# Inogen Inogen One G-3 Report No. INOG0018

Report Prepared By Northwest EMC Inc.



# NORTHWEST EMC – (888) 364-2378 – www.nwemc.com

California

–Minnesota

–Oregon

–New York

–Washington

# 22975 NW Evergreen Parkway Suite 400 Hillsboro, Oregon 97124



# Certificate of Test Last Date of Test: September 29, 2011 Inogen Model: Inogen One G-3 Emissions Test Description Specification Test Method Pass/Fail Radiated Emissions RTCA DO-160F: 2007, Category M RTCA DO-160F:2007 Pass Conducted Emissions RTCA DO-160F:2007, Category M RTCA DO-160F:2007 Pass



# Deviations From Test Standards None



# Approved By:

Victor Ratinoff, Operations Manager

This report must not be used to claim product certification, ap proval, or endorsement by NVLAP, NIST, or any agency of the federal government of the United States of America. Product compliance is the responsibility of the client, therefore the tests and equipment modes of operation represented in this report were agreed upon by the client, prior to testing. This Report may only be duplicated in its entirety. The results of this test pertain only to the sample(s) tested. The specific description is noted in each of the individual sections of the tes t report supporting this certificate of test.

# Revision History Revision 09/01/11



# Revision Number Description Date Page Number

00 None

# Accreditations and Authorizations Revision 09/01/11



# FCC

Accredited by NVLAP for performance of FCC radio, digital, and IS M device testing. Our Open Area Test Sites, certification chambers, and conducted measurement facilities have been fully descri bed in reports filed with the FCC and accepted by the FCC in letters maintained in our files. Northwest EMC has been accredit ed by ANSI to ISO / IEC Guide 65 as a product certifier. We have been designated by the FCC as a Telecommunications Certificat ion Body (TCB). This allows Northwest EMC to certify transmitters to FCC specifications in accordance with 47 CFR 2.960 and 2.962.



# NVLAP

Northwest EMC, Inc. is accredited under the National Voluntary Laboratory Accreditation Program (NVLAP) for satisfactory compliance with the requirements of ISO/IEC 17025 for Testing L aboratories. NVLAP is administered by the National Institute of Standards and Technology (NIST), an agency of the U.S. Co mmerce Department. The NVLAP accreditation encompasses Electromagnetic Compatibility Testing in accordance with t he European Union EMC Directive 2004/108/EC, and ANSI C63.4. Additionally, Northwest EMC is accredited by NVLAP to perform radio testing in accordance with the European Union R&TTE Directive 1999/5/EEC, the requirements of FCC, and the RSS radio standards for Industry Canada.


# Industry Canada

Accredited by NVLAP for performance of Industry Canada RSS and ICES testing. Our Open Area Test Sites and certification chambers comply with RSS-Gen, Issue 2 and have been filed with Industry Canada and accepted. Northwest EMC has been accredited by ANSI to ISO / IEC Guide 65 as a product certif ier. We have been designated by NIST and recognized by Industry Canada as a Certification Body (CB) per the APEC Mutual Recogniti on Arrangement (MRA). This allows Northwest EMC to certify transmitters to Industry Canada technical requirements. ( Site Filing Numbers - Hillsboro: 2834D-1, 2834D-2, Sultan: 2834C-1, Irvine: 2834B-1, 2834B-2, Brooklyn Park: 2834E-1)



# CAB

Designated by NIST and validated by the European Commission as a Conformity Assessment Body (CAB) to conduct tests and approve products to the EMC directive and trans mitters to the R&TTE directive, as descr ibed in the U.S. - EU Mutual Recognition Agreement.



# Australia/New Zealand

The National Association of Testing Authorities (NATA), Aust ralia has been appointed by the ACA as an accreditation body to accredit test laboratories and competent bodies for EMC standards. Accredited test reports or assessments by competent bodies must carry the NATA logo. Test reports made by an overseas l aboratory that has been accredited for the relevant standards by an overseas accreditation body that has a Mut ual Recognition Agreement (MRA) with NATA are also accepted as technical grounds for product conformity. The report should be endorsed with the respective logo of the accreditation body (NVLAP).

# Accreditations and Authorizations Revision 09/01/11



# VCCI

Accepted as an Associate Member to the VCCI, Acceptance No . 564. Conducted and radiated measurement facilities have been registered in accordance with Regulations fo r Voluntary Control Measures, Article 8. (Registration Numbers. - Hillsboro: C-1071, R1025, G-84, C-2687, T-1658, and R-2318, Irvine: R-1943, G-85, C-2766, and T-1659, Sultan: R-871, G-83, C-3265, and T-1511, Brooklyn Park: R-3125, G-86, G-141, C-3464, and T-1634).



# BSMI

Northwest EMC has been designated by NIST and validated by C-Tai pei (BSMI) as a CAB to conduct tests as described in the APEC Mutual Recognition Agreement (US0017).



# GOST

Northwest EMC, Inc. has been assessed and accredited by the Russian Certification bodies Certinform VNIINMASH, CERTINFO, SAMTES, and Federal CHEC, to perform EMC and Hygienic testing for Information Technology Prod ucts. As a result of their laboratory assessment, they will accept test results fr om Northwest EMC, Inc. for product certification



# KCC

Northwest EMC, Inc is a CAB designated by MRA partners and recognized by Korea. ( Assigned Lab Numbers: Hillsboro: US0017, Irvine: US0158, Sultan: US0157, Brooklyn Park: US0175)



# VIETNAM

Vietnam MIC has approved Northwest EMC as an accredited test lab. Per Decision No. 194/QD-QLCL (dated December 15, 2009), Northwest EMC test reports can be used for Vietnam approval submissions.



# SCOPE For details on the Scopes of our Accreditations, please visit: http://www.nwemc.com/accreditations/

# Explanation of NWEMC Performance Criteria Revision 09/01/11



# How important is it to unde rstand performance criteria?

It is the responsibility of the test la boratory to observe the results of the tests thatare performed and to accurately report those results. As the responsibl e party (manufacturer, importer, etc) it is your responsibility to take those results, compare them against the specifications and standards, then, if appropriate make a declaration of conformity. As the responsible party it makes sense that you are fully aware of the requirements, how your device perf orms when tested to those requirements, and what information is being used to declare conformity.
To better assist you in making those conformity dec isions, Northwest EMC has adopted a very simple, yet very clear performance assessment procedure. The following criteria is used when performing immunity or susceptibility tests:


# Performance Criteria 1:

 

# The EUT exhibited no change in performance when o perating as specified by the manufacturer. In this case no changes were observed during the test.
 

# In most cases this would be equivalent to Perf ormance Criteria A. When operating the equipment in the modes or configurations specified by t he responsible party, monitoring the parameters specified, no changes were observed. Basically nothing happened.



# Performance Criteria 2:

 

# The EUT exhibited a change in performance when op erating as specified by the manufacturer. In this case the equipment recovered without any oper ator intervention, once the test signal was removed. The data sheets will detail the exact phenomena observed.
 

# In most cases this would be equivalent to Perf ormance Criteria B. When operating the equipment in the modes or configurations specified by t he responsible party, monitoring the parameters specified, changes were observed. The EUT was able to recover from those changes without any operator intervention, once the test signal was removed.



# Performance Criteria 3:

 

# The EUT exhibited a change in performance when op erating as specified by the manufacturer. In this case the equipment required some operator intervention in order to recover. This intervention may be in the form of changing EUT settings, or even resetting the system. The data sheets will detail the exact phenomena observed.
 

# In most cases this would be equivalent to Performance Criteria C. When operating the equipment in the modes or configurations spec ified by the responsible party, monitoring the parameters specified, changes were observed . The EUT required some sort of operator intervention to recover. There was no permanent damage and the EUT appeared to function normally after completion of test.



# Performance Criteria 4:

 

# The EUT exhibited a change in performance when op erating as specified by the manufacturer. In this case the equipment was damaged and would notrecover. The data sheets will detail the exact phenomena observed.
 

# In most cases there is no specific criterion to co mpare this to; it typically ends the test. When operating the equipment in the modes or configur ations specified by the responsible party, monitoring the parameters specified, changes we re observed. There was no recovery; the equipment would no longer function as intended.

# Locations Revision 09/01/11

Oregon Labs EV01-EV12 22975 NW Evergreen Pkwy Suite 400 Hillsboro, OR 97124 (503) 844-4066 California Labs OC01-OC13 41 Tesla Irvine, CA 92618 (949) 861-8918 Minnesota Labs MN01-MN08 9349 W Broadway Ave. Brooklyn Park, MN 55445 (763) 425-2281 Washington Labs SU01-SU07 14128 339 th 

### Ave. SE Sultan, WA 98294 (360) 793-8675 New York Labs WA01-WA04 4939 Jordan Rd. Elbridge, NY 13060 (315) 685-0796

# Product Description Revision 09/01/11



# Client and Equipment Unde r Test (EUT) Information

Company Name: Inogen Address: 326 Bollay Drive City, State, Zip: Goleta, CA 93117 Test Requested By: Mike Pollack Model: Inogen One G-3 First Date of Test: September 29, 2011 Last Date of Test: September 29, 2011 Receipt Date of Samples: September 29, 2011 Equipment Design Stage: Prototype Equipment Condition: No Damage 

# Information Provided by the Party Requesting the Test

Functional Description of the EUT (Equipment Under Test): Portable Oxygen Concentrator
Testing Objective: Provide the specific EMC testing and any asso ciated documentation requested by the client.

EUT Photo

# Product Description Revision 09/01/11

# Configurations Revision 09/01/11



# Configuration 1 INOG0018

EUT Description Manufacturer Model/Part Number Serial Number Oxygen Concentrator Inogen I nogen One G-3 Prototype 1 DC Power Adapter Digital Power ADP90-119 ADP901195210A0010592

Cables Cable Type Shield Length (m) Ferrite Connection 1 Connection 2 DC Out Cable No 1.8m Yes DC Power Adapter Inogen 1G-3 DC In Cable No .9m No DC Power Adapter DC Battery PA = Cable is permanently attached to the device. Shielding and/or presence of ferrite may be unknown. 

# Configuration 2 INOG0018

EUT Description Manufacturer Model/Part Number Serial Number Oxygen Concentrator Inogen I nogen One G-3 Prototype 1

# Modifications Revision 09/01/11



# Equipment Modifications

Item Date Test Modification Note Disposition of EUT

1 9/29/2011 Radiated Emissions Tested as delivered to Test Station. No EMI suppression devices were added or modified during this test. EUT remained at Northwest EMC following the test.

2 9/29/2011 Conducted Emissions Tested as delivered to Test Station. No EMI suppression devices were added or modified during this test. Scheduled testing was completed.

## descri p tion

MODES OF OPERATION

POWER SETTINGS INVESTIGATED

CONFIGURATIONS INVESTIGATED

FREQUENCY RANGE INVESTIGATED Start Frequency 2 MHz Stop Frequency 6 GHz

SAMPLE CALCULATIONS
Radiated Emissions: Field Strength = Measured Level + Antenna Factor + Cable Factor - Amplifier Gain + Distance Adjustment Factor + External Attenuation



## TEST EQUIPMENT ID Interval CAA 0 CAC 0 LIJ 12 LIL 12 ALA 12 ABJ 12 ACC 12 AIR 24 OCR 12 OCT 12 AOU 12 AFG 12

MEASUREMENT BANDWIDTHS

MEASUREMENT UNCERTAINTY

TEST DESCRIPTION BWI (kHz) 10uF Feedthrough Capacitor Solar 6512-106R NCR 10uF Feedthrough Capacitor Solar 6512-106R NCR

1.0 10.0 100.0 1000.0 Autopulse 17 BPM Setting 4

110VAC/60Hz Battery

Description Manufacturer Model Last Cal. INOG0018 - 1

LISN Fischer Custom Communications FCC-LISN-DO-160 7/27/2011 LISN Fischer Custom Communications FCC LISN DO-160 7/27/2011 Antenna, Log Periodic EMCO 3146 8/8/2011 Antenna, Bicon EMCO 3104C 8/8/2011 Antenna, Active Rod AH Systems SAS-200/550-1B 1/27/2011 Antenna, Horn (DRG) ETS Lindgren 3115 5/26/2011 OC11 Cables N/A 1-8GHz RE Cables 4/8/2011 OC11 Cables N/A Disturbance Power Cables 8/25/2011 Pre-Amplifier Miteq AM-1551 3/4/2011 Spectrum Analyzer Agilent E4440A 4/28/2011 PSA 2011.09.01 

# RADIATED EMISSIONS

Testing was performed using the mode(s) of operation and configuration(s) noted within the report. The individuals and/or the organization requesting the test provided the modes, configurations and settings used to complete the evaluation. The actual test parameters are specified in the test data, this includes items such as investigated frequency range (scanned) and test levels. The testing methods and performance specifications, as well as the test site used for the evaluation are indicated in the test data. The test data represents the configuration / operating mode/ model that produced the highest emission levels as compared to the specification limit.

400.0 - 1000.0

A measurement uncertainty estimation has been performed for each test per our internal quality document WP 342. The estimation is used to compare the measured result with its "true" or theoretically correct value. The expanded measurement uncertainty for radiated emissions measurements is less than +/- 4 dB, and for conducted emissions measurements is less than +/

- 2.7 dB. Our measurement data meets or exceeds the measurement uncertainty requirements of CISPR 16-4; therefore, the test data can be compared directly to the specification limit to determine compliance. The calculations for measurement uncertainty are available upon request. 1000.0 - 6000.0 Frequency Range (MHz) 0.15 - 30.0 30.0 - 400.0

Using the mode of operation and configuration noted within this report, a radiated emissions test was performed. The frequency range tested, is also noted in this report. Radiated emissions measurements were made with the test setup as defined in RTCA DO160 E. A calibrated antenna was positioned at the specified distance and height from the periphery of the EUT. Where polarized antennas are used, emissions were measured with both polarities of the antenna. Sufficient number of antenna locations are used to measure the emissions from the entire test setup.

### scan

EUT: Ambien t Work Order: INOG0018 Serial Number: None Date: Customer: Inogen Temperature: 23C Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb Tested by: Jeremiah Darden Power: 110VAC/60H z Job Site: OC10 

### TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS Antenna Height(s) (m) 1 Test Distance (m) 1 

### COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDAR D

Run # Configuration # Results Signature

Freq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

26.325 8.2 0.0 1.0 0.0 0.8 0.0 V 0.0 9.0 35.4 -26.3 26.534 7.6 0.0 1.0 0.0 0.8 0.0 V 0.0 8.4 35.4 -27.0 26.612 7.3 0.0 1.0 0.0 0.9 0.0 V 0.0 8.2 35.5 -27.3 26.395 6.7 0.0 1.0 0.0 0.8 0.0 V 0.0 7.5 35.4 -27.8 26.893 6.8 0.0 1.0 0.0 0.9 0.0 V 0.0 7.7 35.5 -27.9 26.405 6.6 0.0 1.0 0.0 0.8 0.0 V 0.0 7.4 35.4 -28.0 25.688 6.4 0.0 1.0 0.0 0.8 0.0 V 0.0 7.2 35.2 -28.0 26.683 6.6 0.0 1.0 0.0 0.9 0.0 V 0.0 7.5 35.5 -28.0 26.543 6.4 0.0 1.0 0.0 0.8 0.0 V 0.0 7.2 35.4 -28.2 28.679 6.8 0.0 1.0 0.0 0.9 0.0 V 0.0 7.7 36.0 -28.3 26.753 6.2 0.0 1.0 0.0 0.9 0.0 V 0.0 7.1 35.5 -28.4 26.466 6.0 0.0 1.0 0.0 0.8 0.0 V 0.0 6.8 35.4 -28.6 27.005 6.0 0.0 1.0 0.0 0.9 0.0 V 0.0 6.9 35.6 -28.7 RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007 09/29/11 PSA 2011.09.01
EMI 2009.8.29 

# RADIATED EMISSIONS

1 Pass

2 No deviations. None

None



### -10 0 10 20 30 40 50 60 70 80

1 10 100 

## MHzdBuV/mFreq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

26.044 5.7 0.0 1.0 0.0 0.8 0.0 V 0.0 6.5 35.3 -28.8 26.604 5.4 0.0 1.0 0.0 0.9 0.0 V 0.0 6.3 35.4 -29.2 26.899 5.4 0.0 1.0 0.0 0.9 0.0 V 0.0 6.3 35.5 -29.3 26.744 5.3 0.0 1.0 0.0 0.9 0.0 V 0.0 6.2 35.5 -29.3 27.679 5.5 0.0 1.0 0.0 0.9 0.0 V 0.0 6.4 35.7 -29.3 26.386 5.2 0.0 1.0 0.0 0.8 0.0 V 0.0 6.0 35.4 -29.3

### scan

EUT: Ambien t Work Order: INOG0018 Serial Number: None Date: Customer: Inogen Temperature: 23C Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb Tested by: Jeremiah Darden Power: 110VAC/60H z Job Site: OC11 

### TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS Antenna Height(s) (m) 1 Test Distance (m) 1 

### COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDAR D

Run # Configuration # Results Signature

Freq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

120.303 28.0 41.7 1.0 15.2 1.9 0.0 V 0.0 3.4 34.0 -30.6 111.024 27.8 41.7 1.0 14.4 1.9 0.0 H 0.0 2.4 34.0 -31.6 116.091 27.1 41.6 1.0 14.9 1.9 0.0 H 0.0 2.3 34.0 -31.7 115.493 26.8 41.6 1.0 14.8 1.9 0.0 H 0.0 1.9 34.0 -32.1 119.888 26.3 41.7 1.0 15.2 1.9 0.0 V 0.0 1.8 34.0 -32.2 124.356 26.8 41.7 1.0 14.7 2.0 0.0 V 0.0 1.8 34.0 -32.2 118.704 26.3 41.6 1.0 15.1 1.9 0.0 V 0.0 1.7 34.0 -32.3 125.980 26.9 41.7 1.0 14.4 2.0 0.0 H 0.0 1.6 34.0 -32.4 121.157 26.2 41.7 1.0 15.1 1.9 0.0 H 0.0 1.5 34.0 -32.5 119.070 26.0 41.6 1.0 15.1 1.9 0.0 V 0.0 1.4 34.0 -32.6 123.111 26.2 41.7 1.0 14.8 2.0 0.0 V 0.0 1.3 34.0 -32.7 121.743 26.0 41.7 1.0 15.0 1.9 0.0 H 0.0 1.3 34.0 -32.7 115.627 26.1 41.6 1.0 14.9 1.9 0.0 V 0.0 1.2 34.0 -32.8 RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007 09/28/11 PSA 2011.09.01
EMI 2009.8.29 

# RADIATED EMISSIONS

1 Pass 38 No deviations. None

None



### -10 0 10 20 30 40 50 60 70 80

10 100 1000 

## MHzdBuV/mFreq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

129.142 26.9 41.6 1.0 13.9 2.0 0.0 V 0.0 1.2 34.0 -32.8 114.150 26.1 41.6 1.0 14.7 1.9 0.0 H 0.0 1.1 34.0 -32.9 112.514 26.2 41.6 1.0 14.6 1.9 0.0 V 0.0 1.0 34.0 -33.0 112.892 26.1 41.6 1.0 14.6 1.9 0.0 V 0.0 1.0 34.0 -33.0 114.662 25.9 41.6 1.0 14.8 1.9 0.0 V 0.0 1.0 34.0 -33.0 114.626 25.9 41.6 1.0 14.8 1.9 0.0 V 0.0 1.0 34.0 -33.0

### scan

EUT: Ambien t Work Order: INOG0018 Serial Number: None Date: Customer: Inogen Temperature: 23C Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb Tested by: Jeremiah Darden Power: 110VAC/60H z Job Site: OC11 

### TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS Antenna Height(s) (m) 1 Test Distance (m) 1 

### COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDAR D

Run # Configuration # Results Signature

Freq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

974.289 39.4 42.1 1.0 24.6 6.2 0.0 H 0.0 28.0 60.7 -32.6 982.566 38.2 42.0 1.0 24.7 6.2 0.0 V 0.0 27.0 60.7 -33.7 991.869 38.0 42.0 1.0 24.8 6.2 0.0 H 0.0 27.0 60.8 -33.8 968.649 38.3 42.1 1.0 24.4 6.1 0.0 H 0.0 26.7 60.6 -34.0 981.248 37.9 42.0 1.0 24.7 6.2 0.0 V 0.0 26.7 60.7 -34.0 984.471 37.8 42.0 1.0 24.7 6.2 0.0 H 0.0 26.7 60.8 -34.1 986.522 37.7 42.0 1.0 24.7 6.2 0.0 V 0.0 26.6 60.8 -34.2 986.009 37.7 42.0 1.0 24.7 6.2 0.0 H 0.0 26.6 60.8 -34.2 983.665 37.7 42.0 1.0 24.7 6.2 0.0 H 0.0 26.6 60.8 -34.2 990.844 37.5 42.0 1.0 24.8 6.2 0.0 H 0.0 26.5 60.8 -34.3 944.329 38.7 42.3 1.0 23.7 6.1 0.0 H 0.0 26.2 60.5 -34.3 975.241 37.7 42.1 1.0 24.6 6.2 0.0 V 0.0 26.4 60.7 -34.3 942.058 38.6 42.3 1.0 23.7 6.1 0.0 H 0.0 26.0 60.5 -34.4 RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007 09/29/11 PSA 2011.09.01
EMI 2009.8.29 

# RADIATED EMISSIONS

1 Pass 25 No deviations. None

None



### -10 0 10 20 30 40 50 60 70 80

100 1000 

## MHzdBuV/mFreq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

884.703 38.6 42.3 1.0 23.6 5.7 0.0 H 0.0 25.6 60.0 -34.4 903.968 38.9 42.4 1.0 23.3 5.9 0.0 H 0.0 25.7 60.2 -34.5 976.486 37.5 42.1 1.0 24.6 6.2 0.0 V 0.0 26.2 60.7 -34.5 963.667 37.9 42.2 1.0 24.2 6.1 0.0 H 0.0 26.1 60.6 -34.5 853.498 38.6 42.3 1.0 23.4 5.6 0.0 V 0.0 25.2 59.8 -34.6 878.257 38.3 42.3 1.0 23.7 5.7 0.0 H 0.0 25.4 60.0 -34.6

### scan

EUT: Ambien t Work Order: INOG0018 Serial Number: None Date: Customer: Inogen Temperature: 23C Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb Tested by: Jeremiah Darden Power: 110VAC/60H z Job Site: OC11 

### TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS Antenna Height(s) (m) 1 Test Distance (m) 1 

### COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDAR D

Run # Configuration # Results Signature

Freq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

1079.355 41.6 37.6 1.0 24.0 5.0 0.0 V 0.0 32.9 50.0 -17.1 1094.616 41.3 37.6 1.0 24.1 5.0 0.0 H 0.0 32.7 50.0 -17.3 1083.018 41.4 37.6 1.0 24.0 5.0 0.0 H 0.0 32.7 50.0 -17.3 1033.573 41.5 37.7 1.0 23.6 4.9 0.0 V 0.0 32.3 50.0 -17.7 1092.174 40.5 37.6 1.0 24.0 5.0 0.0 V 0.0 31.9 50.0 -18.1 1045.782 40.9 37.7 1.0 23.7 4.9 0.0 H 0.0 31.9 50.0 -18.1 1054.938 40.7 37.7 1.0 23.8 5.0 0.0 H 0.0 31.7 50.0 -18.3 1058.601 40.5 37.7 1.0 23.8 5.0 0.0 H 0.0 31.6 50.0 -18.4 1039.678 40.6 37.7 1.0 23.7 4.9 0.0 V 0.0 31.5 50.0 -18.5 1072.641 40.2 37.7 1.0 23.9 5.0 0.0 V 0.0 31.4 50.0 -18.6 1061.653 40.3 37.7 1.0 23.8 5.0 0.0 H 0.0 31.4 50.0 -18.6 1071.420 40.2 37.7 1.0 23.9 5.0 0.0 H 0.0 31.4 50.0 -18.6 1028.080 40.6 37.7 1.0 23.6 4.9 0.0 H 0.0 31.4 50.0 -18.6 RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007 09/29/11 PSA 2011.09.01
EMI 2009.8.29 

# RADIATED EMISSIONS

1 Pass 23 No deviations. None

None



### 0 10 20 30 40 50 60 70 80

1000 10000 

## MHzdBuV/mFreq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

1034.794 40.4 37.7 1.0 23.6 4.9 0.0 H 0.0 31.2 50.0 -18.8 1064.705 40.1 37.7 1.0 23.9 5.0 0.0 H 0.0 31.2 50.0 -18.8 1089.733 39.7 37.6 1.0 24.0 5.0 0.0 H 0.0 31.1 50.0 -18.9 1058.601 40.0 37.7 1.0 23.8 5.0 0.0 V 0.0 31.1 50.0 -18.9 1043.951 40.1 37.7 1.0 23.7 4.9 0.0 V 0.0 31.0 50.0 -19.0 1083.018 39.7 37.6 1.0 24.0 5.0 0.0 V 0.0 31.0 50.0 -19.0

### scan

EUT: Inogen 1G-3 Work Order: INOG0018 Serial Number: Prototype 1 Date: Customer: Inogen Temperature: 23C Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb Tested by: Jeremiah Darden Power: Battery Job Site: OC11 

### TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS Antenna Height(s) (m) 1 Test Distance (m) 1 

### COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDAR D

Run # Configuration # Results Signature

Freq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

6.778 22.0 0.0 1.0 0.2 0.4 0.0 V 0.0 22.6 37.6 -15.0 7.162 21.6 0.0 1.0 0.2 0.4 0.0 V 0.0 22.2 37.5 -15.3 7.042 20.4 0.0 1.0 0.2 0.4 0.0 V 0.0 21.0 37.5 -16.5 6.392 20.1 0.0 1.0 0.2 0.4 0.0 V 0.0 20.7 37.7 -17.0 7.233 19.8 0.0 1.0 0.2 0.4 0.0 V 0.0 20.4 37.5 -17.1 6.849 19.5 0.0 1.0 0.2 0.4 0.0 V 0.0 20.1 37.6 -17.5 7.210 19.2 0.0 1.0 0.2 0.4 0.0 V 0.0 19.8 37.5 -17.7 6.968 19.1 0.0 1.0 0.2 0.4 0.0 V 0.0 19.7 37.5 -17.9 2.787 20.9 0.0 1.0 0.3 0.2 0.0 V 0.0 21.4 39.3 -17.9 7.065 19.0 0.0 1.0 0.2 0.4 0.0 V 0.0 19.6 37.5 -17.9 6.991 19.0 0.0 1.0 0.2 0.4 0.0 V 0.0 19.6 37.5 -18.0 6.825 19.0 0.0 1.0 0.2 0.4 0.0 V 0.0 19.6 37.6 -18.0 6.585 19.0 0.0 1.0 0.2 0.4 0.0 V 0.0 19.6 37.6 -18.1 RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007 09/29/11 PSA 2011.09.01
EMI 2009.8.29 

# RADIATED EMISSIONS

2 Pass

6 No deviations. Front, Internal Battery

Autopulse 17 BPM Setting 4



### 0 10 20 30 40 50 60 70 80

1 10 100 

## MHzdBuV/mFreq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

6.705 18.7 0.0 1.0 0.2 0.4 0.0 V 0.0 19.3 37.6 -18.3 6.344 18.8 0.0 1.0 0.2 0.4 0.0 V 0.0 19.4 37.7 -18.3 7.017 18.6 0.0 1.0 0.2 0.4 0.0 V 0.0 19.2 37.5 -18.4 6.273 18.8 0.0 1.0 0.2 0.4 0.0 V 0.0 19.4 37.7 -18.4 5.959 18.9 0.0 1.0 0.2 0.4 0.0 V 0.0 19.5 37.8 -18.4 6.463 18.7 0.0 1.0 0.2 0.4 0.0 V 0.0 19.3 37.7 -18.4

### scan

EUT: Inogen 1G-3 Work Order: INOG0018 Serial Number: Prototype 1 Date: Customer: Inogen Temperature: 23C Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb Tested by: Jeremiah Darden Power: Battery Job Site: OC11 

### TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS Antenna Height(s) (m) 1 Test Distance (m) 1 

### COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDAR D

Run # Configuration # Results Signature

Freq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

2.050 22.5 0.0 1.0 0.3 0.2 0.0 V 0.0 23.0 40.0 -17.0 2.214 21.9 0.0 1.0 0.3 0.2 0.0 V 0.0 22.4 39.8 -17.4 2.132 21.3 0.0 1.0 0.3 0.2 0.0 V 0.0 21.8 39.9 -18.1 2.296 21.0 0.0 1.0 0.3 0.2 0.0 V 0.0 21.5 39.7 -18.2 2.706 20.3 0.0 1.0 0.3 0.2 0.0 V 0.0 20.8 39.4 -18.6 2.542 20.4 0.0 1.0 0.3 0.2 0.0 V 0.0 20.9 39.5 -18.6 2.788 20.1 0.0 1.0 0.3 0.2 0.0 V 0.0 20.6 39.3 -18.7 2.378 20.3 0.0 1.0 0.3 0.2 0.0 V 0.0 20.8 39.7 -18.9 2.460 20.0 0.0 1.0 0.3 0.2 0.0 V 0.0 20.5 39.6 -19.1 4.044 18.7 0.0 1.0 0.3 0.3 0.0 V 0.0 19.3 38.6 -19.3 2.055 19.8 0.0 1.0 0.3 0.2 0.0 V 0.0 20.3 39.9 -19.6 2.141 19.2 0.0 1.0 0.3 0.2 0.0 V 0.0 19.7 39.9 -20.2 4.210 17.7 0.0 1.0 0.3 0.3 0.0 V 0.0 18.3 38.5 -20.2 RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007 09/29/11 PSA 2011.09.01
EMI 2009.8.29 

# RADIATED EMISSIONS

2 Pass

7 No deviations. Right, Internal Battery

Autopulse 17 BPM Setting 4



### 0 10 20 30 40 50 60 70 80

1 10 100 

## MHzdBuV/mFreq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

3.881 17.7 0.0 1.0 0.3 0.3 0.0 V 0.0 18.3 38.7 -20.4 2.624 18.4 0.0 1.0 0.3 0.2 0.0 V 0.0 18.9 39.5 -20.6 3.550 17.7 0.0 1.0 0.3 0.2 0.0 V 0.0 18.2 38.9 -20.6 4.183 17.3 0.0 1.0 0.3 0.3 0.0 V 0.0 17.9 38.5 -20.7 3.939 17.4 0.0 1.0 0.3 0.3 0.0 V 0.0 18.0 38.7 -20.7 3.197 17.9 0.0 1.0 0.3 0.2 0.0 V 0.0 18.4 39.1 -20.7

### scan

EUT: Inogen 1G-3 Work Order: INOG0018 Serial Number: Prototype 1 Date: Customer: Inogen Temperature: 23C Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb Tested by: Jeremiah Darden Power: Battery Job Site: OC11 

### TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS Antenna Height(s) (m) 1 Test Distance (m) 1 

### COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDAR D

Run # Configuration # Results Signature

Freq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

7.402 21.4 0.0 1.0 0.2 0.4 0.0 V 0.0 22.0 37.4 -15.5 7.331 19.5 0.0 1.0 0.2 0.4 0.0 V 0.0 20.1 37.4 -17.4 7.138 19.5 0.0 1.0 0.2 0.4 0.0 V 0.0 20.1 37.5 -17.4 7.089 18.8 0.0 1.0 0.2 0.4 0.0 V 0.0 19.4 37.5 -18.1 7.450 18.6 0.0 1.0 0.2 0.4 0.0 V 0.0 19.2 37.4 -18.2 7.282 18.3 0.0 1.0 0.2 0.4 0.0 V 0.0 18.9 37.4 -18.6 7.258 18.3 0.0 1.0 0.2 0.4 0.0 V 0.0 18.9 37.4 -18.6 7.475 18.1 0.0 1.0 0.2 0.4 0.0 V 0.0 18.7 37.4 -18.7 6.392 18.0 0.0 1.0 0.2 0.4 0.0 V 0.0 18.6 37.7 -19.1 7.619 17.3 0.0 1.0 0.1 0.4 0.0 V 0.0 17.8 37.4 -19.5 7.378 17.2 0.0 1.0 0.2 0.4 0.0 V 0.0 17.8 37.4 -19.7 7.522 17.0 0.0 1.0 0.1 0.4 0.0 V 0.0 17.5 37.4 -19.8 7.762 16.8 0.0 1.0 0.1 0.4 0.0 V 0.0 17.3 37.3 -20.0 RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007 09/29/11 PSA 2011.09.01
EMI 2009.8.29 

# RADIATED EMISSIONS

2 Pass

8 No deviations. Back, Internal Battery

Autopulse 17 BPM Setting 4



### 0 10 20 30 40 50 60 70 80

1 10 100 

## MHzdBuV/mFreq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

6.754 17.0 0.0 1.0 0.2 0.4 0.0 V 0.0 17.6 37.6 -20.0 6.993 16.9 0.0 1.0 0.2 0.4 0.0 V 0.0 17.5 37.5 -20.1 7.231 16.7 0.0 1.0 0.2 0.4 0.0 V 0.0 17.3 37.5 -20.2 6.609 16.7 0.0 1.0 0.2 0.4 0.0 V 0.0 17.3 37.6 -20.4 7.161 16.5 0.0 1.0 0.2 0.4 0.0 V 0.0 17.1 37.5 -20.4 7.186 16.3 0.0 1.0 0.2 0.4 0.0 V 0.0 16.9 37.5 -20.6

### scan

EUT: Inogen 1G-3 Work Order: INOG0018 Serial Number: Prototype 1 Date: Customer: Inogen Temperature: 23C Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb Tested by: Jeremiah Darden Power: Battery Job Site: OC11 

### TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS Antenna Height(s) (m) 1 Test Distance (m) 1 

### COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDAR D

Run # Configuration # Results Signature

Freq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

2.132 16.7 0.0 1.0 0.3 0.2 0.0 V 0.0 17.2 39.9 -22.7 2.214 14.3 0.0 1.0 0.3 0.2 0.0 V 0.0 14.8 39.8 -25.0 2.460 13.9 0.0 1.0 0.3 0.2 0.0 V 0.0 14.4 39.6 -25.2 2.952 13.1 0.0 1.0 0.3 0.2 0.0 V 0.0 13.6 39.2 -25.6 2.296 13.4 0.0 1.0 0.3 0.2 0.0 V 0.0 13.9 39.7 -25.8 6.969 10.9 0.0 1.0 0.2 0.4 0.0 V 0.0 11.5 37.5 -26.1 2.378 12.7 0.0 1.0 0.3 0.2 0.0 V 0.0 13.2 39.7 -26.5 2.824 12.1 0.0 1.0 0.3 0.2 0.0 V 0.0 12.6 39.3 -26.7 3.280 11.4 0.0 1.0 0.3 0.2 0.0 V 0.0 11.9 39.0 -27.1 2.050 12.3 0.0 1.0 0.3 0.2 0.0 V 0.0 12.8 40.0 -27.2 7.956 9.4 0.0 1.0 0.1 0.4 0.0 V 0.0 9.9 37.3 -27.3 6.354 9.7 0.0 1.0 0.2 0.4 0.0 V 0.0 10.3 37.7 -27.4 3.116 11.0 0.0 1.0 0.3 0.2 0.0 V 0.0 11.5 39.1 -27.6 RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007 09/29/11 PSA 2011.09.01
EMI 2009.8.29 

# RADIATED EMISSIONS

2 Pass

9 No deviations. Left, Internal Battery

Autopulse 17 BPM Setting 4



### 0 10 20 30 40 50 60 70 80

1 10 100 

## MHzdBuV/mFreq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

2.787 11.2 0.0 1.0 0.3 0.2 0.0 V 0.0 11.7 39.3 -27.6 3.854 10.4 0.0 1.0 0.3 0.3 0.0 V 0.0 11.0 38.7 -27.7 24.925 6.5 0.0 1.0 0.0 0.8 0.0 V 0.0 7.3 35.0 -27.7 3.198 10.7 0.0 1.0 0.3 0.2 0.0 V 0.0 11.2 39.1 -27.9 2.870 10.9 0.0 1.0 0.3 0.2 0.0 V 0.0 11.4 39.3 -27.9 2.706 11.0 0.0 1.0 0.3 0.2 0.0 V 0.0 11.5 39.4 -27.9

### scan

EUT: Inogen 1G-3 Work Order: INOG0018 Serial Number: Prototype 1 Date: Customer: Inogen Temperature: 23C Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb Tested by: Jeremiah Darden Power: Battery Job Site: OC11 

### TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS Antenna Height(s) (m) 1 Test Distance (m) 1 

### COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDAR D

Run # Configuration # Results Signature

Freq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

119.998 39.3 41.7 1.0 15.2 1.9 0.0 H 0.0 14.8 34.0 -19.2 131.999 39.6 41.5 1.0 13.5 2.0 0.0 H 0.0 13.6 34.0 -20.4 126.004 38.8 41.7 1.0 14.4 2.0 0.0 V 0.0 13.5 34.0 -20.5 131.999 39.0 41.5 1.0 13.5 2.0 0.0 V 0.0 13.0 34.0 -21.0 126.004 38.1 41.7 1.0 14.4 2.0 0.0 H 0.0 12.8 34.0 -21.2 107.997 38.8 41.7 1.0 13.8 1.9 0.0 V 0.0 12.8 34.0 -21.2 114.003 37.6 41.6 1.0 14.7 1.9 0.0 H 0.0 12.6 34.0 -21.4 119.998 36.8 41.7 1.0 15.2 1.9 0.0 V 0.0 12.3 34.0 -21.7 102.002 39.1 41.8 1.0 12.4 1.9 0.0 V 0.0 11.5 34.0 -22.5 138.005 38.1 41.4 1.0 12.8 2.0 0.0 V 0.0 11.5 34.0 -22.5 144.000 38.4 41.4 1.0 12.4 2.0 0.0 V 0.0 11.4 34.0 -22.6 116.396 36.2 41.6 1.0 14.9 1.9 0.0 V 0.0 11.4 34.0 -22.6 114.003 36.3 41.6 1.0 14.7 1.9 0.0 V 0.0 11.3 34.0 -22.7 RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007 09/29/11 PSA 2011.09.01
EMI 2009.8.29 

# RADIATED EMISSIONS

2 Pass 30 No deviations. Front, Internal Battery

Autopulse 17 BPM Setting 4



### 0 10 20 30 40 50 60 70 80

10 100 1000 

## MHzdBuV/mFreq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

145.184 38.3 41.5 1.0 12.4 2.0 0.0 V 0.0 11.3 34.0 -22.7 128.556 36.4 41.6 1.0 14.0 2.0 0.0 H 0.0 10.8 34.0 -23.2 107.997 36.5 41.7 1.0 13.8 1.9 0.0 H 0.0 10.5 34.0 -23.5 131.327 36.2 41.6 1.0 13.6 2.0 0.0 H 0.0 10.3 34.0 -23.7 130.314 35.5 41.6 1.0 13.8 2.0 0.0 H 0.0 9.7 34.0 -24.3 128.458 34.9 41.6 1.0 14.0 2.0 0.0 V 0.0 9.3 34.0 -24.7

### scan

EUT: Inogen 1G-3 Work Order: INOG0018 Serial Number: Prototype 1 Date: Customer: Inogen Temperature: 23C Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb Tested by: Jeremiah Darden Power: Battery Job Site: OC11 

### TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS Antenna Height(s) (m) 1 Test Distance (m) 1 

### COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDAR D

Run # Configuration # Results Signature

Freq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

131.999 40.7 41.5 1.0 13.5 2.0 0.0 V 0.0 14.7 34.0 -19.3 138.030 40.5 41.4 1.0 12.8 2.0 0.0 V 0.0 13.9 34.0 -20.1 126.004 39.0 41.7 1.0 14.4 2.0 0.0 V 0.0 13.7 34.0 -20.3 107.997 38.5 41.7 1.0 13.8 1.9 0.0 V 0.0 12.5 34.0 -21.5 114.003 37.4 41.6 1.0 14.7 1.9 0.0 V 0.0 12.4 34.0 -21.6 119.998 36.8 41.7 1.0 15.2 1.9 0.0 V 0.0 12.3 34.0 -21.7 138.005 38.8 41.4 1.0 12.8 2.0 0.0 V 0.0 12.2 34.0 -21.8 102.002 39.7 41.8 1.0 12.4 1.9 0.0 V 0.0 12.1 34.0 -21.9 109.193 36.5 41.7 1.0 14.1 1.9 0.0 V 0.0 10.8 34.0 -23.2 140.349 36.8 41.4 1.0 12.6 2.0 0.0 V 0.0 10.0 34.0 -24.0 144.000 36.8 41.4 1.0 12.4 2.0 0.0 V 0.0 9.8 34.0 -24.2 123.624 34.4 41.7 1.0 14.8 2.0 0.0 V 0.0 9.4 34.0 -24.6 147.699 35.3 41.5 1.0 12.5 2.0 0.0 V 0.0 8.4 34.0 -25.6 RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007 09/29/11 PSA 2011.09.01
EMI 2009.8.29 

# RADIATED EMISSIONS

2 Pass 31 No deviations. Right, Internal Battery

Autopulse 17 BPM Setting 4



### 0 10 20 30 40 50 60 70 80

10 100 1000 

## MHzdBuV/mFreq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

116.274 32.8 41.6 1.0 14.9 1.9 0.0 V 0.0 8.0 34.0 -26.0 192.003 44.7 41.3 1.0 17.8 2.4 0.0 V 0.0 23.6 49.7 -26.0 116.506 32.3 41.6 1.0 14.9 1.9 0.0 V 0.0 7.5 34.0 -26.5 109.938 32.7 41.7 1.0 14.3 1.9 0.0 V 0.0 7.2 34.0 -26.8 117.666 31.9 41.6 1.0 15.0 1.9 0.0 V 0.0 7.2 34.0 -26.8 124.710 32.0 41.7 1.0 14.6 2.0 0.0 V 0.0 6.9 34.0 -27.1

### scan

EUT: Inogen 1G-3 Work Order: INOG0018 Serial Number: Prototype 1 Date: Customer: Inogen Temperature: 23C Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb Tested by: Jeremiah Darden Power: Battery Job Site: OC11 

### TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS Antenna Height(s) (m) 1 Test Distance (m) 1 

### COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDAR D

Run # Configuration # Results Signature

Freq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

126.004 40.0 41.7 1.0 14.4 2.0 0.0 V 0.0 14.7 34.0 -19.3 149.982 41.2 41.5 1.0 12.6 2.1 0.0 V 0.0 14.4 34.0 -19.6 131.999 40.3 41.5 1.0 13.5 2.0 0.0 V 0.0 14.3 34.0 -19.7 119.998 37.8 41.7 1.0 15.2 1.9 0.0 V 0.0 13.3 34.0 -20.7 114.003 37.9 41.6 1.0 14.7 1.9 0.0 V 0.0 12.9 34.0 -21.1 140.422 39.6 41.4 1.0 12.6 2.0 0.0 V 0.0 12.8 34.0 -21.2 138.005 39.0 41.4 1.0 12.8 2.0 0.0 V 0.0 12.4 34.0 -21.6 101.087 40.2 41.8 1.0 12.2 1.9 0.0 V 0.0 12.4 34.0 -21.6 101.160 39.9 41.8 1.0 12.2 1.9 0.0 V 0.0 12.1 34.0 -21.9 107.997 38.1 41.7 1.0 13.8 1.9 0.0 V 0.0 12.1 34.0 -21.9 102.002 39.1 41.8 1.0 12.4 1.9 0.0 V 0.0 11.5 34.0 -22.5 140.374 37.9 41.4 1.0 12.6 2.0 0.0 V 0.0 11.1 34.0 -22.9 144.000 37.8 41.4 1.0 12.4 2.0 0.0 H 0.0 10.8 34.0 -23.2 RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007 09/29/11 PSA 2011.09.01
EMI 2009.8.29 

# RADIATED EMISSIONS

2 Pass 32 No deviations. Back, Internal Battery

Autopulse 17 BPM Setting 4



### 0 10 20 30 40 50 60 70 80

10 100 1000 

## MHzdBuV/mFreq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

131.999 36.5 41.5 1.0 13.5 2.0 0.0 H 0.0 10.5 34.0 -23.5 144.000 37.2 41.4 1.0 12.4 2.0 0.0 V 0.0 10.2 34.0 -23.8 126.004 35.0 41.7 1.0 14.4 2.0 0.0 H 0.0 9.7 34.0 -24.3 119.680 34.0 41.7 1.0 15.2 1.9 0.0 H 0.0 9.5 34.0 -24.5 138.005 35.9 41.4 1.0 12.8 2.0 0.0 H 0.0 9.3 34.0 -24.7 124.100 34.1 41.7 1.0 14.7 2.0 0.0 H 0.0 9.1 34.0 -24.9

### scan

EUT: Inogen 1G-3 Work Order: INOG0018 Serial Number: Prototype 1 Date: Customer: Inogen Temperature: 23C Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb Tested by: Jeremiah Darden Power: Battery Job Site: OC11 

### TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS Antenna Height(s) (m) 1 Test Distance (m) 1 

### COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDAR D

Run # Configuration # Results Signature

Freq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

131.999 40.1 41.5 1.0 13.5 2.0 0.0 V 0.0 14.1 34.0 -19.9 126.004 39.3 41.7 1.0 14.4 2.0 0.0 V 0.0 14.0 34.0 -20.0 145.233 40.9 41.5 1.0 12.4 2.0 0.0 V 0.0 13.9 34.0 -20.1 137.993 39.5 41.4 1.0 12.8 2.0 0.0 V 0.0 12.9 34.0 -21.1 140.422 39.6 41.4 1.0 12.6 2.0 0.0 V 0.0 12.8 34.0 -21.2 107.997 38.4 41.7 1.0 13.8 1.9 0.0 V 0.0 12.4 34.0 -21.6 119.998 36.9 41.7 1.0 15.2 1.9 0.0 V 0.0 12.4 34.0 -21.6 102.002 39.5 41.8 1.0 12.4 1.9 0.0 V 0.0 11.9 34.0 -22.1 109.181 37.3 41.7 1.0 14.1 1.9 0.0 V 0.0 11.6 34.0 -22.4 114.003 36.4 41.6 1.0 14.7 1.9 0.0 V 0.0 11.4 34.0 -22.6 130.778 35.5 41.6 1.0 13.7 2.0 0.0 V 0.0 9.6 34.0 -24.4 126.627 33.6 41.7 1.0 14.3 2.0 0.0 V 0.0 8.3 34.0 -25.7 149.725 35.0 41.5 1.0 12.6 2.1 0.0 V 0.0 8.2 34.0 -25.8 RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007 09/29/11 PSA 2011.09.01
EMI 2009.8.29 

# RADIATED EMISSIONS

2 Pass 33 No deviations. Left, Internal Battery

Autopulse 17 BPM Setting 4



### 0 10 20 30 40 50 60 70 80

10 100 1000 

## MHzdBuV/mFreq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

144.000 34.9 41.4 1.0 12.4 2.0 0.0 V 0.0 7.9 34.0 -26.1 69.935 48.1 42.1 1.0 8.7 1.5 0.0 H 0.0 16.2 42.4 -26.2 101.099 34.8 41.8 1.0 12.2 1.9 0.0 H 0.0 7.0 34.0 -27.0 148.736 33.8 41.5 1.0 12.5 2.1 0.0 V 0.0 6.9 34.0 -27.1 101.050 34.7 41.8 1.0 12.2 1.9 0.0 H 0.0 6.9 34.0 -27.1 132.243 32.9 41.5 1.0 13.5 2.0 0.0 V 0.0 6.9 34.0 -27.1

### scan

EUT: Inogen 1G-3 Work Order: INOG0018 Serial Number: Prototype 1 Date: Customer: Inogen Temperature: 23C Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb Tested by: Jeremiah Darden Power: Battery Job Site: OC11 

### TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS Antenna Height(s) (m) 1 Test Distance (m) 1 

### COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDAR D

Run # Configuration # Results Signature

Freq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

287.999 53.1 41.3 1.0 16.3 2.9 0.0 V 0.0 31.0 52.4 -21.4 243.755 53.8 40.9 1.0 14.3 2.7 0.0 V 0.0 29.8 51.3 -21.4 246.368 53.6 40.9 1.0 14.4 2.7 0.0 V 0.0 29.8 51.4 -21.6 239.995 53.5 41.0 1.0 14.1 2.6 0.0 V 0.0 29.3 51.2 -21.9 248.175 53.2 40.9 1.0 14.5 2.7 0.0 V 0.0 29.5 51.4 -22.0 246.905 53.2 40.9 1.0 14.5 2.7 0.0 V 0.0 29.4 51.4 -22.0 246.002 53.2 40.9 1.0 14.4 2.7 0.0 V 0.0 29.4 51.4 -22.0 232.450 53.7 41.1 1.0 13.8 2.6 0.0 V 0.0 29.0 51.0 -22.0 249.127 53.1 41.0 1.0 14.6 2.7 0.0 V 0.0 29.4 51.4 -22.0 245.513 53.1 40.9 1.0 14.4 2.7 0.0 V 0.0 29.2 51.3 -22.1 245.074 53.1 40.9 1.0 14.4 2.7 0.0 V 0.0 29.2 51.3 -22.1 246.612 53.0 40.9 1.0 14.4 2.7 0.0 V 0.0 29.2 51.4 -22.2 248.297 52.9 41.0 1.0 14.5 2.7 0.0 V 0.0 29.2 51.4 -22.3 RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007 09/29/11 PSA 2011.09.01
EMI 2009.8.29 

# RADIATED EMISSIONS

2 Pass 26 No deviations. Front, Internal Battery

Autopulse 17 BPM Setting 4



### 0 10 20 30 40 50 60 70 80

100 1000 

## MHzdBuV/mFreq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

247.296 52.8 40.9 1.0 14.5 2.7 0.0 V 0.0 29.0 51.4 -22.4 244.537 52.7 40.9 1.0 14.3 2.7 0.0 V 0.0 28.8 51.3 -22.5 242.950 52.6 40.9 1.0 14.3 2.6 0.0 V 0.0 28.6 51.3 -22.7 251.032 52.4 41.0 1.0 14.6 2.7 0.0 V 0.0 28.7 51.5 -22.7 243.584 52.5 40.9 1.0 14.3 2.6 0.0 V 0.0 28.5 51.3 -22.8 250.031 52.2 41.0 1.0 14.6 2.7 0.0 V 0.0 28.5 51.5 -22.9

### scan

EUT: Inogen 1G-3 Work Order: INOG0018 Serial Number: Prototype 1 Date: Customer: Inogen Temperature: 23C Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb Tested by: Jeremiah Darden Power: Battery Job Site: OC11 

### TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS Antenna Height(s) (m) 1 Test Distance (m) 1 

### COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDAR D

Run # Configuration # Results Signature

Freq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

778.049 55.4 42.1 1.0 21.8 5.2 0.0 V 0.0 40.3 59.2 -18.9 287.999 55.3 41.3 1.0 16.3 2.9 0.0 V 0.0 33.2 52.4 -19.2 246.978 54.8 40.9 1.0 14.5 2.7 0.0 V 0.0 31.0 51.4 -20.4 246.905 54.6 40.9 1.0 14.5 2.7 0.0 V 0.0 30.8 51.4 -20.6 245.489 54.5 40.9 1.0 14.4 2.7 0.0 V 0.0 30.6 51.3 -20.7 252.008 54.4 41.0 1.0 14.6 2.7 0.0 V 0.0 30.8 51.5 -20.8 239.995 54.6 41.0 1.0 14.1 2.6 0.0 V 0.0 30.4 51.2 -20.8 246.417 54.2 40.9 1.0 14.4 2.7 0.0 V 0.0 30.4 51.4 -21.0 245.758 54.0 40.9 1.0 14.4 2.7 0.0 V 0.0 30.1 51.3 -21.2 248.321 53.8 41.0 1.0 14.5 2.7 0.0 V 0.0 30.1 51.4 -21.4 245.294 53.6 40.9 1.0 14.4 2.7 0.0 V 0.0 29.7 51.3 -21.6 241.802 53.7 40.9 1.0 14.2 2.6 0.0 V 0.0 29.6 51.2 -21.6 247.394 53.5 40.9 1.0 14.5 2.7 0.0 V 0.0 29.7 51.4 -21.7 RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007 09/29/11 PSA 2011.09.01
EMI 2009.8.29 

# RADIATED EMISSIONS

2 Pass 27 No deviations. Right, Internal Battery

Autopulse 17 BPM Setting 4



### 0 10 20 30 40 50 60 70 80

100 1000 

## MHzdBuV/mFreq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

246.685 53.5 40.9 1.0 14.4 2.7 0.0 V 0.0 29.7 51.4 -21.7 246.051 53.5 40.9 1.0 14.4 2.7 0.0 V 0.0 29.7 51.4 -21.7 248.810 53.4 41.0 1.0 14.5 2.7 0.0 V 0.0 29.7 51.4 -21.7 245.074 53.3 40.9 1.0 14.4 2.7 0.0 V 0.0 29.4 51.3 -21.9 244.073 53.3 40.9 1.0 14.3 2.7 0.0 V 0.0 29.4 51.3 -21.9 239.800 53.4 41.0 1.0 14.1 2.6 0.0 V 0.0 29.2 51.2 -22.0

### scan

EUT: Inogen 1G-3 Work Order: INOG0018 Serial Number: Prototype 1 Date: Customer: Inogen Temperature: 23C Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb Tested by: Jeremiah Darden Power: Battery Job Site: OC11 

### TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS Antenna Height(s) (m) 1 Test Distance (m) 1 

### COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDAR D

Run # Configuration # Results Signature

Freq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

287.999 56.0 41.3 1.0 16.3 2.9 0.0 V 0.0 33.9 52.4 -18.5 246.441 55.9 40.9 1.0 14.4 2.7 0.0 V 0.0 32.1 51.4 -19.3 245.343 55.7 40.9 1.0 14.4 2.7 0.0 V 0.0 31.8 51.3 -19.5 247.125 55.1 40.9 1.0 14.5 2.7 0.0 V 0.0 31.3 51.4 -20.1 246.002 55.1 40.9 1.0 14.4 2.7 0.0 V 0.0 31.3 51.4 -20.1 244.610 55.1 40.9 1.0 14.3 2.7 0.0 V 0.0 31.2 51.3 -20.1 245.660 54.9 40.9 1.0 14.4 2.7 0.0 V 0.0 31.0 51.3 -20.3 242.535 54.9 40.9 1.0 14.2 2.6 0.0 V 0.0 30.9 51.3 -20.4 246.197 54.7 40.9 1.0 14.4 2.7 0.0 V 0.0 30.9 51.4 -20.5 249.005 54.5 41.0 1.0 14.6 2.7 0.0 V 0.0 30.8 51.4 -20.6 243.755 54.6 40.9 1.0 14.3 2.7 0.0 V 0.0 30.6 51.3 -20.6 248.077 54.5 40.9 1.0 14.5 2.7 0.0 V 0.0 30.7 51.4 -20.7 246.905 54.5 40.9 1.0 14.5 2.7 0.0 V 0.0 30.7 51.4 -20.7 RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007 09/29/11 PSA 2011.09.01
EMI 2009.8.29 

# RADIATED EMISSIONS

2 Pass 28 No deviations. Back, Internal Battery

Autopulse 17 BPM Setting 4



### 0 10 20 30 40 50 60 70 80

100 1000 

## MHzdBuV/mFreq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

244.756 54.5 40.9 1.0 14.3 2.7 0.0 V 0.0 30.6 51.3 -20.7 240.703 54.4 40.9 1.0 14.2 2.6 0.0 V 0.0 30.2 51.2 -21.0 241.558 54.3 40.9 1.0 14.2 2.6 0.0 V 0.0 30.2 51.2 -21.0 248.273 54.1 40.9 1.0 14.5 2.7 0.0 V 0.0 30.4 51.4 -21.1 242.290 54.2 40.9 1.0 14.2 2.6 0.0 V 0.0 30.1 51.3 -21.1 244.439 54.1 40.9 1.0 14.3 2.7 0.0 V 0.0 30.2 51.3 -21.1

### scan

EUT: Inogen 1G-3 Work Order: INOG0018 Serial Number: Prototype 1 Date: Customer: Inogen Temperature: 23C Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb Tested by: Jeremiah Darden Power: Battery Job Site: OC11 

### TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS Antenna Height(s) (m) 1 Test Distance (m) 1 

### COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDAR D

Run # Configuration # Results Signature

Freq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

246.685 56.7 40.9 1.0 14.4 2.7 0.0 V 0.0 32.9 51.4 -18.5 232.792 56.6 41.1 1.0 13.8 2.6 0.0 V 0.0 31.9 51.0 -19.1 244.879 55.8 40.9 1.0 14.4 2.7 0.0 V 0.0 31.9 51.3 -19.4 247.955 55.5 40.9 1.0 14.5 2.7 0.0 V 0.0 31.7 51.4 -19.7 247.638 55.5 40.9 1.0 14.5 2.7 0.0 V 0.0 31.7 51.4 -19.7 287.999 54.8 41.3 1.0 16.3 2.9 0.0 V 0.0 32.7 52.4 -19.7 240.313 55.6 41.0 1.0 14.1 2.6 0.0 V 0.0 31.4 51.2 -19.8 245.806 55.3 40.9 1.0 14.4 2.7 0.0 V 0.0 31.4 51.3 -19.9 231.571 55.8 41.1 1.0 13.7 2.5 0.0 V 0.0 31.0 50.9 -20.0 243.511 55.1 40.9 1.0 14.3 2.6 0.0 V 0.0 31.1 51.3 -20.2 247.003 55.0 40.9 1.0 14.5 2.7 0.0 V 0.0 31.2 51.4 -20.2 244.122 55.0 40.9 1.0 14.3 2.7 0.0 V 0.0 31.1 51.3 -20.2 248.541 54.9 41.0 1.0 14.5 2.7 0.0 V 0.0 31.2 51.4 -20.3 RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007 09/29/11 PSA 2011.09.01
EMI 2009.8.29 

# RADIATED EMISSIONS

2 Pass 29 No deviations. Left, Internal Battery

Autopulse 17 BPM Setting 4



### 0 10 20 30 40 50 60 70 80

100 1000 

## MHzdBuV/mFreq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

243.340 55.0 40.9 1.0 14.3 2.6 0.0 V 0.0 31.0 51.3 -20.3 240.093 55.1 41.0 1.0 14.1 2.6 0.0 V 0.0 30.9 51.2 -20.3 242.144 55.0 40.9 1.0 14.2 2.6 0.0 V 0.0 30.9 51.2 -20.3 241.924 55.0 40.9 1.0 14.2 2.6 0.0 V 0.0 30.9 51.2 -20.3 244.415 54.8 40.9 1.0 14.3 2.7 0.0 V 0.0 30.9 51.3 -20.4 245.709 54.7 40.9 1.0 14.4 2.7 0.0 V 0.0 30.8 51.3 -20.5

### scan

EUT: Inogen 1G-3 Work Order: INOG0018 Serial Number: Prototype 1 Date: Customer: Inogen Temperature: 23C Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb Tested by: Jeremiah Darden Power: Battery Job Site: OC11 

### TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS Antenna Height(s) (m) 1 Test Distance (m) 1 

### COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDAR D

Run # Configuration # Results Signature

Freq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

1051.886 42.6 37.7 1.0 23.8 5.0 0.0 H 0.0 33.6 50.0 -16.4 1093.395 41.8 37.6 1.0 24.1 5.0 0.0 H 0.0 33.2 50.0 -16.8 1053.107 42.1 37.7 1.0 23.8 5.0 0.0 H 0.0 33.1 50.0 -16.9 1078.745 41.6 37.6 1.0 24.0 5.0 0.0 H 0.0 32.9 50.0 -17.1 1047.003 41.6 37.7 1.0 23.7 4.9 0.0 V 0.0 32.6 50.0 -17.4 1073.251 41.0 37.7 1.0 23.9 5.0 0.0 H 0.0 32.2 50.0 -17.8 1047.003 41.2 37.7 1.0 23.7 4.9 0.0 H 0.0 32.2 50.0 -17.8 1086.070 40.7 37.6 1.0 24.0 5.0 0.0 H 0.0 32.1 50.0 -17.9 1065.926 40.9 37.7 1.0 23.9 5.0 0.0 H 0.0 32.1 50.0 -17.9 1095.837 40.6 37.6 1.0 24.1 5.0 0.0 V 0.0 32.1 50.0 -17.9 1034.794 40.9 37.7 1.0 23.6 4.9 0.0 V 0.0 31.7 50.0 -18.3 1081.797 40.4 37.6 1.0 24.0 5.0 0.0 V 0.0 31.7 50.0 -18.3 1073.251 40.4 37.7 1.0 23.9 5.0 0.0 V 0.0 31.6 50.0 -18.4 RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007 09/29/11 PSA 2011.09.01
EMI 2009.8.29 

# RADIATED EMISSIONS

2 Pass 19 No deviations. Front, Internal Battery

Autopulse 17 BPM Setting 4



### 0 10 20 30 40 50 60 70 80

1000 10000 

## MHzdBuV/mFreq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

1088.512 40.2 37.6 1.0 24.0 5.0 0.0 V 0.0 31.6 50.0 -18.4 1070.809 40.3 37.7 1.0 23.9 5.0 0.0 H 0.0 31.5 50.0 -18.5 1028.690 40.6 37.7 1.0 23.6 4.9 0.0 V 0.0 31.4 50.0 -18.6 1056.770 40.3 37.7 1.0 23.8 5.0 0.0 V 0.0 31.4 50.0 -18.6 1031.132 40.5 37.7 1.0 23.6 4.9 0.0 H 0.0 31.3 50.0 -18.7 1026.248 40.5 37.7 1.0 23.6 4.9 0.0 H 0.0 31.3 50.0 -18.7

### scan

EUT: Inogen 1G-3 Work Order: INOG0018 Serial Number: Prototype 1 Date: Customer: Inogen Temperature: 23C Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb Tested by: Jeremiah Darden Power: Battery Job Site: OC11 

### TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS Antenna Height(s) (m) 1 Test Distance (m) 1 

### COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDAR D

Run # Configuration # Results Signature

Freq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

1034.794 43.2 37.7 1.0 23.6 4.9 0.0 V 0.0 34.0 50.0 -16.0 1054.938 42.2 37.7 1.0 23.8 5.0 0.0 V 0.0 33.2 50.0 -16.8 1072.641 41.4 37.7 1.0 23.9 5.0 0.0 H 0.0 32.6 50.0 -17.4 1056.770 41.5 37.7 1.0 23.8 5.0 0.0 V 0.0 32.6 50.0 -17.4 1028.690 41.6 37.7 1.0 23.6 4.9 0.0 V 0.0 32.4 50.0 -17.6 1078.745 41.0 37.6 1.0 24.0 5.0 0.0 V 0.0 32.3 50.0 -17.7 1065.926 41.1 37.7 1.0 23.9 5.0 0.0 H 0.0 32.3 50.0 -17.7 1021.975 41.5 37.8 1.0 23.6 4.9 0.0 H 0.0 32.2 50.0 -17.8 1089.733 40.5 37.6 1.0 24.0 5.0 0.0 H 0.0 31.9 50.0 -18.1 1046.392 40.9 37.7 1.0 23.7 4.9 0.0 V 0.0 31.9 50.0 -18.1 1020.755 41.1 37.8 1.0 23.5 4.9 0.0 V 0.0 31.8 50.0 -18.2 1024.417 41.0 37.8 1.0 23.6 4.9 0.0 V 0.0 31.7 50.0 -18.3 1079.966 40.4 37.6 1.0 24.0 5.0 0.0 H 0.0 31.7 50.0 -18.3 RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007 09/29/11 PSA 2011.09.01
EMI 2009.8.29 

# RADIATED EMISSIONS

2 Pass 20 No deviations. Right, Internal Battery

Autopulse 17 BPM Setting 4



### 0 10 20 30 40 50 60 70 80

1000 10000 

## MHzdBuV/mFreq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

1048.834 40.6 37.7 1.0 23.7 4.9 0.0 V 0.0 31.6 50.0 -18.4 1051.886 40.5 37.7 1.0 23.8 5.0 0.0 H 0.0 31.5 50.0 -18.5 1074.472 40.2 37.7 1.0 23.9 5.0 0.0 H 0.0 31.4 50.0 -18.6 1063.484 40.2 37.7 1.0 23.8 5.0 0.0 V 0.0 31.3 50.0 -18.7 1092.785 39.9 37.6 1.0 24.0 5.0 0.0 V 0.0 31.3 50.0 -18.7 1094.616 39.7 37.6 1.0 24.1 5.0 0.0 H 0.0 31.1 50.0 -18.9

### scan

EUT: Inogen 1G-3 Work Order: INOG0018 Serial Number: Prototype 1 Date: Customer: Inogen Temperature: 23C Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb Tested by: Jeremiah Darden Power: Battery Job Site: OC11 

### TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS Antenna Height(s) (m) 1 Test Distance (m) 1 

### COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDAR D

Run # Configuration # Results Signature

Freq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

1021.975 42.8 37.8 1.0 23.6 4.9 0.0 V 0.0 33.5 50.0 -16.5 1026.859 42.4 37.7 1.0 23.6 4.9 0.0 V 0.0 33.2 50.0 -16.8 1020.755 42.3 37.8 1.0 23.5 4.9 0.0 V 0.0 33.0 50.0 -17.0 1025.027 42.2 37.7 1.0 23.6 4.9 0.0 V 0.0 33.0 50.0 -17.0 1045.782 41.9 37.7 1.0 23.7 4.9 0.0 V 0.0 32.9 50.0 -17.1 1071.420 41.4 37.7 1.0 23.9 5.0 0.0 V 0.0 32.6 50.0 -17.4 1068.368 41.4 37.7 1.0 23.9 5.0 0.0 H 0.0 32.6 50.0 -17.4 1094.006 40.9 37.6 1.0 24.1 5.0 0.0 V 0.0 32.3 50.0 -17.7 1036.626 41.4 37.7 1.0 23.7 4.9 0.0 V 0.0 32.3 50.0 -17.7 1031.132 41.4 37.7 1.0 23.6 4.9 0.0 H 0.0 32.2 50.0 -17.8 1095.226 40.5 37.6 1.0 24.1 5.0 0.0 H 0.0 32.0 50.0 -18.0 1041.509 40.9 37.7 1.0 23.7 4.9 0.0 H 0.0 31.8 50.0 -18.2 1031.132 41.0 37.7 1.0 23.6 4.9 0.0 V 0.0 31.8 50.0 -18.2 RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007 09/29/11 PSA 2011.09.01
EMI 2009.8.29 

# RADIATED EMISSIONS

2 Pass 21 No deviations. Back, Internal Battery

Autopulse 17 BPM Setting 4



### 0 10 20 30 40 50 60 70 80

1000 10000 

## MHzdBuV/mFreq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

1034.794 40.9 37.7 1.0 23.6 4.9 0.0 H 0.0 31.7 50.0 -18.3 1073.862 40.5 37.7 1.0 23.9 5.0 0.0 V 0.0 31.7 50.0 -18.3 1083.018 40.4 37.6 1.0 24.0 5.0 0.0 V 0.0 31.7 50.0 -18.3 1055.549 40.6 37.7 1.0 23.8 5.0 0.0 V 0.0 31.7 50.0 -18.3 1063.484 40.5 37.7 1.0 23.8 5.0 0.0 V 0.0 31.6 50.0 -18.4 1059.211 40.5 37.7 1.0 23.8 5.0 0.0 V 0.0 31.6 50.0 -18.4

### scan

EUT: Inogen 1G-3 Work Order: INOG0018 Serial Number: Prototype 1 Date: Customer: Inogen Temperature: 23C Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb Tested by: Jeremiah Darden Power: Battery Job Site: OC11 

### TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS Antenna Height(s) (m) 1 Test Distance (m) 1 

### COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDAR D

Run # Configuration # Results Signature

Freq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

1072.641 45.4 37.7 1.0 23.9 5.0 0.0 H 0.0 36.6 50.0 -13.4 1037.236 45.1 37.7 1.0 23.7 4.9 0.0 H 0.0 36.0 50.0 -14.0 1047.003 44.0 37.7 1.0 23.7 4.9 0.0 H 0.0 35.0 50.0 -15.0 1053.718 43.9 37.7 1.0 23.8 5.0 0.0 H 0.0 34.9 50.0 -15.1 1070.199 43.0 37.7 1.0 23.9 5.0 0.0 H 0.0 34.2 50.0 -15.8 1021.365 43.2 37.8 1.0 23.5 4.9 0.0 H 0.0 33.9 50.0 -16.1 1067.757 42.7 37.7 1.0 23.9 5.0 0.0 H 0.0 33.9 50.0 -16.1 1058.601 42.4 37.7 1.0 23.8 5.0 0.0 H 0.0 33.5 50.0 -16.5 1084.849 42.1 37.6 1.0 24.0 5.0 0.0 H 0.0 33.4 50.0 -16.6 1040.899 42.4 37.7 1.0 23.7 4.9 0.0 H 0.0 33.3 50.0 -16.7 1024.417 42.2 37.8 1.0 23.6 4.9 0.0 H 0.0 32.9 50.0 -17.1 1050.665 41.8 37.7 1.0 23.8 5.0 0.0 H 0.0 32.8 50.0 -17.2 1032.353 41.8 37.7 1.0 23.6 4.9 0.0 H 0.0 32.6 50.0 -17.4 RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007 09/29/11 PSA 2011.09.01
EMI 2009.8.29 

# RADIATED EMISSIONS

2 Pass 22 No deviations. Left, Internal Battery

Autopulse 17 BPM Setting 4



### 0 10 20 30 40 50 60 70 80

1000 10000 

## MHzdBuV/mFreq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

1054.938 41.4 37.7 1.0 23.8 5.0 0.0 H 0.0 32.4 50.0 -17.6 1065.316 41.2 37.7 1.0 23.9 5.0 0.0 H 0.0 32.4 50.0 -17.6 1081.797 41.0 37.6 1.0 24.0 5.0 0.0 V 0.0 32.3 50.0 -17.7 1023.196 41.5 37.8 1.0 23.6 4.9 0.0 V 0.0 32.2 50.0 -17.8 1079.355 40.9 37.6 1.0 24.0 5.0 0.0 V 0.0 32.2 50.0 -17.8 1070.809 40.8 37.7 1.0 23.9 5.0 0.0 V 0.0 32.0 50.0 -18.0

### scan

EUT: Inogen 1G-3 Work Order: INOG0018 Serial Number: Prototype 1 Date: Customer: Inogen Temperature: 23C Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb Tested by: Jeremiah Darden Power: 110VAC/60H z Job Site: OC11 

### TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS Antenna Height(s) (m) 1 Test Distance (m) 1 

### COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDAR D

Run # Configuration # Results Signature

Freq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

2.231 34.1 0.0 1.0 0.3 0.2 0.0 V 0.0 34.6 39.8 -5.2 2.292 32.4 0.0 1.0 0.3 0.2 0.0 V 0.0 32.9 39.7 -6.8 2.313 32.3 0.0 1.0 0.3 0.2 0.0 V 0.0 32.8 39.7 -6.9 2.067 32.4 0.0 1.0 0.3 0.2 0.0 V 0.0 32.9 39.9 -7.0 2.484 31.8 0.0 1.0 0.3 0.2 0.0 V 0.0 32.3 39.6 -7.3 2.103 31.8 0.0 1.0 0.3 0.2 0.0 V 0.0 32.3 39.9 -7.6 3.255 30.9 0.0 1.0 0.3 0.2 0.0 V 0.0 31.4 39.0 -7.6 3.696 30.5 0.0 1.0 0.3 0.3 0.0 V 0.0 31.1 38.8 -7.7 2.544 31.3 0.0 1.0 0.3 0.2 0.0 V 0.0 31.8 39.5 -7.7 2.181 31.4 0.0 1.0 0.3 0.2 0.0 V 0.0 31.9 39.8 -7.9 2.403 31.2 0.0 1.0 0.3 0.2 0.0 V 0.0 31.7 39.6 -7.9 2.710 30.9 0.0 1.0 0.3 0.2 0.0 V 0.0 31.4 39.4 -8.0 2.565 30.9 0.0 1.0 0.3 0.2 0.0 V 0.0 31.4 39.5 -8.1 RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007 09/29/11 PSA 2011.09.01
EMI 2009.8.29 

# RADIATED EMISSIONS

1 Pass 10 No deviations. Top, DC Power Supply: Digital Power Model ADP90-119 SN ADP901195210A0010592

Autopulse 17 BPM Setting 4



### 0 10 20 30 40 50 60 70 80

1 10 100 

## MHzdBuV/mFreq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

2.927 30.4 0.0 1.0 0.3 0.2 0.0 V 0.0 30.9 39.2 -8.3 2.248 30.9 0.0 1.0 0.3 0.2 0.0 V 0.0 31.4 39.8 -8.4 2.005 31.1 0.0 1.0 0.3 0.2 0.0 V 0.0 31.6 40.0 -8.4 2.827 30.4 0.0 1.0 0.3 0.2 0.0 V 0.0 30.9 39.3 -8.4 2.656 30.5 0.0 1.0 0.3 0.2 0.0 V 0.0 31.0 39.4 -8.4 2.053 31.0 0.0 1.0 0.3 0.2 0.0 V 0.0 31.5 39.9 -8.4

### scan

EUT: Inogen 1G-3 Work Order: INOG0018 Serial Number: Prototype 1 Date: Customer: Inogen Temperature: 23C Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb Tested by: Jeremiah Darden Power: 110VAC/60H z Job Site: OC11 

### TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS Antenna Height(s) (m) 1 Test Distance (m) 1 

### COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDAR D

Run # Configuration # Results Signature

Freq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

11.334 27.4 0.0 1.0 0.1 0.5 0.0 V 0.0 28.0 36.6 -8.6 2.569 30.4 0.0 1.0 0.3 0.2 0.0 V 0.0 30.9 39.5 -8.6 11.366 27.0 0.0 1.0 0.1 0.5 0.0 V 0.0 27.6 36.6 -9.0 11.331 26.9 0.0 1.0 0.1 0.5 0.0 V 0.0 27.5 36.6 -9.1 11.250 26.2 0.0 1.0 0.1 0.5 0.0 V 0.0 26.8 36.6 -9.8 11.392 26.1 0.0 1.0 0.1 0.5 0.0 V 0.0 26.7 36.6 -9.9 2.463 29.2 0.0 1.0 0.3 0.2 0.0 V 0.0 29.7 39.6 -9.9 11.344 26.0 0.0 1.0 0.1 0.5 0.0 V 0.0 26.6 36.6 -10.0 2.998 28.7 0.0 1.0 0.3 0.2 0.0 V 0.0 29.2 39.2 -10.0 11.260 25.9 0.0 1.0 0.1 0.5 0.0 V 0.0 26.5 36.6 -10.1 2.203 29.1 0.0 1.0 0.3 0.2 0.0 V 0.0 29.6 39.8 -10.2 2.544 28.8 0.0 1.0 0.3 0.2 0.0 V 0.0 29.3 39.5 -10.2 2.827 28.5 0.0 1.0 0.3 0.2 0.0 V 0.0 29.0 39.3 -10.3 RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007 09/29/11 PSA 2011.09.01
EMI 2009.8.29 

# RADIATED EMISSIONS

1 Pass 11 No deviations. Bottom, DC Power Supply: Digital Power Model ADP90-119 SN ADP901195210A0010592

Autopulse 17 BPM Setting 4



### 0 10 20 30 40 50 60 70 80

1 10 100 

## MHzdBuV/mFreq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

12.316 25.4 0.0 1.0 0.1 0.5 0.0 V 0.0 26.0 36.4 -10.4 2.142 28.9 0.0 1.0 0.3 0.2 0.0 V 0.0 29.4 39.9 -10.5 2.188 28.8 0.0 1.0 0.3 0.2 0.0 V 0.0 29.3 39.8 -10.5 11.265 25.4 0.0 1.0 0.1 0.5 0.0 V 0.0 26.0 36.6 -10.6 2.116 28.7 0.0 1.0 0.3 0.2 0.0 V 0.0 29.2 39.9 -10.7 11.347 25.3 0.0 1.0 0.1 0.5 0.0 V 0.0 25.9 36.6 -10.7

### scan

EUT: Inogen 1G-3 Work Order: INOG0018 Serial Number: Prototype 1 Date: Customer: Inogen Temperature: 23C Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb Tested by: Jeremiah Darden Power: 110VAC/60H z Job Site: OC11 

### TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS Antenna Height(s) (m) 1 Test Distance (m) 1 

### COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDAR D

Run # Configuration # Results Signature

Freq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

147.198 54.6 41.5 1.0 12.5 2.0 0.0 H 0.0 27.7 34.0 -6.3 147.503 54.2 41.5 1.0 12.5 2.0 0.0 H 0.0 27.3 34.0 -6.7 146.637 54.2 41.5 1.0 12.5 2.0 0.0 H 0.0 27.2 34.0 -6.8 146.270 54.2 41.5 1.0 12.5 2.0 0.0 H 0.0 27.2 34.0 -6.8 146.148 53.8 41.5 1.0 12.4 2.0 0.0 H 0.0 26.8 34.0 -7.2 145.819 53.8 41.5 1.0 12.4 2.0 0.0 H 0.0 26.8 34.0 -7.2 147.357 53.6 41.5 1.0 12.5 2.0 0.0 H 0.0 26.7 34.0 -7.3 145.098 53.6 41.5 1.0 12.4 2.0 0.0 H 0.0 26.6 34.0 -7.4 146.759 53.5 41.5 1.0 12.5 2.0 0.0 H 0.0 26.5 34.0 -7.5 145.550 53.5 41.5 1.0 12.4 2.0 0.0 H 0.0 26.5 34.0 -7.5 145.855 53.4 41.5 1.0 12.4 2.0 0.0 H 0.0 26.4 34.0 -7.6 144.525 53.4 41.4 1.0 12.4 2.0 0.0 H 0.0 26.4 34.0 -7.6 145.281 53.4 41.5 1.0 12.4 2.0 0.0 H 0.0 26.4 34.0 -7.6 RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007 09/29/11 PSA 2011.09.01
EMI 2009.8.29 

# RADIATED EMISSIONS

1 Pass 13 No deviations. Top, DC Power Supply: Digital Power Model ADP90-119 SN ADP901195210A0010592

Autopulse 17 BPM Setting 4



### 0 10 20 30 40 50 60 70 80

10 100 1000 

## MHzdBuV/mFreq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

147.674 53.2 41.5 1.0 12.5 2.0 0.0 H 0.0 26.3 34.0 -7.7 149.908 53.1 41.5 1.0 12.6 2.1 0.0 V 0.0 26.3 34.0 -7.7 149.628 53.1 41.5 1.0 12.6 2.1 0.0 H 0.0 26.3 34.0 -7.7 147.039 53.2 41.5 1.0 12.5 2.0 0.0 H 0.0 26.3 34.0 -7.7 145.245 53.2 41.5 1.0 12.4 2.0 0.0 H 0.0 26.2 34.0 -7.8 144.769 53.2 41.5 1.0 12.4 2.0 0.0 H 0.0 26.2 34.0 -7.8

### scan

EUT: Inogen 1G-3 Work Order: INOG0018 Serial Number: Prototype 1 Date: Customer: Inogen Temperature: 23C Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb Tested by: Jeremiah Darden Power: 110VAC/60H z Job Site: OC11 

### TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS Antenna Height(s) (m) 1 Test Distance (m) 1 

### COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDAR D

Run # Configuration # Results Signature

Freq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

149.652 58.5 41.5 1.0 12.6 2.1 0.0 H 0.0 31.7 34.0 -2.3 149.530 58.5 41.5 1.0 12.6 2.1 0.0 H 0.0 31.7 34.0 -2.3 148.041 58.3 41.5 1.0 12.5 2.1 0.0 H 0.0 31.4 34.0 -2.6 149.384 57.7 41.5 1.0 12.6 2.1 0.0 H 0.0 30.9 34.0 -3.1 148.651 57.4 41.5 1.0 12.5 2.1 0.0 H 0.0 30.5 34.0 -3.5 148.505 57.4 41.5 1.0 12.5 2.1 0.0 H 0.0 30.5 34.0 -3.5 149.957 57.3 41.5 1.0 12.6 2.1 0.0 H 0.0 30.5 34.0 -3.5 149.860 57.1 41.5 1.0 12.6 2.1 0.0 H 0.0 30.3 34.0 -3.7 149.188 57.0 41.5 1.0 12.6 2.1 0.0 H 0.0 30.1 34.0 -3.9 147.577 57.0 41.5 1.0 12.5 2.0 0.0 H 0.0 30.1 34.0 -3.9 148.334 56.8 41.5 1.0 12.5 2.1 0.0 H 0.0 29.9 34.0 -4.1 148.871 56.7 41.5 1.0 12.6 2.1 0.0 H 0.0 29.8 34.0 -4.2 148.981 56.6 41.5 1.0 12.6 2.1 0.0 H 0.0 29.7 34.0 -4.3 RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007 09/29/11 PSA 2011.09.01
EMI 2009.8.29 

# RADIATED EMISSIONS

1 Pass 14 No deviations. Bottom, DC Power Supply: Digital Power Model ADP90-119 SN ADP901195210A0010592

Autopulse 17 BPM Setting 4



### 0 10 20 30 40 50 60 70 80

10 100 1000 

## MHzdBuV/mFreq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

148.370 56.6 41.5 1.0 12.5 2.1 0.0 H 0.0 29.7 34.0 -4.3 147.821 56.5 41.5 1.0 12.5 2.1 0.0 H 0.0 29.6 34.0 -4.4 149.310 56.4 41.5 1.0 12.6 2.1 0.0 H 0.0 29.6 34.0 -4.4 149.066 56.4 41.5 1.0 12.6 2.1 0.0 H 0.0 29.5 34.0 -4.5 148.712 56.4 41.5 1.0 12.5 2.1 0.0 H 0.0 29.5 34.0 -4.5 148.773 56.2 41.5 1.0 12.6 2.1 0.0 H 0.0 29.3 34.0 -4.7

### scan

EUT: Inogen 1G-3 Work Order: INOG0018 Serial Number: Prototype 1 Date: Customer: Inogen Temperature: 23C Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb Tested by: Jeremiah Darden Power: 110VAC/60H z Job Site: OC11 

### TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS Antenna Height(s) (m) 1 Test Distance (m) 1 

### COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDAR D

Run # Configuration # Results Signature

Freq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

287.999 57.9 41.3 1.0 16.3 2.9 0.0 V 0.0 35.8 52.4 -16.6 277.231 54.3 41.2 1.0 15.3 2.8 0.0 V 0.0 31.2 52.2 -21.0 293.981 52.0 41.3 1.0 16.8 2.9 0.0 V 0.0 30.5 52.6 -22.1 291.637 51.5 41.3 1.0 16.6 2.9 0.0 V 0.0 29.8 52.5 -22.7 263.387 52.5 41.1 1.0 14.9 2.8 0.0 V 0.0 29.0 51.8 -22.8 259.333 52.4 41.1 1.0 14.8 2.7 0.0 V 0.0 28.8 51.7 -22.9 262.532 52.3 41.1 1.0 14.9 2.8 0.0 V 0.0 28.8 51.8 -23.0 260.774 52.1 41.1 1.0 14.8 2.7 0.0 V 0.0 28.6 51.8 -23.2 261.043 52.1 41.1 1.0 14.8 2.7 0.0 V 0.0 28.6 51.8 -23.2 261.678 52.0 41.1 1.0 14.8 2.7 0.0 V 0.0 28.5 51.8 -23.3 262.605 52.0 41.1 1.0 14.9 2.8 0.0 V 0.0 28.5 51.8 -23.3 263.924 51.9 41.1 1.0 14.9 2.8 0.0 V 0.0 28.4 51.8 -23.4 246.002 51.7 40.9 1.0 14.4 2.7 0.0 H 0.0 27.9 51.4 -23.5 RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007 09/29/11 PSA 2011.09.01
EMI 2009.8.29 

# RADIATED EMISSIONS

1 Pass 16 No deviations. Top, DC Power Supply: Digital Power Model ADP90-119 SN ADP901195210A0010592

Autopulse 17 BPM Setting 4



### 0 10 20 30 40 50 60 70 80

100 1000 

## MHzdBuV/mFreq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

265.218 51.8 41.2 1.0 14.9 2.8 0.0 V 0.0 28.3 51.9 -23.6 260.115 51.7 41.1 1.0 14.8 2.7 0.0 V 0.0 28.2 51.7 -23.6 263.533 51.7 41.1 1.0 14.9 2.8 0.0 V 0.0 28.2 51.8 -23.6 257.991 51.6 41.1 1.0 14.8 2.7 0.0 V 0.0 28.0 51.7 -23.7 259.871 51.6 41.1 1.0 14.8 2.7 0.0 V 0.0 28.0 51.7 -23.7 261.995 51.6 41.1 1.0 14.8 2.8 0.0 V 0.0 28.1 51.8 -23.7

### scan

EUT: Inogen 1G-3 Work Order: INOG0018 Serial Number: Prototype 1 Date: Customer: Inogen Temperature: 23C Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb Tested by: Jeremiah Darden Power: 110VAC/60H z Job Site: OC11 

### TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS Antenna Height(s) (m) 1 Test Distance (m) 1 

### COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDAR D

Run # Configuration # Results Signature

Freq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

287.999 57.9 41.3 1.0 16.3 2.9 0.0 V 0.0 35.8 52.4 -16.6 845.587 52.6 42.3 1.0 23.2 5.5 0.0 H 0.0 39.0 59.7 -20.7 289.196 53.2 41.3 1.0 16.4 2.9 0.0 V 0.0 31.2 52.5 -21.2 261.726 52.6 41.1 1.0 14.8 2.7 0.0 V 0.0 29.1 51.8 -22.7 260.457 52.5 41.1 1.0 14.8 2.7 0.0 V 0.0 29.0 51.7 -22.8 480.064 51.3 41.2 1.0 18.7 4.1 0.0 V 0.0 32.8 55.9 -23.1 261.800 52.2 41.1 1.0 14.8 2.8 0.0 V 0.0 28.7 51.8 -23.1 263.704 52.2 41.1 1.0 14.9 2.8 0.0 V 0.0 28.7 51.8 -23.1 260.945 52.1 41.1 1.0 14.8 2.7 0.0 V 0.0 28.6 51.8 -23.2 263.558 52.1 41.1 1.0 14.9 2.8 0.0 V 0.0 28.6 51.8 -23.2 262.264 52.0 41.1 1.0 14.8 2.8 0.0 V 0.0 28.5 51.8 -23.3 262.776 52.0 41.1 1.0 14.9 2.8 0.0 V 0.0 28.5 51.8 -23.3 246.881 51.8 40.9 1.0 14.5 2.7 0.0 H 0.0 28.0 51.4 -23.4 RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007 09/29/11 PSA 2011.09.01
EMI 2009.8.29 

# RADIATED EMISSIONS

1 Pass 15 No deviations. Bottom, DC Power Supply: Digital Power Model ADP90-119 SN ADP901195210A0010592

Autopulse 17 BPM Setting 4



### 0 10 20 30 40 50 60 70 80

100 1000 

## MHzdBuV/mFreq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

259.968 51.8 41.1 1.0 14.8 2.7 0.0 V 0.0 28.2 51.7 -23.5 261.336 51.8 41.1 1.0 14.8 2.7 0.0 V 0.0 28.3 51.8 -23.5 262.654 51.5 41.1 1.0 14.9 2.8 0.0 V 0.0 28.0 51.8 -23.8 264.144 51.5 41.1 1.0 14.9 2.8 0.0 V 0.0 28.0 51.8 -23.8 256.184 51.3 41.0 1.0 14.7 2.7 0.0 V 0.0 27.7 51.6 -23.9 257.282 51.3 41.1 1.0 14.7 2.7 0.0 V 0.0 27.7 51.7 -23.9

### scan

EUT: Inogen 1G-3 Work Order: INOG0018 Serial Number: Prototype 1 Date: Customer: Inogen Temperature: 23C Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb Tested by: Jeremiah Darden Power: 110VAC/60H z Job Site: OC11 

### TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS Antenna Height(s) (m) 1 Test Distance (m) 1 

### COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDAR D

Run # Configuration # Results Signature

Freq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

1089.733 42.2 37.6 1.0 24.0 5.0 0.0 H 0.0 33.6 50.0 -16.4 1025.027 42.0 37.7 1.0 23.6 4.9 0.0 H 0.0 32.8 50.0 -17.2 1092.785 40.9 37.6 1.0 24.0 5.0 0.0 H 0.0 32.3 50.0 -17.7 1051.886 41.2 37.7 1.0 23.8 5.0 0.0 H 0.0 32.2 50.0 -17.8 1096.447 40.6 37.6 1.0 24.1 5.0 0.0 V 0.0 32.1 50.0 -17.9 1045.782 41.1 37.7 1.0 23.7 4.9 0.0 V 0.0 32.1 50.0 -17.9 1049.444 41.0 37.7 1.0 23.7 4.9 0.0 V 0.0 32.0 50.0 -18.0 1048.834 41.0 37.7 1.0 23.7 4.9 0.0 H 0.0 32.0 50.0 -18.0 1092.785 40.3 37.6 1.0 24.0 5.0 0.0 V 0.0 31.7 50.0 -18.3 1081.797 40.4 37.6 1.0 24.0 5.0 0.0 H 0.0 31.7 50.0 -18.3 1083.628 40.3 37.6 1.0 24.0 5.0 0.0 H 0.0 31.6 50.0 -18.4 1053.107 40.6 37.7 1.0 23.8 5.0 0.0 V 0.0 31.6 50.0 -18.4 1042.119 40.7 37.7 1.0 23.7 4.9 0.0 H 0.0 31.6 50.0 -18.4 RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007 09/29/11 PSA 2011.09.01
EMI 2009.8.29 

# RADIATED EMISSIONS

1 Pass 17 No deviations. Top, DC Power Supply: Digital Power Model ADP90-119 SN ADP901195210A0010592

Autopulse 17 BPM Setting 4



### 0 10 20 30 40 50 60 70 80

1000 10000 

## MHzdBuV/mFreq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

1020.755 40.8 37.8 1.0 23.5 4.9 0.0 H 0.0 31.5 50.0 -18.5 1656.818 42.1 37.9 1.0 25.2 5.0 0.0 V 0.0 34.4 53.0 -18.6 1076.914 40.1 37.6 1.0 23.9 5.0 0.0 V 0.0 31.4 50.0 -18.6 1085.460 40.0 37.6 1.0 24.0 5.0 0.0 V 0.0 31.4 50.0 -18.6 1070.199 40.1 37.7 1.0 23.9 5.0 0.0 H 0.0 31.3 50.0 -18.7 1056.159 40.2 37.7 1.0 23.8 5.0 0.0 H 0.0 31.3 50.0 -18.7

### scan

EUT: Inogen 1G-3 Work Order: INOG0018 Serial Number: Prototype 1 Date: Customer: Inogen Temperature: 23C Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb Tested by: Jeremiah Darden Power: 110VAC/60H z Job Site: OC11 

### TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS Antenna Height(s) (m) 1 Test Distance (m) 1 

### COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDAR D

Run # Configuration # Results Signature

Freq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

1026.859 42.0 37.7 1.0 23.6 4.9 0.0 H 0.0 32.8 50.0 -17.2 1063.484 41.3 37.7 1.0 23.8 5.0 0.0 V 0.0 32.4 50.0 -17.6 1070.809 41.1 37.7 1.0 23.9 5.0 0.0 V 0.0 32.3 50.0 -17.7 1041.509 41.3 37.7 1.0 23.7 4.9 0.0 V 0.0 32.2 50.0 -17.8 1078.745 40.9 37.6 1.0 24.0 5.0 0.0 V 0.0 32.2 50.0 -17.8 1021.975 41.4 37.8 1.0 23.6 4.9 0.0 H 0.0 32.1 50.0 -17.9 1090.343 40.5 37.6 1.0 24.0 5.0 0.0 H 0.0 31.9 50.0 -18.1 1098.279 40.4 37.6 1.0 24.1 5.0 0.0 H 0.0 31.9 50.0 -18.1 1036.015 41.0 37.7 1.0 23.7 4.9 0.0 V 0.0 31.9 50.0 -18.1 1028.690 41.0 37.7 1.0 23.6 4.9 0.0 H 0.0 31.8 50.0 -18.2 1061.653 40.6 37.7 1.0 23.8 5.0 0.0 H 0.0 31.7 50.0 -18.3 1037.846 40.8 37.7 1.0 23.7 4.9 0.0 H 0.0 31.7 50.0 -18.3 1082.407 40.2 37.6 1.0 24.0 5.0 0.0 V 0.0 31.5 50.0 -18.5 RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007 09/29/11 PSA 2011.09.01
EMI 2009.8.29 

# RADIATED EMISSIONS

1 Pass 18 No deviations. Bottom, DC Power Supply: Digital Power Model ADP90-119 SN ADP901195210A0010592

Autopulse 17 BPM Setting 4



### 0 10 20 30 40 50 60 70 80

1000 10000 

## MHzdBuV/mFreq Amplitude Preamp Antenna Height Transducer Cable External Attenuation Polarity/ Transducer

Type Detector Distance Adjustment Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks [PK] from scan) (dB) dBuV/m dBuV/m (dB)

1081.187 40.2 37.6 1.0 24.0 5.0 0.0 H 0.0 31.5 50.0 -18.5 1065.926 40.3 37.7 1.0 23.9 5.0 0.0 V 0.0 31.5 50.0 -18.5 1091.564 40.0 37.6 1.0 24.0 5.0 0.0 H 0.0 31.4 50.0 -18.6 1039.678 40.5 37.7 1.0 23.7 4.9 0.0 H 0.0 31.4 50.0 -18.6 1077.524 40.1 37.6 1.0 23.9 5.0 0.0 H 0.0 31.4 50.0 -18.6 1065.316 40.2 37.7 1.0 23.9 5.0 0.0 H 0.0 31.4 50.0 -18.6

## photo

re (9).JPG re (1).JPG 

# RADIATED EMISSIONS PSA 2011.09.01

## photo

re (2).JPG re (3).JPG 

# RADIATED EMISSIONS PSA 2011.09.01

## photo

re (4).JPG re (5).JPG 

# RADIATED EMISSIONS PSA 2011.09.01

## photo

re (6).JPG re (7).JPG 

# RADIATED EMISSIONS PSA 2011.09.01

## photo

re (8).JPG 

# RADIATED EMISSIONS PSA 2011.09.01

## description

MODES OF OPERATION

POWER SETTINGS INVESTIGATED

CONFIGURATIONS INVESTIGATED

FREQUENCY RANGE INVESTIGATED Start Frequency 150 kHz Stop Frequency 30 MHz

SAMPLE CALCULATIONS

Conducted Emissions: Adjusted Level = Measured Level + Transducer Factor + Cable Attenuation Factor + External Attenuator



## TEST EQUIPMENT ID Interval OCT 12 CAA 0 CAC 0 LIJ 12 LIL 12 NCE 24 AFG 12

MEASUREMENT BANDWIDTHS

MEASUREMENT UNCERTAINTY

TEST DESCRIPTION Autopulse 17 BPM Setting 4

Description Manufacturer Model Last Cal. Battery 110VAC/60Hz

INOG0018 - 1

OC11 Cables N/A Disturbance Power Cables 8/25/2011 10uF Feedthrough Capacitor Solar 6512-106R NCR 10uF Feedthrough Capacitor Solar 6512-106R NCR LISN Fischer Custom FCC-LISN-DO-160 7/27/2011 LISN Fischer Custom FCC LISN DO-160 7/27/2011 Current Probe Fischer Custom F-33-4 6/3/2011

Ameasurement uncertainty estimation has been performed for each test per our internal quality document WP 342. The estimation is used to compare the measured result with its "true" or theoretically correct value. The expanded measurement uncertainty for radiated emissions measurements is less than +/- 4 dB, and for conducted emissions measurements is less than +/- 2.7 dB. Our measurement data meets or exceeds the measurement uncertainty requirements of CISPR 16-4; therefore, the test data can be compared directly to the specification limit to determine compliance. The calculations for measurement uncertainty are available upon request. 400.0 - 1000.0 1000.0 - 6000.0 100.0 1000.0 0.15 - 30.0 30.0 - 400.0 1.0 10.0 Frequency Range (MHz) BWI (kHz) PSA 2011.09.01



## Testing was performed using the mode(s) of operation and configuration(s) noted within the report. The individuals and/or the organization requesting the test provided the modes, configurations and settings used to complete the evaluation. The actual test parameters are specified in the test data, this includes items such as investigated frequency range (scanned) and test levels. The testing methods and performance specifications, as well as the test site used for the evaluation are indicated in the test data. 

# CONDUCTED EMISSIONS

Spectrum Analyzer Agilent E4440A 4/28/2011

Using the mode of operation and configuration noted within this report, a conducted emissions test was performed. The frequency range tested, is also noted on the data sheets in this report. Conducted emissions measurements were made with the test setup as defined in RTCA DO160 E. A calibrated current probe was positioned within 5 cm of the EUT to measure the conducted disturbances while the EUT was powered. Where applicable the conducted disturbances on both the power leads and interconnecting cable bundles were measured.

### cond

EUT: Ambien t Work Order: INOG0018 Serial Number: None Date: Customer: Inogen Temperature: 23C Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb Tested by: Jeremiah Darden Power: 110VAC/60H z Job Site: OC11 

### TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS Cable or Line Tested Ambient 

### COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDAR D

Run # Configuration # Results Signature

Freq Amplitude Transducer Cable External Attenuation Detector Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (dB) (dB) (blank equal peaks [PK] from scan) dBuA dBuA (dB)

1.932 2.1 0.1 0.2 0.0 2.4 40.4 -38.0 2.000 0.2 0.1 0.2 0.0 0.5 40.0 -39.5 4.398 0.1 0.1 0.3 0.0 0.5 40.0 -39.5 3.222 0.1 0.1 0.2 0.0 0.4 40.0 -39.6 12.146 -0.3 0.1 0.5 0.0 0.3 40.0 -39.7 2.768 0.0 0.1 0.2 0.0 0.3 40.0 -39.7 6.373 -0.2 0.1 0.4 0.0 0.3 40.0 -39.7 2.021 -0.2 0.1 0.2 0.0 0.1 40.0 -39.9 6.289 -0.4 0.1 0.4 0.0 0.1 40.0 -39.9 6.839 -0.4 0.1 0.4 0.0 0.1 40.0 -39.9 23.706 -0.9 0.1 0.8 0.0 0.0 40.0 -40.0 2.499 -0.3 0.1 0.2 0.0 0.0 40.0 -40.0 16.643 -0.8 0.1 0.7 0.0 0.0 40.0 -40.0 3.620 -0.4 0.1 0.3 0.0 0.0 40.0 -40.0 4 No deviations. None

None

1 Pass RTCA DO-160F Category M: 2007 RTCA DO-160F Interconnecting Cables: 2007 09/29/11 PSA 2011.09.01
EMI 2009.8.29 

# CONDUCTED EMISSIONS

-10

0 10 20 30 40 50 60 70 80

0.1 1 10 100 

## MHzdBuAFreq Amplitude Transducer Cable External Attenuation Detector Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (dB) (dB) (blank equal peaks [PK] from scan) dBuA dBuA (dB)

2.719 -0.4 0.1 0.2 0.0 -0.1 40.0 -40.1 3.037 -0.4 0.1 0.2 0.0 -0.1 40.0 -40.1 3.784 -0.5 0.1 0.3 0.0 -0.1 40.0 -40.1 23.010 -1.1 0.1 0.8 0.0 -0.2 40.0 -40.2 3.900 -0.6 0.1 0.3 0.0 -0.2 40.0 -40.2

### cond

EUT: Inogen 1G-3 Work Order: INOG0018 Serial Number: Prototype 1 Date: Customer: Inogen Temperature: 23C Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb Tested by: Jeremiah Darden Power: 110VAC/60H z Job Site: OC11 

### TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS Cable or Line Tested DC Cable 

### COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDAR D

Run # Configuration # Results Signature

Freq Amplitude Transducer Cable External Attenuation Detector Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (dB) (dB) (blank equal peaks [PK] from scan) dBuA dBuA (dB)

11.989 7.3 0.1 0.5 0.0 7.9 40.0 -32.1 9.507 6.6 0.1 0.5 0.0 7.2 40.0 -32.8 11.906 6.5 0.1 0.5 0.0 7.1 40.0 -32.9 13.382 6.1 0.1 0.6 0.0 6.8 40.0 -33.2 13.386 5.9 0.1 0.6 0.0 6.6 40.0 -33.4 11.816 4.9 0.1 0.5 0.0 5.5 40.0 -34.5 13.713 4.6 0.1 0.6 0.0 5.3 40.0 -34.7 13.056 4.5 0.1 0.6 0.0 5.2 40.0 -34.8 12.075 4.5 0.1 0.5 0.0 5.1 40.0 -34.9 11.733 4.3 0.1 0.5 0.0 4.9 40.0 -35.1 12.729 4.2 0.1 0.6 0.0 4.9 40.0 -35.1 12.483 4.2 0.1 0.6 0.0 4.9 40.0 -35.1 12.975 4.0 0.1 0.6 0.0 4.7 40.0 -35.3 11.333 4.0 0.1 0.5 0.0 4.6 40.0 -35.4 1 No deviations. Probe position on DC In Side of PS, DC Power Supply: Digital Power Model ADP90-119 SN ADP901195210A0010592

Autopulse 17 BPM Setting 4

1 Pass RTCA DO-160F Category M: 2007 RTCA DO-160F Interconnecting Cables: 2007 09/29/11 PSA 2011.09.01
EMI 2009.8.29 

# CONDUCTED EMISSIONS

-10

0 10 20 30 40 50 60 70 80

0.1 1 10 100 

## MHzdBuAFreq Amplitude Transducer Cable External Attenuation Detector Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (dB) (dB) (blank equal peaks [PK] from scan) dBuA dBuA (dB)

12.562 3.9 0.1 0.6 0.0 4.6 40.0 -35.4 11.579 3.9 0.1 0.5 0.0 4.5 40.0 -35.5 12.236 3.7 0.1 0.5 0.0 4.3 40.0 -35.7 11.646 3.7 0.1 0.5 0.0 4.3 40.0 -35.7 12.153 3.6 0.1 0.5 0.0 4.2 40.0 -35.8

### cond

EUT: Inogen 1G-3 Work Order: INOG0018 Serial Number: Prototype 1 Date: Customer: Inogen Temperature: 23C Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb Tested by: Jeremiah Darden Power: 110VAC/60H z Job Site: OC11 

### TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS Cable or Line Tested DC Cable 

### COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDAR D

Run # Configuration # Results Signature

Freq Amplitude Transducer Cable External Attenuation Detector Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (dB) (dB) (blank equal peaks [PK] from scan) dBuA dBuA (dB)

9.761 4.2 0.1 0.5 0.0 4.8 40.0 -35.2 11.661 4.0 0.1 0.5 0.0 4.6 40.0 -35.4 12.154 3.7 0.1 0.5 0.0 4.3 40.0 -35.7 12.975 3.5 0.1 0.6 0.0 4.2 40.0 -35.8 12.730 3.4 0.1 0.6 0.0 4.1 40.0 -35.9 13.302 3.3 0.1 0.6 0.0 4.0 40.0 -36.0 7.620 3.3 0.1 0.4 0.0 3.8 40.0 -36.2 11.744 3.2 0.1 0.5 0.0 3.8 40.0 -36.2 12.401 3.1 0.1 0.6 0.0 3.8 40.0 -36.2 13.220 3.0 0.1 0.6 0.0 3.7 40.0 -36.3 13.445 2.9 0.1 0.6 0.0 3.6 40.0 -36.4 13.468 2.9 0.1 0.6 0.0 3.6 40.0 -36.4 13.795 2.9 0.1 0.6 0.0 3.6 40.0 -36.4 11.908 2.6 0.1 0.5 0.0 3.2 40.0 -36.8 2 No deviations. Probe position on DC Out Side of PS, DC Power Supply: Digital Power Model ADP90-119 SN ADP901195210A0010592

Autopulse 17 BPM Setting 4

1 Pass RTCA DO-160F Category M: 2007 RTCA DO-160F Interconnecting Cables: 2007 09/29/11 PSA 2011.09.01
EMI 2009.8.29 

# CONDUCTED EMISSIONS

-10

0 10 20 30 40 50 60 70 80

0.1 1 10 100 

## MHzdBuAFreq Amplitude Transducer Cable External Attenuation Detector Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (dB) (dB) (blank equal peaks [PK] from scan) dBuA dBuA (dB)

13.139 2.4 0.1 0.6 0.0 3.1 40.0 -36.9 9.772 2.4 0.1 0.5 0.0 3.0 40.0 -37.0 11.086 2.4 0.1 0.5 0.0 3.0 40.0 -37.0 11.417 2.4 0.1 0.5 0.0 3.0 40.0 -37.0 13.271 2.3 0.1 0.6 0.0 3.0 40.0 -37.0

### cond

EUT: Inogen 1G-3 Work Order: INOG0018 Serial Number: Prototype 1 Date: Customer: Inogen Temperature: 23C Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb Tested by: Jeremiah Darden Power: 110VAC/60H z Job Site: OC11 

### TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS Cable or Line Tested DC Cable 

### COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDAR D

Run # Configuration # Results Signature

Freq Amplitude Transducer Cable External Attenuation Detector Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (dB) (dB) (blank equal peaks [PK] from scan) dBuA dBuA (dB)

8.236 6.7 0.1 0.4 0.0 7.2 40.0 -32.8 11.047 4.1 0.1 0.5 0.0 4.7 40.0 -35.3 11.004 3.5 0.1 0.5 0.0 4.1 40.0 -35.9 12.159 3.3 0.1 0.5 0.0 3.9 40.0 -36.1 11.818 3.3 0.1 0.5 0.0 3.9 40.0 -36.1 8.240 3.2 0.1 0.4 0.0 3.7 40.0 -36.3 9.610 3.0 0.1 0.5 0.0 3.6 40.0 -36.4 8.790 2.8 0.1 0.4 0.0 3.3 40.0 -36.7 11.302 2.7 0.1 0.5 0.0 3.3 40.0 -36.7 12.588 2.6 0.1 0.6 0.0 3.3 40.0 -36.7 11.331 2.6 0.1 0.5 0.0 3.2 40.0 -36.8 7.064 2.6 0.1 0.4 0.0 3.1 40.0 -36.9 7.533 2.6 0.1 0.4 0.0 3.1 40.0 -36.9 10.841 2.5 0.1 0.5 0.0 3.1 40.0 -36.9 3 No deviations. Probe position on DC Out Side of EUT, DC Power Supply: Digital Power Model ADP90-119 SN ADP901195210A0010592

Autopulse 17 BPM Setting 4

1 Pass RTCA DO-160F Category M: 2007 RTCA DO-160F Interconnecting Cables: 2007 09/29/11 PSA 2011.09.01
EMI 2009.8.29 

# CONDUCTED EMISSIONS

-10

0 10 20 30 40 50 60 70 80

0.1 1 10 100 

## MHzdBuAFreq Amplitude Transducer Cable External Attenuation Detector Adjusted Spec. Limit Compared to Spec. (MHz) (dBuV) (dB) (dB) (dB) (blank equal peaks [PK] from scan) dBuA dBuA (dB)

2.055 2.7 0.1 0.2 0.0 3.0 40.0 -37.0 13.221 2.3 0.1 0.6 0.0 3.0 40.0 -37.0 11.076 2.3 0.1 0.5 0.0 2.9 40.0 -37.1 10.910 2.2 0.1 0.5 0.0 2.8 40.0 -37.2 11.577 2.2 0.1 0.5 0.0 2.8 40.0 -37.2

## photo

ce (4).JPG ce (1).JPG 

# CONDUCTED EMISSIONS PSA 2011.09.01

## photo

ce (2).JPG ce (3).JPG 

# CONDUCTED EMISSIONS PSA 2011.09.01