.9

GLP10

- 01

- 02

- 30

- 027 JPAL

- GW

- MW7R

- FMS

- 120405 2.49 97.7 3.68 90.5 NR (1) 

## NC 96.4

**Average Concentration (ng/mL) ±**  **%RPD /%RSD**  **0.536 ng/mL**  **± 0.75%**  **1.87 ng/mL**  **± 1.1%**  **22.5 ng/mL**  **± 2.2%**  **94.4% ±**  **2.4%**

NA = Not Applicabl e

NC = Not Calculated; the sample concentration is greater than 2x the spike level.

(1) NR = Not Repor ted ; sample concentration exceeded the calibration range of 23.2 ng/mL for PFOS

.



# Table 19

. JPAL GW MW8R 120405

**PF BS**  **PF HS**  **PF OS**  **13** 

## C **4** 

## - **PFOS**

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **%Recovery**

GLP10

- 01

- 02

- 30

- 028 JPAL

- GW

- MW8R

- 0

- 120405 0.0492 NA 0.124 NA 0.396 NA 91.7

GLP10

- 01

- 02

- 30

- 029 JPAL

- GW

- MW8R

- DB

- 120405 0.0524 NA 0.131 NA 0.423 NA 94.6

GLP10

- 01

- 02

- 30

- 030 JPAL

- GW

- MW8R

- FMS

- 120405 1.93 94.0 2.06 96.6 2.30 94.5 94.7

**Average Concentration (ng/mL) ±**  **%RPD /%RSD**  **0.0508 ng/mL**  **± 6.3**  **% 0.128**  **ng/mL ±**  **5.5 %**  **0.410 ng/mL**  **± 6.6**  **% 93.7**  **% ±**  **1.8 %** 

NA = Not Applicable

# Table 20

. JPAL GW M W9R 120405

**PF BS**  **PF HS**  **PF OS**  **13** 

## C **4** 

## - **PFOS**

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **%Recovery**

GLP10

- 01

- 02

- 30

- 031 JPAL

- GW

- MW9R

- 0

- 120405 <0.0250 NA 0.0653 NA 0.0652 NA 96.0

GLP10

- 01

- 02

- 30

- 032 JPAL

- GW

- MW9R

- DB

- 120405 <0.0250 NA 0.0646 NA 0.0588 NA 97.6

GLP10

- 01

- 02

- 30

- 033 JPAL

- GW

- MW9R

- FMS

- 120405 2.01 101 2.14 104 1.84 88.9 94.9

**Average Concentration (ng/mL) ±**  **%RPD /%RSD**  **<0.0250 ng/mL**  **0.0650 ng/mL**  **± 1.1%**  **0.0620 ng/mL**  **± 10**  **% 96.2**  **% ±**  **1.4 %** 

NA = Not Applicable



# Table 21

. JPAL GW MW9L 120405

**PF BS**  **PF HS**  **PF OS**  **13** 

## C **4** 

## - **PFOS**

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **%Recovery**

GLP10

- 01

- 02

- 30

- 034 JPAL

- GW

- MW9L

- 0

- 12040 5 <0.0250 NA <0.0250 NA 0.0423 NA 90.6

GLP10

- 01

- 02

- 30

- 035 JPAL

- GW

- MW9L

- DB

- 120405 <0.0250 NA <0.0250 NA 0.0426 NA 94.4

GLP10

- 01

- 02

- 30

- 036 JPAL

- GW

- MW9L

- FMS

- 120405 1.94 97.0 1.99 99.5 1.81 88.4 91.1

**Average Concentration (ng/mL) ±**  **%RPD /%RSD**  **<0.0250 ng/mL**  **< 0.0250**  **ng/mL 0.0425**  **ng/mL ±**  **0.71 %**  **92.0 %**  **± 2.2**  **%**

NA = Not Applicable

# Table 22

. JPAL GW MW10R 120405

**PF BS**  **PF HS**  **PF OS**  **13** 

## C **4** 

## - **PFOS**

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **%Recovery**

GLP10

- 01

- 02

- 30

- 037 JPAL

- GW

- MW10R

- 0

- 120405 

## 0.0602 NA 0.269 NA 0.232 NA 93.8

GLP10

- 01

- 02

- 30

- 038 JPAL

- GW

- MW10R

- DB

- 120405 

## 0.0555 NA 0.258 NA 0.244 NA 94.8

GLP10

- 01

- 02

- 30

- 039 JPAL

- GW

- MW10R

- FMS

- 120405 

## 2.05 99.6 2.20 96.8 2.13 94.6 97.6

**Average Conc entration (ng/mL)**  **± %RPD**  **/%RSD 0.0579**  **ng/mL ±**  **8.1% 0.264**  **ng/mL ±**  **4.2% 0.238**  **ng/mL ±**  **5.0 %**  **95.4 %**  **± 2.1**  **%**

NA = Not Applicable



# Table 23

. JPAL GW MW11L 120405

**PF BS**  **PF HS**  **PF OS**  **13** 

## C **4** 

## - **PFOS**

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**( ng/mL)**  **%Recovery Concentration** 

**(ng/mL) %Recovery**  **%Recovery**

GLP10

- 01

- 02

- 30

- 040 JPAL

- GW

- MW11L

- 0

- 120405 

## <0.0250 NA 0.0919 NA 0.512 NA 86.6

GLP10

- 01

- 02

- 30

- 041 JPAL

- GW

- MW11L

- DB

- 120405 

## <0.0250 NA 0.0852 NA 0.480 NA 98.5

GLP10

- 01

- 02

- 30

- 042 JPAL

- GW

- MW11L

- FMS -1 20405 

## 2.03 102 2.16 104 2.46 98.2 100

**Average Concentration (ng/mL) ±**  **%RPD /%RSD**  **<0.0250 ng/mL**  **0.0886 ng/mL**  **± 7.6%**  **0.496 ng/mL**  **± 6.5**  **% 95.0**  **% ±**  **7.7 %** 

NA = Not Applicable

# Table 24

. Rinseate Blank

**PF BS**  **PF HS**  **PF OS**  **13** 

## C **4** 

## - **PFOS**

**3M LIMS ID Description**  **Conc entration** 

**(ng/mL) Concentration** 

**(ng/mL) Concentration** 

**(ng/mL) %Recovery** 

GLP10

- 01

- 02

- 30

- 043 JPAL

- GW

- MW3R

- RB

- 120405 <0.0250 <0.0250 <0.0232 92.1

NA = Not Applicable



# Table 25

. Trip Blank

**PF BS**  **PF HS**  **PF OS**  **13** 

## C **4** 

## - **PFOS**

**3M LIMS ID Description**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **Concentration**

**(ng/mL) %Recovery**  **%Recovery**

GLP10

- 01

- 02

- 30

- 044 JPAL

- GW

- TRIP01

- 0

- 120316 

## <0.0250 NA <0.0250 NA <0.0232 NA 92.8

GLP10

- 01

- 02

- 30

- 045 JPAL

- GW

- TRIP01

- FMS

- 1 20316 

## 2.00 100 2.00 100 1.82 91.0 97.5

NA = Not Applicable

# 10 **Conclusion**

Laboratory control spikes and field matrix spikes were used to determine the analytical method

accuracy and precision for PFBS, PFHS, and PFOS. A nalysis was successfully completed fo llowing

3M Environmental Laboratory method ETS -8- 044. 1 described herein.



# 11 **Data/Sample Retention**

All remaining samples and associated project data (hardcopy and electronic) will be archived according

to 3M Environmental Laboratory standard operating procedures.



# 12 **Attachments**

At tachment A: Protocol Amendment 30 (General Project Outline)

Attachment B: Representative Chromatograms and Calibration Curves

Attachment C: Analytical Method

- ETS -8- 044. 1

### GLP10-01-02 ; Interi m Repor t 3 0

Analysi s o f PFBS

, PFHS

, an d PFO S i n Groundwate r Sample s

Ber t Jeffrie s Propert y - Apri l 201 2



# 13 Signatures

Clesto n Lange

, Ph.D.

, 3 M Principa l Analytica l Investigato r Dat e

Willia m K

. Reagen

, Ph.D.

, 3 M Environmenta l Laborator y Departmen t Manage r Dat e

Gar y Hohenstein

, Sponso r Representativ e Dat e

Jaisimh a Kesari

, Stud y Directo r ' / Dat e

Pag e 2 9 o f 11 2

# A TTACHMENT 

# A: P ROTOCOL 

# A MENDMENT

# A TTACHMENT 

# B: R EPRESENTATIVE 

# S AMPLE 

# C HROMATOGRAMS AND



# C ALIBRATION 

# C URVE 

# ( S 

# )b120423a.rdb (13C4PFOS-surr): "Quadratic" Regression ("1 / x" weighting): y = -0.0189 x^2 + 0.986 x + 0.0166 (r = 0.9999)0.0 0.5 1.0 1.5 2.0 2.5 3.0 3.5 4.0 4.5 5.0 5.5 6.0 6.5 7.0 7.5 8.0 8.5 9.0 9.5 10.0Analyte Conc. / IS Conc.0.00.51.01.52.02.53.03.54.04.55.05.56.06.57.07.58.0

Analyte Area / IS AreaWorkstation: ETSBUSTERBatch Name: b120423a.dab*Data worked up by KJUPage 1 of 1Printing Time: 2:06:53 PMPrinting Date: Thursday, April 26, 2012b120423a.rdb (PFOS): "Quadratic" Regression ("1 / x" weighting): y = -0.0124 x^2 + 1.53 x + 0.00798 (r = 0.9999)0 1 2 3 4 5 6 7 8 9 10 11 12 13 14 15 16 17 18 19 20 21 22 23 24Analyte Conc. / IS Conc.024681012141618202224262830

Analyte Area / IS AreaWorkstation: ETSBUSTERBatch Name: b120423a.dab*Data worked up by KJUPage 1 of 1Printing Time: 2:06:40 PMPrinting Date: Thursday, April 26, 2012b120423a.rdb (PFHS): "Quadratic" Regression ("1 / x" weighting): y = -0.07 x^2 + 5.49 x + 0.0397 (r = 0.9995)0 2 4 6 810 12 14 16 18 20 22 24 26Analyte Conc. / IS Conc.05101520253035404550556065707580859095100

Analyte Area / IS AreaWorkstation: ETSBUSTERBatch Name: b120423a.dab*Data worked up by KJUPage 1 of 1Printing Time: 2:06:15 PMPrinting Date: Thursday, April 26, 2012b120423a.rdb (PFBS): "Quadratic" Regression ("1 / x" weighting): y = -0.0169 x^2 + 1.27 x + 0.0105 (r = 0.9995)0 1 2 3 4 5 6 7 8 9 10 11 12 13 14 15 16 17 18 19 20 21 22 23 24 25Analyte Conc. / IS Conc.012345678910111213141516171819202122

Analyte Area / IS AreaWorkstation: ETSBUSTERBatch Name: b120423a.dab*Data worked up by KJUPage 1 of 1Printing Time: 2:06:24 PMPrinting Date: Thursday, April 26, 2012Sample Name: "b120423a017" Sample ID: "11012-62-8" File: "b120423a.wiff"Peak Name: "13C3PFHS-IS(IS)" Mass(es): "402.0/99.0 amu"Comment: "1.0 ng/mL FC std in Synth. Water" Annotation: ""Sample Index: 17Sample Type: StandardConcentration: 0.975 ng/mLCalculated Conc: N/AAcq. Date: 4/23/2012Acq. Time: 7:37:05 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 361554 countsHeight: 1.02e+005 cpsStart Time: 15.0 minEnd Time: 15.7 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e5

Intensity, cps15.12Sample Name: "b120423a024" Sample ID: "12001-160" File: "b120423a.wiff"Peak Name: "13C3PFHS-IS(IS)" Mass(es): "402.0/99.0 amu"Comment: "Method Blank" Annotation: ""Sample Index: 24Sample Type: UnknownConcentration: 0.975 ng/mLCalculated Conc: N/AAcq. Date: 4/23/2012Acq. Time: 10:07:34 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 389935 countsHeight: 1.18e+005 cpsStart Time: 15.0 minEnd Time: 15.4 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e51.05e51.10e51.15e5

Intensity, cps15.12Sample Name: "b120423a026" Sample ID: "LCS-120416-1" File: "b120423a.wiff"Peak Name: "13C3PFHS-IS(IS)" Mass(es): "402.0/99.0 amu"Comment: "0.2ppb LCS" Annotation: ""Sample Index: 26Sample Type: QCConcentration: 0.966 ng/mLCalculated Conc: N/AAcq. Date: 4/23/2012Acq. Time: 10:50:29 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 359640 countsHeight: 1.07e+005 cpsStart Time: 15.0 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e51.05e5

Intensity, cps15.12Sample Name: "b120423a029" Sample ID: "LCS-120416-4" File: "b120423a.wiff"Peak Name: "13C3PFHS-IS(IS)" Mass(es): "402.0/99.0 amu"Comment: "2ppb LCS" Annotation: ""Sample Index: 29Sample Type: QCConcentration: 0.966 ng/mLCalculated Conc: N/AAcq. Date: 4/23/2012Acq. Time: 11:54:49 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 370701 countsHeight: 1.09e+005 cpsStart Time: 15.0 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e51.05e5

Intensity, cps15.11*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:54 PMPrinting Date: Thursday, May 03, 2012Page 1 of 48Sample Name: "b120423a032" Sample ID: "LCS-120416-7" File: "b120423a.wiff"Peak Name: "13C3PFHS-IS(IS)" Mass(es): "402.0/99.0 amu"Comment: "10ppb LCS" Annotation: ""Sample Index: 32Sample Type: QCConcentration: 0.969 ng/mLCalculated Conc: N/AAcq. Date: 4/24/2012Acq. Time: 12:59:13 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 303348 countsHeight: 8.43e+004 cpsStart Time: 15.0 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e48.0e4

Intensity, cps15.11Sample Name: "b120423a129" Sample ID: "GLP10-01-02-30-001" File: "b120423a.wiff"Peak Name: "13C3PFHS-IS(IS)" Mass(es): "402.0/99.0 amu"Comment: "JPAL-GW-MW1R-0-" Annotation: ""Sample Index: 129Sample Type: UnknownConcentration: 0.975 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 11:44:06 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 367959 countsHeight: 9.50e+004 cpsStart Time: 15.0 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e48.0e48.5e49.0e49.5e4

Intensity, cps15.06Sample Name: "b120423a131" Sample ID: "GLP10-01-02-30-003" File: "b120423a.wiff"Peak Name: "13C3PFHS-IS(IS)" Mass(es): "402.0/99.0 amu"Comment: "JPAL-GW-MW1R-FMS-" Annotation: ""Sample Index: 131Sample Type: UnknownConcentration: 0.975 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 12:27:03 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 308761 countsHeight: 8.17e+004 cpsStart Time: 15.0 minEnd Time: 15.5 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e48.0e4

Intensity, cps15.05Sample Name: "b120423a135" Sample ID: "GLP10-01-02-30-004" File: "b120423a.wiff"Peak Name: "13C3PFHS-IS(IS)" Mass(es): "402.0/99.0 amu"Comment: "JPAL-GW-MW3R-0-" Annotation: ""Sample Index: 135Sample Type: UnknownConcentration: 0.975 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 1:53:05 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 324283 countsHeight: 8.61e+004 cpsStart Time: 14.9 minEnd Time: 15.5 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e48.0e48.5e4

Intensity, cps15.06*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:54 PMPrinting Date: Thursday, May 03, 2012Page 2 of 48Sample Name: "b120423a138" Sample ID: "GLP10-01-02-30-007" File: "b120423a.wiff"Peak Name: "13C3PFHS-IS(IS)" Mass(es): "402.0/99.0 amu"Comment: "JPAL-GW-MW4R-0-" Annotation: ""Sample Index: 138Sample Type: UnknownConcentration: 0.975 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 2:57:37 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 352639 countsHeight: 9.05e+004 cpsStart Time: 14.9 minEnd Time: 15.4 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e48.0e48.5e49.0e4

Intensity, cps15.06Sample Name: "b120423a141" Sample ID: "GLP10-01-02-30-010" File: "b120423a.wiff"Peak Name: "13C3PFHS-IS(IS)" Mass(es): "402.0/99.0 amu"Comment: "JPAL-GW-MW4L-0-" Annotation: ""Sample Index: 141Sample Type: UnknownConcentration: 0.975 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 4:02:04 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 377376 countsHeight: 9.96e+004 cpsStart Time: 15.0 minEnd Time: 15.5 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e48.0e48.5e49.0e49.5e4

Intensity, cps15.08Sample Name: "b120423a143" Sample ID: "GLP10-01-02-30-012" File: "b120423a.wiff"Peak Name: "13C3PFHS-IS(IS)" Mass(es): "402.0/99.0 amu"Comment: "JPAL-GW-MW4L-FMS-" Annotation: ""Sample Index: 143Sample Type: UnknownConcentration: 0.975 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 4:44:58 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 367607 countsHeight: 1.00e+005 cpsStart Time: 15.0 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e5

Intensity, cps15.08Sample Name: "b120423a147" Sample ID: "GLP10-01-02-30-013" File: "b120423a.wiff"Peak Name: "13C3PFHS-IS(IS)" Mass(es): "402.0/99.0 amu"Comment: "JPAL-GW-MW5R-0-" Annotation: ""Sample Index: 147Sample Type: UnknownConcentration: 0.975 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 6:11:06 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: Base To BaseRetention Time: 15.1 minArea: 351670 countsHeight: 1.03e+005 cpsStart Time: 15.0 minEnd Time: 15.4 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e5

Intensity, cps15.07*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:54 PMPrinting Date: Thursday, May 03, 2012Page 3 of 48Sample Name: "b120423a150" Sample ID: "GLP10-01-02-30-016" File: "b120423a.wiff"Peak Name: "13C3PFHS-IS(IS)" Mass(es): "402.0/99.0 amu"Comment: "JPAL-GW-MW5L-0-" Annotation: ""Sample Index: 150Sample Type: UnknownConcentration: 0.975 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 7:15:38 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: Base To BaseRetention Time: 15.1 minArea: 382714 countsHeight: 1.02e+005 cpsStart Time: 14.9 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e5

Intensity, cps15.05Sample Name: "b120423a152" Sample ID: "GLP10-01-02-30-018" File: "b120423a.wiff"Peak Name: "13C3PFHS-IS(IS)" Mass(es): "402.0/99.0 amu"Comment: "JPAL-GW-MW5L-FMS-" Annotation: ""Sample Index: 152Sample Type: UnknownConcentration: 0.975 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 7:58:43 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 368497 countsHeight: 9.97e+004 cpsStart Time: 14.9 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e48.0e48.5e49.0e49.5e4

Intensity, cps15.06Sample Name: "b120423a153" Sample ID: "GLP10-01-02-30-019" File: "b120423a.wiff"Peak Name: "13C3PFHS-IS(IS)" Mass(es): "402.0/99.0 amu"Comment: "JPAL-GW-MW6R-0-" Annotation: ""Sample Index: 153Sample Type: UnknownConcentration: 0.975 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 8:20:17 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 380468 countsHeight: 1.01e+005 cpsStart Time: 15.0 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e5

Intensity, cps15.06Sample Name: "b120423a159" Sample ID: "GLP10-01-02-30-022" File: "b120423a.wiff"Peak Name: "13C3PFHS-IS(IS)" Mass(es): "402.0/99.0 amu"Comment: "JPAL-GW-MW6L-0-" Annotation: ""Sample Index: 159Sample Type: UnknownConcentration: 0.975 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 10:29:17 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 370567 countsHeight: 9.57e+004 cpsStart Time: 14.9 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e48.0e48.5e49.0e49.5e4

Intensity, cps15.06*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:54 PMPrinting Date: Thursday, May 03, 2012Page 4 of 48Sample Name: "b120423a162" Sample ID: "GLP10-01-02-30-025" File: "b120423a.wiff"Peak Name: "13C3PFHS-IS(IS)" Mass(es): "402.0/99.0 amu"Comment: "JPAL-GW-MW7R-0-" Annotation: ""Sample Index: 162Sample Type: UnknownConcentration: 0.975 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 11:33:51 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 383125 countsHeight: 9.91e+004 cpsStart Time: 15.0 minEnd Time: 15.5 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e48.0e48.5e49.0e49.5e4

Intensity, cps15.08Sample Name: "b120423a165" Sample ID: "GLP10-01-02-30-028" File: "b120423a.wiff"Peak Name: "13C3PFHS-IS(IS)" Mass(es): "402.0/99.0 amu"Comment: "JPAL-GW-MW8R-0-" Annotation: ""Sample Index: 165Sample Type: UnknownConcentration: 0.975 ng/mLCalculated Conc: N/AAcq. Date: 4/26/2012Acq. Time: 12:38:33 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 407311 countsHeight: 1.04e+005 cpsStart Time: 15.0 minEnd Time: 15.5 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e5

Intensity, cps15.09Sample Name: "b120423a171" Sample ID: "GLP10-01-02-30-031" File: "b120423a.wiff"Peak Name: "13C3PFHS-IS(IS)" Mass(es): "402.0/99.0 amu"Comment: "JPAL-GW-MW9R-0-" Annotation: ""Sample Index: 171Sample Type: UnknownConcentration: 0.975 ng/mLCalculated Conc: N/AAcq. Date: 4/26/2012Acq. Time: 2:47:37 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 415777 countsHeight: 1.10e+005 cpsStart Time: 15.0 minEnd Time: 15.8 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e51.05e51.10e5

Intensity, cps15.10Sample Name: "b120423a173" Sample ID: "GLP10-01-02-30-033" File: "b120423a.wiff"Peak Name: "13C3PFHS-IS(IS)" Mass(es): "402.0/99.0 amu"Comment: "JPAL-GW-MW9R-FMS-" Annotation: ""Sample Index: 173Sample Type: UnknownConcentration: 0.975 ng/mLCalculated Conc: N/AAcq. Date: 4/26/2012Acq. Time: 3:30:38 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 372399 countsHeight: 9.94e+004 cpsStart Time: 15.0 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e48.0e48.5e49.0e49.5e4

Intensity, cps15.08*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:54 PMPrinting Date: Thursday, May 03, 2012Page 5 of 48Sample Name: "b120423a174" Sample ID: "GLP10-01-02-30-034" File: "b120423a.wiff"Peak Name: "13C3PFHS-IS(IS)" Mass(es): "402.0/99.0 amu"Comment: "JPAL-GW-MW9L-0-" Annotation: ""Sample Index: 174Sample Type: UnknownConcentration: 0.975 ng/mLCalculated Conc: N/AAcq. Date: 4/26/2012Acq. Time: 3:52:08 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 395626 countsHeight: 1.03e+005 cpsStart Time: 14.9 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e5

Intensity, cps15.07Sample Name: "b120423a177" Sample ID: "GLP10-01-02-30-037" File: "b120423a.wiff"Peak Name: "13C3PFHS-IS(IS)" Mass(es): "402.0/99.0 amu"Comment: "JPAL-GW-MW10R-0-" Annotation: ""Sample Index: 177Sample Type: UnknownConcentration: 0.975 ng/mLCalculated Conc: N/AAcq. Date: 4/26/2012Acq. Time: 4:56:43 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 396365 countsHeight: 1.05e+005 cpsStart Time: 15.0 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e51.05e5

Intensity, cps15.07Sample Name: "b120423a183" Sample ID: "GLP10-01-02-30-040" File: "b120423a.wiff"Peak Name: "13C3PFHS-IS(IS)" Mass(es): "402.0/99.0 amu"Comment: "JPAL-GW-MW11L-0-" Annotation: ""Sample Index: 183Sample Type: UnknownConcentration: 0.975 ng/mLCalculated Conc: N/AAcq. Date: 4/26/2012Acq. Time: 7:05:50 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 366895 countsHeight: 9.52e+004 cpsStart Time: 14.9 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e48.0e48.5e49.0e49.5e4

Intensity, cps15.06Sample Name: "b120423a185" Sample ID: "GLP10-01-02-30-042" File: "b120423a.wiff"Peak Name: "13C3PFHS-IS(IS)" Mass(es): "402.0/99.0 amu"Comment: "JPAL-GW-MW11L-FMS-" Annotation: ""Sample Index: 185Sample Type: UnknownConcentration: 0.975 ng/mLCalculated Conc: N/AAcq. Date: 4/26/2012Acq. Time: 7:48:55 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 361079 countsHeight: 9.66e+004 cpsStart Time: 15.0 minEnd Time: 15.6 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e48.0e48.5e49.0e49.5e4

Intensity, cps*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:54 PMPrinting Date: Thursday, May 03, 2012Page 6 of 48Sample Name: "b120423a186" Sample ID: "GLP10-01-02-30-043" File: "b120423a.wiff"Peak Name: "13C3PFHS-IS(IS)" Mass(es): "402.0/99.0 amu"Comment: "JPAL-GW-MW3R-RB-" Annotation: ""Sample Index: 186Sample Type: UnknownConcentration: 0.975 ng/mLCalculated Conc: N/AAcq. Date: 4/26/2012Acq. Time: 8:10:35 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 404766 countsHeight: 1.07e+005 cpsStart Time: 14.9 minEnd Time: 15.4 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e51.05e5

Intensity, cps15.06Sample Name: "b120423a187" Sample ID: "GLP10-01-02-30-044" File: "b120423a.wiff"Peak Name: "13C3PFHS-IS(IS)" Mass(es): "402.0/99.0 amu"Comment: "JPAL-GW-TRIP01-0-" Annotation: ""Sample Index: 187Sample Type: UnknownConcentration: 0.975 ng/mLCalculated Conc: N/AAcq. Date: 4/26/2012Acq. Time: 8:32:03 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 406347 countsHeight: 1.08e+005 cpsStart Time: 15.0 minEnd Time: 15.5 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e51.05e5

Intensity, cps15.06Sample Name: "b120423a188" Sample ID: "GLP10-01-02-30-045" File: "b120423a.wiff"Peak Name: "13C3PFHS-IS(IS)" Mass(es): "402.0/99.0 amu"Comment: "JPAL-GW-TRIP01-FMS-" Annotation: ""Sample Index: 188Sample Type: UnknownConcentration: 0.975 ng/mLCalculated Conc: N/AAcq. Date: 4/26/2012Acq. Time: 8:53:31 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 15.1 minArea: 389518 countsHeight: 1.07e+005 cpsStart Time: 15.0 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e51.05e5

Intensity, cps15.06Sample Name: "b120423a017" Sample ID: "11012-62-8" File: "b120423a.wiff"Peak Name: "13C4PFOS-surr" Mass(es): "503.0/80.0 amu"Comment: "1.0 ng/mL FC std in Synth. Water" Annotation: ""Sample Index: 17Sample Type: StandardConcentration: 0.956 ng/mLCalculated Conc: 0.954 ng/mLAcq. Date: 4/23/2012Acq. Time: 7:37:05 PMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.1 minArea: 770277 countsHeight: 2.33e+005 cpsStart Time: 16.0 minEnd Time: 16.3 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e51.9e52.0e52.1e52.2e52.3e5

Intensity, cps16.07*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:54 PMPrinting Date: Thursday, May 03, 2012Page 7 of 48Sample Name: "b120423a024" Sample ID: "12001-160" File: "b120423a.wiff"Peak Name: "13C4PFOS-surr" Mass(es): "503.0/80.0 amu"Comment: "Method Blank" Annotation: ""Sample Index: 24Sample Type: UnknownConcentration: N/ACalculated Conc: 0.000165 ng/mLAcq. Date: 4/23/2012Acq. Time: 10:07:34 PMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.1 minArea: 22816 countsHeight: 6.54e+003 cpsStart Time: 16.0 minEnd Time: 16.3 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0500100015002000250030003500400045005000550060006500

Intensity, cpsSample Name: "b120423a026" Sample ID: "LCS-120416-1" File: "b120423a.wiff"Peak Name: "13C4PFOS-surr" Mass(es): "503.0/80.0 amu"Comment: "0.2ppb LCS" Annotation: ""Sample Index: 26Sample Type: QCConcentration: 0.189 ng/mLCalculated Conc: 0.202 ng/mLAcq. Date: 4/23/2012Acq. Time: 10:50:29 PMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.1 minArea: 236159 countsHeight: 7.00e+004 cpsStart Time: 16.0 minEnd Time: 16.4 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e4

Intensity, cps16.06Sample Name: "b120423a029" Sample ID: "LCS-120416-4" File: "b120423a.wiff"Peak Name: "13C4PFOS-surr" Mass(es): "503.0/80.0 amu"Comment: "2ppb LCS" Annotation: ""Sample Index: 29Sample Type: QCConcentration: 1.89 ng/mLCalculated Conc: 1.91 ng/mLAcq. Date: 4/23/2012Acq. Time: 11:54:49 PMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.1 minArea: 2311600 countsHeight: 6.78e+005 cpsStart Time: 15.9 minEnd Time: 16.3 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e5

Intensity, cps16.06Sample Name: "b120423a032" Sample ID: "LCS-120416-7" File: "b120423a.wiff"Peak Name: "13C4PFOS-surr" Mass(es): "503.0/80.0 amu"Comment: "10ppb LCS" Annotation: ""Sample Index: 32Sample Type: QCConcentration: 0.00 ng/mLCalculated Conc: 0.00606 ng/mLAcq. Date: 4/24/2012Acq. Time: 12:59:13 AMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.1 minArea: 25882 countsHeight: 7.52e+003 cpsStart Time: 15.9 minEnd Time: 16.3 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min050010001500200025003000350040004500500055006000650070007500

Intensity, cps*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:54 PMPrinting Date: Thursday, May 03, 2012Page 8 of 48Sample Name: "b120423a129" Sample ID: "GLP10-01-02-30-001" File: "b120423a.wiff"Peak Name: "13C4PFOS-surr" Mass(es): "503.0/80.0 amu"Comment: "JPAL-GW-MW1R-0-" Annotation: ""Sample Index: 129Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0916 ng/mLAcq. Date: 4/25/2012Acq. Time: 11:44:06 AMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 96575 countsHeight: 2.47e+004 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01000.02000.03000.04000.05000.06000.07000.08000.09000.01.0e41.1e41.2e41.3e41.4e41.5e41.6e41.7e41.8e41.9e42.0e42.1e42.2e42.3e42.4e4

Intensity, cps16.00Sample Name: "b120423a131" Sample ID: "GLP10-01-02-30-003" File: "b120423a.wiff"Peak Name: "13C4PFOS-surr" Mass(es): "503.0/80.0 amu"Comment: "JPAL-GW-MW1R-FMS-" Annotation: ""Sample Index: 131Sample Type: UnknownConcentration: N/ACalculated Conc: 0.105 ng/mLAcq. Date: 4/25/2012Acq. Time: 12:27:03 PMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 108462 countsHeight: 2.76e+004 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01000.02000.03000.04000.05000.06000.07000.08000.09000.01.0e41.1e41.2e41.3e41.4e41.5e41.6e41.7e41.8e41.9e42.0e42.1e42.2e42.3e42.4e42.5e42.6e42.7e4

Intensity, cps16.00Sample Name: "b120423a135" Sample ID: "GLP10-01-02-30-004" File: "b120423a.wiff"Peak Name: "13C4PFOS-surr" Mass(es): "503.0/80.0 amu"Comment: "JPAL-GW-MW3R-0-" Annotation: ""Sample Index: 135Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0935 ng/mLAcq. Date: 4/25/2012Acq. Time: 1:53:05 PMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 125837 countsHeight: 3.22e+004 cpsStart Time: 15.9 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e4

Intensity, cps16.00Sample Name: "b120423a138" Sample ID: "GLP10-01-02-30-007" File: "b120423a.wiff"Peak Name: "13C4PFOS-surr" Mass(es): "503.0/80.0 amu"Comment: "JPAL-GW-MW4R-0-" Annotation: ""Sample Index: 138Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0890 ng/mLAcq. Date: 4/25/2012Acq. Time: 2:57:37 PMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 142835 countsHeight: 3.70e+004 cpsStart Time: 15.8 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e43.4e43.6e4

Intensity, cps16.00*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:54 PMPrinting Date: Thursday, May 03, 2012Page 9 of 48Sample Name: "b120423a141" Sample ID: "GLP10-01-02-30-010" File: "b120423a.wiff"Peak Name: "13C4PFOS-surr" Mass(es): "503.0/80.0 amu"Comment: "JPAL-GW-MW4L-0-" Annotation: ""Sample Index: 141Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0928 ng/mLAcq. Date: 4/25/2012Acq. Time: 4:02:04 PMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 105745 countsHeight: 2.97e+004 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e4

Intensity, cps16.02Sample Name: "b120423a143" Sample ID: "GLP10-01-02-30-012" File: "b120423a.wiff"Peak Name: "13C4PFOS-surr" Mass(es): "503.0/80.0 amu"Comment: "JPAL-GW-MW4L-FMS-" Annotation: ""Sample Index: 143Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0878 ng/mLAcq. Date: 4/25/2012Acq. Time: 4:44:58 PMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 94551 countsHeight: 2.65e+004 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01000.02000.03000.04000.05000.06000.07000.08000.09000.01.0e41.1e41.2e41.3e41.4e41.5e41.6e41.7e41.8e41.9e42.0e42.1e42.2e42.3e42.4e42.5e42.6e4

Intensity, cps16.02Sample Name: "b120423a147" Sample ID: "GLP10-01-02-30-013" File: "b120423a.wiff"Peak Name: "13C4PFOS-surr" Mass(es): "503.0/80.0 amu"Comment: "JPAL-GW-MW5R-0-" Annotation: ""Sample Index: 147Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0926 ng/mLAcq. Date: 4/25/2012Acq. Time: 6:11:06 PMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 148548 countsHeight: 3.98e+004 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e43.4e43.6e43.8e4

Intensity, cps16.01Sample Name: "b120423a150" Sample ID: "GLP10-01-02-30-016" File: "b120423a.wiff"Peak Name: "13C4PFOS-surr" Mass(es): "503.0/80.0 amu"Comment: "JPAL-GW-MW5L-0-" Annotation: ""Sample Index: 150Sample Type: UnknownConcentration: N/ACalculated Conc: 0.102 ng/mLAcq. Date: 4/25/2012Acq. Time: 7:15:38 PMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 93055 countsHeight: 2.55e+004 cpsStart Time: 15.8 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01000.02000.03000.04000.05000.06000.07000.08000.09000.01.0e41.1e41.2e41.3e41.4e41.5e41.6e41.7e41.8e41.9e42.0e42.1e42.2e42.3e42.4e42.5e4

Intensity, cps16.00*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:54 PMPrinting Date: Thursday, May 03, 2012Page 10 of 48Sample Name: "b120423a152" Sample ID: "GLP10-01-02-30-018" File: "b120423a.wiff"Peak Name: "13C4PFOS-surr" Mass(es): "503.0/80.0 amu"Comment: "JPAL-GW-MW5L-FMS-" Annotation: ""Sample Index: 152Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0952 ng/mLAcq. Date: 4/25/2012Acq. Time: 7:58:43 PMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 86234 countsHeight: 2.40e+004 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01000.02000.03000.04000.05000.06000.07000.08000.09000.01.0e41.1e41.2e41.3e41.4e41.5e41.6e41.7e41.8e41.9e42.0e42.1e42.2e42.3e4

Intensity, cps16.00Sample Name: "b120423a153" Sample ID: "GLP10-01-02-30-019" File: "b120423a.wiff"Peak Name: "13C4PFOS-surr" Mass(es): "503.0/80.0 amu"Comment: "JPAL-GW-MW6R-0-" Annotation: ""Sample Index: 153Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0966 ng/mLAcq. Date: 4/25/2012Acq. Time: 8:20:17 PMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 137119 countsHeight: 3.82e+004 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e43.4e43.6e43.8e4

Intensity, cps16.00Sample Name: "b120423a159" Sample ID: "GLP10-01-02-30-022" File: "b120423a.wiff"Peak Name: "13C4PFOS-surr" Mass(es): "503.0/80.0 amu"Comment: "JPAL-GW-MW6L-0-" Annotation: ""Sample Index: 159Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0964 ng/mLAcq. Date: 4/25/2012Acq. Time: 10:29:17 PMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 73755 countsHeight: 1.95e+004 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01000.02000.03000.04000.05000.06000.07000.08000.09000.01.0e41.1e41.2e41.3e41.4e41.5e41.6e41.7e41.8e41.9e4

Intensity, cps16.00Sample Name: "b120423a162" Sample ID: "GLP10-01-02-30-025" File: "b120423a.wiff"Peak Name: "13C4PFOS-surr" Mass(es): "503.0/80.0 amu"Comment: "JPAL-GW-MW7R-0-" Annotation: ""Sample Index: 162Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0905 ng/mLAcq. Date: 4/25/2012Acq. Time: 11:33:51 PMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 123445 countsHeight: 3.19e+004 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e4

Intensity, cps16.02*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:54 PMPrinting Date: Thursday, May 03, 2012Page 11 of 48Sample Name: "b120423a165" Sample ID: "GLP10-01-02-30-028" File: "b120423a.wiff"Peak Name: "13C4PFOS-surr" Mass(es): "503.0/80.0 amu"Comment: "JPAL-GW-MW8R-0-" Annotation: ""Sample Index: 165Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0877 ng/mLAcq. Date: 4/26/2012Acq. Time: 12:38:33 AMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 129232 countsHeight: 3.44e+004 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e43.4e4

Intensity, cps16.02Sample Name: "b120423a171" Sample ID: "GLP10-01-02-30-031" File: "b120423a.wiff"Peak Name: "13C4PFOS-surr" Mass(es): "503.0/80.0 amu"Comment: "JPAL-GW-MW9R-0-" Annotation: ""Sample Index: 171Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0918 ng/mLAcq. Date: 4/26/2012Acq. Time: 2:47:37 AMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 140890 countsHeight: 3.73e+004 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e43.4e43.6e4

Intensity, cps16.04Sample Name: "b120423a173" Sample ID: "GLP10-01-02-30-033" File: "b120423a.wiff"Peak Name: "13C4PFOS-surr" Mass(es): "503.0/80.0 amu"Comment: "JPAL-GW-MW9R-FMS-" Annotation: ""Sample Index: 173Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0907 ng/mLAcq. Date: 4/26/2012Acq. Time: 3:30:38 AMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 146911 countsHeight: 3.84e+004 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e43.4e43.6e43.8e4

Intensity, cps16.02Sample Name: "b120423a174" Sample ID: "GLP10-01-02-30-034" File: "b120423a.wiff"Peak Name: "13C4PFOS-surr" Mass(es): "503.0/80.0 amu"Comment: "JPAL-GW-MW9L-0-" Annotation: ""Sample Index: 174Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0866 ng/mLAcq. Date: 4/26/2012Acq. Time: 3:52:08 AMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 129142 countsHeight: 3.22e+004 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e4

Intensity, cps16.01*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:55 PMPrinting Date: Thursday, May 03, 2012Page 12 of 48Sample Name: "b120423a177" Sample ID: "GLP10-01-02-30-037" File: "b120423a.wiff"Peak Name: "13C4PFOS-surr" Mass(es): "503.0/80.0 amu"Comment: "JPAL-GW-MW10R-0-" Annotation: ""Sample Index: 177Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0897 ng/mLAcq. Date: 4/26/2012Acq. Time: 4:56:43 AMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 144417 countsHeight: 3.69e+004 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e43.4e43.6e4

Intensity, cps16.01Sample Name: "b120423a183" Sample ID: "GLP10-01-02-30-040" File: "b120423a.wiff"Peak Name: "13C4PFOS-surr" Mass(es): "503.0/80.0 amu"Comment: "JPAL-GW-MW11L-0-" Annotation: ""Sample Index: 183Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0828 ng/mLAcq. Date: 4/26/2012Acq. Time: 7:05:50 AMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 75053 countsHeight: 1.97e+004 cpsStart Time: 15.8 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01000.02000.03000.04000.05000.06000.07000.08000.09000.01.0e41.1e41.2e41.3e41.4e41.5e41.6e41.7e41.8e41.9e4

Intensity, cps16.00Sample Name: "b120423a185" Sample ID: "GLP10-01-02-30-042" File: "b120423a.wiff"Peak Name: "13C4PFOS-surr" Mass(es): "503.0/80.0 amu"Comment: "JPAL-GW-MW11L-FMS-" Annotation: ""Sample Index: 185Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0956 ng/mLAcq. Date: 4/26/2012Acq. Time: 7:48:55 AMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 96873 countsHeight: 2.53e+004 cpsStart Time: 15.9 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01000.02000.03000.04000.05000.06000.07000.08000.09000.01.0e41.1e41.2e41.3e41.4e41.5e41.6e41.7e41.8e41.9e42.0e42.1e42.2e42.3e42.4e42.5e4

Intensity, cps16.00Sample Name: "b120423a186" Sample ID: "GLP10-01-02-30-043" File: "b120423a.wiff"Peak Name: "13C4PFOS-surr" Mass(es): "503.0/80.0 amu"Comment: "JPAL-GW-MW3R-RB-" Annotation: ""Sample Index: 186Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0880 ng/mLAcq. Date: 4/26/2012Acq. Time: 8:10:35 AMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 154375 countsHeight: 4.20e+004 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e43.4e43.6e43.8e44.0e44.2e4

Intensity, cps16.00*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:55 PMPrinting Date: Thursday, May 03, 2012Page 13 of 48Sample Name: "b120423a187" Sample ID: "GLP10-01-02-30-044" File: "b120423a.wiff"Peak Name: "13C4PFOS-surr" Mass(es): "503.0/80.0 amu"Comment: "JPAL-GW-TRIP01-0-" Annotation: ""Sample Index: 187Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0887 ng/mLAcq. Date: 4/26/2012Acq. Time: 8:32:03 AMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 143722 countsHeight: 3.71e+004 cpsStart Time: 15.9 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e43.4e43.6e4

Intensity, cps16.00Sample Name: "b120423a188" Sample ID: "GLP10-01-02-30-045" File: "b120423a.wiff"Peak Name: "13C4PFOS-surr" Mass(es): "503.0/80.0 amu"Comment: "JPAL-GW-TRIP01-FMS-" Annotation: ""Sample Index: 188Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0932 ng/mLAcq. Date: 4/26/2012Acq. Time: 8:53:31 AMModified: NoProc. Algorithm: IntelliQuan - IQAMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 181700 countsHeight: 4.59e+004 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e43.4e43.6e43.8e44.0e44.2e44.4e4

Intensity, cps16.00Sample Name: "b120423a017" Sample ID: "11012-62-8" File: "b120423a.wiff"Peak Name: "13C8PFOS-IS(IS)" Mass(es): "507.0/80.0 amu"Comment: "1.0 ng/mL FC std in Synth. Water" Annotation: ""Sample Index: 17Sample Type: StandardConcentration: 0.960 ng/mLCalculated Conc: N/AAcq. Date: 4/23/2012Acq. Time: 7:37:05 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.1 minArea: 787582 countsHeight: 2.37e+005 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e51.9e52.0e52.1e52.2e52.3e5

Intensity, cps16.07Sample Name: "b120423a024" Sample ID: "12001-160" File: "b120423a.wiff"Peak Name: "13C8PFOS-IS(IS)" Mass(es): "507.0/80.0 amu"Comment: "Method Blank" Annotation: ""Sample Index: 24Sample Type: UnknownConcentration: 0.960 ng/mLCalculated Conc: N/AAcq. Date: 4/23/2012Acq. Time: 10:07:34 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.1 minArea: 1359480 countsHeight: 4.10e+005 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e5

Intensity, cps16.07*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:55 PMPrinting Date: Thursday, May 03, 2012Page 14 of 48Sample Name: "b120423a026" Sample ID: "LCS-120416-1" File: "b120423a.wiff"Peak Name: "13C8PFOS-IS(IS)" Mass(es): "507.0/80.0 amu"Comment: "0.2ppb LCS" Annotation: ""Sample Index: 26Sample Type: QCConcentration: 0.951 ng/mLCalculated Conc: N/AAcq. Date: 4/23/2012Acq. Time: 10:50:29 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.1 minArea: 1050554 countsHeight: 3.29e+005 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e5

Intensity, cps16.06Sample Name: "b120423a029" Sample ID: "LCS-120416-4" File: "b120423a.wiff"Peak Name: "13C8PFOS-IS(IS)" Mass(es): "507.0/80.0 amu"Comment: "2ppb LCS" Annotation: ""Sample Index: 29Sample Type: QCConcentration: 0.951 ng/mLCalculated Conc: N/AAcq. Date: 4/23/2012Acq. Time: 11:54:49 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.1 minArea: 1202973 countsHeight: 3.55e+005 cpsStart Time: 16.0 minEnd Time: 16.3 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e5

Intensity, cps16.06Sample Name: "b120423a032" Sample ID: "LCS-120416-7" File: "b120423a.wiff"Peak Name: "13C8PFOS-IS(IS)" Mass(es): "507.0/80.0 amu"Comment: "10ppb LCS" Annotation: ""Sample Index: 32Sample Type: QCConcentration: 0.954 ng/mLCalculated Conc: N/AAcq. Date: 4/24/2012Acq. Time: 12:59:13 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.1 minArea: 1131136 countsHeight: 3.23e+005 cpsStart Time: 16.0 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e5

Intensity, cps16.05Sample Name: "b120423a129" Sample ID: "GLP10-01-02-30-001" File: "b120423a.wiff"Peak Name: "13C8PFOS-IS(IS)" Mass(es): "507.0/80.0 amu"Comment: "JPAL-GW-MW1R-0-" Annotation: ""Sample Index: 129Sample Type: UnknownConcentration: 0.960 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 11:44:06 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 873731 countsHeight: 2.24e+005 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e51.9e52.0e52.1e52.2e5

Intensity, cps16.00*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:55 PMPrinting Date: Thursday, May 03, 2012Page 15 of 48Sample Name: "b120423a131" Sample ID: "GLP10-01-02-30-003" File: "b120423a.wiff"Peak Name: "13C8PFOS-IS(IS)" Mass(es): "507.0/80.0 amu"Comment: "JPAL-GW-MW1R-FMS-" Annotation: ""Sample Index: 131Sample Type: UnknownConcentration: 0.960 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 12:27:03 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 874182 countsHeight: 2.27e+005 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e51.9e52.0e52.1e52.2e5

Intensity, cps16.00Sample Name: "b120423a135" Sample ID: "GLP10-01-02-30-004" File: "b120423a.wiff"Peak Name: "13C8PFOS-IS(IS)" Mass(es): "507.0/80.0 amu"Comment: "JPAL-GW-MW3R-0-" Annotation: ""Sample Index: 135Sample Type: UnknownConcentration: 0.960 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 1:53:05 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 1118749 countsHeight: 2.95e+005 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e5

Intensity, cps16.00Sample Name: "b120423a138" Sample ID: "GLP10-01-02-30-007" File: "b120423a.wiff"Peak Name: "13C8PFOS-IS(IS)" Mass(es): "507.0/80.0 amu"Comment: "JPAL-GW-MW4R-0-" Annotation: ""Sample Index: 138Sample Type: UnknownConcentration: 0.960 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 2:57:37 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 1324151 countsHeight: 3.26e+005 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e5

Intensity, cps16.00Sample Name: "b120423a141" Sample ID: "GLP10-01-02-30-010" File: "b120423a.wiff"Peak Name: "13C8PFOS-IS(IS)" Mass(es): "507.0/80.0 amu"Comment: "JPAL-GW-MW4L-0-" Annotation: ""Sample Index: 141Sample Type: UnknownConcentration: 0.960 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 4:02:04 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 945843 countsHeight: 2.58e+005 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e51.9e52.0e52.1e52.2e52.3e52.4e52.5e5

Intensity, cps16.02*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:55 PMPrinting Date: Thursday, May 03, 2012Page 16 of 48Sample Name: "b120423a143" Sample ID: "GLP10-01-02-30-012" File: "b120423a.wiff"Peak Name: "13C8PFOS-IS(IS)" Mass(es): "507.0/80.0 amu"Comment: "JPAL-GW-MW4L-FMS-" Annotation: ""Sample Index: 143Sample Type: UnknownConcentration: 0.960 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 4:44:58 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 886356 countsHeight: 2.49e+005 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e51.9e52.0e52.1e52.2e52.3e52.4e5

Intensity, cps16.02Sample Name: "b120423a147" Sample ID: "GLP10-01-02-30-013" File: "b120423a.wiff"Peak Name: "13C8PFOS-IS(IS)" Mass(es): "507.0/80.0 amu"Comment: "JPAL-GW-MW5R-0-" Annotation: ""Sample Index: 147Sample Type: UnknownConcentration: 0.960 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 6:11:06 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 1331542 countsHeight: 3.67e+005 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e5

Intensity, cps16.01Sample Name: "b120423a150" Sample ID: "GLP10-01-02-30-016" File: "b120423a.wiff"Peak Name: "13C8PFOS-IS(IS)" Mass(es): "507.0/80.0 amu"Comment: "JPAL-GW-MW5L-0-" Annotation: ""Sample Index: 150Sample Type: UnknownConcentration: 0.960 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 7:15:38 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 768371 countsHeight: 2.16e+005 cpsStart Time: 15.9 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e51.9e52.0e52.1e5

Intensity, cps16.00Sample Name: "b120423a152" Sample ID: "GLP10-01-02-30-018" File: "b120423a.wiff"Peak Name: "13C8PFOS-IS(IS)" Mass(es): "507.0/80.0 amu"Comment: "JPAL-GW-MW5L-FMS-" Annotation: ""Sample Index: 152Sample Type: UnknownConcentration: 0.960 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 7:58:43 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 755129 countsHeight: 2.10e+005 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e51.9e52.0e52.1e5

Intensity, cps16.00*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:55 PMPrinting Date: Thursday, May 03, 2012Page 17 of 48Sample Name: "b120423a153" Sample ID: "GLP10-01-02-30-019" File: "b120423a.wiff"Peak Name: "13C8PFOS-IS(IS)" Mass(es): "507.0/80.0 amu"Comment: "JPAL-GW-MW6R-0-" Annotation: ""Sample Index: 153Sample Type: UnknownConcentration: 0.960 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 8:20:17 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 1185585 countsHeight: 3.17e+005 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e5

Intensity, cps16.00Sample Name: "b120423a159" Sample ID: "GLP10-01-02-30-022" File: "b120423a.wiff"Peak Name: "13C8PFOS-IS(IS)" Mass(es): "507.0/80.0 amu"Comment: "JPAL-GW-MW6L-0-" Annotation: ""Sample Index: 159Sample Type: UnknownConcentration: 0.960 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 10:29:17 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 638706 countsHeight: 1.69e+005 cpsStart Time: 15.9 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e5

Intensity, cps16.00Sample Name: "b120423a162" Sample ID: "GLP10-01-02-30-025" File: "b120423a.wiff"Peak Name: "13C8PFOS-IS(IS)" Mass(es): "507.0/80.0 amu"Comment: "JPAL-GW-MW7R-0-" Annotation: ""Sample Index: 162Sample Type: UnknownConcentration: 0.960 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 11:33:51 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 1127839 countsHeight: 2.89e+005 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e5

Intensity, cps16.02Sample Name: "b120423a165" Sample ID: "GLP10-01-02-30-028" File: "b120423a.wiff"Peak Name: "13C8PFOS-IS(IS)" Mass(es): "507.0/80.0 amu"Comment: "JPAL-GW-MW8R-0-" Annotation: ""Sample Index: 165Sample Type: UnknownConcentration: 0.960 ng/mLCalculated Conc: N/AAcq. Date: 4/26/2012Acq. Time: 12:38:33 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 1212507 countsHeight: 3.25e+005 cpsStart Time: 15.9 minEnd Time: 16.3 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e5

Intensity, cps16.02*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:55 PMPrinting Date: Thursday, May 03, 2012Page 18 of 48Sample Name: "b120423a171" Sample ID: "GLP10-01-02-30-031" File: "b120423a.wiff"Peak Name: "13C8PFOS-IS(IS)" Mass(es): "507.0/80.0 amu"Comment: "JPAL-GW-MW9R-0-" Annotation: ""Sample Index: 171Sample Type: UnknownConcentration: 0.960 ng/mLCalculated Conc: N/AAcq. Date: 4/26/2012Acq. Time: 2:47:37 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 1271805 countsHeight: 3.28e+005 cpsStart Time: 15.9 minEnd Time: 16.3 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e5

Intensity, cps16.04Sample Name: "b120423a173" Sample ID: "GLP10-01-02-30-033" File: "b120423a.wiff"Peak Name: "13C8PFOS-IS(IS)" Mass(es): "507.0/80.0 amu"Comment: "JPAL-GW-MW9R-FMS-" Annotation: ""Sample Index: 173Sample Type: UnknownConcentration: 0.960 ng/mLCalculated Conc: N/AAcq. Date: 4/26/2012Acq. Time: 3:30:38 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 1340231 countsHeight: 3.56e+005 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e5

Intensity, cps16.02Sample Name: "b120423a174" Sample ID: "GLP10-01-02-30-034" File: "b120423a.wiff"Peak Name: "13C8PFOS-IS(IS)" Mass(es): "507.0/80.0 amu"Comment: "JPAL-GW-MW9L-0-" Annotation: ""Sample Index: 174Sample Type: UnknownConcentration: 0.960 ng/mLCalculated Conc: N/AAcq. Date: 4/26/2012Acq. Time: 3:52:08 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 1224729 countsHeight: 3.13e+005 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e5

Intensity, cps16.01Sample Name: "b120423a177" Sample ID: "GLP10-01-02-30-037" File: "b120423a.wiff"Peak Name: "13C8PFOS-IS(IS)" Mass(es): "507.0/80.0 amu"Comment: "JPAL-GW-MW10R-0-" Annotation: ""Sample Index: 177Sample Type: UnknownConcentration: 0.960 ng/mLCalculated Conc: N/AAcq. Date: 4/26/2012Acq. Time: 4:56:43 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 1329768 countsHeight: 3.54e+005 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e5

Intensity, cps16.01*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:55 PMPrinting Date: Thursday, May 03, 2012Page 19 of 48Sample Name: "b120423a183" Sample ID: "GLP10-01-02-30-040" File: "b120423a.wiff"Peak Name: "13C8PFOS-IS(IS)" Mass(es): "507.0/80.0 amu"Comment: "JPAL-GW-MW11L-0-" Annotation: ""Sample Index: 183Sample Type: UnknownConcentration: 0.960 ng/mLCalculated Conc: N/AAcq. Date: 4/26/2012Acq. Time: 7:05:50 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 738955 countsHeight: 1.93e+005 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e51.9e5

Intensity, cps16.00Sample Name: "b120423a185" Sample ID: "GLP10-01-02-30-042" File: "b120423a.wiff"Peak Name: "13C8PFOS-IS(IS)" Mass(es): "507.0/80.0 amu"Comment: "JPAL-GW-MW11L-FMS-" Annotation: ""Sample Index: 185Sample Type: UnknownConcentration: 0.960 ng/mLCalculated Conc: N/AAcq. Date: 4/26/2012Acq. Time: 7:48:55 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 845166 countsHeight: 2.20e+005 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e51.9e52.0e52.1e5

Intensity, cps16.00Sample Name: "b120423a186" Sample ID: "GLP10-01-02-30-043" File: "b120423a.wiff"Peak Name: "13C8PFOS-IS(IS)" Mass(es): "507.0/80.0 amu"Comment: "JPAL-GW-MW3R-RB-" Annotation: ""Sample Index: 186Sample Type: UnknownConcentration: 0.960 ng/mLCalculated Conc: N/AAcq. Date: 4/26/2012Acq. Time: 8:10:35 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 1445002 countsHeight: 3.80e+005 cpsStart Time: 15.9 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e5

Intensity, cps16.00Sample Name: "b120423a187" Sample ID: "GLP10-01-02-30-044" File: "b120423a.wiff"Peak Name: "13C8PFOS-IS(IS)" Mass(es): "507.0/80.0 amu"Comment: "JPAL-GW-TRIP01-0-" Annotation: ""Sample Index: 187Sample Type: UnknownConcentration: 0.960 ng/mLCalculated Conc: N/AAcq. Date: 4/26/2012Acq. Time: 8:32:03 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 1335725 countsHeight: 3.44e+005 cpsStart Time: 15.9 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e5

Intensity, cps16.00*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:55 PMPrinting Date: Thursday, May 03, 2012Page 20 of 48Sample Name: "b120423a188" Sample ID: "GLP10-01-02-30-045" File: "b120423a.wiff"Peak Name: "13C8PFOS-IS(IS)" Mass(es): "507.0/80.0 amu"Comment: "JPAL-GW-TRIP01-FMS-" Annotation: ""Sample Index: 188Sample Type: UnknownConcentration: 0.960 ng/mLCalculated Conc: N/AAcq. Date: 4/26/2012Acq. Time: 8:53:31 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: NoInt. Type: ValleyRetention Time: 16.0 minArea: 1619010 countsHeight: 4.10e+005 cpsStart Time: 15.9 minEnd Time: 16.3 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e5

Intensity, cps16.00Sample Name: "b120423a017" Sample ID: "11012-62-8" File: "b120423a.wiff"Peak Name: "18O2PFBS-IS(IS)" Mass(es): "303.0/84.0 amu"Comment: "1.0 ng/mL FC std in Synth. Water" Annotation: ""Sample Index: 17Sample Type: StandardConcentration: 0.945 ng/mLCalculated Conc: N/AAcq. Date: 4/23/2012Acq. Time: 7:37:05 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.6 minArea: 1822166 countsHeight: 4.98e+005 cpsStart Time: 13.5 minEnd Time: 14.3 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e54.4e54.6e54.8e5

Intensity, cps13.60Sample Name: "b120423a024" Sample ID: "12001-160" File: "b120423a.wiff"Peak Name: "18O2PFBS-IS(IS)" Mass(es): "303.0/84.0 amu"Comment: "Method Blank" Annotation: ""Sample Index: 24Sample Type: UnknownConcentration: 0.945 ng/mLCalculated Conc: N/AAcq. Date: 4/23/2012Acq. Time: 10:07:34 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: Base To BaseRetention Time: 13.6 minArea: 1846579 countsHeight: 5.06e+005 cpsStart Time: 13.5 minEnd Time: 14.5 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e54.4e54.6e54.8e55.0e5

Intensity, cps13.61Sample Name: "b120423a026" Sample ID: "LCS-120416-1" File: "b120423a.wiff"Peak Name: "18O2PFBS-IS(IS)" Mass(es): "303.0/84.0 amu"Comment: "0.2ppb LCS" Annotation: ""Sample Index: 26Sample Type: QCConcentration: 0.937 ng/mLCalculated Conc: N/AAcq. Date: 4/23/2012Acq. Time: 10:50:29 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.6 minArea: 1679916 countsHeight: 4.55e+005 cpsStart Time: 13.5 minEnd Time: 13.8 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e54.4e5

Intensity, cps13.61*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:55 PMPrinting Date: Thursday, May 03, 2012Page 21 of 48Sample Name: "b120423a029" Sample ID: "LCS-120416-4" File: "b120423a.wiff"Peak Name: "18O2PFBS-IS(IS)" Mass(es): "303.0/84.0 amu"Comment: "2ppb LCS" Annotation: ""Sample Index: 29Sample Type: QCConcentration: 0.937 ng/mLCalculated Conc: N/AAcq. Date: 4/23/2012Acq. Time: 11:54:49 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.6 minArea: 1605602 countsHeight: 4.40e+005 cpsStart Time: 13.5 minEnd Time: 14.3 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e54.4e5

Intensity, cps13.60Sample Name: "b120423a032" Sample ID: "LCS-120416-7" File: "b120423a.wiff"Peak Name: "18O2PFBS-IS(IS)" Mass(es): "303.0/84.0 amu"Comment: "10ppb LCS" Annotation: ""Sample Index: 32Sample Type: QCConcentration: 0.939 ng/mLCalculated Conc: N/AAcq. Date: 4/24/2012Acq. Time: 12:59:13 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.6 minArea: 1457678 countsHeight: 3.78e+005 cpsStart Time: 13.5 minEnd Time: 14.8 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e5

Intensity, cps13.60Sample Name: "b120423a129" Sample ID: "GLP10-01-02-30-001" File: "b120423a.wiff"Peak Name: "18O2PFBS-IS(IS)" Mass(es): "303.0/84.0 amu"Comment: "JPAL-GW-MW1R-0-" Annotation: ""Sample Index: 129Sample Type: UnknownConcentration: 0.945 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 11:44:06 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.5 minArea: 1724691 countsHeight: 4.43e+005 cpsStart Time: 13.4 minEnd Time: 13.8 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e54.4e5

Intensity, cps13.55Sample Name: "b120423a131" Sample ID: "GLP10-01-02-30-003" File: "b120423a.wiff"Peak Name: "18O2PFBS-IS(IS)" Mass(es): "303.0/84.0 amu"Comment: "JPAL-GW-MW1R-FMS-" Annotation: ""Sample Index: 131Sample Type: UnknownConcentration: 0.945 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 12:27:03 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.5 minArea: 1406986 countsHeight: 3.42e+005 cpsStart Time: 13.4 minEnd Time: 13.9 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e5

Intensity, cps13.55*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:55 PMPrinting Date: Thursday, May 03, 2012Page 22 of 48Sample Name: "b120423a135" Sample ID: "GLP10-01-02-30-004" File: "b120423a.wiff"Peak Name: "18O2PFBS-IS(IS)" Mass(es): "303.0/84.0 amu"Comment: "JPAL-GW-MW3R-0-" Annotation: ""Sample Index: 135Sample Type: UnknownConcentration: 0.945 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 1:53:05 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: Base To BaseRetention Time: 13.6 minArea: 1664943 countsHeight: 4.13e+005 cpsStart Time: 13.4 minEnd Time: 14.5 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e5

Intensity, cps13.56Sample Name: "b120423a138" Sample ID: "GLP10-01-02-30-007" File: "b120423a.wiff"Peak Name: "18O2PFBS-IS(IS)" Mass(es): "303.0/84.0 amu"Comment: "JPAL-GW-MW4R-0-" Annotation: ""Sample Index: 138Sample Type: UnknownConcentration: 0.945 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 2:57:37 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.5 minArea: 1754198 countsHeight: 4.23e+005 cpsStart Time: 13.4 minEnd Time: 14.3 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e5

Intensity, cps13.55Sample Name: "b120423a141" Sample ID: "GLP10-01-02-30-010" File: "b120423a.wiff"Peak Name: "18O2PFBS-IS(IS)" Mass(es): "303.0/84.0 amu"Comment: "JPAL-GW-MW4L-0-" Annotation: ""Sample Index: 141Sample Type: UnknownConcentration: 0.945 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 4:02:04 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.6 minArea: 1719797 countsHeight: 4.49e+005 cpsStart Time: 13.4 minEnd Time: 13.9 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e54.4e5

Intensity, cps13.57Sample Name: "b120423a143" Sample ID: "GLP10-01-02-30-012" File: "b120423a.wiff"Peak Name: "18O2PFBS-IS(IS)" Mass(es): "303.0/84.0 amu"Comment: "JPAL-GW-MW4L-FMS-" Annotation: ""Sample Index: 143Sample Type: UnknownConcentration: 0.945 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 4:44:58 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.6 minArea: 1694331 countsHeight: 4.27e+005 cpsStart Time: 13.4 minEnd Time: 14.2 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e5

Intensity, cps13.57*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:55 PMPrinting Date: Thursday, May 03, 2012Page 23 of 48Sample Name: "b120423a147" Sample ID: "GLP10-01-02-30-013" File: "b120423a.wiff"Peak Name: "18O2PFBS-IS(IS)" Mass(es): "303.0/84.0 amu"Comment: "JPAL-GW-MW5R-0-" Annotation: ""Sample Index: 147Sample Type: UnknownConcentration: 0.945 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 6:11:06 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.6 minArea: 1773708 countsHeight: 4.44e+005 cpsStart Time: 13.4 minEnd Time: 13.8 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e54.4e5

Intensity, cps13.55Sample Name: "b120423a150" Sample ID: "GLP10-01-02-30-016" File: "b120423a.wiff"Peak Name: "18O2PFBS-IS(IS)" Mass(es): "303.0/84.0 amu"Comment: "JPAL-GW-MW5L-0-" Annotation: ""Sample Index: 150Sample Type: UnknownConcentration: 0.945 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 7:15:38 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.5 minArea: 1724822 countsHeight: 4.44e+005 cpsStart Time: 13.4 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e54.4e5

Intensity, cps13.55Sample Name: "b120423a152" Sample ID: "GLP10-01-02-30-018" File: "b120423a.wiff"Peak Name: "18O2PFBS-IS(IS)" Mass(es): "303.0/84.0 amu"Comment: "JPAL-GW-MW5L-FMS-" Annotation: ""Sample Index: 152Sample Type: UnknownConcentration: 0.945 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 7:58:43 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.5 minArea: 1681214 countsHeight: 4.39e+005 cpsStart Time: 13.4 minEnd Time: 13.8 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e5

Intensity, cps13.55Sample Name: "b120423a153" Sample ID: "GLP10-01-02-30-019" File: "b120423a.wiff"Peak Name: "18O2PFBS-IS(IS)" Mass(es): "303.0/84.0 amu"Comment: "JPAL-GW-MW6R-0-" Annotation: ""Sample Index: 153Sample Type: UnknownConcentration: 0.945 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 8:20:17 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.5 minArea: 1760165 countsHeight: 4.60e+005 cpsStart Time: 13.4 minEnd Time: 13.9 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e54.4e5

Intensity, cps13.55*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:55 PMPrinting Date: Thursday, May 03, 2012Page 24 of 48Sample Name: "b120423a159" Sample ID: "GLP10-01-02-30-022" File: "b120423a.wiff"Peak Name: "18O2PFBS-IS(IS)" Mass(es): "303.0/84.0 amu"Comment: "JPAL-GW-MW6L-0-" Annotation: ""Sample Index: 159Sample Type: UnknownConcentration: 0.945 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 10:29:17 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.6 minArea: 1772369 countsHeight: 4.48e+005 cpsStart Time: 13.4 minEnd Time: 14.3 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e54.4e5

Intensity, cps13.56Sample Name: "b120423a162" Sample ID: "GLP10-01-02-30-025" File: "b120423a.wiff"Peak Name: "18O2PFBS-IS(IS)" Mass(es): "303.0/84.0 amu"Comment: "JPAL-GW-MW7R-0-" Annotation: ""Sample Index: 162Sample Type: UnknownConcentration: 0.945 ng/mLCalculated Conc: N/AAcq. Date: 4/25/2012Acq. Time: 11:33:51 PMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.6 minArea: 1766370 countsHeight: 4.38e+005 cpsStart Time: 13.4 minEnd Time: 13.9 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e5

Intensity, cps13.56Sample Name: "b120423a165" Sample ID: "GLP10-01-02-30-028" File: "b120423a.wiff"Peak Name: "18O2PFBS-IS(IS)" Mass(es): "303.0/84.0 amu"Comment: "JPAL-GW-MW8R-0-" Annotation: ""Sample Index: 165Sample Type: UnknownConcentration: 0.945 ng/mLCalculated Conc: N/AAcq. Date: 4/26/2012Acq. Time: 12:38:33 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.6 minArea: 1858581 countsHeight: 4.55e+005 cpsStart Time: 13.4 minEnd Time: 13.8 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e54.4e5

Intensity, cps13.58Sample Name: "b120423a171" Sample ID: "GLP10-01-02-30-031" File: "b120423a.wiff"Peak Name: "18O2PFBS-IS(IS)" Mass(es): "303.0/84.0 amu"Comment: "JPAL-GW-MW9R-0-" Annotation: ""Sample Index: 171Sample Type: UnknownConcentration: 0.945 ng/mLCalculated Conc: N/AAcq. Date: 4/26/2012Acq. Time: 2:47:37 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.6 minArea: 1813291 countsHeight: 4.47e+005 cpsStart Time: 13.5 minEnd Time: 13.9 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e54.4e5

Intensity, cps13.60*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:55 PMPrinting Date: Thursday, May 03, 2012Page 25 of 48Sample Name: "b120423a173" Sample ID: "GLP10-01-02-30-033" File: "b120423a.wiff"Peak Name: "18O2PFBS-IS(IS)" Mass(es): "303.0/84.0 amu"Comment: "JPAL-GW-MW9R-FMS-" Annotation: ""Sample Index: 173Sample Type: UnknownConcentration: 0.945 ng/mLCalculated Conc: N/AAcq. Date: 4/26/2012Acq. Time: 3:30:38 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.6 minArea: 1676658 countsHeight: 4.10e+005 cpsStart Time: 13.4 minEnd Time: 14.0 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e5

Intensity, cps13.58Sample Name: "b120423a174" Sample ID: "GLP10-01-02-30-034" File: "b120423a.wiff"Peak Name: "18O2PFBS-IS(IS)" Mass(es): "303.0/84.0 amu"Comment: "JPAL-GW-MW9L-0-" Annotation: ""Sample Index: 174Sample Type: UnknownConcentration: 0.945 ng/mLCalculated Conc: N/AAcq. Date: 4/26/2012Acq. Time: 3:52:08 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.6 minArea: 1780242 countsHeight: 4.52e+005 cpsStart Time: 13.4 minEnd Time: 14.2 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e54.4e5

Intensity, cps13.57Sample Name: "b120423a177" Sample ID: "GLP10-01-02-30-037" File: "b120423a.wiff"Peak Name: "18O2PFBS-IS(IS)" Mass(es): "303.0/84.0 amu"Comment: "JPAL-GW-MW10R-0-" Annotation: ""Sample Index: 177Sample Type: UnknownConcentration: 0.945 ng/mLCalculated Conc: N/AAcq. Date: 4/26/2012Acq. Time: 4:56:43 AMModified: YesProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.6 minArea: 1777112 countsHeight: 4.62e+005 cpsStart Time: 13.4 minEnd Time: 14.1 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e54.4e54.6e5

Intensity, cps13.56Sample Name: "b120423a183" Sample ID: "GLP10-01-02-30-040" File: "b120423a.wiff"Peak Name: "18O2PFBS-IS(IS)" Mass(es): "303.0/84.0 amu"Comment: "JPAL-GW-MW11L-0-" Annotation: ""Sample Index: 183Sample Type: UnknownConcentration: 0.945 ng/mLCalculated Conc: N/AAcq. Date: 4/26/2012Acq. Time: 7:05:50 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.6 minArea: 1654303 countsHeight: 4.05e+005 cpsStart Time: 13.4 minEnd Time: 13.8 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e5

Intensity, cps13.56*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:55 PMPrinting Date: Thursday, May 03, 2012Page 26 of 48Sample Name: "b120423a185" Sample ID: "GLP10-01-02-30-042" File: "b120423a.wiff"Peak Name: "18O2PFBS-IS(IS)" Mass(es): "303.0/84.0 amu"Comment: "JPAL-GW-MW11L-FMS-" Annotation: ""Sample Index: 185Sample Type: UnknownConcentration: 0.945 ng/mLCalculated Conc: N/AAcq. Date: 4/26/2012Acq. Time: 7:48:55 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.6 minArea: 1693072 countsHeight: 4.08e+005 cpsStart Time: 13.4 minEnd Time: 13.9 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e5

Intensity, cps13.55Sample Name: "b120423a186" Sample ID: "GLP10-01-02-30-043" File: "b120423a.wiff"Peak Name: "18O2PFBS-IS(IS)" Mass(es): "303.0/84.0 amu"Comment: "JPAL-GW-MW3R-RB-" Annotation: ""Sample Index: 186Sample Type: UnknownConcentration: 0.945 ng/mLCalculated Conc: N/AAcq. Date: 4/26/2012Acq. Time: 8:10:35 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.5 minArea: 1669827 countsHeight: 3.79e+005 cpsStart Time: 13.4 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e5

Intensity, cps13.54Sample Name: "b120423a187" Sample ID: "GLP10-01-02-30-044" File: "b120423a.wiff"Peak Name: "18O2PFBS-IS(IS)" Mass(es): "303.0/84.0 amu"Comment: "JPAL-GW-TRIP01-0-" Annotation: ""Sample Index: 187Sample Type: UnknownConcentration: 0.945 ng/mLCalculated Conc: N/AAcq. Date: 4/26/2012Acq. Time: 8:32:03 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.6 minArea: 1679510 countsHeight: 4.20e+005 cpsStart Time: 13.4 minEnd Time: 13.8 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e5

Intensity, cps13.56Sample Name: "b120423a188" Sample ID: "GLP10-01-02-30-045" File: "b120423a.wiff"Peak Name: "18O2PFBS-IS(IS)" Mass(es): "303.0/84.0 amu"Comment: "JPAL-GW-TRIP01-FMS-" Annotation: ""Sample Index: 188Sample Type: UnknownConcentration: 0.945 ng/mLCalculated Conc: N/AAcq. Date: 4/26/2012Acq. Time: 8:53:31 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 2Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: NoInt. Type: ValleyRetention Time: 13.6 minArea: 1763978 countsHeight: 4.39e+005 cpsStart Time: 13.4 minEnd Time: 14.0 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e5

Intensity, cps13.56*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:55 PMPrinting Date: Thursday, May 03, 2012Page 27 of 48Sample Name: "b120423a017" Sample ID: "11012-62-8" File: "b120423a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "1.0 ng/mL FC std in Synth. Water" Annotation: ""Sample Index: 17Sample Type: StandardConcentration: 1.00 ng/mLCalculated Conc: 0.993 ng/mLAcq. Date: 4/23/2012Acq. Time: 7:37:05 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: YesInt. Type: ValleyRetention Time: 13.6 minArea: 2340818 countsHeight: 6.28e+005 cpsStart Time: 13.5 minEnd Time: 13.8 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e5

Intensity, cps13.60Sample Name: "b120423a024" Sample ID: "12001-160" File: "b120423a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "Method Blank" Annotation: ""Sample Index: 24Sample Type: UnknownConcentration: N/ACalculated Conc: < 0Acq. Date: 4/23/2012Acq. Time: 10:07:34 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: YesInt. Type: ValleyRetention Time: 13.6 minArea: 11199 countsHeight: 2.03e+003 cpsStart Time: 13.5 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0100200300400500600700800900100011001200130014001500160017001800190020002100

Intensity, cpsSample Name: "b120423a026" Sample ID: "LCS-120416-1" File: "b120423a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "0.2ppb LCS" Annotation: ""Sample Index: 26Sample Type: QCConcentration: 0.198 ng/mLCalculated Conc: 0.203 ng/mLAcq. Date: 4/23/2012Acq. Time: 10:50:29 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: YesInt. Type: ValleyRetention Time: 13.6 minArea: 463904 countsHeight: 1.25e+005 cpsStart Time: 13.5 minEnd Time: 13.8 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e51.05e51.10e51.15e51.20e51.25e5

Intensity, cps13.6114.2311.7615.4815.8113.7915.4214.7612.56Sample Name: "b120423a029" Sample ID: "LCS-120416-4" File: "b120423a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "2ppb LCS" Annotation: ""Sample Index: 29Sample Type: QCConcentration: 1.98 ng/mLCalculated Conc: 2.13 ng/mLAcq. Date: 4/23/2012Acq. Time: 11:54:49 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: YesInt. Type: ValleyRetention Time: 13.6 minArea: 4384603 countsHeight: 1.20e+006 cpsStart Time: 13.5 minEnd Time: 13.8 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e61.10e61.15e61.20e6

Intensity, cps13.61*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:55 PMPrinting Date: Thursday, May 03, 2012Page 28 of 48Sample Name: "b120423a032" Sample ID: "LCS-120416-7" File: "b120423a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "10ppb LCS" Annotation: ""Sample Index: 32Sample Type: QCConcentration: 9.94 ng/mLCalculated Conc: 9.83 ng/mLAcq. Date: 4/24/2012Acq. Time: 12:59:13 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: YesInt. Type: ValleyRetention Time: 13.6 minArea: 16228398 countsHeight: 3.91e+006 cpsStart Time: 13.5 minEnd Time: 13.8 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e54.0e56.0e58.0e51.0e61.2e61.4e61.6e61.8e62.0e62.2e62.4e62.6e62.8e63.0e63.2e63.4e63.6e63.8e6

Intensity, cps13.60Sample Name: "b120423a129" Sample ID: "GLP10-01-02-30-001" File: "b120423a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "JPAL-GW-MW1R-0-" Annotation: ""Sample Index: 129Sample Type: UnknownConcentration: N/ACalculated Conc: 0.114 ng/mLAcq. Date: 4/25/2012Acq. Time: 11:44:06 AMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: YesInt. Type: ManualRetention Time: 13.5 minArea: 272727 countsHeight: 6.62e+004 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e4

Intensity, cpsSample Name: "b120423a131" Sample ID: "GLP10-01-02-30-003" File: "b120423a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "JPAL-GW-MW1R-FMS-" Annotation: ""Sample Index: 131Sample Type: UnknownConcentration: N/ACalculated Conc: 2.48 ng/mLAcq. Date: 4/25/2012Acq. Time: 12:27:03 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: YesInt. Type: ManualRetention Time: 13.6 minArea: 4393315 countsHeight: 1.08e+006 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e6

Intensity, cps13.5613.41Sample Name: "b120423a135" Sample ID: "GLP10-01-02-30-004" File: "b120423a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "JPAL-GW-MW3R-0-" Annotation: ""Sample Index: 135Sample Type: UnknownConcentration: N/ACalculated Conc: 1.78 ng/mLAcq. Date: 4/25/2012Acq. Time: 1:53:05 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: YesInt. Type: ManualRetention Time: 13.6 minArea: 3770318 countsHeight: 9.03e+005 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e58.5e59.0e5

Intensity, cps13.56*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:55 PMPrinting Date: Thursday, May 03, 2012Page 29 of 48Sample Name: "b120423a138" Sample ID: "GLP10-01-02-30-007" File: "b120423a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "JPAL-GW-MW4R-0-" Annotation: ""Sample Index: 138Sample Type: UnknownConcentration: N/ACalculated Conc: 0.188 ng/mLAcq. Date: 4/25/2012Acq. Time: 2:57:37 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: YesInt. Type: ManualRetention Time: 13.6 minArea: 447307 countsHeight: 1.08e+005 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e51.05e5

Intensity, cps13.55Sample Name: "b120423a141" Sample ID: "GLP10-01-02-30-010" File: "b120423a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "JPAL-GW-MW4L-0-" Annotation: ""Sample Index: 141Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0759 ng/mLAcq. Date: 4/25/2012Acq. Time: 4:02:04 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: YesInt. Type: ManualRetention Time: 13.6 minArea: 187664 countsHeight: 4.83e+004 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e43.4e43.6e43.8e44.0e44.2e44.4e44.6e44.8e4

Intensity, cps13.5711.78Sample Name: "b120423a143" Sample ID: "GLP10-01-02-30-012" File: "b120423a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "JPAL-GW-MW4L-FMS-" Annotation: ""Sample Index: 143Sample Type: UnknownConcentration: N/ACalculated Conc: 2.05 ng/mLAcq. Date: 4/25/2012Acq. Time: 4:44:58 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: YesInt. Type: ManualRetention Time: 13.6 minArea: 4419368 countsHeight: 1.11e+006 cpsStart Time: 13.3 minEnd Time: 13.8 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e61.10e6

Intensity, cps13.5811.7813.43Sample Name: "b120423a147" Sample ID: "GLP10-01-02-30-013" File: "b120423a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "JPAL-GW-MW5R-0-" Annotation: ""Sample Index: 147Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0400 ng/mLAcq. Date: 4/25/2012Acq. Time: 6:11:06 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: YesInt. Type: ManualRetention Time: 13.6 minArea: 110762 countsHeight: 2.67e+004 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.01000.02000.03000.04000.05000.06000.07000.08000.09000.01.0e41.1e41.2e41.3e41.4e41.5e41.6e41.7e41.8e41.9e42.0e42.1e42.2e42.3e42.4e42.5e42.6e4

Intensity, cps*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:55 PMPrinting Date: Thursday, May 03, 2012Page 30 of 48Sample Name: "b120423a150" Sample ID: "GLP10-01-02-30-016" File: "b120423a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "JPAL-GW-MW5L-0-" Annotation: ""Sample Index: 150Sample Type: UnknownConcentration: N/ACalculated Conc: 0.259 ng/mLAcq. Date: 4/25/2012Acq. Time: 7:15:38 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: YesInt. Type: ManualRetention Time: 13.5 minArea: 596616 countsHeight: 1.44e+005 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e5

Intensity, cps13.5511.7713.3914.16Sample Name: "b120423a152" Sample ID: "GLP10-01-02-30-018" File: "b120423a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "JPAL-GW-MW5L-FMS-" Annotation: ""Sample Index: 152Sample Type: UnknownConcentration: N/ACalculated Conc: 2.23 ng/mLAcq. Date: 4/25/2012Acq. Time: 7:58:43 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: YesInt. Type: ManualRetention Time: 13.5 minArea: 4752996 countsHeight: 1.19e+006 cpsStart Time: 13.3 minEnd Time: 13.8 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e61.10e61.15e6

Intensity, cps13.55Sample Name: "b120423a153" Sample ID: "GLP10-01-02-30-019" File: "b120423a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "JPAL-GW-MW6R-0-" Annotation: ""Sample Index: 153Sample Type: UnknownConcentration: N/ACalculated Conc: 0.336 ng/mLAcq. Date: 4/25/2012Acq. Time: 8:20:17 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: YesInt. Type: ManualRetention Time: 13.5 minArea: 783297 countsHeight: 1.97e+005 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e51.9e5

Intensity, cps13.55Sample Name: "b120423a159" Sample ID: "GLP10-01-02-30-022" File: "b120423a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "JPAL-GW-MW6L-0-" Annotation: ""Sample Index: 159Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0548 ng/mLAcq. Date: 4/25/2012Acq. Time: 10:29:17 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: YesInt. Type: ManualRetention Time: 13.6 minArea: 144903 countsHeight: 3.37e+004 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e4

Intensity, cps13.56*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:55 PMPrinting Date: Thursday, May 03, 2012Page 31 of 48Sample Name: "b120423a162" Sample ID: "GLP10-01-02-30-025" File: "b120423a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "JPAL-GW-MW7R-0-" Annotation: ""Sample Index: 162Sample Type: UnknownConcentration: N/ACalculated Conc: 0.534 ng/mLAcq. Date: 4/25/2012Acq. Time: 11:33:51 PMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: YesInt. Type: ManualRetention Time: 13.6 minArea: 1236990 countsHeight: 2.96e+005 cpsStart Time: 13.3 minEnd Time: 13.8 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e5

Intensity, cps13.5613.41Sample Name: "b120423a165" Sample ID: "GLP10-01-02-30-028" File: "b120423a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "JPAL-GW-MW8R-0-" Annotation: ""Sample Index: 165Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0492 ng/mLAcq. Date: 4/26/2012Acq. Time: 12:38:33 AMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: YesInt. Type: ManualRetention Time: 13.6 minArea: 138471 countsHeight: 3.31e+004 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e4

Intensity, cps13.59Sample Name: "b120423a171" Sample ID: "GLP10-01-02-30-031" File: "b120423a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "JPAL-GW-MW9R-0-" Annotation: ""Sample Index: 171Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0188 ng/mLAcq. Date: 4/26/2012Acq. Time: 2:47:37 AMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: YesInt. Type: ManualRetention Time: 13.6 minArea: 63280 countsHeight: 1.36e+004 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.0500.01000.01500.02000.02500.03000.03500.04000.04500.05000.05500.06000.06500.07000.07500.08000.08500.09000.09500.01.0e41.1e41.1e41.2e41.2e41.3e41.3e41.4e4

Intensity, cps13.6011.81Sample Name: "b120423a173" Sample ID: "GLP10-01-02-30-033" File: "b120423a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "JPAL-GW-MW9R-FMS-" Annotation: ""Sample Index: 173Sample Type: UnknownConcentration: N/ACalculated Conc: 2.01 ng/mLAcq. Date: 4/26/2012Acq. Time: 3:30:38 AMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: YesInt. Type: ManualRetention Time: 13.6 minArea: 4280247 countsHeight: 1.05e+006 cpsStart Time: 13.3 minEnd Time: 13.8 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e6

Intensity, cps13.5813.43*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:55 PMPrinting Date: Thursday, May 03, 2012Page 32 of 48Sample Name: "b120423a174" Sample ID: "GLP10-01-02-30-034" File: "b120423a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "JPAL-GW-MW9L-0-" Annotation: ""Sample Index: 174Sample Type: UnknownConcentration: N/ACalculated Conc: 0.00124 ng/mLAcq. Date: 4/26/2012Acq. Time: 3:52:08 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: YesInt. Type: ValleyRetention Time: 13.6 minArea: 21523 countsHeight: 5.03e+003 cpsStart Time: 13.4 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min02004006008001000120014001600180020002200240026002800300032003400360038004000420044004600480050005200

Intensity, cpsSample Name: "b120423a177" Sample ID: "GLP10-01-02-30-037" File: "b120423a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "JPAL-GW-MW10R-0-" Annotation: ""Sample Index: 177Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0602 ng/mLAcq. Date: 4/26/2012Acq. Time: 4:56:43 AMModified: YesRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: YesInt. Type: ManualRetention Time: 13.6 minArea: 157709 countsHeight: 3.89e+004 cpsStart Time: 13.3 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e43.4e43.6e43.8e4

Intensity, cps13.5714.1811.7913.4015.3814.6315.1712.5811.3611.9412.90Sample Name: "b120423a183" Sample ID: "GLP10-01-02-30-040" File: "b120423a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "JPAL-GW-MW11L-0-" Annotation: ""Sample Index: 183Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0209 ng/mLAcq. Date: 4/26/2012Acq. Time: 7:05:50 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: YesInt. Type: ValleyRetention Time: 13.6 minArea: 62300 countsHeight: 1.48e+004 cpsStart Time: 13.5 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.01000.02000.03000.04000.05000.06000.07000.08000.09000.01.0e41.1e41.2e41.3e41.4e41.5e4

Intensity, cps13.5611.79Sample Name: "b120423a185" Sample ID: "GLP10-01-02-30-042" File: "b120423a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "JPAL-GW-MW11L-FMS-" Annotation: ""Sample Index: 185Sample Type: UnknownConcentration: N/ACalculated Conc: 2.03 ng/mLAcq. Date: 4/26/2012Acq. Time: 7:48:55 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: YesInt. Type: ValleyRetention Time: 13.6 minArea: 4365678 countsHeight: 1.07e+006 cpsStart Time: 13.4 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e6

Intensity, cps13.56*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:56 PMPrinting Date: Thursday, May 03, 2012Page 33 of 48Sample Name: "b120423a186" Sample ID: "GLP10-01-02-30-043" File: "b120423a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "JPAL-GW-MW3R-RB-" Annotation: ""Sample Index: 186Sample Type: UnknownConcentration: N/ACalculated Conc: < 0Acq. Date: 4/26/2012Acq. Time: 8:10:35 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: YesInt. Type: ValleyRetention Time: 13.6 minArea: 10341 countsHeight: 2.35e+003 cpsStart Time: 13.5 minEnd Time: 13.6 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min01002003004005006007008009001000110012001300140015001600170018001900200021002200230024002500

Intensity, cpsSample Name: "b120423a187" Sample ID: "GLP10-01-02-30-044" File: "b120423a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "JPAL-GW-TRIP01-0-" Annotation: ""Sample Index: 187Sample Type: UnknownConcentration: N/ACalculated Conc: < 0Acq. Date: 4/26/2012Acq. Time: 8:32:03 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: YesInt. Type: ValleyRetention Time: 13.6 minArea: 10730 countsHeight: 2.64e+003 cpsStart Time: 13.5 minEnd Time: 13.7 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min02004006008001000120014001600180020002200240026002800

Intensity, cps13.5614.1811.7411.6914.8415.3414.3813.9115.4912.1113.4412.5213.1111.64Sample Name: "b120423a188" Sample ID: "GLP10-01-02-30-045" File: "b120423a.wiff"Peak Name: "PFBS" Mass(es): "299.0/99.0 amu,299.0/80.0 amu"Comment: "JPAL-GW-TRIP01-FMS-" Annotation: ""Sample Index: 188Sample Type: UnknownConcentration: N/ACalculated Conc: 2.00 ng/mLAcq. Date: 4/26/2012Acq. Time: 8:53:31 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 13.6 minUse Relative RT: YesInt. Type: ValleyRetention Time: 13.6 minArea: 4473984 countsHeight: 1.08e+006 cpsStart Time: 13.4 minEnd Time: 13.8 min11.5 12.0 12.5 13.0 13.5 14.0 14.5 15.0 15.5 16.0Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e6

Intensity, cps13.5613.41Sample Name: "b120423a017" Sample ID: "11012-62-8" File: "b120423a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "1.0 ng/mL FC std in Synth. Water" Annotation: ""Sample Index: 17Sample Type: StandardConcentration: 0.998 ng/mLCalculated Conc: 1.06 ng/mLAcq. Date: 4/23/2012Acq. Time: 7:37:05 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: NoMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: YesInt. Type: ValleyRetention Time: 15.1 minArea: 2201803 countsHeight: 6.49e+005 cpsStart Time: 15.0 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e5

Intensity, cps15.12*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:56 PMPrinting Date: Thursday, May 03, 2012Page 34 of 48Sample Name: "b120423a024" Sample ID: "12001-160" File: "b120423a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "Method Blank" Annotation: ""Sample Index: 24Sample Type: UnknownConcentration: N/ACalculated Conc: < 0Acq. Date: 4/23/2012Acq. Time: 10:07:34 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: NoMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: YesInt. Type: ValleyRetention Time: 15.1 minArea: 2711 countsHeight: 5.04e+002 cpsStart Time: 15.1 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min01002003004005006007008009001000110012001300140015001600170018001900200021002200230024002500

Intensity, cpsSample Name: "b120423a026" Sample ID: "LCS-120416-1" File: "b120423a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "0.2ppb LCS" Annotation: ""Sample Index: 26Sample Type: QCConcentration: 0.198 ng/mLCalculated Conc: 0.201 ng/mLAcq. Date: 4/23/2012Acq. Time: 10:50:29 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: NoMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: YesInt. Type: ValleyRetention Time: 15.1 minArea: 436474 countsHeight: 1.26e+005 cpsStart Time: 15.0 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e51.05e51.10e51.15e51.20e51.25e5

Intensity, cps15.1217.2815.5616.84Sample Name: "b120423a029" Sample ID: "LCS-120416-4" File: "b120423a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "2ppb LCS" Annotation: ""Sample Index: 29Sample Type: QCConcentration: 1.98 ng/mLCalculated Conc: 2.10 ng/mLAcq. Date: 4/23/2012Acq. Time: 11:54:49 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: NoMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: YesInt. Type: ValleyRetention Time: 15.1 minArea: 4450237 countsHeight: 1.29e+006 cpsStart Time: 15.0 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e61.10e61.15e61.20e61.25e6

Intensity, cps15.1217.28Sample Name: "b120423a032" Sample ID: "LCS-120416-7" File: "b120423a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "10ppb LCS" Annotation: ""Sample Index: 32Sample Type: QCConcentration: 9.92 ng/mLCalculated Conc: 10.0 ng/mLAcq. Date: 4/24/2012Acq. Time: 12:59:13 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: NoMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: YesInt. Type: ValleyRetention Time: 15.1 minArea: 15372831 countsHeight: 3.97e+006 cpsStart Time: 14.9 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.02.0e54.0e56.0e58.0e51.0e61.2e61.4e61.6e61.8e62.0e62.2e62.4e62.6e62.8e63.0e63.2e63.4e63.6e63.8e6

Intensity, cps15.11*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:56 PMPrinting Date: Thursday, May 03, 2012Page 35 of 48Sample Name: "b120423a129" Sample ID: "GLP10-01-02-30-001" File: "b120423a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "JPAL-GW-MW1R-0-" Annotation: ""Sample Index: 129Sample Type: UnknownConcentration: N/ACalculated Conc: 0.865 ng/mLAcq. Date: 4/25/2012Acq. Time: 11:44:06 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 15.1 minArea: 1840960 countsHeight: 4.05e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e5

Intensity, cps15.06Sample Name: "b120423a131" Sample ID: "GLP10-01-02-30-003" File: "b120423a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "JPAL-GW-MW1R-FMS-" Annotation: ""Sample Index: 131Sample Type: UnknownConcentration: N/ACalculated Conc: 3.31 ng/mLAcq. Date: 4/25/2012Acq. Time: 12:27:03 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 15.1 minArea: 5675949 countsHeight: 1.39e+006 cpsStart Time: 14.8 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e58.5e59.0e59.5e51.0e61.1e61.1e61.2e61.2e61.3e61.3e61.4e6

Intensity, cps15.0614.92Sample Name: "b120423a135" Sample ID: "GLP10-01-02-30-004" File: "b120423a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "JPAL-GW-MW3R-0-" Annotation: ""Sample Index: 135Sample Type: UnknownConcentration: N/ACalculated Conc: 9.04 ng/mLAcq. Date: 4/25/2012Acq. Time: 1:53:05 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 15.1 minArea: 14949811 countsHeight: 2.92e+006 cpsStart Time: 14.7 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.02.0e54.0e56.0e58.0e51.0e61.2e61.4e61.6e61.8e62.0e62.2e62.4e62.6e62.8e6

Intensity, cps15.0614.92Sample Name: "b120423a138" Sample ID: "GLP10-01-02-30-007" File: "b120423a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "JPAL-GW-MW4R-0-" Annotation: ""Sample Index: 138Sample Type: UnknownConcentration: N/ACalculated Conc: 4.95 ng/mLAcq. Date: 4/25/2012Acq. Time: 2:57:37 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 15.1 minArea: 9472177 countsHeight: 1.98e+006 cpsStart Time: 14.7 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.01.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e61.6e61.7e61.8e61.9e6

Intensity, cps15.0614.92*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:56 PMPrinting Date: Thursday, May 03, 2012Page 36 of 48Sample Name: "b120423a141" Sample ID: "GLP10-01-02-30-010" File: "b120423a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "JPAL-GW-MW4L-0-" Annotation: ""Sample Index: 141Sample Type: UnknownConcentration: N/ACalculated Conc: 0.200 ng/mLAcq. Date: 4/25/2012Acq. Time: 4:02:04 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 15.1 minArea: 452562 countsHeight: 9.80e+004 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e48.0e48.5e49.0e49.5e4

Intensity, cpsSample Name: "b120423a143" Sample ID: "GLP10-01-02-30-012" File: "b120423a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "JPAL-GW-MW4L-FMS-" Annotation: ""Sample Index: 143Sample Type: UnknownConcentration: N/ACalculated Conc: 2.15 ng/mLAcq. Date: 4/25/2012Acq. Time: 4:44:58 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 15.1 minArea: 4470646 countsHeight: 1.17e+006 cpsStart Time: 14.8 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e61.10e61.15e6

Intensity, cps15.0914.93Sample Name: "b120423a147" Sample ID: "GLP10-01-02-30-013" File: "b120423a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "JPAL-GW-MW5R-0-" Annotation: ""Sample Index: 147Sample Type: UnknownConcentration: N/ACalculated Conc: 0.133 ng/mLAcq. Date: 4/25/2012Acq. Time: 6:11:06 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 15.1 minArea: 285723 countsHeight: 6.44e+004 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e4

Intensity, cps15.07Sample Name: "b120423a150" Sample ID: "GLP10-01-02-30-016" File: "b120423a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "JPAL-GW-MW5L-0-" Annotation: ""Sample Index: 150Sample Type: UnknownConcentration: N/ACalculated Conc: 0.399 ng/mLAcq. Date: 4/25/2012Acq. Time: 7:15:38 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 15.1 minArea: 896158 countsHeight: 1.95e+005 cpsStart Time: 14.7 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e51.9e5

Intensity, cps15.0614.9217.45*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:56 PMPrinting Date: Thursday, May 03, 2012Page 37 of 48Sample Name: "b120423a152" Sample ID: "GLP10-01-02-30-018" File: "b120423a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "JPAL-GW-MW5L-FMS-" Annotation: ""Sample Index: 152Sample Type: UnknownConcentration: N/ACalculated Conc: 2.33 ng/mLAcq. Date: 4/25/2012Acq. Time: 7:58:43 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 15.1 minArea: 4846977 countsHeight: 1.27e+006 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e61.10e61.15e61.20e61.25e6

Intensity, cps15.0614.92Sample Name: "b120423a153" Sample ID: "GLP10-01-02-30-019" File: "b120423a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "JPAL-GW-MW6R-0-" Annotation: ""Sample Index: 153Sample Type: UnknownConcentration: N/ACalculated Conc: 1.09 ng/mLAcq. Date: 4/25/2012Acq. Time: 8:20:17 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 15.1 minArea: 2379737 countsHeight: 5.42e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e54.4e54.6e54.8e55.0e55.2e5

Intensity, cps15.0614.92Sample Name: "b120423a159" Sample ID: "GLP10-01-02-30-022" File: "b120423a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "JPAL-GW-MW6L-0-" Annotation: ""Sample Index: 159Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0949 ng/mLAcq. Date: 4/25/2012Acq. Time: 10:29:17 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 15.1 minArea: 218532 countsHeight: 4.50e+004 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e43.4e43.6e43.8e44.0e44.2e44.4e4

Intensity, cps15.06Sample Name: "b120423a162" Sample ID: "GLP10-01-02-30-025" File: "b120423a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "JPAL-GW-MW7R-0-" Annotation: ""Sample Index: 162Sample Type: UnknownConcentration: N/ACalculated Conc: 1.88 ng/mLAcq. Date: 4/25/2012Acq. Time: 11:33:51 PMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 15.1 minArea: 4097791 countsHeight: 9.10e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e58.0e58.5e5

Intensity, cps15.0814.9417.47*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:56 PMPrinting Date: Thursday, May 03, 2012Page 38 of 48Sample Name: "b120423a165" Sample ID: "GLP10-01-02-30-028" File: "b120423a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "JPAL-GW-MW8R-0-" Annotation: ""Sample Index: 165Sample Type: UnknownConcentration: N/ACalculated Conc: 0.124 ng/mLAcq. Date: 4/26/2012Acq. Time: 12:38:33 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 15.1 minArea: 309504 countsHeight: 6.86e+004 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e4

Intensity, cps15.09Sample Name: "b120423a171" Sample ID: "GLP10-01-02-30-031" File: "b120423a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "JPAL-GW-MW9R-0-" Annotation: ""Sample Index: 171Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0653 ng/mLAcq. Date: 4/26/2012Acq. Time: 2:47:37 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 15.1 minArea: 174004 countsHeight: 3.74e+004 cpsStart Time: 14.8 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e43.4e43.6e4

Intensity, cps15.1114.9717.49Sample Name: "b120423a173" Sample ID: "GLP10-01-02-30-033" File: "b120423a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "JPAL-GW-MW9R-FMS-" Annotation: ""Sample Index: 173Sample Type: UnknownConcentration: N/ACalculated Conc: 2.14 ng/mLAcq. Date: 4/26/2012Acq. Time: 3:30:38 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 15.1 minArea: 4506299 countsHeight: 1.17e+006 cpsStart Time: 14.9 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e61.10e61.15e6

Intensity, cps15.0814.9517.46Sample Name: "b120423a174" Sample ID: "GLP10-01-02-30-034" File: "b120423a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "JPAL-GW-MW9L-0-" Annotation: ""Sample Index: 174Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0129 ng/mLAcq. Date: 4/26/2012Acq. Time: 3:52:08 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 15.1 minArea: 45270 countsHeight: 9.79e+003 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0500100015002000250030003500400045005000550060006500700075008000850090009500

Intensity, cps*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:56 PMPrinting Date: Thursday, May 03, 2012Page 39 of 48Sample Name: "b120423a177" Sample ID: "GLP10-01-02-30-037" File: "b120423a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "JPAL-GW-MW10R-0-" Annotation: ""Sample Index: 177Sample Type: UnknownConcentration: N/ACalculated Conc: 0.269 ng/mLAcq. Date: 4/26/2012Acq. Time: 4:56:43 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 15.1 minArea: 632621 countsHeight: 1.47e+005 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e5

Intensity, cps15.0714.9417.4615.4817.12Sample Name: "b120423a183" Sample ID: "GLP10-01-02-30-040" File: "b120423a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "JPAL-GW-MW11L-0-" Annotation: ""Sample Index: 183Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0919 ng/mLAcq. Date: 4/26/2012Acq. Time: 7:05:50 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 15.1 minArea: 210029 countsHeight: 4.79e+004 cpsStart Time: 14.8 minEnd Time: 15.2 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e43.4e43.6e43.8e44.0e44.2e44.4e44.6e4

Intensity, cps15.07Sample Name: "b120423a185" Sample ID: "GLP10-01-02-30-042" File: "b120423a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "JPAL-GW-MW11L-FMS-" Annotation: ""Sample Index: 185Sample Type: UnknownConcentration: N/ACalculated Conc: 2.16 ng/mLAcq. Date: 4/26/2012Acq. Time: 7:48:55 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 15.1 minArea: 4405466 countsHeight: 1.13e+006 cpsStart Time: 14.8 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e61.10e6

Intensity, cps15.0614.9317.45Sample Name: "b120423a186" Sample ID: "GLP10-01-02-30-043" File: "b120423a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "JPAL-GW-MW3R-RB-" Annotation: ""Sample Index: 186Sample Type: UnknownConcentration: N/ACalculated Conc: < 0Acq. Date: 4/26/2012Acq. Time: 8:10:35 AMModified: YesRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 15.1 minArea: 2999 countsHeight: 4.74e+002 cpsStart Time: 15.0 minEnd Time: 15.1 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min020040060080010001200140016001800200022002400260028003000320034003600380040004200440046004800

Intensity, cps*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:56 PMPrinting Date: Thursday, May 03, 2012Page 40 of 48Sample Name: "b120423a187" Sample ID: "GLP10-01-02-30-044" File: "b120423a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "JPAL-GW-TRIP01-0-" Annotation: ""Sample Index: 187Sample Type: UnknownConcentration: N/ACalculated Conc: < 0Acq. Date: 4/26/2012Acq. Time: 8:32:03 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: NoMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: YesInt. Type: ValleyRetention Time: 15.1 minArea: 1190 countsHeight: 4.04e+002 cpsStart Time: 15.0 minEnd Time: 15.1 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0200400600800100012001400160018002000220024002600280030003200340036003800400042004400

Intensity, cps17.4715.4817.1615.3816.3117.0515.2414.5215.8614.9814.4613.7815.9414.20Sample Name: "b120423a188" Sample ID: "GLP10-01-02-30-045" File: "b120423a.wiff"Peak Name: "PFHS" Mass(es): "399.0/99.0 amu,399.0/80.0 amu"Comment: "JPAL-GW-TRIP01-FMS-" Annotation: ""Sample Index: 188Sample Type: UnknownConcentration: N/ACalculated Conc: 2.00 ng/mLAcq. Date: 4/26/2012Acq. Time: 8:53:31 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 1Report Largest Peak: NoMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 15.1 minUse Relative RT: YesInt. Type: ValleyRetention Time: 15.1 minArea: 4420469 countsHeight: 1.13e+006 cpsStart Time: 14.9 minEnd Time: 15.3 min13.0 13.5 14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5Time, min0.005.00e41.00e51.50e52.00e52.50e53.00e53.50e54.00e54.50e55.00e55.50e56.00e56.50e57.00e57.50e58.00e58.50e59.00e59.50e51.00e61.05e61.10e6

Intensity, cps15.0717.47Sample Name: "b120423a017" Sample ID: "11012-62-8" File: "b120423a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "1.0 ng/mL FC std in Synth. Water" Annotation: ""Sample Index: 17Sample Type: StandardConcentration: 0.928 ng/mLCalculated Conc: 0.897 ng/mLAcq. Date: 4/23/2012Acq. Time: 7:37:05 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.1 minArea: 1126109 countsHeight: 2.44e+005 cpsStart Time: 15.7 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e51.9e52.0e52.1e52.2e52.3e52.4e5

Intensity, cps16.0715.93Sample Name: "b120423a024" Sample ID: "12001-160" File: "b120423a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "Method Blank" Annotation: ""Sample Index: 24Sample Type: UnknownConcentration: N/ACalculated Conc: < 0Acq. Date: 4/23/2012Acq. Time: 10:07:34 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 15.9 minArea: 8654 countsHeight: 1.14e+003 cpsStart Time: 15.9 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.0500.01000.01500.02000.02500.03000.03500.04000.04500.05000.05500.06000.06500.07000.07500.08000.08500.09000.09500.01.0e41.1e41.1e41.2e41.2e41.3e41.3e41.4e4

Intensity, cps16.44*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:56 PMPrinting Date: Thursday, May 03, 2012Page 41 of 48Sample Name: "b120423a026" Sample ID: "LCS-120416-1" File: "b120423a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "0.2ppb LCS" Annotation: ""Sample Index: 26Sample Type: QCConcentration: 0.184 ng/mLCalculated Conc: 0.179 ng/mLAcq. Date: 4/23/2012Acq. Time: 10:50:29 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 8Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ValleyRetention Time: 16.1 minArea: 311104 countsHeight: 6.65e+004 cpsStart Time: 15.7 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e4

Intensity, cps16.0715.9316.4415.7517.76Sample Name: "b120423a029" Sample ID: "LCS-120416-4" File: "b120423a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "2ppb LCS" Annotation: ""Sample Index: 29Sample Type: QCConcentration: 1.84 ng/mLCalculated Conc: 1.84 ng/mLAcq. Date: 4/23/2012Acq. Time: 11:54:49 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.1 minArea: 3517781 countsHeight: 7.69e+005 cpsStart Time: 15.7 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e5

Intensity, cps16.0715.9216.4315.75Sample Name: "b120423a032" Sample ID: "LCS-120416-7" File: "b120423a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "10ppb LCS" Annotation: ""Sample Index: 32Sample Type: QCConcentration: 9.22 ng/mLCalculated Conc: 9.16 ng/mLAcq. Date: 4/24/2012Acq. Time: 12:59:13 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.1 minArea: 15367081 countsHeight: 3.03e+006 cpsStart Time: 15.6 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e54.0e56.0e58.0e51.0e61.2e61.4e61.6e61.8e62.0e62.2e62.4e62.6e62.8e63.0e6

Intensity, cps16.0615.9215.74Sample Name: "b120423a129" Sample ID: "GLP10-01-02-30-001" File: "b120423a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "JPAL-GW-MW1R-0-" Annotation: ""Sample Index: 129Sample Type: UnknownConcentration: N/ACalculated Conc: 1.62 ng/mLAcq. Date: 4/25/2012Acq. Time: 11:44:06 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 8Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ValleyRetention Time: 16.0 minArea: 2234130 countsHeight: 2.68e+005 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e42.0e43.0e44.0e45.0e46.0e47.0e48.0e49.0e41.0e51.1e51.2e51.3e51.4e51.5e51.6e51.7e51.8e51.9e52.0e52.1e52.2e52.3e52.4e52.5e52.6e5

Intensity, cps16.0015.85*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:56 PMPrinting Date: Thursday, May 03, 2012Page 42 of 48Sample Name: "b120423a131" Sample ID: "GLP10-01-02-30-003" File: "b120423a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "JPAL-GW-MW1R-FMS-" Annotation: ""Sample Index: 131Sample Type: UnknownConcentration: N/ACalculated Conc: 4.05 ng/mLAcq. Date: 4/25/2012Acq. Time: 12:27:03 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 5469408 countsHeight: 7.96e+005 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e5

Intensity, cps16.0015.8515.69Sample Name: "b120423a135" Sample ID: "GLP10-01-02-30-004" File: "b120423a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "JPAL-GW-MW3R-0-" Annotation: ""Sample Index: 135Sample Type: UnknownConcentration: N/ACalculated Conc: 9.53 ng/mLAcq. Date: 4/25/2012Acq. Time: 1:53:05 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 15680798 countsHeight: 1.97e+006 cpsStart Time: 15.5 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e61.6e61.7e61.8e61.9e6

Intensity, cps16.0115.8515.70Sample Name: "b120423a138" Sample ID: "GLP10-01-02-30-007" File: "b120423a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "JPAL-GW-MW4R-0-" Annotation: ""Sample Index: 138Sample Type: UnknownConcentration: N/ACalculated Conc: 5.87 ng/mLAcq. Date: 4/25/2012Acq. Time: 2:57:37 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 11808280 countsHeight: 1.77e+006 cpsStart Time: 15.6 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e61.6e61.7e6

Intensity, cps16.0015.85Sample Name: "b120423a141" Sample ID: "GLP10-01-02-30-010" File: "b120423a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "JPAL-GW-MW4L-0-" Annotation: ""Sample Index: 141Sample Type: UnknownConcentration: N/ACalculated Conc: 0.347 ng/mLAcq. Date: 4/25/2012Acq. Time: 4:02:04 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 8Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ValleyRetention Time: 15.9 minArea: 530778 countsHeight: 5.47e+004 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e43.4e43.6e43.8e44.0e44.2e44.4e44.6e44.8e45.0e45.2e45.4e4

Intensity, cps*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:56 PMPrinting Date: Thursday, May 03, 2012Page 43 of 48Sample Name: "b120423a143" Sample ID: "GLP10-01-02-30-012" File: "b120423a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "JPAL-GW-MW4L-FMS-" Annotation: ""Sample Index: 143Sample Type: UnknownConcentration: N/ACalculated Conc: 2.21 ng/mLAcq. Date: 4/25/2012Acq. Time: 4:44:58 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 8Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ValleyRetention Time: 16.0 minArea: 3071219 countsHeight: 4.90e+005 cpsStart Time: 15.6 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e54.4e54.6e54.8e5

Intensity, cps16.0215.8715.7216.38Sample Name: "b120423a147" Sample ID: "GLP10-01-02-30-013" File: "b120423a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "JPAL-GW-MW5R-0-" Annotation: ""Sample Index: 147Sample Type: UnknownConcentration: N/ACalculated Conc: 0.125 ng/mLAcq. Date: 4/25/2012Acq. Time: 6:11:06 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 8Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ValleyRetention Time: 16.0 minArea: 276063 countsHeight: 4.02e+004 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e43.4e43.6e43.8e44.0e4

Intensity, cps16.02Sample Name: "b120423a150" Sample ID: "GLP10-01-02-30-016" File: "b120423a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "JPAL-GW-MW5L-0-" Annotation: ""Sample Index: 150Sample Type: UnknownConcentration: N/ACalculated Conc: 2.14 ng/mLAcq. Date: 4/25/2012Acq. Time: 7:15:38 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 8Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ValleyRetention Time: 16.0 minArea: 2588539 countsHeight: 3.68e+005 cpsStart Time: 15.5 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e5

Intensity, cps16.0015.8515.6916.36Sample Name: "b120423a152" Sample ID: "GLP10-01-02-30-018" File: "b120423a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "JPAL-GW-MW5L-FMS-" Annotation: ""Sample Index: 152Sample Type: UnknownConcentration: N/ACalculated Conc: 4.00 ng/mLAcq. Date: 4/25/2012Acq. Time: 7:58:43 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 4664843 countsHeight: 7.23e+005 cpsStart Time: 15.5 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e5

Intensity, cps16.0015.8515.70*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:56 PMPrinting Date: Thursday, May 03, 2012Page 44 of 48Sample Name: "b120423a153" Sample ID: "GLP10-01-02-30-019" File: "b120423a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "JPAL-GW-MW6R-0-" Annotation: ""Sample Index: 153Sample Type: UnknownConcentration: N/ACalculated Conc: 7.89 ng/mLAcq. Date: 4/25/2012Acq. Time: 8:20:17 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 13961086 countsHeight: 1.93e+006 cpsStart Time: 15.5 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01.0e52.0e53.0e54.0e55.0e56.0e57.0e58.0e59.0e51.0e61.1e61.2e61.3e61.4e61.5e61.6e61.7e61.8e61.9e6

Intensity, cps16.0015.8515.69Sample Name: "b120423a159" Sample ID: "GLP10-01-02-30-022" File: "b120423a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "JPAL-GW-MW6L-0-" Annotation: ""Sample Index: 159Sample Type: UnknownConcentration: N/ACalculated Conc: 0.365 ng/mLAcq. Date: 4/25/2012Acq. Time: 10:29:17 PMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 8Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ValleyRetention Time: 16.0 minArea: 375924 countsHeight: 4.02e+004 cpsStart Time: 15.5 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02000.04000.06000.08000.01.0e41.2e41.4e41.6e41.8e42.0e42.2e42.4e42.6e42.8e43.0e43.2e43.4e43.6e43.8e44.0e4

Intensity, cpsSample Name: "b120423a162" Sample ID: "GLP10-01-02-30-025" File: "b120423a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "JPAL-GW-MW7R-0-" Annotation: ""Sample Index: 162Sample Type: UnknownConcentration: N/ACalculated Conc: 22.7 ng/mLAcq. Date: 4/25/2012Acq. Time: 11:33:51 PMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 33044400 countsHeight: 4.02e+006 cpsStart Time: 15.6 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e54.0e56.0e58.0e51.0e61.2e61.4e61.6e61.8e62.0e62.2e62.4e62.6e62.8e63.0e63.2e63.4e63.6e63.8e64.0e6

Intensity, cps16.0215.8715.72Sample Name: "b120423a165" Sample ID: "GLP10-01-02-30-028" File: "b120423a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "JPAL-GW-MW8R-0-" Annotation: ""Sample Index: 165Sample Type: UnknownConcentration: N/ACalculated Conc: 0.396 ng/mLAcq. Date: 4/26/2012Acq. Time: 12:38:33 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 8Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ValleyRetention Time: 16.0 minArea: 774026 countsHeight: 1.06e+005 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.005000.001.00e41.50e42.00e42.50e43.00e43.50e44.00e44.50e45.00e45.50e46.00e46.50e47.00e47.50e48.00e48.50e49.00e49.50e41.00e51.05e5

Intensity, cps*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:56 PMPrinting Date: Thursday, May 03, 2012Page 45 of 48Sample Name: "b120423a171" Sample ID: "GLP10-01-02-30-031" File: "b120423a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "JPAL-GW-MW9R-0-" Annotation: ""Sample Index: 171Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0652 ng/mLAcq. Date: 4/26/2012Acq. Time: 2:47:37 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 8Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ValleyRetention Time: 16.0 minArea: 142433 countsHeight: 2.10e+004 cpsStart Time: 15.6 minEnd Time: 16.2 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01000.02000.03000.04000.05000.06000.07000.08000.09000.01.0e41.1e41.2e41.3e41.4e41.5e41.6e41.7e41.8e41.9e42.0e42.1e4

Intensity, cps16.0416.4015.90Sample Name: "b120423a173" Sample ID: "GLP10-01-02-30-033" File: "b120423a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "JPAL-GW-MW9R-FMS-" Annotation: ""Sample Index: 173Sample Type: UnknownConcentration: N/ACalculated Conc: 1.84 ng/mLAcq. Date: 4/26/2012Acq. Time: 3:30:38 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 3886088 countsHeight: 6.59e+005 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e5

Intensity, cps16.0215.8715.7016.38Sample Name: "b120423a174" Sample ID: "GLP10-01-02-30-034" File: "b120423a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "JPAL-GW-MW9L-0-" Annotation: ""Sample Index: 174Sample Type: UnknownConcentration: N/ACalculated Conc: 0.0423 ng/mLAcq. Date: 4/26/2012Acq. Time: 3:52:08 AMModified: NoProc. Algorithm: IntelliQuan - MQIINoise Percentage: 50Base. Sub. Window: 1.00 minPeak-Split. Factor: 8Report Largest Peak: YesMin. Peak Height: 0.00 cpsMin. Peak Width: 0.00 secSmoothing Width: 3 pointsRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ValleyRetention Time: 16.0 minArea: 92538 countsHeight: 1.13e+004 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01000.02000.03000.04000.05000.06000.07000.08000.09000.01.0e41.1e41.2e41.3e41.4e4

Intensity, cpsSample Name: "b120423a177" Sample ID: "GLP10-01-02-30-037" File: "b120423a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "JPAL-GW-MW10R-0-" Annotation: ""Sample Index: 177Sample Type: UnknownConcentration: N/ACalculated Conc: 0.232 ng/mLAcq. Date: 4/26/2012Acq. Time: 4:56:43 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 501449 countsHeight: 6.17e+004 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e4

Intensity, cps16.0115.8616.3715.7117.65*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:56 PMPrinting Date: Thursday, May 03, 2012Page 46 of 48Sample Name: "b120423a183" Sample ID: "GLP10-01-02-30-040" File: "b120423a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "JPAL-GW-MW11L-0-" Annotation: ""Sample Index: 183Sample Type: UnknownConcentration: N/ACalculated Conc: 0.512 ng/mLAcq. Date: 4/26/2012Acq. Time: 7:05:50 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 607500 countsHeight: 8.69e+004 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05000.01.0e41.5e42.0e42.5e43.0e43.5e44.0e44.5e45.0e45.5e46.0e46.5e47.0e47.5e48.0e48.5e4

Intensity, cps16.0015.8516.3615.71Sample Name: "b120423a185" Sample ID: "GLP10-01-02-30-042" File: "b120423a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "JPAL-GW-MW11L-FMS-" Annotation: ""Sample Index: 185Sample Type: UnknownConcentration: N/ACalculated Conc: 2.46 ng/mLAcq. Date: 4/26/2012Acq. Time: 7:48:55 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 3263041 countsHeight: 5.04e+005 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.02.0e44.0e46.0e48.0e41.0e51.2e51.4e51.6e51.8e52.0e52.2e52.4e52.6e52.8e53.0e53.2e53.4e53.6e53.8e54.0e54.2e54.4e54.6e54.8e55.0e5

Intensity, cps16.0015.8515.6916.36Sample Name: "b120423a186" Sample ID: "GLP10-01-02-30-043" File: "b120423a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "JPAL-GW-MW3R-RB-" Annotation: ""Sample Index: 186Sample Type: UnknownConcentration: N/ACalculated Conc: 0.00188 ng/mLAcq. Date: 4/26/2012Acq. Time: 8:10:35 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 15854 countsHeight: 1.75e+003 cpsStart Time: 15.8 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01000.02000.03000.04000.05000.06000.07000.08000.09000.01.0e41.1e41.2e41.3e41.4e41.5e41.6e4

Intensity, cpsSample Name: "b120423a187" Sample ID: "GLP10-01-02-30-044" File: "b120423a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "JPAL-GW-TRIP01-0-" Annotation: ""Sample Index: 187Sample Type: UnknownConcentration: N/ACalculated Conc: < 0Acq. Date: 4/26/2012Acq. Time: 8:32:03 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 15.9 minArea: 8444 countsHeight: 1.46e+003 cpsStart Time: 15.8 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.01000.02000.03000.04000.05000.06000.07000.08000.09000.01.0e41.1e41.2e41.3e41.4e41.5e4

Intensity, cps16.3616.2216.1815.8717.68*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:56 PMPrinting Date: Thursday, May 03, 2012Page 47 of 48Sample Name: "b120423a188" Sample ID: "GLP10-01-02-30-045" File: "b120423a.wiff"Peak Name: "PFOS" Mass(es): "499.0/130.0 amu,499.0/99.0 amu,499.0/80.0 amu"Comment: "JPAL-GW-TRIP01-FMS-" Annotation: ""Sample Index: 188Sample Type: UnknownConcentration: N/ACalculated Conc: 1.82 ng/mLAcq. Date: 4/26/2012Acq. Time: 8:53:31 AMModified: YesRT Window: 30.0 secExpected RT: 16.1 minUse Relative RT: YesInt. Type: ManualRetention Time: 16.0 minArea: 4648601 countsHeight: 7.92e+005 cpsStart Time: 15.6 minEnd Time: 16.1 min14.0 14.5 15.0 15.5 16.0 16.5 17.0 17.5 18.0 18.5Time, min0.05.0e41.0e51.5e52.0e52.5e53.0e53.5e54.0e54.5e55.0e55.5e56.0e56.5e57.0e57.5e5

Intensity, cps16.0015.8515.7016.36*** Buster J2930203Batch Name: b120423a.dabData printed by STWPrinting Time: 2:31:56 PMPrinting Date: Thursday, May 03, 2012Page 48 of 48

# A TTACHMENT 

# C: A NALYTICAL 

# M ETHOD 

# ( S 

# )

# ETS -8- 044. 1 Page 2 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS;

Direct Injection Analysis 

# 1 Scope and Application

This method describes the direct injection analysis of perfluorinated compounds (PFCs) from

water matrices using high

- performance liquid chromatography tandem mass spectrometry

(HPLC/MS/MS). The method is generally applicable but not limited to the measurement of

perfluoroalkyl sulfonamides and perfluorinated alkyl acids (PFAAs) such as perfluorosulfonic

acids (PFSAs) and perfluorocarboxylic acids (PFCAs) (Table 1). Water samples containing

heavy particulate may require preparation by an alternate method such as ETS -8- 154

“ Determination of Perfluorinated Acids, Alcohols, Amides, and Sulfonates In Water By Solid

Phase Extraction and High Performance Liquid Chromatography/Mass Spectrometry ”. The

method is applicable to both external standard and internal standard calibration 1



# .

Table 1. Representative Tar get Analytes

**Acronym Analyte**  **Chemical Abstract Services**

**Registry Number (CASRN)**

PFBA (C4 Acid) Perfluorobutanoic acid 375

- 22

- 4

PFPeA (C5 Acid) Perfluoropentanoic acid 2706

- 90

- 3

PFHxA (C6 Acid) Perfluorohexanoic acid 307

- 24

- 4

PFHpA (C7 Acid) Perfluoroheptanoic acid 375

- 85

- 9

PFOA (C8 Acid) Perfluorooctanoic acid 335

- 67

- 1

PFNA (C9 Acid) Perfluorononanoic acid 375

- 95

- 1

PFDA (C10 Acid) Perfluorodecanoic acid 335

- 76

- 2

PFUnA (C11 Acid) Perfluoroundecanoic acid 2058

- 94

- 8

PFDoA (C12 Acid) Perfluorododecanoic acid 307

- 55

- 1

PF TrD A (C1 3 Acid) Perfluorotridecanoic acid 72629

- 94

- 8

PFBS (C4 Sulfonate) Perfluorobutanesulfonic acid 375

- 73

- 5

PFHS (C6 Sulfonate) Perfluorohexanesulfonic acid 355

- 46

- 4

PFOS (C8 Sulfonate) Perfluorooctanesulfonic acid 1763

- 23

- 1

FBSA (C4 Sulfonamide Perfluorobutanesulfonamide 30334

- 69

- 1

FOSA (C8 Sulfonamide) Pefluorooctanesulfonamide 754

- 91

- 6



# The Minimum Reporting Level (MRL) is the Limit of Quantitation (LOQ) that meets Data

Quality Objectives (DQOs) that are developed based on the intended use of this method.

Method Flexibility

– This is a performance

- based method and may be generally applied to the

determination of perfluorinated compounds in water matrices when analysis batch quality

c ontrol ( QC ) criteria are met 2



# . Each set of samples are prepared in an analysis batch with

calibration standards, LCS s, blanks, and continuing calibration check standards analyzed on

the same instrument during a time period that begins and ends with the analysis of the

appropriate continuing calibration check standards. The laboratory is permitted to modify the

LC column, mobile phase composition, LC conditions, and MS/MS conditions. Method

modifications should be considered to improve method performance or to meet data quality

objectives for the study. In all cases where method modifications are implemented, the batch

1 

## The method is supported by validation with internal standard calibration for C4

- C13 PFCAs, C4, C6, and C8 PFSAs,

and C8 perfluoroalkane sulfonamide in laboratory control samples under 3M method validation E11

- 0667.

2 

## Guidance for establishing method QC Criteria based on a.) FDA May 2001, “Guidance for Industry, Bioanalytical

Method Vali dation”, b.) EPA Method 537, and c.) European Commission: Guidance for Generating and Reporting

Methods of Analysis in Support of Pre

- registration Data Requirements for Annex II (Part A, section 4) and Annex III

(Part A,section 5) of Directive 91/414, SANCO/3029/99 rev. 4 (11/07/00).

# ETS -8- 044. 1 Page 3 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS;

Direct Injection Analysis analytical QCs ( section 9) must be completed and pass QC acceptance criteria ( section 13 ) if

the data from the analytical batch are to be reported

.



# 2 Method Summary

Water samples are analyzed as neat aqueous sample or as solvent diluted aqueous samples

by direct injection using LC/ MS/MS. Samples containing heavy particulate may not be

suitable for analysis by this method. Samples containing suspended particulate should be

centrifuge d or filtered prior to removing a sample aliquot or diluting with solvent

. The water

sample is mixe d well prior to removing an aliquot or diluting, if necessary, with ASTM Type I

water, HPLC water, other suitable water, or solvent (methanol).

Quantitation is by stable isotope internal standard calibration in laboratory reagent water. All

perfluorinat ed compounds ( PFC s) target analyte concentrations of perfluorosulfonic acids

(PFSAs) and perfluorocarboxylic acids (PFCAs) are reported as anions and corrected for

their salt or free acid forms. Alternatively, quantitation may be performed by external

st andard calibration.

This is a performance

- based method. Method uncertainty for each target analyte is

determined for each analytical batch using multiple laboratory control spikes at multiple

concentrations. This method also requires that the precision and accuracy for each sample

be determined using field matrix spikes to verify that the method is applicable to each sample

matrix.

Calibration standards for PFUnA, PFDoA, PFTrDA, and FOSA have been found to be

un stable for more than 2 days in 100% water. Samples requiring analysis for these

compounds by this method should be diluted 1:1 with methanol and analyzed against a

calibration curve prepared in 1:1 synthetic groundwater:MeOH.



# 3 Definitions

3.1 Analysis Batch

A set of study samples that are prepared with calibration standards, laboratory control

samples, and procedural blanks , and analyzed on the same instrument during a time period

that begins and ends with the analysis of the appropriate continuing calibration check

standards.



# 3.2 Analytical Sample

A portion of a laboratory sample prepared for analysis.



# 3.3 Calibration Standard

A solution prepared by spiking a known volume of the Working Standard (WS) into a

predete rmined amount of ASTM Type I, HPLC grade water, or other suitable water (i.e.

matrix water), and analyzed according to this method. Calibration standards are used to

calibrate the instrument response with respect to analyte concentration.



# 3.4 Laboratory Duplicate Sample (LDS, or Lab Dup)

A laboratory duplicate sample is a separate aliquot of a sample taken in the analytical

laboratory that is analyzed separately with identical procedures. Analysis of LDSs compared

to that of the first aliquot give a measure of the precision associated with laboratory

procedures, but not with sample collection, preservation, or storage procedures.

# ETS -8- 044. 1 Page 4 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS;

Direct Injection Analysis 

# 3.5 Field Blank (FB)/Trip Blank (TB)

ASTM Type I, HPLC grade water, or other suitable water, placed in a sample container in the

laboratory and treated as a sample in all respects, including exposure to samp ling site

conditions, storage, preservation and all analytical procedures. The purpose of the TB is to

determine if test substances or other interferences are present in the field environment. This

sample is also referred to as a Trip Blank.



# 3.6 Field Duplicate Sample (FDS, Field Dup)

A sample collected in duplicate at the same time from the same location as the sample. The

FDS is handled under identical circumstances and treated exactly the same throughout field

and laboratory procedures. Analysis of the FDS compared to that of the first sample gives a

measure of the precision associated with sample collection, preservation and storage, as well

as with laboratory procedures.



# 3.7 Field Matrix Spike (FMS)

A sample to which known quantities of the target analyte s , ISs and SRSs are added to the

sample bottle in the laboratory before the bottles are sent to the field for collection of aqueous

samples. A known, specific volume of sample must be added to the sample container

without rinsing. This may be accomplished by making a “fill to this level” line on the outside of

the sample container. The FMS is analyzed to ascertain if any matrix effects, interferences,

or stability issues may complicate the interpretation of the sample analysis.



# 3.8 Trip Blank Matrix Spike ( T BMS )

An aliquot of ASTM Type I, HPLC grade water, or other suitable water, to which known

quantities of the target analytes , ISs and SRSs are added in the laboratory prior to the

shipment of the collection bottles. The TBMS is analyzed exactly like a study sample to help

determine if the method is in control and whether a loss of analyte or analytical bias could be

attributed to sample holding time, sample storage and/or shipment issues. A low and high

TBMS are appropriate when expected sample concentrations are not known or may vary.



# 3.9 Internal Standard (IS)

A compound added to each study sample, calibration standard, laboratory control samples,

and procedural blanks at a consistent level (typically around 1 ng/mL). T he internal

standard (s) are stable isotope labeled versions of the target analyte s . The area count ratio of

the target analyte to the internal standard is used for calibration. Surrogate ISs are applied

when stable isotope ISs of target analytes are unavailable. A surrogate IS is not necessar ily

a stable isotope labeled version of the target analyte, but is treated as an internal standard for

quantitation.



# 3.10 Laboratory Control Sample (LCS)

An aliquot of control matrix to which known quantities of the target analytes , ISs and SRSs

(when applicabl e) are added in the laboratory at the time when samples are aliquotted . At

least three levels (two levels for SRSs) in triplicate are included, one generally at the low end

of the calibration curve and one near the mid range and the upper end of the curve. The

LCSs are analyzed exactly like a laboratory sample to determine whether the stability of the

standards . LCSs should be prepared each day samples are aliquoted

.



# 3.11 Laboratory Matrix Spike (LMS)

A laboratory matrix spike is an aliquot of a sample to which known quantities of target

analytes , ISs and SRSs (when applicable) are added in the laboratory. The LMS is analyzed

exactly like a laboratory sample to determine whether the sample matrix contributes bias to

the analytical results. The endogenous concentrations of the analytes in the sample matrix

must be determined in a separate aliquot and the measured values in the LMS corrected for

these concentrations. LMSs are optional for analysis of aqueous samples.

# ETS -8- 044. 1 Page 5 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS;

Direct Injection Analysis 

# 3.12 Laboratory Sample

A portion or aliquot of a sample received from the field for testing.



# 3.13 Limit of Quantitation (LOQ)

The lower limit of quantitation (LLOQ) for an analytical batch is the lowest concentration that

can be reliably quantitated within the specified limits of precision and accuracy. The LLOQ is

generally selected as the lowest non

- zero standard in the calibration curve that meets method

acceptance criteria. The LLOQ for each target analyte is established for each analysis batch

as the lowest calibration standard with area counts at least twice that of the average area

counts of the procedural blanks.

The upper limit of quantitation (ULOQ) for an analytical batch is the highest concentration that

can be reliably quantitated within the specified limits of precision and accuracy. The highest

standard in the calibration curve that meets method acceptance criteria is defined as the

ULOQ.



# 3.14 Method /Procedural Blank

An aliquot of control matrix that is treated exactly like a laboratory sample including exposure

to all glassware, equipment, solvents, and reagents that are used with other laboratory

samples. The method blank is used to determine if test substances or other interferences are

present in the laboratory environment, the reagents, or the apparatus.



# 3.15 Sample

A sample is an aliquot removed from a larger quantity of material intended to represent the

original source material.



# 3.16 Stock Standard Solution (SSS)

A concentrated solution of a single

- analyte prepared in the laboratory with an assayed

reference compound.



# 3.17 Surrogate Internal Standard

An IS that is not necessarily a stable isotopically labeled target analyte, but is treated as an

internal standard for quantitation. Surrogate ISs are used when isotopically labeled

counter parts of the target analyte are not commercially or re adily available.



# 3.18 Surrogate Recovery Standard (SRS)

An isotopically labeled standard , not used as an internal standard, that is added to each

sample and appropriate QC sample as a means to evaluate the method performance for a

chemical class of compounds (e .g., PFSAs, PFCAs)

.



# 3.19 Working Standard (WS)

A solution of several analytes prepared in the laboratory from SSSs and diluted as needed to

prepare calibration standards and other required analyte solutions.



# 4 Warnings and Cautions

4.1 Health and Safety

The acute and chronic toxicity of the standards for this method have not been precisely

determined; however, each should be treated as a potential health hazard. The analyst

should wear gloves, a lab coat, and safety glasses to prevent exposure to chemicals that

might be present.

# ETS -8- 044. 1 Page 6 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS;

Direct Injection Analysis The laboratory is responsible for maintaining a safe work environment and a current

awareness of local regulations regarding the handling of the chemicals used in this method.

A reference file of material safety data sheets (MSDS) should be available to all personnel

involved in these analyses.



# 4.2 Cautions

The analyst must be familiar with the laboratory equipment and potential hazards including,

but not limited to, the use of solvents, pressurized gas and solvent lines, high voltage, and

vacuum systems. Refer to the appropriate equipment procedure or operator manual for

additional information and cautions.



# 5 Interferences

During sample preparation and analysis, major potential contaminant sources are reagents

and glassware. All materials used in the analyses shall be demonstrated to be free from

interferences under conditions of analysis by running method blanks.

Parts and supplies that contain Teflon® should be avoided or minimized due to the possibility

of interference and/or contamination. These may include, but are not limited to: wash bottles,

Teflon ® 

# lined caps, autovial caps, HPLC parts, etc.

The use of disposable micropipettes or pipettes to aliquot standard solutions is recommended

to make calibration standards and matrix spikes.



# 6 Instrumen tation, Supplies, and Materials

6.1 Instrumentation

Analytical balance capable of reading to 0.0001g

HPLC/MS/MS or HPLC/MS system, as described in Section 10

.



# 6.2 Supplies and Materials

Sample collection bottles

— HDPE (e.g., NalgeneTM) wide

- mouth bottles with screw cap.

Note: Do not use fluorinated or Teflon® bottles or lined caps.

Coolers or boxes for sample shipment.

15

- mL and 50

- mL disposable polypropylene centrifuge tubes.

Class A pipettes and volumetric flasks, various.

2 mL HPLC auto vial s

Disposable pipettes, polypropylene or glass as appropriate

Centrifuge capable of spinning 15

- mL and 50

- mL polypropylene tubes at 3000 rpm.



# 7 Reagents and Standards

Note: Suppliers and catalog numbers are for illustrative purposes only. Equivalent

performance may be ach ieved using chemicals obtained from other suppliers. Do not use a

lesser grade of chemical than those listed.



# 7.1 Chemicals

Water

– Milli

- Q, HPLC grade, or other suitably appropriate sources

Calcium Acetate

- A.C.S. Reagent Grade

# ETS -8- 044. 1 Page 7 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS;

Direct Injection Analysis Magnesium Acetate

- A.C.S. Re agent Grade

Methanol

– HPLC grade

Ammonium Acetate

– A.C.S. Reagent Grade



# 7.2 Representative Target Analytes, ISs, and SRSs

PFBA, Heptafluorobutyric Acid, (C 4 

# Perfluorinated Acid)

PFPeA , Nonafluoropentanoic Acid (C 5 

# Perfluorinated Acid)

PFHxA, Perfluorohexanoi c Acid (C 6 

# Perfluorinated Acid)

PFHpA, Tridecafluoroheptanoic Acid, (C 7 

# Perfluorinated Acid)

PFOA, Ammonium perfluorooctanoate, (C 8 

# Perfluorinated Acid)

PFNA, Heptadecafluorononanoic Acid, (C 9 

# Perfluorinated Acid)

PFDA, Nonadecafluorodecanoic Acid (C 10 

# Per fluorinated Acid)

PFUnA, Perfluoroundecanoic Acid, (C 11 

# Perfluorinated Acid)

PFDoA, Perfluorododecanoic Acid, (C 12 

# Perfluorinated Acid)

PFTrDA, Perfluorotridecanoic Acid, (C 13 

# Perfluorinated Acid)

FBSA, Perfluorobutanesulfonamide

FOSA, Perfluorooctanesulfo nylamide

PFBS, Potassium Perfluorobutane sulfonate

PFHS, Perfluorohexanesulfonate

PFOS, Potassium perfluorooctanesulfonate

PFOA [1,2, 3, 4

- 13 

# C], 13 

# C 4 

# - isotopically labeled perfluorooctanoic acid (SRS)

PFOS [1,2, 3, 4

- 13 

# C], 13 

# C 4 

# - isotopically labeled Perfluorooctanesulfonate (SRS)

PFUnA [1,2

- 13 

# C], 13 

# C 2 

# - isotopically labeled Perfluoroundecanoic acid (SRS)

A custom mix of ISs in a methanolic solution containing ([1,2,3,4

- 13 

# C 4 

# ]PFBA, [1,2

-

13 

# C 2 

# ]PFHxA, [1,2,3,4 ,5,6,7,8

- 13 

# C 8 

# ]PFOA, [1,2,3,4,5 ,6,7,8,9

- 13 

# C 9 

# ]PFN A, [1,2

- 13 

# C 2 

# ]PFDA,

[1,2 ,3,4,5,6,7

- 13 

# C 7 

# ]PFUnA, [1,2

- 13 

# C 2 

# ]PFDoA, [1,2 ,3

- 13 

# C 3 

# ]PFHS, [1,2,3,4 ,5,6,7,8

- 13 

# C 8 

# ]PFOS

, and [1,2,3,4,5,6,7,8

- 13 

# C 8 

# ]PFOSA (Wellington Laboratories, Guelph, ON) in combination with

added ([1,2,3,4 ,5

- 13 

# C 5 

# ]PFPe A

, ([1,2,3,4

- 13 

# C 4 

# ]PFHpA, and [ 18 

# O 2 

# ]PFBS can be used to prepare

a stock IS solution. Alternatively, individual stable isotope ISs can be used to prepare a stock

IS mixture.

Other ISs can be applied.



# 7.3 Reagent Preparation

2 mM A mmonium acetate solution (Analysis)

— Weigh 0.3 g of A mmonium acetate and

dissolve in 2.0 L of reagent water.

Synthetic Groundwater (containing 25 ppm Ca and Mg)

– Weigh 0.61 g of Calcium Acetate

and 0.92 g of Magnesium Acetate and dissolve in 6.0 L of reagent water.

Note: Alternative volumes may be prepared as long as the ratios of the solvent to solute ratios

are maintained.

# ETS -8- 044. 1 Page 8 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS;

Direct Injection Analysis 

# 7.4 Stock Standard Solution (SSS) and Working Standard Solution

Preparation

The following standard preparation procedure serves as an example. Weighed amounts and

final volumes may be changed to suit the needs of a particular study. For example, μL

volumes may be spiked into volumetric flasks when diluting stock solutions to appropriate

levels.

100 μg/mL target analyte SSSs

— Weigh out 10 mg of analytical standard **(corrected for**

**percen t salt**  **, acid [ETS -4-**  **031] and purity**  ) and dilute to 100 mL with methanol or other

suitable solvent, in a 100 mL volumetric flask. Transfer to a 125 mL LDPE bottle or other

suitable container. Prepare a separate solution for each analyte. Expiration dates and

storage conditions of stock solutions should be assigned in accordance with laboratory

standard operating procedure. An example of purity and salt correction is given below for

PFOS.



# salt of weight moclecular anion of weight molecular factor correction salt

=

9275.0

538 499 factor correction )salt (K PFOS

== +



# 10 mg C 8 

# F 17 

# S0 3 - 

# K + 

# with purity 90% = 8.35 mg C 8 

# F 17 

# S0 3 – 

# (10 mg*0.90*0.9275=8.35 mg)

10 μg/mL (10,000 ng/mL) mixed working standard

— Add 5.0 mL each of the 100 μg/mL

SSSs to a 50 mL volumetric flask and bring up to volume with solvent.

1 μg/mL (1,000 ng/mL) mixed working stan dard

— Add 0.5 mL of the 100 μg/mL SSSs to a

50 mL volumetric flask and bring up to volume with solvent.

0.1 μg/mL (100 ng/mL) mixed standard

— Add 0.05 mL of the 100 μg/mL SSSs to a 50 mL

volumetric flask and bring up to volume with solvent.

Storage Conditi ons

— Store all SSSs and working standards in accordance with laboratory

standard operating procedure or in a refrigerator at 4°±2°C for a maximum period of 6 months

from the date of preparation.



# 7.5 Calibration Standards

Calibration can be performed by IS or external calibration. Using the working standards

described above, prepare calibration solutions in ASTM Type I water, HPLC water, other

suitable water, or a mixture of solvent and water using the information in Table 2 as a

guideline. Note: Volumes of wa ter or water/solvent mixtures and working standards may be

adjusted to meet the data quality objectives addressed in the general project outline.

Calibration levels other than those listed below can be prepared as needed.

For the quantitation of PFOA and PFOS, reference materials of certified mixed linear and

branched isomer are preferred. Alternately, reference materials of primarily linear isomers of

PFOA and/or PFOS may be used, however, when quantitating with predominantly linear

reference standards, additional LCS samples containing both linear and branched isomers of

PFOA and PFOS are required 3



# .

7.5.1 Internal Standard (IS) and Surrogate Recovery Standard (SRS)

For IS calibration, stable isotope internal standards of each target analyte or appropriate

surrogate ISs should be spiked at the same level in all calibration standards. Once the

calibration standards have been prepared as stated above in Section 7.5, all calibration

standards are spiked with a separate internal standard spiking solution. Typically the

3 

### A report summarizing an assessment of the use of reference standards containing certified linear and branched isomers of

PFOA/PFOS can be found in 3M report E11

- 0560.

# ETS -8- 044. 1 Page 9 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS;

Direct Injection Analysis concentration of the internal standard is consistent with the internal standard concentration

expected in the samples being prepared, usually 1 ng/mL. The concentration of the internal

standard spiking solution is typically 2 μg/ mL. A separate zero point or method blank is

typically prepared at the same time as the calibration standards, using the same solution

used to prepare the standards (ASTM Type I water, HPLC water, other suitable water, or a

solvent/water mixture), and is spiked with the internal standard at the same concentration as

the calibration curve, typically at 1 ng/mL.

If the sample s being analzyed w ere pre

- spiked with SRSs , the calibration curve prepared in

Section 7.5 is spiked with a separate SRS spiking solution. Typically, the sample bottles are

spiked with a SRS at 0.1 ng/mL. The final calibration curve must consist of at least six

calibration points after analysis. The following table provides an example of spike

concentrations and volumes used to achieve a multi

- point extracted calibration curve with

internal standard and surrogate standard.

Table 1 lists recommended stable isotope internal standards for several PFSA and PFCA

target compounds. A custom mix of isotopically labeled target analytes in a methanolic

solution containing ([1,2,3,4

- 13 

# C 4 

# ]PFBA, [1,2

- 13 

# C 2 

# ]PFHxA, [1,2,3,4,5,6,7,8

- 13 

# C 8 

# ]PFOA,

[1,2,3,4,5,6,7,8,9

- 13 

# C 9 

# ]PFNA, [1,2,3,4,5,6

- 13 

# C 6 

# ]PFDA, [1,2,3,4,5,6,7

- 13 

# C 7 

# ]PFUnA, [1,2

-

13 

# C 2 

# ]PFDoA, [ 1,2,3

- 13 

# C 3 

# ]PFHS, [1,2,3,4,5,6,7,8

- 13 

# C 8 

# ]PFOS, and [1,2,3,4,5,6,7,8

- 13 

# C 8 

# ]FOSA

(Wellington Laboratories, Guelph, ON) in combination with added ([1,2,3,4 ,5

- 13 

# C 5 

# ]PFPe A

, ([1,2,3,4

- 13 

# C 4 

# ]PFHp A , and [ 18 

# O 2 

# ]PFBS can be used to prepare a stock IS solution.

Alternative sources of certified stable isotope labeled target analytes are applicable.

Alternatively, individual stable isotope ISs can be used to prepare a stock IS mixture. The

table below lists the recommended stable isotope ISs and SRSs applied in the method.

Other stable isotope ISs and SRSs of target analytes not listed in the table may be used if

supported by validation and/or analysis batch QCs meeting method acceptance criteria (e.g.,

[ 13 

# C 2 

# ]- PFOA) . The same internal standard should be used for a given analyte throughout the

entire project/study. Note: some of the compounds listed below are appropriate to use as

surrogate ISs when a stable isotope IS of a target analyte is not available. Generally,

surrogate isotopically labeled PFCAs are used for PFCAs, and surrogate i sotopically labeled

PFSA s are used for PFSAs.

Table 2 provides examples of spike concentrations and volumes used to achieve a multi

- point

calibration curve with ISs and SRSs

.

# ETS -8- 044. 1 Page 10 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS;

Direct Injection Analysis Table 1 . Stable Isotope PFCAs and PFSAs used for ISs and SRSs

Compound Name Synonym or Acronym Analytical Purpose Reference Standard

Source

13 

### C 4 

### - Perfluorobutanoic acid [1,2,3,4

- 13 

### C 4 

### ]PFBA IS for PFBA Wellington Labs

(Mix or Individual)

13 

### C 4 

### - Perfluoro pentanoic acid [1,2,3,4 ,5

- 13 

### C 5 

### ]PFPe A IS for PFPeA Wellington Labs

(Mix or Individual)

13 

### C 2 

### - Perfluorohexanoic acid [1,2

- 13 

### C 2 

### ]PFHxA IS for PFHxA Wellington Labs

(Mix or Individual)

13 

### C 4 

### - Perfluoro hept anoic acid [1,2,3,4

- 13 

### C 4 

### ]PFHp A IS for PF HpA Wellington Labs

(Mix or Individual)

13 

### C 8 

### - Perfluorooctanoic acid [1,2,3,4,5,6,7,8

- 13 

### C 8 

### ]PFOA IS for PFOA and [1,2,3,4

13 

### C 4 

### ]PFOA Wellington Labs

(Mix or Individual)

13 

### C 9 

### - Perfluorononanoic acid [1,2,3,4,5,6,7,8,9

- 13 

### C 9 

### ]PFNA IS for PFNA Wellington Labs

(Mix or Individual)

13 

### C 6 

### - Perfluorodecanoic acid [1,2,3,4,5,6

- 13 

### C 6 

### ]PFDA IS for PFDA Wellington Labs

(Mix or Individual)

13 

### C 7 

### - Perfluoroundecanoic acid [1,2,3,4,5,6,7

- 13 

### C 7 

### ]PFUnA IS for PFUnA Wellington Labs

(Mix or Individual)

13 

### C 2 

### - Perfluorododecanoic acid [1,2

- 13 

### C 2 

### ]PFDoA IS for PFDoA, *PFTA Wellington Labs

(Mix or Individual)

18 

### O 2 

### - Ammonium Perfluorobutane sulfonate [ 18 

### O 2 

### ]PFBS IS for PFBS RTI International

(Individual)

13 

### C 3 

### - Ammonium Perfluorohexane sulfonate [ 1,2,3

- 13 

### C 3 

### ]PFHS IS for PFHS Wellington Labs

(Mix or Individual)

13 

### C 8 

### - Sodium Perfluorooctane sulfonate [1,2,3,4,5,6,7,8

- 13 

### C 8 

### ]PFOS IS for PFOS and

PFOS[1,2,3,4 13 

### C 4 

### ], Wellington Labs

(Mix or Individual)

13 

### C 8 

### - Perfluorooctanesulfonamide [1,2,3,4,5,6,7,8

- 13 

### C 8 

### ]FOSA IS for FOSA Wellington Labs (mix)

RTI International

(Individual)

13 

### C 4 

### - Perfluorooctanoic acid [1,2,3,4

- 13 

### C 4 

### ]PFOA SRS for all PFCAs : C4

- C8 Wellington

13 

### C 2 

### - Perfluoroundecanoic acid [1,2

- 13 

### C 2 

### ]PFUnA SRS for all PFCAs C9

- C13 Wellington

13 

### C 8 

### - Perfluorooctane sulfonate [1,2,3,4

- 13 

### C 4 

### ]PFOS SRS for all PFSA s: C4,

C6, and C8 Wellington

# ETS -8- 044. 1 Page 11 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Water by LC/MS/MS; Direct Injection Analysis Table 2 . Example Preparation of Calibration Curve with ISs and SRSs

**Sample Description Concentration** 

**of WS, μ**  **g/mL Volume of** 

**WS,** μ **L Volume of IS** 

**(2 μ**  **g/mL),** μ **L Concentration of** 

**Surrogate, μ**  **g/mL Volume of** 

**Surrogate,** μ **L Volume of ASTM Type I**  **Water, or**

**other suitable solvent**  **(1)** 

### , mL

0.025 ng/mL curve point 0.10 25 50 0.20 12.5 100

0.030 ng/mL curve point 0.10 30 50 0.20 15 100

0.04 ng/mL curve point 0.10 40 50 0.20 20 100

0.05 ng/mL curve point 0.10 50 50 0.20 25 100

0.1 ng/mL curve point 0.10 100 50 0.20 50 100

0.25 ng/mL curve point 0.10 250 50 0.20 125 100

0.5 ng/mL curve point 1.0 50 50 0.20 250 100

1 ng/mL curve point 1.0 100 50 0.20 500 100

2.5 ng/mL curve point 10.0 25 50 10.0 25 100

5.0 ng/mL curve point 10.0 50 50 10.0 50 100

10.0 ng/mL curve point 10.0 100 50 10.0 100 100

25.0 ng/mL curve point 10.0 250 50 10.0 NA 100

50.0 ng/mL curve point 10.0 500 50 10.0 NA 100

75.0 ng/mL curve point 10.0 750 50 10.0 NA 100

100 ng/mL curve point 10.0 1000 50 10.0 NA 100

N/A

- Not Applicable

(1) Samples requiring analysis for PFUnA, PFDoA, PFTrDA, and FOSA should be analyzed against a calibration curve prepared in 1:1 synthetic groundwa ter:MeOH.

# ETS -8- 044. 1 Page 12 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS; Direct Injection

Analysis 

# 8 Sample Collection and Bottle Preparation

Sample collection bottles are prepared by 3M Environmental Laboratory (or subcontract supplier) personnel for

shipment at ambient temperature to the collection site. Typically, four separate collection bottles are

associated with a single collection site: sample, field duplicate sample, low field matrix spike, and high field

matrix spike. Alternatively, the sample and field duplicate sample may contain SRSs in lieu of additional target

analyte low field matrix spike and target analyte high field matrix spike samples. Depending on the scope of

the project, additional replicates of the field sample and field matrix spikes may be added. Also, it is not

uncommon for additional mid

- level field matrix spikes to be collected if the expected sample concentrations are

truly unknown or could span a large concentration range.

High

- density polyethylene (HDPE) wide

- mouth Nalgene bottles are used for the sample collection containers.

(Volumes of the bottles may vary depending on how much sample is required to meet data quality objectives.)

Sample collection volumes are project specific and based on data quality objectives. The Nalgene bottles do

not require any pretreatment prior to use. Typically, placement of a sample bottle volumetric “fill to here” line is

done by using a sample bottle marker template. Alternatively, bottles may be weighed prior to bottle

preparation and weighed again after samples have been collected.

All bottles should be clearly labeled to indicate its intended use as a sample, field sample duplicate, low field

matrix spike, high field matrix spike, sample/SRS field matrix spike, field duplicate sample/SRS field matrix

spike, trip blank, or trip blank matrix spike. If each location has different designated spike levels, the label

should also clearly indicate the sample location designation. Generally, a set of bottles for a given collection

site are prepared then grouped together in plastic bags for organizational purposes. For each sample

collection event, at least one set of trip blank and trip blank matrix spikes are prepared.

Bottle preparation should be documented in a Note to File or on a sample preparation worksheet and should

include the following information: date prepared, total number of bottles prepared, number of sample sites, the

standard identification numbers and spike volumes used to prepare spiked bottles, the “fill to here” volume, and

any other pertinent information needed for reconstructibility of the data. The Note to File will be included in the

final data package for the project.

Samples are collected in the field and shipped to the laboratory at ambient temperature.



# 8.1 Field Matrix Spike Sample (FMS)

Field matrix spike samples are a requirement of the method. A FMS sample is defined as a QC sample to

which known quantities of appropriate target analytes are added to the sample bottle in the field or in the

laboratory before the bottles are sent to the field. The sample and field duplicate sample may contain

appropriate SRSs in lieu of target analyte FMS samples. Sample quantities are determined vol umetrically or

gravimetrically. A known, specific volume or weight of sample is added to the sample container without

rinsing. Volumetric sample measurements may be acquired by a laboratory applied “fill to this level” line on the

outside of the sample container. Target analyte FMS samples should be spiked at approximately 0.5

- 10 times

the expected analyte concentration in the sample. If the expected range of analyte concentrations is unknown,

multiple spikes at varying levels may be prepared to increase the likelihood that a spike at an appropriate level

is made. Typically a low and a high target analyte spike are prepared for each sampling location. In those

instances where SRSs are to be used in lieu of target analyte FMS samples, the sample and field duplicate

sample are spiked at approximately 2

- 5 times the target LOQ. The FMS is analyzed to ascertain if matrix

effects or sample holding time contributes bias to the analytical results. For the sample bottles designated for

matrix spikes, an appropriate volume of matrix spiking solution is added to the empty bottle prior to sampling.

The volume of spike solution added should produce the desired final concentration of target analytes once the

bottle is filled with sample to the “fill to here line”. The matrix spiking solution(s) should be prepared in a

suitable solvent and contain all of the appropriate target analytes, ISs, and SRSs. The target analyte matrix

spiking solution is often the same as the working standards used to create the calibration standards. An

example of a bottle spike is given below.

“Fill to here” volume = 200 mL (A 250 mL Nalgene bottle is used)

Desired Field Spike Concentration = 0.25 ng/mL

# ETS -8- 044. 1 Page 13 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS; Direct Injection

Analysis 500 μ L of a 0.1 μ g/mL spiking solution (containing the target analytes) is added to the bottle and the bottle cap

promptly sealed.



# 8.2 Internal Standard and Surrogate Recovery Standard

If analysis of a surrogate recovery standard (SRS) is included in the project objectives, an appropriate volume

of a surrogate standard solution is added to all the bottles prior to sampling and SPE. Typically sample bottles

are spiked with surrogate recovery standards at a final desired spike concentration of 0.1 ng/mL.

If quantitation by internal standard (IS) is included in the project objective, an appropriate volume of internal

standard solution is added to all the bottles prior to sampling and SPE . Typically sample bottles are spiked

with internal standard at a final desired spike concentration of 1 ng/mL.

For the trip blank, the SRS spike and IS spike i s added to the bottle and then ASTM Type I water (HPLC grade

reagent water or other suitable water may used) is added to the “fill to here” line. The bottle is capped and

sealing tape may be placed around the outer edge of the cap. Trip blank matrix spikes are prepared by adding

the appropriate volume of target analyte spiking solution, IS, and SRS spiking solutions and filling the bottle to

the desired volume with the appropriate water and capping and sealing the cap.



# 9 Quality Control and Data Quality Objectives

9.1 Data Quality Objectives

This method and required quality control samples is designed to generate data accurate to ±30% with a

targeted LOQ of 0.025 ng/mL. Any deviations from the quality control measures spelled out below will be

documented in the raw data and footnoted in the final report.



# 9.2 Method/Procedural Blanks

The method/procedural blank is zero point calibration standard (which includes ISs) analyzed in a regular basis

with each analysis batch. At a minimum, method blanks are analyzed prior to instrument calibration, prior to

the analysis of CCV samples, after every 10 sample injections, and at the end of the analytical run.

The mean area count or area ratios when using internal standard calibration, for each analyte in the method

blanks m ust be less than 50% of the area count counts or area ratios when using internal standard calibration,

of the LOQ standard. The standard deviation of the area counts, or area ratios when using internal standard

calibration, of these method blanks should be calculated. A specific %RSD acceptance criteria is not specified

but is assessed on an analytical batch basis. If the mean area counts or area ratios when using internal

standard calibration, of the method blanks exceed 50% of the LOQ standard, then the LOQ must be raised to

the first standard level in the curve that meets criteria. Method blanks may be eliminated if technical

justification can be provided (e.g. the procedural blank was analyzed after an unexpectedly high level sample).

If any procedural blanks are removed from the LOQ determination, document in the raw data and report as

appropriate. Laboratory Sample Replicates / Field Duplicate Sample

Typically, samples are collected in duplicates in the field. The relative percent difference (RPD) of duplicate

samples should be ≤20% for the precision of sample preparation and analysis to be considered in control.

Replicate samples not meeting the ≤20% RPD criteria are flagged and reported as outside of QC acceptance

criteria.



# 9.3 Laboratory Matrix S pikes (LMSs)

LMSs may be performed in lieu of FMSs if FMSs have previously been performed for the sample matrix.

Additionally, LMSs may be performed in lieu of FMSs for a sample matrix if the FMS levels were not

appropriate for determining spike recoveries relative to endogenous levels of target analytes and appropriate

SRSs. Generally, each sample location represents a different sample and sample matrix. LMSs are prepared

for each sample and analyzed to determine the matrix effect on spike recovery efficiency of each target analyte

and appropriate SRSs. LMSs should be prepared at a minimum of one level and in duplicate. LMS

concentrations should be prepared at approximately 0.5

- 10 times the endogenous concentration or

approximately 4

- 10 times the LOQ concentration of each target analyte.

# ETS -8- 044. 1 Page 14 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS; Direct Injection

Analysis Lab matrix spike recoveries should fall within ± 30% of expected values. Sample data with LMS recovery

outside of ±30% but within ±50% of the expected value are flagged and reported as outside of QC acceptance

criter ia. Data with LMS recovery outside of ±50% of the expected value are reported as NR, where NR is

defined as “Not Reportable” data outside of QC acceptance criteria.



# 9.4 Lab Control Sample

Lab control spikes are prepared for each analysis batch to determine method accuracy and precision. LCSs

should be prepared at three levels in triplicate for each target analyte and at a minimum of two levels in

triplicate for appropriate SRSs. Low lab control spikes should be prepared at a concentration in the range of

a pproximately four to ten times higher than the targeted lower LOQ, the mid lab control spikes should be

prepared at a concentration near the mid

- point of the calibration curve and the high lab control spikes at

approximately 80% of the upper LOQ. For each target analyte and SRSs, the percent relative standard

deviation (method precision) for each control spike level must be less than or equal to 20% and the average

recovery (method accuracy) for each control spike level must be 80

- 120%. Sample data for target analytes

outside of the laboratory control spike acceptance criteria will be handled as follows:

If the average recovery of a spiking level falls outside method acceptance, but at least 67% (6 out of 9 ) o f LCS

samples are within 20 % of their respective nominal value (33% of the QC samples, not all replicates at the

same concentration, may be outside 20% of nominal value), the average recovery will be flagged as outside

method acceptance criteria. All LCS samples will be control charted as per ETS -4- 02 6. If the average

recovery of one of the spiking levels exceeded the analytical method uncertainty as determined by ETS

- 12

- 012, that analytical batch uncertainty will be expanded for that particular study.

If more than 67% of the LCS samples fail to meet method acceptance criteria, the data will not be reported.

Calibration standards consisting of mixed branched and linear isomer PFOS/PFOA are preferred. However,

for PFOS/PFOA target analytes, if the calibration standards are comprised of predominantly linear isomers

only, at least one level of triplicate LCSs should be prepared using PFOS/PFOA which contains a mix of linear

and branched isomers. These LCSs will be used to demonstrate quantitative equivalency (or quantitative bias)

of the isomeric mix when using a predominantly linear standard for calibration. The mixed linear and branched

isomer PFOS/PFOA LCSs recoveries should fall within ± 30% of expected values. Alternatively, in lieu of

mixed branched and linear isomer PFOS/PFOA LCSs, mixed branched and linear isomer PFOS/PFOA

TBMSs may be applied to demonstrate method accuracy and precision.



# 9.5 Field Matrix Spikes (FMSs) / Surrogate Recovery Standards (SRSs)

FMSs are prepared for each sampling location and analyzed to determine the matrix effect and sample holding

time on the spike recovery of each target analyte and/or appropriate SRS. Generally, each sample location

represents a different sample and sample matrix.

FMSs are QC samples to which known quantities of appropriate target analytes are added to the sample bottle

in the laboratory before the bottles are sent to the field. Typically a low and a high target analyte FMS are

prepared for each sampling location. The sample and field duplicate sample may contain appropriate SRSs in

lieu of target analyte low field matrix spike and target analyte high field matrix spike samples.

Field matrix spike method acceptance criteria are recoveries within ±30% of the expected value. If FMS

recovery (target analyte or SRS spike) is outside of ±30% of the expected value or could not be assessed

because the FMS (target analyte) was spiked at an inappropriate level, the sample result is reported as follows:

1.) If target analyte FMS recovery could not be assessed because the FMS's were at an inappropriate level,

then Laboratory Matrix Spikes (LMS) may be substituted. If LMS recoveries are within ±30% the data are

reportable and flagged to indicate that the FMS spikes levels were inappropriate.

2.) If multiple target analyte FMS's were prepared on a sample and the closest FMS level to the reported

sample meets the ±30% acceptance criteria but additional FMS's are outside the ±30% acceptance range, the

data are reportable and flagged to indicate that while there were failing FMS’s, the uncertainty will not be

expanded since the most appropriate spike level passed.

# ETS -8- 044. 1 Page 15 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS; Direct Injection

Analysis 3.) If the target analyte FMS recoveries are outside of the ±30% acceptance range but at least 30 acceptable

historical reportable FMS sample results are available, the data may be reported but flagged with an expanded

uncertainty and as not meeting FMS criteria.

4.) Sample data with FMS recovery outside of ±30% but within ±50% of the expected value are flagged and

reported as outside of QC acceptance criteria with an expanded uncertainty.

5.) If FMS recovery is outside of ±50%, the sample result is reported as NR, where NR is defined as “Not

Reportable” due to noncompliant QC results.

The targeted fortification levels should be at least 50% of the endogenous level and less than 10 times the

e ndogenous level to be used without justification to determine the statement of accuracy for analytical results.

Note: It is possible for bottles utilized for Field Matrix Spike samples to be under

- filled or over

- filled during

sample collection. Since this scenario will effect the actual concentration of the FMS sample (surrogate and

internal standard concentrations will also be effected, if used), it is important that any obvious under

- filling or

over

- filling of sample bottles be documented in the data package and taken into account in the FMS , ISs, or

SRSs recovery calculations. Samples over

- filled or under

- filled by more than 10% will be require recalculation

of the FMS, ISs, and SRS true values.

The average of the sample and the field duplicate should be used to calculate the recovery.



# 10 Procedures

10.1 Water Sample Preparation

This method is applicable to water samples. Samples containing heavy particulate may not be suitable for

analysis by this method. Samples containing suspended particulate should be centrifuge prior to removing a

sample aliquot, or filtered.

• Thoroughly mix sample before removing a n aliquot and placing in a labeled autovial.

• Dilute sample, if necessary, with ASTM Type I water , HPLC water, other suitable water , or solvent

(methanol)

.

• L ab control spikes are prepared for each analysis batch to determine method accuracy and precision.

LCSs should be prepared at three levels in triplicate for each target analyte and at a minimum of two

levels in triplicate for appropriate SRSs. Low lab control spikes should be prepared at a concentration

in the range of approximately four to ten times higher than the targeted lower LOQ, the mid lab control

spikes should be prepared at a concentration near the mid

- point of the calibration curve and the high

lab control spikes at approximately 80% of the upper LOQ. For IS quantitation, stable isotope internal

standards of each target analyte or appropriate surrogate ISs should be spiked at the same level as the

samples being analyzed, in all LCSs

.

• If LCSs are being prepared us ing synthetic groundwater, allow the LCSs samples to equilibrate for a

minimum of 4 hours before aliquoting for analysis or diluting with solvent (methanol).



# 11 Sample Analysis

- LC/MS/MS

11.1 Instrument Setup

Note: In this example, an Applied Biosystems Sciex API 4000 (API 5000 or API 5500) Tandem Mass

Spectrometer (LC/MS/MS) is used. Other brands/models of LC/MS/MS instruments as well as single

quadrupole mass spectrometers (LC/MS) may be used as long as the method acceptance criteria are met.

Brand names, suppliers, part numbers, and models are for illustrative purposes only. Equivalent performance

may be achieved using apparatus and materials other than those specified here, but demonstration of

equivalent performance that meets the requirements of this method is the responsibility of the laboratory. The

operator must optimize and document the equipment and settings used.

# ETS -8- 044. 1 Page 16 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS; Direct Injection

Analysis Establish the LC/MS/MS system and operating conditions equivalent to the following:

Mass Spec: Applied Biosystems API 4000, API 5000, or API 5500

Ion Source: Turbo Ion Spray (ABS)

Mode: Electrospray Negative

Scan Type: MRM (Multiple Reaction Monitoring)

Computer: Dell DHM

Software: Windows 2000 or Windows XP, Analyst 1.4.2 or higher versi ons

HPLC: Agilent Series 1100,1200, or 1290

Agilent Quaternary Pump

Agilent Vacuum Degasser

Agilent Autosampler

Agilent Column Oven

Note: One or more C18 HPLC analytical columns (2.1 mm x 100 mm, 5 μ m or 2.1 mm x 50 mm, 5 μ m) may be

attached on

- line after the purge valve and before the sample injection port to retard and separate any residue

contaminants that may be in the mobile phase and/or HPLC system.

HPLC Column: Betasil C18, 4.6mm x 100mm, 5 μ m (ThermoElectron Corporation)

Column Temperature: 35 ° C

Inje ction Volume: 5μL

Mobile Phase (A): 2mM Ammonium Acetate in ASTM Type I water (See 7.3)

Mobile Phase (B): Methanol

Table 3

. Liquid Chromatography Gradient Program

.

**Step**

**Number Total Time** 

**(min) Flow Rate** 

**(** μ

**L/min) Percent A** 

**(2 mM ammonium** 

**acetate) Percent B** 

**(Methanol)**

0 0 750 97.0 3.0

1 2.0 750 97.0 3.0

2 14.5 750 5.0 95.0

3 15.5 750 5.0 95.0

4 16.5 750 97.0 3.0

5 20.0 750 97.0 3.0



# Note: Other HPLC gradients may be used as long as the method criteria and project data quality objectives are

met.

It may be necessary to adjust the HPLC gradient in order to optimize instrument performance. Columns with

different dimensions (e.g. 2.1mm x 30mm) and columns from different manufacturers (Keystone Betasil C18

etc.) may be used.

# ETS -8- 044. 1 Page 17 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS; Direct Injection

Analysis Table 4 Suggested MRM Transitions for Target Analytes, Surrogates, and Internal Standards

Analyte Analyte Description Mass Transition

Q1 (amu) Mass Transition

Q3 (amu)

PFBA (C4 Acid) Target 213 169

PFPeA (C5 Acid) Target 263 219

PFHxA (C6 Acid) Target 313 269, 119

PFHpA (C7 Acid) Target 363 319, 169

PFOA (C8 Acid) Target 413 369, 219, 169

PFNA (C9 Acid) Target 463 419, 169, 219

PFDA (C10 Acid) Target 513 469, 269, 219

PFUnA (C11 Acid) Target 563 519, 269, 219

PFDoA (C12 Acid) Target 613 569, 169, 319

PFTA (C13 Acid) Target 663 619, 369, 319

FBSA (C4 Sulfonamide) Target 298 78

FOSA (C8 Sulfonamide) Target 498 78

PFBS (C4 Sulfonate) Target 299 99, 80

PFHS (C6 Sulfonate) Target 399 99, 80

PFOS (C8 Sulfonate) Target 499 80, 99, 130

[1,2,3,4

- 13 

### C 4 

### ]PFBA IS for PFBA 217 172

[1,2 ,3,4,5

- 13 

### C 5 

### ]PFPe A IS for PFPeA 268 223

[1,2

- 13 

### C 2 

### ]PFHxA IS for PFHxA 315 270

[1,2,3,4

- 13 

### C 4 

### ]PFHpA IS for PFHpA 367 322

[1,2,3,4,5,6,7,8

- 13 

### C 8 

### ]PFOA IS for PFOA 421 376

[1,2,3,4,5,6,7,8,9

- 13 

### C 9 

### ]PFNA IS for PFNA 472 427

[1,2,3,4,5,6

- 13 

### C 6 

### ]PFDA IS for PFDA 519 474

[1,2,3,4,5,6,7

- 13 

### C 7 

### ]PFUnA IS for PFUnA 570 525

[1,2

- 13 

### C 2 

### ]PFDoA IS for PFDoA and PFTA 615 570

[ 18 

### O 2 

### ]PFBS IS for PFBS 303 84

[ 1,2,3

- 13 

### C 3 

### ]PFHS IS for PFHS 402 80

[1,2,3,4

- 13 

### C 4 

### ]PFOS IS for PFOS 503 80

[1,2,3,4,5,6,7,8

- 13 

### C 8 

### ]FOSA IS for FOSA 507 80

[1,2,3,4

- 13 

### C 4 

### ]PFOA Surrogate (C4

- C8 Acids) 417 372

[1,2,3,4

- 13 

### C 4 

### ]PFOS Surrogate(Sulfonates, FOSA) 503 80

[1,2

- 13 

### C 2 

### ]PFUnA Surrogate (C9

- C13 Acids) 565 520



# Multiple transitions for monitoring the analytes is an option. The use of one daughter ion is acceptable if data

sensitivity and selectivity is achieved and provided that retention time criteria are met to assure adequate

specificity. While the daughter ions may be chosen at the discretion of the analyst, mass transition 99 is

suggested for PFOS. Quantitation may be performed using the total ion chromatogram (TIC, or summed

MRMs) for a given analyte. For example, the PFOA TIC would sum all three of the monitored transitions. Use

of the suggested primary ion is recommended. Retention times may vary slightly, on a day

- to

- day basis,

depending on the batch of mobile phase and the gradient, column, guard column(s) used etc. Drift in retention

times is acce ptable within an analytical run, as long as the drift continues through the entire analysis and the

standards are interspersed throughout the analytical run.



# 11.2 Calibration Curve

Quantitation is by internal standard or external standard calibration. Calibration standards may be prepared in

ASTM Type I, HPLC water, other suitable water, or a solvent/water mixture. If internal standard calibration

does not meet calibration acceptance criteria, external calibration can be applied. See Table 1 for

# ETS -8- 044. 1 Page 18 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS; Direct Injection

Analysis recom mended application of available internal standards. Quantitation of PFOA and PFOS is by summed

analyte

- specific mass transitions.

Analyze the standard curve prior to each set of samples. If internal standards were added to the calibration

standards area ratios are used to generate the calibration curve. The standard curve may be plotted using a

linear regression (y = mx + b), weighted 1/x or unweighted, or by quadratic fi t (y = ax 2 

# + bx + c), weighted 1/x or

unweighted

, using suitable software

. The mathematical method used to calculate the calibration curve should

be applied consistently throughout a study. Any change should be thoroughly documented in the raw data.

High and/or low points may be excluded from the calibration curves to provide a better fit over the range

appropriate to the data or because they did not meet the pre

- determined acceptance criteria. Low

- level curve

points should also be excluded if their area counts (or area ratio if quantitating by IS) are not at least twice that

of the ave rage area counts (or area ratio if quantitating by IS) of method and/or solvent blanks. The coefficient

of determination (r 2 

# ) value for the calibration curve must be greater than or equal to 0.990 (or a correlation

coefficient (r) of 0.995). Each point in the curve must be within ± 25% of the theoretical concentration with the

exception of the LLOQ, which may be within ± 30%. Justification for exclusion of calibration curve points will be

noted in the raw data. A minimum of 6 points will be used to construct the calibration curve.

If the calibration curve does not meet acceptance criteria, perform routine maintenance or prepare a new

standard curve (if necessary) and reanalyze.



# 11.3 Continuing Calibration Verification (CCV)

Continuing calibration verifications (CCV) are analyzed to verify the accuracy of the calibration curve. Analyze

a mid

- range calibration standard, one of the same standards used to construct the calibration curve, at a

minimum after every tenth sample, not including solvent blanks, with a minimum of one per sample set.

Calibration verification injections must be within ± 25% to be considered acceptable. The calibration curve and

the last passing CCV will then bracket acceptable samples. Multiple CCV levels may be used. Samples must

be bracketed by passing CCVs or the calibration curve and a passing CCV to be reportable.



# 11.4 System Suitability

A minimum of three system suitability samples should be injected at the beginning of each analytical run, prior

to th e analysis of the calibration curve. Typically these samples are at a concentration near the mid

- level of the

calibration curve and are repeated injections from one autosampler vial. It is suggested that the system

suitability injections have area counts or area ratios when using internal standard calibration, with a target RSD

of ≤5% and a target retention time RSD of ≤2%. There is no defined acceptability limit on these results as the

%RSD value is dependent on the number of MRM transitions being monitored in the LC/MS/MS run or time

period. Ultimately, any effects on these parameters for the System Suitability samples will also be evident on

all standards and QC samples analyzed as part of the analysis batch. Any effect of system suitability is

incorp orated within QC acceptance criteria. 4



# 11.5 Sample Analysis and QCs

For each analysis batch, the instrument analysis run sequence should include an initial calibration curve,

samples, FDSs, interspersed blanks, interspersed CCVs, appropriate QCs (i.e., LCSs, LMSs, FMSs, TBMSs,

and TBs), and a final CCV or calibration curve bracketing samples and appropriate QCs

Inject the same volume (between 5

- 100μL) of each standard, analytical sample and blank into the instrument

(unless an on

- instrument sample dilution is desired).

Samples containing analytes that are quantitated above the concentration of the highest standard in the curve

should be further diluted and reanalyzed.

4 

### 3M Environmental Laboratory study E08

- 0096 evaluated the effect on these results as a function of the number of MRMs being monitored.

# ETS -8- 044. 1 Page 19 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS; Direct Injection

Analysis 

# 12 Data Analysis and Calculations

The chromatography analysis software will typically calculate the amount of target analyte in the sample

extracts using the established calibration curve.

Calculate the percent recovery of the LCS using the following equation:



# 100%*

)

mL ng (ion Concentrat Spike )

mL ng (ion Concentrat LCS

recovery LCS%

=

Calculate the percent recovery of the LMS using the following equation:

100%*

)

mL ng ( ion Concentrat Spike )

mL ng ( Sample of ion Concentrat -)

mL ng ( ion Concentrat LMS

recovery % LMS

=



# For samples fortified with known amounts of analyte prior to extraction, use the following equation to calculate

the percent recovery.



# 100 (ng/mL) added Analyte (ng/mL) sample in found analyte Average- (ng/mL) found analyte Total Recovery × =



# 13 Analysis Batch Method Performance Criteria

Any method performance parameters that are not achieved must be considered in the evaluation of the data.

Nonconformance to any specified parameters must be described and discussed in the final report if the

Technical Manager (non

- GLP study) or Study Director (GLP study) chooses to report the data.

If criteria listed in this method performance section are not met, maintenance may be performed on the system

and samples reanalyzed, or other actions taken as appropriate. Document all actions in the raw data.

If data are to be reported when performance criteria have not been met, the data must be footnoted on tables

and discussed in the text of the report.



# 13.1 System Suitability

– Analysis Batch

A minimum of three system suitability samples should be injected at the beginning of each analytical run.

These samples are run prior to the calibration curve. It is suggested that the system suitability injections have

area counts with a target RSD of ≤5% and a target retention time RSD of ≤2%. There is no defined

acceptability limit on these results as the %RSDs are dependent on the number of MRM transitions being

monitored in the LC/MS/MS run or time period. Any effect of system suitability is incorporated in the QC

acceptance criteria.



# 13.2 Calibration and Limit of Quantitation (LOQ)

– Analysis Batch

Calibration Curve: The coefficient of determination (r 2 

# ) value for the calibration curve must be greater than or

equal to 0 .990 corresponding to a correlation coefficient (r) = 0.995. Each point in the curve must be within

± 25% of the theoretical concentration with the exception of the LLOQ, which may be within ± 30%.

CCV Performance: The calibration standards that are interspersed throughout the analytical sequence are

evaluated as continuing calibration verifications in addition to being part of the calibration curve. The accuracy

# ETS -8- 044. 1 Page 20 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS; Direct Injection

Analysis of each curve point must be within 25% of the theoretical value ( within 30% for lowest curve point ) . Samples

that are bracketed by CCVs not meeting these criteria must be reanalyzed.

Limits of Quantitation (LOQ): The lower LOQ (LLOQ) is the lowest non

- zero active standard in the

calibration curve; the peak area of the LLOQ must be at least 2X that of the average area counts for all

prepared procedural blank(s). By definition, the measured value of the LLOQ must be within 30% of the

theoretical value.

Demonstration of Specificity: Specificity is demonstrated by chromatographic retention time (within 4% of

standard) and the mass spectral response of unique ions.



# 13.3 Blanks

– Method/Procedural Blanks and Trip

Method/ Procedural Blanks: Multiple procedural blanks should be interspersed throughout the analysis batch

and the analytical sequence. At a minimum, method blanks are analyzed prior to instrument calibration, prior

to the analysis of CCV samples, after every 10 sample injections, and at the end of the analytical run.

The mean area counts (or area ratios when using IS calibration) for each analyte must be less than 50% of the

area count of the LOQ standard. If the area counts of the procedural blanks exceed 50% of the LOQ standard,

then the LOQ must be raised to the first standard level that meets criteria.

Trip Blank: A t rip blank of ASTM Type I water (or lab equivalent) is prepared in a sample container in the

laboratory and treated as a sample, including exposure to shipping, sampling site conditions, storage,

preservation and all analytical procedures. The trip blanks results for each analyte are included with the

repo rted sample results.



# 13.4 Data Accuracy and Precision

– Analysis Batch

Lab Control Spikes: The average recovery at each LCS level for each target analyte and appropriate SRS

should be within 80

- 120% and the percent relative standard deviation of the recoveries must be less than or

equal to 20%. If the average recovery of a spiking level falls outside method acceptance, but at least 67% (6

out of 9 ) o f LCS samples are within 20 % of their respective nominal value (33% of the QC samples, not all

replicates at the same concentration, may be outside 20% of nominal value), the average recovery will be

flagged as outside method acceptance criteria. All LCS samples will be control charted as per ETS

- 12

- 012. If

the average recovery of one of the spiking levels exceeded the analytical method uncertainty as determined by

ETS

- 12

- 012, that analytical batch uncertainty will be expanded for that particular study. The average recovery

at each LCS level for mixed branched/linear isomer PFOA and PFOS should be within 70

- 130% and the

percent relative standard deviation of the recoveries must be less than or equal to 20%.

Field Duplicates: The relative percent difference (RPD) of duplicate samples should be less than 20% for the

precision of sample preparation and analysis to be considered in control. Replicate samples not meeting the

20% RPD criteria are flagged and reported as outside of QC acceptance criteria.

Field Matrix Spikes: FMS acceptance criteria are recoveries within ±30% of the expected value for each

target analyte and appropriate SRS. Sample data with FMS recovery outside of ±30% but within ±50% of the

expected value are flagged and reported as outside of QC acceptance criteria. Data with FMS recovery

outside of ±50% of the expected value are reported as NR, where NR is defined as “Not Reportable” data

outside of QC acceptance criteria. If FMS recovery could not be assessed because FMSs were at an

inappropriate level, then Laboratory Matrix Spikes (LMSs) may be substituted. If LMS recoveries are within

±30% for each target analyte and SRSs the data are reportable but flagged as not meeting the FMS method

acceptance criteria.



# 13.5 Analytical Method Uncertainty

Analytical method uncertainty for each target analyte and SRS is determined with control charted historica l

analysis batch LCS data for the method and reported with each analysis batch. 5 

# Uncertainty determinations

5 

## Method uncertainty based on INTERNATIONAL ANS/ISO/IED STANDARD 17025 reference (GUM, Guide to the Expression of

Uncert ainty in Measurement). Method application demonstrated in ETS

- 12

- 012, citing references: a.) EURACHEM/CITAC Guide,

“Quantifying Uncertainty in Analytical Measurement,” Second Edition; Editors: S.L.R. Ellison, M. Rosslein, and A. Williams

. b.)Georgian, Thomas, “Estimation of Laboratory Analytical Uncertainty Using Laboratory Control Samples,” Environmental Testing &

# ETS -8- 044. 1 Page 21 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS; Direct Injection

Analysis are based on INTERNATIONAL ANS/ISO/IED STANDARD 17025 reference (GUM, Guide to the Expression

of Uncertainty in Measurement) and described in ETS -12- 012 . At least thirty data points are required for

determining analytical method uncertainty. The method uncertainty is defined as 2x the standard deviation of

the percent recoveries of the pooled lab control spikes. While all LCS data points are control charted, only the

most recent fifty data points are used for determining the method uncertainty.

When less than thirty LCS data points have been generated for a given analyte, the analysis batch LCSs are

used to determine the data uncertainty. If FMSs meet the ±30% recovery criteria at a level appropriate to the

endogenous level, and the LCS meet the ±20% recovery criteria, then the uncertainty of the data is determined

as within 100±20%.

Analysis batch sample data with FMS recovery outside of ±30% but within ±50% of the expected value are

flagged and reported as outside of QC acceptance criteria with expanded uncertainties. Data with FMS

recovery outside of ±50% of the expected value are reported as NR, where NR is defined as “Not Reportable”

data outside of QC acceptance criteria. If FMS recovery could not be assessed because FMSs were at an

inappropriate level, then Laboratory Matrix Spikes (LMSs) may be substituted. If LMS recoveries are within

±30% for each target analyte and appropriate SRSs the data are reportable but flagged as not meeting the

FMS method acceptance criteria with uncertainties of ±30%. If FMS do not meet the ±30% recovery criteria,

and historical FMS data does not exist, the analytical uncertainty is evaluated on a sample

- by -s ample basis,

the data may be reported with expanded uncertainty and are flagged.



# 13.6 Quantitation of PFOA/PFOS

– Analysis Batch

Calibration standards consisting of mixed branched and linear isomer PFOS/PFOA are preferred.

Quantitation is performed by integrating the linear and branched isomers together. Alternately, the linear and

branched isomers can be integrated separately, applying the appropriate true value to each calibration curve

po int for each isomer. The LCS and samples are then quantitated by integrating the linear and branched

isomers separately (requires separate analytical results files) and quantitating the resulting peak against the

linear or branched calibration curve

. The results from both integrations are then summed to produce the final

result. Integrating the linear and branched isomers separately may be helpful for those samples where the

linear/branched ratios do not closely match those of the reference standards.

Ho wever, for PFOS/PFOA target analytes, if the calibration standards are comprised of predominantly linear

isomers only t he method requires the addition of LCS s of mixed branched/linear isomer PFOS/PFOA. The

purpose of including these LCSs is to demonstrate quantitative equivalency (or quantitative bias) of the

isomeric mix when using a predominantly linear PFOS or PFOA standard for calibration. Alternatively, in lieu

of mixed branched and linear isomer PFOS/PFOA LCSs, mixed branched and linear isomer PFOS/ PFOA

TBMSs may be applied to demonstrate method accuracy and precision.

An alternate method of quantitation can be performed whereby only the linear isomer of PFOS/PFOA is

integrated and used for generating the calibration curve. The LCS and samples are then quantitated by

integrating the linear and branched isomers separately (requires separate analytical results files) and

quantitating the resulting peak against the linear calibration curve. The results from both integrations are then

summed to produce the final result. Integrating the linear and branched isomers separately reduces the on

column concentration for those samples that contain both linear and branched isomers of PFOA/PFOS. This

ensures that the concentration detected is within the a range of the calibration curve that is comparable

regardless of whether the calibration curve was generated using predominantly linear isomers of PFOS/PFOA

or linear plus branched isomers of PFOS/PFOA.



# 14 Pollution Prevention and Waste Management

Waste generated when performing this method will be disposed of appropriately. The original samples will be

archived at the 3M Environmental Laboratory in accordance with internal procedures.

Analysis, November/December 2000. c.)Taylor, B.N. and CE. Kuyatt, NIST Technical Note 1297, 1994 Edition: “Guidelines for

Evaluating and Expressing the Uncertainty of NIST Measurement Results.”d.)Adams, T.M., “A2LA Guide for the Estimation of

Measurement Uncertainty in Testing”, July 2002.

# ETS -8- 044. 1 Page 22 of 22

Method of Analysis for the Determination of Perfluorinated Compounds in Wate r by LC/MS/MS; Direct Injection

Analysis 

# 15 Records

Each data package generated for a study must include all supporting information for reconstruction of the data.

Information for the data package must include, but is not limited to the following items: study or project

number, sample and standard prep sheets/records, instrument run log (instrument batch records, instrument

ac quisition method, summary pages), instrument results files, chromatograms, calibration curves, and data

calculations.



# 16 Affected Documents

None.



# 17 Revisions

Revision

Number Summary of Changes

1 Section 1

. Included the use of internal standard calibration by this method.

Section 2. Included the use of internal standard calibration by this method. Included the

use of a solvent/water mixture when analyzing for PFUnA, PFDoA, PFTrDA, and FOSA.

Section 3

. Added definitions for internal standard, surrogate internal standard, and

surrogate recovery standard.

Section 6.Removed the details regarding the instrument parameters to section 10 of the

method

. Section 7. Updated reference standards to include internal standards and surrogates.

Changed concentration levels for working standards and included the use of internal

standards and surrogates.

Section 8. Inserted a new section on sample bottle preparation.

Se ction 9 Quality Control. This section was previously section 10 in ETS -8- 044.0. Updated

QC criteria to be consistent with method ETS -8- 154.4.

Section 10 Procedures . This section was previously section 8 (Sample Handling) in E TS -8-

044.0. Added detail regarding the preparation of LCSs. Included the use of methanol as a

dilution solvent.

Section 11 Samp le Analysis

. This section was previously section 10 in ETS -8- 044.0.

Included the details regarding the instrument parameters.

Section 12 Data Analysis and Calculations. This section was previously section 11 in ETS

8- 044.0. Removed the equation for calculating the analytes concentration, indicating that

this is done by the instrument software.

Section 13 Method Performance. This section was previously section 12 in ETS -8- 044.0.

Updated QC criteria to be consistent with ETS -8- 154.4. Added information on the

determination of analytical method uncertainty and quantitation of PFOA/PFOS.

Section 14 Pollution Prevention. This section was previously section 13 in ETS -8- 044.0.

Section 15 Records. This section was previously section 14 in ETS -8- 044.0.

Section 16 Affected Documents. This section was previously section 15 in ETS -8- 044.0.

Section 17 Revisions. This section was previously section 16 in ETS

- 8

- 044.0.