### 13339 Hagers Ferry Road Huntersville, NC 28078-7929 McGuire Nuclear Complex - MG03A2 Phone: 980-875-5245 Fax: 980-875-4349



# Order Summary Report 

# Analytical Laboratory

Order Number: J12020345

Lab Contact: Jason C Perkins 

### Date: 3/5/2012 Customer Address: 3195 Pine Hall Rd Customer Name(s): Bill Kennedy, Meloni e Martin, Wayne Chapman, Tom Johnson

Mailcode: Belews Steam Station Belews Creek, NC 28012

Phone: 980-875-5348 Report Authorized By: (Signature)


### Program Comments: Please contact the Program Manager (Jason C Perkins) with any questions regarding this report. Data Flags & Calculations: Any analytical tests or individual analytes within a test flagged with a Qualifier indicate a deviation from the method quality system or quality control requirement. The qualifier description is found at the end of the Cert ificate of Analysis (sample re sults) under the qualifiers heading. All results are repo rted on a dry weight basis unless otherwise noted. Data Package: This data package includes analytical results that are applicable only to the samples described in this narrative. An estimatio n of the uncertainty of measurement for the results in the report is available upon request. This report shall not be reproduced, ex cept in full, without the written consent of t he Analytical Laboratory. Please contact the Analytical laboratory with any questions. The order of individual sections within this report is as follows: Job Summary Report, Sample Identification, Technical Validation of Data Package, A nalytical Laboratory Certificate of Analysis, Analytical Laboratory QC Reports, Sub-contracted Laboratory Results, Customer Specific Data Sheets, Reports & Documentation, Customer Database Entries, Te st Case Narratives, Chain of Custody (COC) Certification: The Analytical Laboratory holds the follow ing State Certifications : North Carolina (DENR) Certificate #248, South Carolina (DHEC) Laboratory ID # 99005. Contact the Analytical Laboratory for definitive information about the certification status of specific methods.

## Sample ID's & Descriptions:
Sample ID Plant/Station Collection Date and Time Collected By Sample Description 2012004159 BELEWS 20-Feb-12 9:00 AM P. GASSETT FGD Purge Eff 2012004160 BELEWS 20-Feb-12 9:00 AM P. GASSETT BIOREACTOR 1 INF. 2012004161 BELEWS 20-Feb-12 9:00 AM P. GASSETT BIOREACTOR 1 INF. BLANK 2012004162 BELEWS 20-Feb-12 9:00 AM P. GASSETT BIOREACTOR 2 EFF. 2012004163 BELEWS 20-Feb-12 9:00 AM P. GASSETT BIOREACTOR 2 EFF. BLANK 2012004164 BELEWS 20-Feb-12 9:00 AM P. GASSETT FILTER BLANK 2012004165 BELEWS 20-Feb-12 9:00 AM P. GASSETT Trip Blank 7 Total Samples

### COC and .pdf report are in agreement with sample totals and analyses (compliance programs and procedures). All Results are less than the laboratory reporting limits. All laboratory QA/QC requirements are acceptable. The Vendor Laboratories have been qualified by the Analytical Laboratory Yes No 

# Technical Validation Review Checklist:

Yes No

Yes No



## Report Sections Included: Job Summary Report Sub-contracted Laboratory Results Sample Identification Customer Specific Data Sheets, Reports, & Documentation Technical Validation of Data Package Customer Database Entries Analytical Laboratory Certificate of Analysis Analytical Laboratory QC Report Chain of Custody

Reviewed By: 

### Mary Ann Ogle Date: 

### 3/5/2012 Yes

Electronic Data Deliverable (EDD) Sent Separately

# Certificate of Laboratory Analysis This re p ort shall not be rep roduced, excep t in full. 

## Order # J12020345

2012004159 Collection Date: 20-Feb-12 9:00 AM 

### Site: FGD Purge Eff Matrix: OTHER

Analyte Analysis Date/Time Method Units Qualifiers RDL Result 

### Sam p le #:

Analyst DF Carbonate, Bicarbonate, and Hydroxide Alkalinity Carbonate (CO3) Complete V_PRISM 1 Bicarbonate (HCO3) Complete V_PRISM 1 Hydroxide (OH) Complete V_PRISM 1 NITRITE + NITRATE (COLORIMETRIC) Nitrite + Nitrate (Colorimetric) 13 mg-N/L EPA 353.2 21-Feb-12 14:21 BGN9034 0.25 25 INORGANIC IONS BY IC Bromide 110 mg/L EPA 300.0 24-Feb-12 20:47 JAHERMA 5 50 Chloride 7400 mg/L EPA 300.0 24-Feb-12 20:47 JAHERMA 100 1000 Sulfate 1300 mg/L EPA 300.0 24-Feb-12 20:47 JAHERMA 100 1000 MERCURY (COLD VAPOR) IN WATER Mercury (Hg) 266 ug/L EPA 245.1 24-Feb-12 09:45 AGIBBS 5 100 Mercury Dissolved (cold vapor) in Water (Filtered) Mercury (Hg) < 2.50 ug/L EPA 245.1 02-Mar-12 10:07 AGIBBS 2.5 50 DISSOLVED METALS BY ICP Manganese (Mn) 6.23 mg/L EPA 200.7 29-Feb-12 10:00 MHH7131 0.05 10 TOTAL RECOVERABLE METALS BY ICP Boron (B) 243 mg/L EPA 200.7 27-Feb-12 11:12 MHH7131 0.5 10 Calcium (Ca) 4490 mg/L EPA 200.7 27-Feb-12 11:12 MHH7131 0.1 10 Iron (Fe) 111 mg/L EPA 200.7 27-Feb-12 11:12 MHH7131 0.1 10 Lithium (Li) 0.144 mg/L EPA 200.7 27-Feb-12 11:12 MHH7131 0.05 10 Magnesium (Mg) 755 mg/L EPA 200.7 27-Feb-12 11:12 MHH7131 0.05 10 Manganese (Mn) 7.28 mg/L EPA 200.7 27-Feb-12 11:12 MHH7131 0.05 10 Potassium (K) 60.4 mg/L EPA 200.7 27-Feb-12 11:12 MHH7131 1 10 Sodium (Na) 43.7 mg/L EPA 200.7 27-Feb-12 11:12 MHH7131 0.5 10 DISSOLVED METALS BY ICP-MS Selenium (Se) 286 ug/L EPA 200.8 22-Feb-12 14:53 MHH7131 10 10

# Certificate of Laboratory Analysis This re p ort shall not be rep roduced, excep t in full. 

## Order # J12020345

2012004159 Collection Date: 20-Feb-12 9:00 AM 

### Site: FGD Purge Eff Matrix: OTHER

Analyte Analysis Date/Time Method Units Qualifiers RDL Result 

### Sam p le #:

Analyst DF TOTAL RECOVERABLE METALS BY ICP-MS Arsenic (As) 165 ug/L EPA 200.8 29-Feb-12 09:51 DJSULL1 10 10 Cadmium (Cd) < 10.0 ug/L EPA 200.8 29-Feb-12 09:51 DJSULL1 10 10 Chromium (Cr) 212 ug/L EPA 200.8 29-Feb-12 09:51 DJSULL1 10 10 Copper (Cu) 93.0 ug/L EPA 200.8 29-Feb-12 09:51 DJSULL1 10 10 Nickel (Ni) 159 ug/L EPA 200.8 29-Feb-12 09:51 DJSULL1 10 10 Selenium (Se) 4980 ug/L EPA 200.8 29-Feb-12 09:51 DJSULL1 10 10 Silver (Ag) < 10.0 ug/L EPA 200.8 29-Feb-12 09:51 DJSULL1 10 10 Zinc (Zn) 186 ug/L EPA 200.8 29-Feb-12 09:51 DJSULL1 10 10 SELENIUM SPECIATION Vendor Parameter complete V_AS&C 1 TOTAL DISSOLVED SOLIDS TDS 20000 mg/L SM2540C 28-Feb-12 16:00 TJA7067 200 1 TOTAL SUSPENDED SOLIDS TSS 3500 mg/L SM2540D 250 1


### 2012004160 Collection Date: 20-Feb-12 9:00 AM 

### Site: BIOREACTOR 1 INF. Matrix: OTHER

Analyte Analysis Date/Time Method Units Qualifiers RDL Result 

### Sam p le #:

Analyst DF Carbonate, Bicarbonate, and Hydroxide Alkalinity Carbonate (CO3) Complete V_PRISM 1 Bicarbonate (HCO3) Complete V_PRISM 1 Hydroxide (OH) Complete V_PRISM 1 NITRITE + NITRATE (COLORIMETRIC) Nitrite + Nitrate (Colorimetric) 15 mg-N/L EPA 353.2 21-Feb-12 14:24 BGN9034 0.25 25 INORGANIC IONS BY IC Bromide 120 mg/L EPA 300.0 24-Feb-12 21:02 JAHERMA 5 50 Chloride 7900 mg/L EPA 300.0 24-Feb-12 21:02 JAHERMA 100 1000 Sulfate 1400 mg/L EPA 300.0 24-Feb-12 21:02 JAHERMA 100 1000 MERCURY 1631 Vendor Parameter Complete V_BRAND 1 MERCURY (COLD VAPOR) IN WATER Mercury (Hg) < 2.50 ug/L EPA 245.1 24-Feb-12 09:48 AGIBBS 2.5 50

# Certificate of Laboratory Analysis This re p ort shall not be rep roduced, excep t in full. 

## Order # J12020345

2012004160 Collection Date: 20-Feb-12 9:00 AM 

### Site: BIOREACTOR 1 INF. Matrix: OTHER

Analyte Analysis Date/Time Method Units Qualifiers RDL Result 

### Sam p le #:

Analyst DF Mercury Dissolved (cold vapor) in Water (Filtered) Mercury (Hg) < 2.50 ug/L EPA 245.1 02-Mar-12 10:09 AGIBBS 2.5 50 DISSOLVED METALS BY ICP Manganese (Mn) 5.80 mg/L EPA 200.7 29-Feb-12 10:03 MHH7131 0.05 10 TOTAL RECOVERABLE METALS BY ICP Boron (B) 257 mg/L EPA 200.7 27-Feb-12 11:16 MHH7131 0.5 10 Calcium (Ca) 3760 mg/L EPA 200.7 27-Feb-12 11:16 MHH7131 0.1 10 Iron (Fe) < 0.100 mg/L EPA 200.7 27-Feb-12 11:16 MHH7131 0.1 10 Lithium (Li) < 0.050 mg/L EPA 200.7 27-Feb-12 11:16 MHH7131 0.05 10 Magnesium (Mg) 807 mg/L EPA 200.7 27-Feb-12 11:16 MHH7131 0.05 10 Manganese (Mn) 6.47 mg/L EPA 200.7 27-Feb-12 11:16 MHH7131 0.05 10 Potassium (K) 24.8 mg/L EPA 200.7 27-Feb-12 11:16 MHH7131 1 10 Sodium (Na) 46.5 mg/L EPA 200.7 27-Feb-12 11:16 MHH7131 0.5 10 DISSOLVED METALS BY ICP-MS Selenium (Se) 112 ug/L EPA 200.8 22-Feb-12 14:57 MHH7131 10 10 TOTAL RECOVERABLE METALS BY ICP-MS Arsenic (As) 21.1 ug/L EPA 200.8 29-Feb-12 09:54 DJSULL1 10 10 Cadmium (Cd) < 10.0 ug/L EPA 200.8 29-Feb-12 09:54 DJSULL1 10 10 Chromium (Cr) < 10.0 ug/L EPA 200.8 29-Feb-12 09:54 DJSULL1 10 10 Copper (Cu) < 10.0 ug/L EPA 200.8 29-Feb-12 09:54 DJSULL1 10 10 Nickel (Ni) 58.7 ug/L EPA 200.8 29-Feb-12 09:54 DJSULL1 10 10 Selenium (Se) 93.0 ug/L EPA 200.8 29-Feb-12 09:54 DJSULL1 10 10 Silver (Ag) < 10.0 ug/L EPA 200.8 29-Feb-12 09:54 DJSULL1 10 10 Zinc (Zn) < 10.0 ug/L EPA 200.8 29-Feb-12 09:54 DJSULL1 10 10 SELENIUM SPECIATION Vendor Parameter complete V_AS&C 1


### 2012004161 Collection Date: 20-Feb-12 9:00 AM 

### Site: BIOREACTOR 1 INF. BLANK Matrix: OTHER

Analyte Analysis Date/Time Method Units Qualifiers RDL Result 

### Sam p le #:

Analyst DF MERCURY 1631 Vendor Parameter Complete V_BRAND 1

# Certificate of Laboratory Analysis This re p ort shall not be rep roduced, excep t in full. 

## Order # J12020345

2012004162 Collection Date: 20-Feb-12 9:00 AM 

### Site: BIOREACTOR 2 EFF. Matrix: OTHER

Analyte Analysis Date/Time Method Units Qualifiers RDL Result 

### Sam p le #:

Analyst DF Carbonate, Bicarbonate, and Hydroxide Alkalinity Carbonate (CO3) Complete V_PRISM 1 Bicarbonate (HCO3) Complete V_PRISM 1 Hydroxide (OH) Complete V_PRISM 1 NITRITE + NITRATE (COLORIMETRIC) Nitrite + Nitrate (Colorimetric) 0.018 mg-N/L EPA 353.2 21-Feb-12 14:25 BGN9034 0.01 1 INORGANIC IONS BY IC Bromide 110 mg/L EPA 300.0 24-Feb-12 21:18 JAHERMA 5 50 Chloride 7500 mg/L EPA 300.0 24-Feb-12 21:18 JAHERMA 100 1000 Sulfate 1500 mg/L EPA 300.0 24-Feb-12 21:18 JAHERMA 100 1000 MERCURY 1631 Vendor Parameter Complete V_BRAND 1 MERCURY (COLD VAPOR) IN WATER Mercury (Hg) < 1.00 ug/L EPA 245.1 24-Feb-12 09:50 AGIBBS 1 20 DISSOLVED METALS BY ICP Manganese (Mn) 5.28 mg/L EPA 200.7 29-Feb-12 10:07 MHH7131 0.05 10 TOTAL RECOVERABLE METALS BY ICP Boron (B) 246 mg/L EPA 200.7 27-Feb-12 11:20 MHH7131 0.5 10 Calcium (Ca) 3580 mg/L EPA 200.7 27-Feb-12 11:20 MHH7131 0.1 10 Iron (Fe) < 0.100 mg/L EPA 200.7 27-Feb-12 11:20 MHH7131 0.1 10 Lithium (Li) < 0.050 mg/L EPA 200.7 27-Feb-12 11:20 MHH7131 0.05 10 Magnesium (Mg) 785 mg/L EPA 200.7 27-Feb-12 11:20 MHH7131 0.05 10 Manganese (Mn) 6.07 mg/L EPA 200.7 27-Feb-12 11:20 MHH7131 0.05 10 Potassium (K) 28.9 mg/L EPA 200.7 27-Feb-12 11:20 MHH7131 1 10 Sodium (Na) 44.2 mg/L EPA 200.7 27-Feb-12 11:20 MHH7131 0.5 10 TOTAL RECOVERABLE METALS BY ICP-MS Arsenic (As) 24.0 ug/L EPA 200.8 29-Feb-12 09:57 DJSULL1 5 5 Cadmium (Cd) < 5.00 ug/L EPA 200.8 29-Feb-12 09:57 DJSULL1 5 5 Chromium (Cr) < 5.00 ug/L EPA 200.8 29-Feb-12 09:57 DJSULL1 5 5 Copper (Cu) < 5.00 ug/L EPA 200.8 29-Feb-12 09:57 DJSULL1 5 5 Nickel (Ni) 8.02 ug/L EPA 200.8 29-Feb-12 09:57 DJSULL1 5 5 Selenium (Se) < 5.00 ug/L EPA 200.8 29-Feb-12 09:57 DJSULL1 5 5 Silver (Ag) < 5.00 ug/L EPA 200.8 29-Feb-12 09:57 DJSULL1 5 5 Zinc (Zn) < 5.00 ug/L EPA 200.8 29-Feb-12 09:57 DJSULL1 5 5 SELENIUM SPECIATION Vendor Parameter complete V_AS&C 1

# Certificate of Laboratory Analysis This re p ort shall not be rep roduced, excep t in full. 

## Order # J12020345

2012004162 Collection Date: 20-Feb-12 9:00 AM 

### Site: BIOREACTOR 2 EFF. Matrix: OTHER

Analyte Analysis Date/Time Method Units Qualifiers RDL Result 

### Sam p le #:

Analyst DF


### 2012004163 Collection Date: 20-Feb-12 9:00 AM 

### Site: BIOREACTOR 2 EFF. BLANK Matrix: OTHER

Analyte Analysis Date/Time Method Units Qualifiers RDL Result 

### Sam p le #:

Analyst DF MERCURY 1631 Vendor Parameter Complete V_BRAND 1


### 2012004164 Collection Date: 20-Feb-12 9:00 AM 

### Site: FILTER BLANK Matrix: OTHER

Analyte Analysis Date/Time Method Units Qualifiers RDL Result 

### Sam p le #:

Analyst DF Mercury Dissolved (cold vapor) in Water (Filtered) Mercury (Hg) < 0.05 ug/L EPA 245.1 02-Mar-12 10:14 AGIBBS 0.05 1 DISSOLVED METALS BY ICP Manganese (Mn) < 0.005 mg/L EPA 200.7 29-Feb-12 09:28 MHH7131 0.005 1 DISSOLVED METALS BY ICP-MS Selenium (Se) < 1.00 ug/L EPA 200.8 22-Feb-12 13:08 MHH7131 1 1


### 2012004165 Collection Date: 20-Feb-12 9:00 AM 

### Site: Trip Blank Matrix: OTHER

Analyte Analysis Date/Time Method Units Qualifiers RDL Result 

### Sam p le #:

Analyst DF TOTAL RECOVERABLE METALS BY ICP Boron (B) < 0.050 mg/L EPA 200.7 27-Feb-12 11:00 MHH7131 0.05 1 Calcium (Ca) < 0.010 mg/L EPA 200.7 27-Feb-12 11:00 MHH7131 0.01 1 Iron (Fe) < 0.010 mg/L EPA 200.7 27-Feb-12 11:00 MHH7131 0.01 1 Lithium (Li) < 0.005 mg/L EPA 200.7 27-Feb-12 11:00 MHH7131 0.005 1 Magnesium (Mg) < 0.005 mg/L EPA 200.7 27-Feb-12 11:00 MHH7131 0.005 1 Manganese (Mn) < 0.005 mg/L EPA 200.7 27-Feb-12 11:00 MHH7131 0.005 1 Potassium (K) < 0.100 mg/L EPA 200.7 27-Feb-12 11:00 MHH7131 0.1 1 Sodium (Na) < 0.050 mg/L EPA 200.7 27-Feb-12 11:00 MHH7131 0.05 1

# Certificate of Laboratory Analysis This re p ort shall not be rep roduced, excep t in full. 

## Order # J12020345

2012004165 Collection Date: 20-Feb-12 9:00 AM 

### Site: Trip Blank Matrix: OTHER

Analyte Analysis Date/Time Method Units Qualifiers RDL Result 

### Sam p le #:

Analyst DF TOTAL RECOVERABLE METALS BY ICP-MS Arsenic (As) < 1.00 ug/L EPA 200.8 29-Feb-12 09:45 DJSULL1 1 1 Cadmium (Cd) < 1.00 ug/L EPA 200.8 29-Feb-12 09:45 DJSULL1 1 1 Chromium (Cr) < 1.00 ug/L EPA 200.8 29-Feb-12 09:45 DJSULL1 1 1 Copper (Cu) < 1.00 ug/L EPA 200.8 29-Feb-12 09:45 DJSULL1 1 1 Nickel (Ni) < 1.00 ug/L EPA 200.8 29-Feb-12 09:45 DJSULL1 1 1 Selenium (Se) < 1.00 ug/L EPA 200.8 29-Feb-12 09:45 DJSULL1 1 1 Silver (Ag) < 1.00 ug/L EPA 200.8 29-Feb-12 09:45 DJSULL1 1 1 Zinc (Zn) < 1.00 ug/L EPA 200.8 29-Feb-12 09:45 DJSULL1 1 1 SELENIUM SPECIATION Vendor Parameter complete V_AS&C 1

## February 2 8 , 201 2

Jay Perkins Duke Energy Analytical Laboratory Mail Code MGO3A2 (Building 7405) 13339 Hagers Ferry Rd. Huntersville, NC 28078 (704) 875-5 245 Project: HAPS/MACT Testing Belews Creek (LIMS # J1 2020 34 5 )

Dear Mr. Perkins, Attached is the report associated with four (4 ) aqueous samples submitted for selenium speciation analysis on February 2 1 , 2012

. The samples were received in a sealed cooler at- 0.3 ° C on February 22 , 2012

. Selenium speciation analysis was performed via ionchromatography inductively coupled plasma collision reaction cell mass spectrometry (IC-ICPCRC -MS). Any issues associated with the analysis are addressed in the following report. If you have any questions, please feel free to contact me at your convenience.
Sincerely,

Russell Gerads Vice President Applied Speciation and Consulting, LLC

## Applied Speciation and Consulting, LLC Report prepared for: Jay Perkins Duke Energy Analytical Laboratory Mail Code MGO3A2 (Building 7405) 13339 Hagers Ferry Rd. Huntersville, NC 28078 Project: HAPS/MACT Testing Belews Creek (LIMS # J12020345 )

February 2 8 , 2012

1. Sample Reception Four (4 ) aqueous samples in125mL HDPE bottles (provided by Applied Speciation and Consulting ) were submitted for selenium speciation analysi s on February 21 , 2012 . The samples were received on February 2 2 , 201 2 in a sealed container at -0.3 °C. The samples were received in a laminar flow clean hood, void of trace metals contamination and ultra-violet radiation, and was designated a discrete sample identifier. An aliquot of each sample was filtered (0.45μm) and each filtrate w as stored in a secure, monitored cryofreezer (maintained at a temperature of -80 o 

## C) until selenium speciation analysis could be performed via ion chromatography inductively coupled plasma collision reaction cell mass spectrometry (IC-ICP- CRC -MS).

2. Sample Preparation All sample preparation is performed in laminar flow clean hoods known to be free from trace metals contamination. All applied water for dilutions and sample preservatives are monitored for contamination to account for any biases associated with the sample results. Selenium Speciation Analysis by IC -ICP-CRC- MS Prior to analysis, an aliquot of each sample was filtered with a syringe filter (0.45μm) and injected directly into a sealed autosampler vial. No further sample preparation was performed as any chemical alteration of a sample may shift the equilibrium of the system, resulting in changes in speciation ratios. 3. Sample Analysis All sample analysis is preceded by a minimum of a five-point calibration curve spanning the entire concentration range of interest. Calibration curves are performed at the beginning of each analytical day. All calibration curves, associated with each species of interest, are

## standardized by linear regression resulting in a response factor. All sample results are

instrument blank corrected to account for any operational biases associated with the analytical platform. Prior to sample analysis, all calibration curves are verified using second source standard s which are identified as initial calibration verification standards (ICV). Ongoing instrument performance is identified by the analysis of continuing calibration verification standards (CCV) and continuing calibration blanks (CCB) at a minimal interval of every ten analytical runs. Selenium Speciation Analysis by IC -ICP-CRC- MS Eachsample for selenium speciation analysis w as analyzed by ion chromatography inductively coupled plasma collision reaction cell mass spectrometry (IC-ICP- CRC -MS) on February 23, 2012. An aliquot of each sample is injected onto an anion exchange column and mobilized by a basic (pH > 7) gradient. The eluting selenium species are then introduced into a radio frequency (RF) plasma where energy-transfer processes cause desolvation, atomization, and ionization. The ions are extracted from the plasma through a differentially-pumped vacuum interface and travel through a pressurized chamber ( CRC ) containing hydrogen gas which preferentially reacts with interfering ions of the same target mass to charge ratios (m/z). A solid-state detector detects ions transmitted through the mass analyzer and the resulting current is processed by a data handling system. Re tention times for each eluting species are compared to known standards for species identification.

4. Analytical Issues The overall analyses went well and no significant analytical issues were encountered. All quality control parameters associated with this sample were within acceptance limits. The estimated method detection limits (eMDLs) for selenite, selenate, and selenocyanate are generated from replicate analyses of the lowest standard in the calibration curve. Not all selenium species are present in preparation blanks; therefore, eMDL calculations based on preparation blanks are artificially biased low. The eMDL for methylseleninic acid and selenomethionine is calculated from the average eMDL of selenite, selenate, and selenocyanate. The calibration does not contain methylseleninic acid or selenomethionine due to impurities in these standards which would bias the results for other selenium species.

## If you have any questions or concerns regarding this report, please feel free to contact m e.

Sincerely,

Russell Gerads Vice President Applied Speciation and Consulting, LLC

### Sample Results
Sample ID Se(IV) Se(VI) SeCN MeSe(IV) SeMe Unknown Se Species (n) FGd Purge Eff 234 53.3 ND (<1.6) ND (<9.4) ND (<9.4) 0 (0) BioReactor 1 Inf 20.4 67.2 ND (<0.39) 4.0 ND (<2.4) 0 (0) BioReactor 2 Eff ND (<2.0) ND (<4.7) ND (<0.39) ND (<2.4) ND (<2.4) 0 (0) Metals Trip Blk ND (<0.079) ND (<0.19) ND (<0.016) ND (<0.094) ND (<0.094) 0 (0) All results reflect the applied dilution and are reported in μg/L ND = Not detected at the applied dilution SeCN = Selenocyanate MeSe(IV) = Methylseleninic acid SeMe = Selenomethionine Unknown Se Species = Total concentration of all unknown Se species observed by IC-ICP-MS n = number of unknown Se species observed Report Generated by: Russell Gerads Applied Speciation and Consulting, LLC Selenium Speciation Results for Duke Energy Project Name: HAPS/MACT Testing Belews Creek

Contact: Jay Perkins
Date: February 28, 2012 LIMS #J12020342

### Quality Control Summary - Preparation Blank Summary

Analyte (μg/L) PBW1 PBW2 PBW3 PBW4 Mean StdDev eMDL* eMDL 10x eMDL 250x eMDL 1000x Se(IV) 0.000 0.000 0.000 0.000 0.000 0.000 0.008 0.079 2.0 7.9 Se(VI) 0.019 0.000 0.000 0.000 0.005 0.009 0.019 0.19 4.7 19 SeCN 0.000 0.000 0.000 0.000 0.000 0.000 0.002 0.016 0.39 1.6 MeSe(IV) 0.000 0.000 0.095 0.000 0.024 0.048 0.009 0.094 2.4 9.4 SeMe 0.000 0.000 0.000 0.000 0.000 0.000 0.009 0.094 2.4 9.4 eMDL = Estimated Method Detection Limit *Please see narrative regarding eMDL calculations
Quality Control Summary - Certified Reference Materials Analyte (μg/L) CRM True Value Result Recovery Se(IV) LCS 9.57 9.38 98.0 Se(VI) LCS 9.48 9.09 95.9 SeCN LCS 8.92 8.54 95.7 MeSe(IV) LCS 6.47 5.79 89.4 SeMe LCS 9.32 8.48 91.0 Applied Speciation and Consulting, LLC Selenium Speciation Results for Duke Energy Project Name: HAPS/MACT Testing Belews Creek

Contact: Jay Perkins LIMS #J12020342 Date: February 28, 2012 Report Generated by: Russell Gerads

### Quality Control Summary - Matrix Duplicates

Analyte (μg/L) Sample ID Rep 1 Rep 2 Mean RPD Se(IV) BioReactor 2 Eff ND (<2.0) ND (<2.0) NC NC Se(VI) BioReactor 2 Eff ND (<4.7) ND (<4.7) NC NC SeCN BioReactor 2 Eff ND (<0.39) ND (<0.39) NC NC MeSe(IV) BioReactor 2 Eff ND (<2.4) ND (<2.4) NC NC SeMe BioReactor 2 Eff ND (<2.4) ND (<2.4) NC NC ND = Not detected at the applied dilution NC = Value was not calculated due to one or more concentrations below the eMDL

Quality Control Summary - Matrix Spike/ Matrix Spike Duplicate Analyte (μg/L) Sample ID Spike Conc MS Result Recovery Spike Conc MSD Result Recovery RPD Se(IV) BioReactor 2 Eff 1390 1592 114.6 1390 1594 114.7 0.1 Se(VI) BioReactor 2 Eff 1261 1279 101.4 1261 1284 101.8 0.4 SeCN BioReactor 2 Eff 1144 812.1 71.0* 1144 821.9 71.9* 1.2 *Low recovery is attributed to matrix induced species conversion Applied Speciation and Consulting, LLC Selenium Speciation Results for Duke Energy Project Name: HAPS/MACT Testing Belews Creek

Contact: Jay Perkins LIMS #J12020342 Date: February 28, 2012 Report Generated by: Russell Gerads

### 3958 6 th 

### Ave NW • Seattle, WA 98107 • T: 206-632-6206 • F: 206-632-6017 • www.brooksrand.com • brl@brooksrand.com 

## February 29, 2012
Duke Energy ATTN: Jay Perkins Scientific Support-Laboratory 13339 Hagers Ferry Road Huntersville NC 28078 jcperkins@duke-energy.com labcustomer@duke-energy.com
RE: Project DUK-HV1201 Client Project: J12020345 Dear Mr. Perkins, On February 22, 2012, Brooks Rand Labs (BRL) received two (2) wastewater samples and two (2) corresponding field blanks. Samples were logged-in for total mercury (Hg) analysis. All samples were received, prepared, analyzed, and stored according to BRL SOPs and EPA methodology. The results were blank-corrected as described in the calculations section of the applicable SOP(s) and may be evaluated using adjusted reporting limits to account for sample aliquot size. Please refer to the Sample Results page for sample-specific detection limits and other details. No qualification of the data was warranted, aside from concentration qualifiers, and all associated quality control sample results met the acceptance criteria. BRL, an accredited laboratory, certifies the reported results of all analyses for which BRL is NELAP accredited meet all NELAP requirements. For more details, see the Report Information page of the report. Please feel free to contact me if you have any questions regarding this report. Sincerely,

Tiffany Stilwater Project Manager tiffany@brooksrand.comProject ID: 

### DUK-HV1201

PM: 

### Tiffany Stilwater Client PM: 

### Jay Perkins Client PO: 

### 141391



### Definition of Data Qualifiers

(Effective 9/23/09) 

### Laboratory Accreditation BRL is accredited by the National Environmental Laboratory Accreditation Program (NELAP) through the State of Florida Department of Health, Bureau of Laboratories (E87982) and is certified to perform many environmental analyses. BRL is also certified by many other states to perform environmental analyses. For a current list of our accreditations /certifications, please visit our website at <http://www.brooksrand.com/default.asp?contentID=586>. Results reported relate only to the samples listed in the report. 

# Report Information

BLK BRL BS CAL CCV CRM D DUP ICV MSD ND NR PS REC RPD RSD SCV SOP 

### method blank Brooks Rand Labs laboratory fortified blank calibration standard continuing calibration verification certified reference material dissolved fraction duplicate initial calibration verification matrix spike duplicate non-detect non-reportable post preparation spike percent recovery relative percent difference relative standard deviation secondary calibration verification standard operating procedure MDL MRL MS



### method detection limit method reporting limit matrix spike

SRM T COC



### standard reference material total recoverable fraction chain of custody record 

### Common Abbreviations

These qualifiers are based on those previously utilized by Brooks Rand, Ltd., those found in the EPA SOW ILM03.0, Exhibit B, Section III, pg. B-18, and the USEPA Laboratory Data Validation Functional Guidelines for Evaluating Inorganic Analyses; USEPA; July 2002. These supersede all previous qualifiers ever employed by BRL. Detected by the instrument, the result is > the MDL but ≤ the MRL. Result is reported and considered an estimate. B E 

### An estimated value due to the presence of interferences. A full explanation is presented in the narrative. H 

### Holding time and/or preservation requirements not met. Result is estimated. J 

### Estimated value. A full explanation is presented in the narrative. J-M 

### Duplicate precision (RPD) for associated QC sample was not within acceptance criteria. Result is estimated. J-N 

### Spike recovery for associated QC sample was not within acceptance criteria. Result is estimated. M 

### Duplicate precision (RPD) was not within acceptance criteria. Result is estimated. N 

### Spike recovery was not within acceptance criteria. Result is estimated. R 

### Rejected, unusable value. A full explanation is presented in the narrative. U 

### Result is ≤ the MDL or client requested reporting limit (CRRL). Result reported as the MDL or CRRL. X 

### Result is not BLK-corrected and is within 10x the absolute value of the highest detectable BLK in the batch. Result is estimated. 

### Field Quality Control Samples Please be notified that certain EPA methods require the collection of field quality control samples of an appropriate type and frequency; failure to do so is considered a deviation from some methods and for compliance purposes should only be done with the approval of regulatory authorities. Please see the specific EPA methods for details regarding required field quality control samples.

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.comProject ID: 

### DUK-HV1201

PM: 

### Tiffany Stilwater Client PM: 

### Jay Perkins Client PO: 

### 141391



# Sample Information

Report Matrix Type Received Sampled Sample Lab ID 

### 1208015-01 BioReactor 1 Inf 02/20/2012 02/22/2012 Sample Water 1208015-02 BioReactor 1 Inf Hg Blk 02/20/2012 02/22/2012 Sample Water 1208015-03 BioReactor 2 Eff 02/20/2012 02/22/2012 Sample Water 1208015-04 BioReactor 2 Inf Hg Blk 02/20/2012 02/22/2012 Sample Water 

# Batch Summary

Analyte Prepared Analyzed Sequence Batch Lab Matrix Method 

### B120297 02/25/2012 02/27/2012 1200129 Water EPA 1631 Hg



# Sample Results

Sample Sequence Result MDL MRL Unit Batch

Qualifier Analyte Fraction

Report Matrix BioReactor 1 Inf, 

### ng/L 426 T 1200129 B120297 Water 40.4 15.2 1208015-01 Hg
BioReactor 1 Inf Hg Blk, 

### ng/L 0.15 T U 1200129 B120297 Water 0.40 0.15 1208015-02 Hg
BioReactor 2 Eff, 

### ng/L 15.3 T 1200129 B120297 Water 1.56 0.58 1208015-03 Hg
BioReactor 2 Inf Hg Blk, 

### ng/L 0.15 T U 1200129 B120297 Water 0.40 0.15 1208015-04 Hg

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.comProject ID: 

### DUK-HV1201

PM: 

### Tiffany Stilwater Client PM: 

### Jay Perkins Client PO: 

### 141391



# Accuracy & Precision Summary

Batch: 

### B120297

Analyte Result Units Native Spike REC & Limits RPD & Limits Method: 

### EPA 1631 Lab Matrix: 

### Water

Sample Certified Reference Material (1209009, NIST 1641d 1000x dilution) B120297-SRM1 

### 15.68 94% ng/L 85-115 Hg 14.73
Matrix Spike (1208004-01) B120297-MS1 

### 3535 114% ng/L 71-125 Hg 4800

758.5
Matrix Spike Duplicate (1208004-01) B120297-MSD1 

### 3535 108% ng/L 71-125 245% Hg 4565

758.5
Matrix Spike (1208004-03) B120297-MS2 

### 139.0 100% ng/L 71-125 Hg 168.1

29.10
Matrix Spike Duplicate (1208004-03) B120297-MSD2 

### 140.2 105% ng/L 71-125 245% Hg 177.0

29.10

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.comProject ID: 

### DUK-HV1201

PM: 

### Tiffany Stilwater Client PM: 

### Jay Perkins Client PO: 

### 141391



# Method Blanks & Reporting Limits

Batch: 

### B120297
Method: 

### EPA 1631 Matrix: 

### Water Analyte: 

### Hg Result Units Sample 

### B120297-BLK1 ng/L 0.11 B120297-BLK2 ng/L 0.04 B120297-BLK3 ng/L 0.04 B120297-BLK4 ng/L 0.04 MDL: 

### 0.15 Average: 

### 0.06 Standard Deviation: 

### 0.04 Limit: 

### 0.10 Limit: 

### 0.50 MRL: 

### 0.40

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.comProject ID: 

### DUK-HV1201

PM: 

### Tiffany Stilwater Client PM: 

### Jay Perkins Client PO: 

### 141391



# Instrument Calibration

Sequence: 

### 1200129 Method: 

### EPA 1631 Instrument: 

### THG-10 Date: 

### 02/27/2012 Total Mercury and Mercury Speciation by CVAFS

Analyte: 

### Hg REC & Limits Result True Value Lab ID Units 

### 1200129-IBL1 4.20 pg of Hg 1200129-IBL2 6.75 pg of Hg 1200129-IBL3 4.72 pg of Hg 1200129-IBL4 5.95 pg of Hg 1200129-CAL1 25.00 24.83 pg of Hg 99% 1200129-CAL2 100.0 90.79 pg of Hg 91% 1200129-CAL3 500.0 481.7 pg of Hg 96% 1200129-CAL4 2500 2727 pg of Hg 109% 1200129-CAL5 10000 10670 pg of Hg 107% 1200129-ICV1 1568 1473 pg of Hg 94% 85-115 1200129-CCB1 4.56 pg of Hg 1200129-CCV1 500.0 512.9 pg of Hg 103% 77-123 1200129-CCV2 500.0 512.7 pg of Hg 103% 77-123 1200129-CCV3 500.0 444.8 pg of Hg 89% 77-123

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.comProject ID: 

### DUK-HV1201

PM: 

### Tiffany Stilwater Client PM: 

### Jay Perkins Client PO: 

### 141391



# Sample Containers

Lab ID: 

### 1208015-01 Report Matrix: 

### Water Sample Type: 

### Sample Received: 

### 02/22/2012 Sample: 

### BioReactor 1 Inf Collected: 

### 02/20/2012 Ship. Cont. pH P-Lot Preservation Lot Size Container Des 

### A 71470160 10 Cooler

none n/a 250mL Bottle FLPE Hg-T

Lab ID: 

### 1208015-02 Report Matrix: 

### Water Sample Type: 

### Sample Received: 

### 02/22/2012 Sample: 

### BioReactor 1 Inf Hg Blk Collected: 

### 02/20/2012 Ship. Cont. pH P-Lot Preservation Lot Size Container Des 

### A 71470160 10 Cooler

none n/a 250mL Bottle FLPE Hg-T

Lab ID: 

### 1208015-03 Report Matrix: 

### Water Sample Type: 

### Sample Received: 

### 02/22/2012 Sample: 

### BioReactor 2 Eff Collected: 

### 02/20/2012 Ship. Cont. pH P-Lot Preservation Lot Size Container Des 

### A 71490150 70 Cooler

none n/a 500mL Bottle FLPE Hg-T

Lab ID: 

### 1208015-04 Report Matrix: 

### Water Sample Type: 

### Sample Received: 

### 02/22/2012 Sample: 

### BioReactor 2 Inf Hg Blk Collected: 

### 02/20/2012 Ship. Cont. pH P-Lot Preservation Lot Size Container Des 

### A 71470160 10 Cooler

none n/a 250mL Bottle FLPE Hg-T



# Shipping Containers

Cooler Tracking No: 

### 472679668573 via FedEx Temperature: 

### 0.4 °C Coolant Type: 

### Ice Description: 

### Cooler Damaged in transit? 

### No Returned to client? 

### No Custody seals present? 

### No Custody seals intact? 

### No COC present? 

### Yes Received: 

### February 22, 2012 9:00

3958 6th Avenue NW Seattle WA 98107 · P(206) 632-6206 · F(206) 632-6017 · brl@brooksrand.com · www.brooksrand.com

### This data package contains the analytical results for the project identified above and includes a Case Narrative, Sample Results and Chain of Custody. Unless otherwise noted, all samples were received in acceptable condition and processed according to the referenced methods. Data qualifiers are flagged individually on each sample. A key reference for the data qualifiers appears at the end of this case narrative. NC Certification No. 402 SC Certification No. 99012 NC Drinking Water Cert No. 37735 

# Case Narrative 02/27/2012

Duke Energy Corporation (04) Jay Perkins 13339 Hagers Ferry Road Huntersville, NC 28078 Project: HAPS/MACT Testing Belews Creek Project No.: J12020345 Lab Submittal Date: 02/21/2012 Prism Work Order: 2020468 VA Certification No. 1287



### Please call if you have any questions relating to this analytical report. Respectfully, PRISM LABORATORIES, INC.



### VP Laboratory Services Reviewed By

Data Qualifiers Key Reference: 

### HT Sample received and analyzed outside of the hold time.

Method Detection Limit MDL Below Reporting Limit BRL
Relative Percent Difference RPD


## * Results reported to the reporting limit. All other results are reported to the MDL with values between MDL and reporting limit indicated with a J.

This report should not be reproduced, except in its entirety, without the written consent of Prism Laboratories, Inc.

449 Springbrook Road - P.O. Box 240543 - Charlotte, NC 28224-0543 Phone: 704/529-6364 - Toll Free Number: 1-800/529-6364 - Fax: 704/525-0409 

### Page 1 of 8

# Sample Receipt Summary 02/27/2012 

### Prism Work Order: 2020468 Client Sample ID Matrix Lab Sample ID Date Sampled Date Received 

### 2020468-01 02/20/12 2012004159/FGD Purge Eff 02/21/12 Water 2020468-02 02/20/12 2012004160/BioReactor 1 Inf 02/21/12 Water 2020468-03 02/20/12 2012004162/BioReactor 2 Eff 02/21/12 Water
Samples received in good condition at 2.5 degrees C unless otherwise noted.

This report should not be reproduced, except in its entirety, without the written consent of Prism Laboratories, Inc.

449 Springbrook Road - P.O. Box 240543 - Charlotte, NC 28224-0543 Phone: 704/529-6364 - Toll Free Number: 1-800/529-6364 - Fax: 704/525-0409 

### Page 2 of 8

# Laboratory Report 02/27/2012

Result Parameter Batch ID Report Limit Analysis Date/Time Method Dilution Factor Units MDL Analyst 

### 13339 Hagers Ferry Road Project No.: Project: HAPS/MACT Testing Belews Creek Duke Energy Corporation (04)

Huntersville, NC 28078 Attn: Jay Perkins Sample Matrix: Water Client Sample ID: Prism Sample ID: 2020468-01 Prism Work Order: 2020468 Time Collected: 02/20/12 09:00 Time Submitted: 02/21/12 15:05 2012004159/FGD Purge Eff


### J12020345

General Chemistry Parameters 6.9 *SM4500-H B P2B0426 1 2/22/12 14:00 JAB HT pH Units pH 64 *SM2320 B P2B0484 1 5.0 2/24/12 11:00 JAB mg/L 0.66 Total Alkalinity BRL *SM2320 B P2B0485 1 5.0 2/24/12 11:00 JAB mg/L 0.66 Carbonate Alkalinity 64 *SM2320 B P2B0487 1 5.0 2/24/12 11:00 JAB mg/L 0.66 Bicarbonate Alkalinity

This report should not be reproduced, except in its entirety, without the written consent of Prism Laboratories, Inc. 449 Springbrook Road - P.O. Box 240543 - Charlotte, NC 28224-0543 Phone: 704/529-6364 - Toll Free Number: 1-800/529-6364 - Fax: 704/525-0409 

### Page 3 of 8

# Laboratory Report 02/27/2012

Result Parameter Batch ID Report Limit Analysis Date/Time Method Dilution Factor Units MDL Analyst 

### 13339 Hagers Ferry Road Project No.: Project: HAPS/MACT Testing Belews Creek Duke Energy Corporation (04)

Huntersville, NC 28078 Attn: Jay Perkins Sample Matrix: Water Client Sample ID: Prism Sample ID: 2020468-02 Prism Work Order: 2020468 Time Collected: 02/20/12 09:00 Time Submitted: 02/21/12 15:05 2012004160/BioReactor 1 Inf


### J12020345

General Chemistry Parameters 7.0 *SM4500-H B P2B0426 1 2/22/12 14:00 JAB HT pH Units pH 56 *SM2320 B P2B0484 1 5.0 2/24/12 11:00 JAB mg/L 0.66 Total Alkalinity BRL *SM2320 B P2B0485 1 5.0 2/24/12 11:00 JAB mg/L 0.66 Carbonate Alkalinity 56 *SM2320 B P2B0487 1 5.0 2/24/12 11:00 JAB mg/L 0.66 Bicarbonate Alkalinity

This report should not be reproduced, except in its entirety, without the written consent of Prism Laboratories, Inc. 449 Springbrook Road - P.O. Box 240543 - Charlotte, NC 28224-0543 Phone: 704/529-6364 - Toll Free Number: 1-800/529-6364 - Fax: 704/525-0409 

### Page 4 of 8

# Laboratory Report 02/27/2012

Result Parameter Batch ID Report Limit Analysis Date/Time Method Dilution Factor Units MDL Analyst 

### 13339 Hagers Ferry Road Project No.: Project: HAPS/MACT Testing Belews Creek Duke Energy Corporation (04)

Huntersville, NC 28078 Attn: Jay Perkins Sample Matrix: Water Client Sample ID: Prism Sample ID: 2020468-03 Prism Work Order: 2020468 Time Collected: 02/20/12 09:00 Time Submitted: 02/21/12 15:05 2012004162/BioReactor 2 Eff


### J12020345

General Chemistry Parameters 6.9 *SM4500-H B P2B0426 1 2/22/12 14:00 JAB HT pH Units pH 130 *SM2320 B P2B0484 1 5.0 2/24/12 11:00 JAB mg/L 0.66 Total Alkalinity BRL *SM2320 B P2B0485 1 5.0 2/24/12 11:00 JAB mg/L 0.66 Carbonate Alkalinity 130 *SM2320 B P2B0487 1 5.0 2/24/12 11:00 JAB mg/L 0.66 Bicarbonate Alkalinity

This report should not be reproduced, except in its entirety, without the written consent of Prism Laboratories, Inc. 449 Springbrook Road - P.O. Box 240543 - Charlotte, NC 28224-0543 Phone: 704/529-6364 - Toll Free Number: 1-800/529-6364 - Fax: 704/525-0409 

### Page 5 of 82/27/12 

### Level II QC Report

13339 Hagers Ferry Road Project No: J12020345 Project: HAPS/MACT Testing Belews Creek Duke Energy Corporation (04)

Huntersville, NC 28078 Attn: Jay Perkins Prism Work Order: 2020468 Time Submitted: 2/21/2012 3:05:00PM

Result Limit

Reporting Units Level Spike Result

Source %REC %REC Limits RPD RPD Limit Notes Analyte General Chemistry Parameters - Quality Control

Batch P2B0426 - NO PREP LCS (P2B0426-BS1) Prepared & Analyzed: 02/22/12 pH pH Units 6.85 6.860 99-101 100 Batch P2B0484 - NO PREP Blank (P2B0484-BLK1) Prepared & Analyzed: 02/24/12 Total Alkalinity mg/L BRL 5.0 LCS (P2B0484-BS1) Prepared & Analyzed: 02/24/12 Total Alkalinity mg/L 260 5.0 250.0 90-110 104 LCS Dup (P2B0484-BSD1) Prepared & Analyzed: 02/24/12 Total Alkalinity mg/L 259 5.0 250.0 200 90-110 103 0.4 Batch P2B0485 - NO PREP Blank (P2B0485-BLK1) Prepared & Analyzed: 02/24/12 Carbonate Alkalinity mg/L BRL 5.0 LCS (P2B0485-BS1) Prepared & Analyzed: 02/24/12 Carbonate Alkalinity mg/L 260 5.0 90-110 LCS Dup (P2B0485-BSD1) Prepared & Analyzed: 02/24/12 Carbonate Alkalinity mg/L 259 5.0 200 90-110 0.4 Batch P2B0487 - NO PREP Blank (P2B0487-BLK1) Prepared & Analyzed: 02/24/12 Bicarbonate Alkalinity mg/L BRL 5.0

This report should not be reproduced, except in its entirety, without the written consent of Prism Laboratories, Inc.

449 Springbrook Road - P.O. Box 240543 - Charlotte, NC 28224-0543 Phone: 704/529-6364 - Toll Free Number: 1-800/529-6364 - Fax: 704/525-0409 

### Page 6 of 82/27/12 

### Level II QC Report

13339 Hagers Ferry Road Project No: J12020345 Project: HAPS/MACT Testing Belews Creek Duke Energy Corporation (04)

Huntersville, NC 28078 Attn: Jay Perkins Prism Work Order: 2020468 Time Submitted: 2/21/2012 3:05:00PM

Result Limit

Reporting Units Level Spike Result

Source %REC %REC Limits RPD RPD Limit Notes Analyte General Chemistry Parameters - Quality Control

Batch P2B0487 - NO PREP LCS (P2B0487-BS1) Prepared & Analyzed: 02/24/12 Bicarbonate Alkalinity mg/L 260 5.0 250.0 90-110 104 LCS Dup (P2B0487-BSD1) Prepared & Analyzed: 02/24/12 Bicarbonate Alkalinity mg/L 259 5.0 250.0 200 90-110 103 0.4

This report should not be reproduced, except in its entirety, without the written consent of Prism Laboratories, Inc.

449 Springbrook Road - P.O. Box 240543 - Charlotte, NC 28224-0543 Phone: 704/529-6364 - Toll Free Number: 1-800/529-6364 - Fax: 704/525-0409 

### Page 7 of 8Page 8 of 8