Inogen 

Inogen One G-3 

Report No. INOG0018 

Report Prepared By Northwest EMC Inc. 

NORTHWEST EMC – (888) 364-2378 – www.nwemc.com

California – Minnesota – Oregon – New York – Washington 
22975 NW Evergreen Parkway 

Suite 400 

Hillsboro, Oregon 97124 

Certificate of Test 

Last Date of Test: September 29, 2011 

Inogen 

Model: Inogen One G-3 

Emissions 

Test Description Specification Test Method Pass/Fail 

Radiated Emissions RTCA DO-160F:2007, Category M RTCA DO-160F:2007 Pass 

Conducted Emissions RTCA DO-160F:2007, Category M RTCA DO-160F:2007 Pass 

Deviations From Test Standards 

None 

Approved By: 

Victor Ratinoff, Operations Manager 

 

This report must not be used to claim product certification, approval, or endorsement by NVLAP, NIST, or any agency of the 

federal government of the United States of America. 

Product compliance is the responsibility of the client, therefore the tests and equipment modes of operation represented in 

this report were agreed upon by the client, prior to testing. This Report may only be duplicated in its entirety. The results of 

this test pertain only to the sample(s) tested. The specific description is noted in each of the individual sections of the test 

report supporting this certificate of test. 
Revision History

Revision 09/01/11 

Revision 

Number Description Date Page Number 

 

00 None 
Accreditations and 

Authorizations

Revision 09/01/11 

FCC 

Accredited by NVLAP for performance of FCC radio, digital, and ISM device testing. Our Open Area Test Sites, certification 

chambers, and conducted measurement facilities have been fully described in reports filed with the FCC and accepted by the FCC 

in letters maintained in our files. Northwest EMC has been accredited by ANSI to ISO / IEC Guide 65 as a product certifier. We

have been designated by the FCC as a Telecommunications Certification Body (TCB). This allows Northwest EMC to certify 

transmitters to FCC specifications in accordance with 47 CFR 2.960 and 2.962. 

NVLAP 

Northwest EMC, Inc. is accredited under the National Voluntary Laboratory Accreditation Program (NVLAP) for satisfactory 

compliance with the requirements of ISO/IEC 17025 for Testing Laboratories. NVLAP is administered by the National Institute of 

Standards and Technology (NIST), an agency of the U.S. Commerce Department. The NVLAP accreditation encompasses 

Electromagnetic Compatibility Testing in accordance with the European Union EMC Directive 2004/108/EC, and ANSI C63.4. 

Additionally, Northwest EMC is accredited by NVLAP to perform radio testing in accordance with the European Union R&TTE 

Directive 1999/5/EEC, the requirements of FCC, and the RSS radio standards for Industry Canada. 

Industry Canada 

Accredited by NVLAP for performance of Industry Canada RSS and ICES testing. Our Open Area Test Sites and certification 

chambers comply with RSS-Gen, Issue 2 and have been filed with Industry Canada and accepted. Northwest EMC has been 

accredited by ANSI to ISO / IEC Guide 65 as a product certifier. We have been designated by NIST and recognized by Industry 

Canada as a Certification Body (CB) per the APEC Mutual Recognition Arrangement (MRA). This allows Northwest EMC to certify 

transmitters to Industry Canada technical requirements. (Site Filing Numbers - Hillsboro: 2834D-1, 2834D-2, Sultan: 2834C-1, 

Irvine: 2834B-1, 2834B-2, Brooklyn Park: 2834E-1) 

CAB 

Designated by NIST and validated by the European Commission as a Conformity Assessment Body (CAB) to conduct tests and 

approve products to the EMC directive and transmitters to the R&TTE directive, as described in the U.S. - EU Mutual Recognition

Agreement. 

Australia/New Zealand 

The National Association of Testing Authorities (NATA), Australia has been appointed by the ACA as an accreditation body to 

accredit test laboratories and competent bodies for EMC standards. Accredited test reports or assessments by competent bodies 

must carry the NATA logo. Test reports made by an overseas laboratory that has been accredited for the relevant standards by an

overseas accreditation body that has a Mutual Recognition Agreement (MRA) with NATA are also accepted as technical grounds for 

product conformity. The report should be endorsed with the respective logo of the accreditation body (NVLAP). 
Accreditations and 

Authorizations

Revision 09/01/11 

VCCI 

Accepted as an Associate Member to the VCCI, Acceptance No. 564. Conducted and radiated measurement facilities have been 

registered in accordance with Regulations for Voluntary Control Measures, Article 8. (Registration Numbers. - Hillsboro: C-1071, R1025, G-84, C-2687, T-1658, and R-2318, Irvine: R-1943, G-85, C-2766, and T-1659, Sultan: R-871, G-83, C-3265, and T-1511, 

Brooklyn Park: R-3125, G-86, G-141, C-3464, and T-1634). 

BSMI 

Northwest EMC has been designated by NIST and validated by C-Taipei (BSMI) as a CAB to conduct tests as described in the 

APEC Mutual Recognition Agreement (US0017). 

GOST 

Northwest EMC, Inc. has been assessed and accredited by the Russian Certification bodies Certinform VNIINMASH, CERTINFO, 

SAMTES, and Federal CHEC, to perform EMC and Hygienic testing for Information Technology Products. As a result of their 

laboratory assessment, they will accept test results from Northwest EMC, Inc. for product certification 

KCC 

Northwest EMC, Inc is a CAB designated by MRA partners and recognized by Korea. (Assigned Lab Numbers: Hillsboro: US0017, 

Irvine: US0158, Sultan: US0157, Brooklyn Park: US0175) 

VIETNAM 

Vietnam MIC has approved Northwest EMC as an accredited test lab. Per Decision No. 194/QD-QLCL (dated December 15, 2009), 

Northwest EMC test reports can be used for Vietnam approval submissions. 

SCOPE 

For details on the Scopes of our Accreditations, please visit: 

http://www.nwemc.com/accreditations/
Explanation of NWEMC 

Performance Criteria

Revision 09/01/11 

How important is it to understand performance criteria? 

It is the responsibility of the test laboratory to observe the results of the tests that are performed and to 

accurately report those results. As the responsible party (manufacturer, importer, etc) it is your 

responsibility to take those results, compare them against the specifications and standards, then, if 

appropriate make a declaration of conformity. As the responsible party it makes sense that you are fully 

aware of the requirements, how your device performs when tested to those requirements, and what 

information is being used to declare conformity. 

To better assist you in making those conformity decisions, Northwest EMC has adopted a very simple, yet 

very clear performance assessment procedure. The following criteria is used when performing immunity 

or susceptibility tests: 

Performance Criteria 1: 

 The EUT exhibited no change in performance when operating as specified by the manufacturer. 

In this case no changes were observed during the test. 

 In most cases this would be equivalent to Performance Criteria A. When operating the equipment 

in the modes or configurations specified by the responsible party, monitoring the parameters 

specified, no changes were observed. Basically nothing happened. 

Performance Criteria 2: 

 The EUT exhibited a change in performance when operating as specified by the manufacturer. In 

this case the equipment recovered without any operator intervention, once the test signal was 

removed. The data sheets will detail the exact phenomena observed. 

 In most cases this would be equivalent to Performance Criteria B. When operating the equipment 

in the modes or configurations specified by the responsible party, monitoring the parameters 

specified, changes were observed. The EUT was able to recover from those changes without 

any operator intervention, once the test signal was removed. 

Performance Criteria 3: 

 The EUT exhibited a change in performance when operating as specified by the manufacturer. In 

this case the equipment required some operator intervention in order to recover. This 

intervention may be in the form of changing EUT settings, or even resetting the system. The data 

sheets will detail the exact phenomena observed. 

 In most cases this would be equivalent to Performance Criteria C. When operating the 

equipment in the modes or configurations specified by the responsible party, monitoring the 

parameters specified, changes were observed. The EUT required some sort of operator 

intervention to recover. There was no permanent damage and the EUT appeared to function 

normally after completion of test. 

Performance Criteria 4: 

 The EUT exhibited a change in performance when operating as specified by the manufacturer. In 

this case the equipment was damaged and would not recover. The data sheets will detail the 

exact phenomena observed. 

 In most cases there is no specific criterion to compare this to; it typically ends the test. When 

operating the equipment in the modes or configurations specified by the responsible party, 

monitoring the parameters specified, changes were observed. There was no recovery; the 

equipment would no longer function as intended. 
Locations

Revision 09/01/11 

Oregon 

Labs EV01-EV12 

22975 NW Evergreen Pkwy 

Suite 400 

Hillsboro, OR 97124 

(503) 844-4066 

California 

Labs OC01-OC13 

41 Tesla 

Irvine, CA 92618 

(949) 861-8918 

Minnesota 

Labs MN01-MN08 

9349 W Broadway Ave. 

Brooklyn Park, 

MN 55445 

(763) 425-2281 

Washington 

Labs SU01-SU07 

14128 339th Ave. SE 

Sultan, WA 98294 

(360) 793-8675 

New York 

Labs WA01-WA04 

4939 Jordan Rd. 

Elbridge, NY 13060 

(315) 685-0796 
Product Description

Revision 09/01/11 

Client and Equipment Under Test (EUT) Information 

Company Name: Inogen 

Address: 326 Bollay Drive 

City, State, Zip: Goleta, CA 93117 

Test Requested By: Mike Pollack 

Model: Inogen One G-3 

First Date of Test: September 29, 2011 

Last Date of Test: September 29, 2011 

Receipt Date of Samples: September 29, 2011 

Equipment Design Stage: Prototype 

Equipment Condition: No Damage 

Information Provided by the Party Requesting the Test 

Functional Description of the EUT (Equipment Under Test):

Portable Oxygen Concentrator 

Testing Objective: 

Provide the specific EMC testing and any associated documentation requested by the client. 

EUT Photo 
Product Description

Revision 09/01/11 

 

 
Configurations

Revision 09/01/11 

Configuration 1 INOG0018 

EUT 

Description Manufacturer Model/Part Number Serial Number 

Oxygen Concentrator Inogen Inogen One G-3 Prototype 1 

DC Power Adapter Digital Power ADP90-119 ADP901195210A0010592 

Cables 

Cable Type Shield Length (m) Ferrite Connection 1 Connection 2 

DC Out Cable No 1.8m Yes DC Power Adapter Inogen 1G-3 

DC In Cable No .9m No DC Power Adapter DC Battery 

PA = Cable is permanently attached to the device. Shielding and/or presence of ferrite may be unknown.

Configuration 2 INOG0018 

EUT 

Description Manufacturer Model/Part Number Serial Number 

Oxygen Concentrator Inogen Inogen One G-3 Prototype 1 
Modifications

Revision 09/01/11 

Equipment Modifications 

Item Date Test Modification Note Disposition of EUT 

 1 9/29/2011 Radiated 

Emissions 

Tested as 

delivered to 

Test Station. 

No EMI suppression 

devices were added or 

modified during this test. 

EUT remained at 

Northwest EMC 

following the test. 

 2 9/29/2011 Conducted 

Emissions 

Tested as 

delivered to 

Test Station. 

No EMI suppression 

devices were added or 

modified during this test. 

Scheduled testing 

was completed. 
description

MODES OF OPERATION

POWER SETTINGS INVESTIGATED

CONFIGURATIONS INVESTIGATED

FREQUENCY RANGE INVESTIGATED

Start Frequency 2 MHz Stop Frequency 6 GHz

SAMPLE CALCULATIONS

Radiated Emissions: Field Strength = Measured Level + Antenna Factor + Cable Factor - Amplifier Gain + Distance Adjustment Factor + External Attenuation

TEST EQUIPMENT

ID Interval

CAA 0

CAC 0

LIJ 12

LIL 12

ALA 12

ABJ 12

ACC 12

AIR 24

OCR 12

OCT 12

AOU 12

AFG 12

MEASUREMENT BANDWIDTHS

MEASUREMENT UNCERTAINTY

TEST DESCRIPTION

BWI

(kHz)

10uF Feedthrough Capacitor Solar 6512-106R NCR

10uF Feedthrough Capacitor Solar 6512-106R NCR

1.0

10.0

100.0

1000.0

Autopulse 17 BPM Setting 4

110VAC/60Hz

Battery

Description Manufacturer Model Last Cal.

INOG0018 - 1

LISN Fischer Custom Communications FCC-LISN-DO-160 7/27/2011

LISN Fischer Custom Communications FCC LISN DO-160 7/27/2011

Antenna, Log Periodic EMCO 3146 8/8/2011

Antenna, Bicon EMCO 3104C 8/8/2011

Antenna, Active Rod AH Systems SAS-200/550-1B 1/27/2011

Antenna, Horn (DRG) ETS Lindgren 3115 5/26/2011

OC11 Cables N/A 1-8GHz RE Cables 4/8/2011

OC11 Cables N/A Disturbance Power Cables 8/25/2011

Pre-Amplifier Miteq AM-1551 3/4/2011

Spectrum Analyzer Agilent E4440A 4/28/2011

PSA 2011.09.01

RADIATED EMISSIONS

Testing was performed using the mode(s) of operation and configuration(s) noted within the report. The individuals and/or the 

organization requesting the test provided the modes, configurations and settings used to complete the evaluation. The actual test 

parameters are specified in the test data, this includes items such as investigated frequency range (scanned) and test levels. The 

testing methods and performance specifications, as well as the test site used for the evaluation are indicated in the test data. The 

test data represents the configuration / operating mode/ model that produced the highest emission levels as compared to the 

specification limit.

400.0 - 1000.0

A measurement uncertainty estimation has been performed for each test per our internal quality document WP 342. The 

estimation is used to compare the measured result with its "true" or theoretically correct value. The expanded measurement 

uncertainty for radiated emissions measurements is less than +/- 4 dB, and for conducted emissions measurements is less than +/-

2.7 dB. Our measurement data meets or exceeds the measurement uncertainty requirements of CISPR 16-4; therefore, the test 

data can be compared directly to the specification limit to determine compliance. The calculations for measurement uncertainty are 

available upon request.

1000.0 - 6000.0

Frequency Range

(MHz)

0.15 - 30.0

30.0 - 400.0

Using the mode of operation and configuration noted within this report, a radiated emissions test was performed. The frequency 

range tested, is also noted in this report. Radiated emissions measurements were made with the test setup as defined in RTCA 

DO160 E. A calibrated antenna was positioned at the specified distance and height from the periphery of the EUT. Where 

polarized antennas are used, emissions were measured with both polarities of the antenna. Sufficient number of antenna 

locations are used to measure the emissions from the entire test setup.
scan

EUT: Ambient Work Order: INOG0018

Serial Number: None Date:

Customer: Inogen Temperature: 23C

Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb

Tested by: Jeremiah Darden Power: 110VAC/60Hz Job Site: OC10

TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS

Antenna Height(s) (m) 1 Test Distance (m) 1

COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDARD

Run #

Configuration #

Results Signature

Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

26.325 8.2 0.0 1.0 0.0 0.8 0.0 V 0.0 9.0 35.4 -26.3

26.534 7.6 0.0 1.0 0.0 0.8 0.0 V 0.0 8.4 35.4 -27.0

26.612 7.3 0.0 1.0 0.0 0.9 0.0 V 0.0 8.2 35.5 -27.3

26.395 6.7 0.0 1.0 0.0 0.8 0.0 V 0.0 7.5 35.4 -27.8

26.893 6.8 0.0 1.0 0.0 0.9 0.0 V 0.0 7.7 35.5 -27.9

26.405 6.6 0.0 1.0 0.0 0.8 0.0 V 0.0 7.4 35.4 -28.0

25.688 6.4 0.0 1.0 0.0 0.8 0.0 V 0.0 7.2 35.2 -28.0

26.683 6.6 0.0 1.0 0.0 0.9 0.0 V 0.0 7.5 35.5 -28.0

26.543 6.4 0.0 1.0 0.0 0.8 0.0 V 0.0 7.2 35.4 -28.2

28.679 6.8 0.0 1.0 0.0 0.9 0.0 V 0.0 7.7 36.0 -28.3

26.753 6.2 0.0 1.0 0.0 0.9 0.0 V 0.0 7.1 35.5 -28.4

26.466 6.0 0.0 1.0 0.0 0.8 0.0 V 0.0 6.8 35.4 -28.6

27.005 6.0 0.0 1.0 0.0 0.9 0.0 V 0.0 6.9 35.6 -28.7

RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007

09/29/11

PSA 2011.09.01

EMI 2009.8.29

RADIATED EMISSIONS

1

Pass

2

No deviations.

None

None

-10

0

10

20

30

40

50

60

70

80

1 10 100

MHz

dBuV/m
Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

26.044 5.7 0.0 1.0 0.0 0.8 0.0 V 0.0 6.5 35.3 -28.8

26.604 5.4 0.0 1.0 0.0 0.9 0.0 V 0.0 6.3 35.4 -29.2

26.899 5.4 0.0 1.0 0.0 0.9 0.0 V 0.0 6.3 35.5 -29.3

26.744 5.3 0.0 1.0 0.0 0.9 0.0 V 0.0 6.2 35.5 -29.3

27.679 5.5 0.0 1.0 0.0 0.9 0.0 V 0.0 6.4 35.7 -29.3

26.386 5.2 0.0 1.0 0.0 0.8 0.0 V 0.0 6.0 35.4 -29.3
scan

EUT: Ambient Work Order: INOG0018

Serial Number: None Date:

Customer: Inogen Temperature: 23C

Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb

Tested by: Jeremiah Darden Power: 110VAC/60Hz Job Site: OC11

TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS

Antenna Height(s) (m) 1 Test Distance (m) 1

COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDARD

Run #

Configuration #

Results Signature

Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

120.303 28.0 41.7 1.0 15.2 1.9 0.0 V 0.0 3.4 34.0 -30.6

111.024 27.8 41.7 1.0 14.4 1.9 0.0 H 0.0 2.4 34.0 -31.6

116.091 27.1 41.6 1.0 14.9 1.9 0.0 H 0.0 2.3 34.0 -31.7

115.493 26.8 41.6 1.0 14.8 1.9 0.0 H 0.0 1.9 34.0 -32.1

119.888 26.3 41.7 1.0 15.2 1.9 0.0 V 0.0 1.8 34.0 -32.2

124.356 26.8 41.7 1.0 14.7 2.0 0.0 V 0.0 1.8 34.0 -32.2

118.704 26.3 41.6 1.0 15.1 1.9 0.0 V 0.0 1.7 34.0 -32.3

125.980 26.9 41.7 1.0 14.4 2.0 0.0 H 0.0 1.6 34.0 -32.4

121.157 26.2 41.7 1.0 15.1 1.9 0.0 H 0.0 1.5 34.0 -32.5

119.070 26.0 41.6 1.0 15.1 1.9 0.0 V 0.0 1.4 34.0 -32.6

123.111 26.2 41.7 1.0 14.8 2.0 0.0 V 0.0 1.3 34.0 -32.7

121.743 26.0 41.7 1.0 15.0 1.9 0.0 H 0.0 1.3 34.0 -32.7

115.627 26.1 41.6 1.0 14.9 1.9 0.0 V 0.0 1.2 34.0 -32.8

RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007

09/28/11

PSA 2011.09.01

EMI 2009.8.29

RADIATED EMISSIONS

1

Pass

38

No deviations.

None

None

-10

0

10

20

30

40

50

60

70

80

10 100 1000

MHz

dBuV/m
Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

129.142 26.9 41.6 1.0 13.9 2.0 0.0 V 0.0 1.2 34.0 -32.8

114.150 26.1 41.6 1.0 14.7 1.9 0.0 H 0.0 1.1 34.0 -32.9

112.514 26.2 41.6 1.0 14.6 1.9 0.0 V 0.0 1.0 34.0 -33.0

112.892 26.1 41.6 1.0 14.6 1.9 0.0 V 0.0 1.0 34.0 -33.0

114.662 25.9 41.6 1.0 14.8 1.9 0.0 V 0.0 1.0 34.0 -33.0

114.626 25.9 41.6 1.0 14.8 1.9 0.0 V 0.0 1.0 34.0 -33.0
scan

EUT: Ambient Work Order: INOG0018

Serial Number: None Date:

Customer: Inogen Temperature: 23C

Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb

Tested by: Jeremiah Darden Power: 110VAC/60Hz Job Site: OC11

TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS

Antenna Height(s) (m) 1 Test Distance (m) 1

COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDARD

Run #

Configuration #

Results Signature

Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

974.289 39.4 42.1 1.0 24.6 6.2 0.0 H 0.0 28.0 60.7 -32.6

982.566 38.2 42.0 1.0 24.7 6.2 0.0 V 0.0 27.0 60.7 -33.7

991.869 38.0 42.0 1.0 24.8 6.2 0.0 H 0.0 27.0 60.8 -33.8

968.649 38.3 42.1 1.0 24.4 6.1 0.0 H 0.0 26.7 60.6 -34.0

981.248 37.9 42.0 1.0 24.7 6.2 0.0 V 0.0 26.7 60.7 -34.0

984.471 37.8 42.0 1.0 24.7 6.2 0.0 H 0.0 26.7 60.8 -34.1

986.522 37.7 42.0 1.0 24.7 6.2 0.0 V 0.0 26.6 60.8 -34.2

986.009 37.7 42.0 1.0 24.7 6.2 0.0 H 0.0 26.6 60.8 -34.2

983.665 37.7 42.0 1.0 24.7 6.2 0.0 H 0.0 26.6 60.8 -34.2

990.844 37.5 42.0 1.0 24.8 6.2 0.0 H 0.0 26.5 60.8 -34.3

944.329 38.7 42.3 1.0 23.7 6.1 0.0 H 0.0 26.2 60.5 -34.3

975.241 37.7 42.1 1.0 24.6 6.2 0.0 V 0.0 26.4 60.7 -34.3

942.058 38.6 42.3 1.0 23.7 6.1 0.0 H 0.0 26.0 60.5 -34.4

RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007

09/29/11

PSA 2011.09.01

EMI 2009.8.29

RADIATED EMISSIONS

1

Pass

25

No deviations.

None

None

-10

0

10

20

30

40

50

60

70

80

100 1000

MHz

dBuV/m
Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

884.703 38.6 42.3 1.0 23.6 5.7 0.0 H 0.0 25.6 60.0 -34.4

903.968 38.9 42.4 1.0 23.3 5.9 0.0 H 0.0 25.7 60.2 -34.5

976.486 37.5 42.1 1.0 24.6 6.2 0.0 V 0.0 26.2 60.7 -34.5

963.667 37.9 42.2 1.0 24.2 6.1 0.0 H 0.0 26.1 60.6 -34.5

853.498 38.6 42.3 1.0 23.4 5.6 0.0 V 0.0 25.2 59.8 -34.6

878.257 38.3 42.3 1.0 23.7 5.7 0.0 H 0.0 25.4 60.0 -34.6
scan

EUT: Ambient Work Order: INOG0018

Serial Number: None Date:

Customer: Inogen Temperature: 23C

Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb

Tested by: Jeremiah Darden Power: 110VAC/60Hz Job Site: OC11

TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS

Antenna Height(s) (m) 1 Test Distance (m) 1

COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDARD

Run #

Configuration #

Results Signature

Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

1079.355 41.6 37.6 1.0 24.0 5.0 0.0 V 0.0 32.9 50.0 -17.1

1094.616 41.3 37.6 1.0 24.1 5.0 0.0 H 0.0 32.7 50.0 -17.3

1083.018 41.4 37.6 1.0 24.0 5.0 0.0 H 0.0 32.7 50.0 -17.3

1033.573 41.5 37.7 1.0 23.6 4.9 0.0 V 0.0 32.3 50.0 -17.7

1092.174 40.5 37.6 1.0 24.0 5.0 0.0 V 0.0 31.9 50.0 -18.1

1045.782 40.9 37.7 1.0 23.7 4.9 0.0 H 0.0 31.9 50.0 -18.1

1054.938 40.7 37.7 1.0 23.8 5.0 0.0 H 0.0 31.7 50.0 -18.3

1058.601 40.5 37.7 1.0 23.8 5.0 0.0 H 0.0 31.6 50.0 -18.4

1039.678 40.6 37.7 1.0 23.7 4.9 0.0 V 0.0 31.5 50.0 -18.5

1072.641 40.2 37.7 1.0 23.9 5.0 0.0 V 0.0 31.4 50.0 -18.6

1061.653 40.3 37.7 1.0 23.8 5.0 0.0 H 0.0 31.4 50.0 -18.6

1071.420 40.2 37.7 1.0 23.9 5.0 0.0 H 0.0 31.4 50.0 -18.6

1028.080 40.6 37.7 1.0 23.6 4.9 0.0 H 0.0 31.4 50.0 -18.6

RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007

09/29/11

PSA 2011.09.01

EMI 2009.8.29

RADIATED EMISSIONS

1

Pass

23

No deviations.

None

None

0

10

20

30

40

50

60

70

80

1000 10000

MHz

dBuV/m
Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

1034.794 40.4 37.7 1.0 23.6 4.9 0.0 H 0.0 31.2 50.0 -18.8

1064.705 40.1 37.7 1.0 23.9 5.0 0.0 H 0.0 31.2 50.0 -18.8

1089.733 39.7 37.6 1.0 24.0 5.0 0.0 H 0.0 31.1 50.0 -18.9

1058.601 40.0 37.7 1.0 23.8 5.0 0.0 V 0.0 31.1 50.0 -18.9

1043.951 40.1 37.7 1.0 23.7 4.9 0.0 V 0.0 31.0 50.0 -19.0

1083.018 39.7 37.6 1.0 24.0 5.0 0.0 V 0.0 31.0 50.0 -19.0
scan

EUT: Inogen 1G-3 Work Order: INOG0018

Serial Number: Prototype 1 Date:

Customer: Inogen Temperature: 23C

Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb

Tested by: Jeremiah Darden Power: Battery Job Site: OC11

TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS

Antenna Height(s) (m) 1 Test Distance (m) 1

COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDARD

Run #

Configuration #

Results Signature

Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

6.778 22.0 0.0 1.0 0.2 0.4 0.0 V 0.0 22.6 37.6 -15.0

7.162 21.6 0.0 1.0 0.2 0.4 0.0 V 0.0 22.2 37.5 -15.3

7.042 20.4 0.0 1.0 0.2 0.4 0.0 V 0.0 21.0 37.5 -16.5

6.392 20.1 0.0 1.0 0.2 0.4 0.0 V 0.0 20.7 37.7 -17.0

7.233 19.8 0.0 1.0 0.2 0.4 0.0 V 0.0 20.4 37.5 -17.1

6.849 19.5 0.0 1.0 0.2 0.4 0.0 V 0.0 20.1 37.6 -17.5

7.210 19.2 0.0 1.0 0.2 0.4 0.0 V 0.0 19.8 37.5 -17.7

6.968 19.1 0.0 1.0 0.2 0.4 0.0 V 0.0 19.7 37.5 -17.9

2.787 20.9 0.0 1.0 0.3 0.2 0.0 V 0.0 21.4 39.3 -17.9

7.065 19.0 0.0 1.0 0.2 0.4 0.0 V 0.0 19.6 37.5 -17.9

6.991 19.0 0.0 1.0 0.2 0.4 0.0 V 0.0 19.6 37.5 -18.0

6.825 19.0 0.0 1.0 0.2 0.4 0.0 V 0.0 19.6 37.6 -18.0

6.585 19.0 0.0 1.0 0.2 0.4 0.0 V 0.0 19.6 37.6 -18.1

RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007

09/29/11

PSA 2011.09.01

EMI 2009.8.29

RADIATED EMISSIONS

2

Pass

6

No deviations.

Front, Internal Battery

Autopulse 17 BPM Setting 4

0

10

20

30

40

50

60

70

80

1 10 100

MHz

dBuV/m
Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

6.705 18.7 0.0 1.0 0.2 0.4 0.0 V 0.0 19.3 37.6 -18.3

6.344 18.8 0.0 1.0 0.2 0.4 0.0 V 0.0 19.4 37.7 -18.3

7.017 18.6 0.0 1.0 0.2 0.4 0.0 V 0.0 19.2 37.5 -18.4

6.273 18.8 0.0 1.0 0.2 0.4 0.0 V 0.0 19.4 37.7 -18.4

5.959 18.9 0.0 1.0 0.2 0.4 0.0 V 0.0 19.5 37.8 -18.4

6.463 18.7 0.0 1.0 0.2 0.4 0.0 V 0.0 19.3 37.7 -18.4
scan

EUT: Inogen 1G-3 Work Order: INOG0018

Serial Number: Prototype 1 Date:

Customer: Inogen Temperature: 23C

Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb

Tested by: Jeremiah Darden Power: Battery Job Site: OC11

TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS

Antenna Height(s) (m) 1 Test Distance (m) 1

COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDARD

Run #

Configuration #

Results Signature

Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

2.050 22.5 0.0 1.0 0.3 0.2 0.0 V 0.0 23.0 40.0 -17.0

2.214 21.9 0.0 1.0 0.3 0.2 0.0 V 0.0 22.4 39.8 -17.4

2.132 21.3 0.0 1.0 0.3 0.2 0.0 V 0.0 21.8 39.9 -18.1

2.296 21.0 0.0 1.0 0.3 0.2 0.0 V 0.0 21.5 39.7 -18.2

2.706 20.3 0.0 1.0 0.3 0.2 0.0 V 0.0 20.8 39.4 -18.6

2.542 20.4 0.0 1.0 0.3 0.2 0.0 V 0.0 20.9 39.5 -18.6

2.788 20.1 0.0 1.0 0.3 0.2 0.0 V 0.0 20.6 39.3 -18.7

2.378 20.3 0.0 1.0 0.3 0.2 0.0 V 0.0 20.8 39.7 -18.9

2.460 20.0 0.0 1.0 0.3 0.2 0.0 V 0.0 20.5 39.6 -19.1

4.044 18.7 0.0 1.0 0.3 0.3 0.0 V 0.0 19.3 38.6 -19.3

2.055 19.8 0.0 1.0 0.3 0.2 0.0 V 0.0 20.3 39.9 -19.6

2.141 19.2 0.0 1.0 0.3 0.2 0.0 V 0.0 19.7 39.9 -20.2

4.210 17.7 0.0 1.0 0.3 0.3 0.0 V 0.0 18.3 38.5 -20.2

RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007

09/29/11

PSA 2011.09.01

EMI 2009.8.29

RADIATED EMISSIONS

2

Pass

7

No deviations.

Right, Internal Battery

Autopulse 17 BPM Setting 4

0

10

20

30

40

50

60

70

80

1 10 100

MHz

dBuV/m
Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

3.881 17.7 0.0 1.0 0.3 0.3 0.0 V 0.0 18.3 38.7 -20.4

2.624 18.4 0.0 1.0 0.3 0.2 0.0 V 0.0 18.9 39.5 -20.6

3.550 17.7 0.0 1.0 0.3 0.2 0.0 V 0.0 18.2 38.9 -20.6

4.183 17.3 0.0 1.0 0.3 0.3 0.0 V 0.0 17.9 38.5 -20.7

3.939 17.4 0.0 1.0 0.3 0.3 0.0 V 0.0 18.0 38.7 -20.7

3.197 17.9 0.0 1.0 0.3 0.2 0.0 V 0.0 18.4 39.1 -20.7
scan

EUT: Inogen 1G-3 Work Order: INOG0018

Serial Number: Prototype 1 Date:

Customer: Inogen Temperature: 23C

Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb

Tested by: Jeremiah Darden Power: Battery Job Site: OC11

TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS

Antenna Height(s) (m) 1 Test Distance (m) 1

COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDARD

Run #

Configuration #

Results Signature

Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

7.402 21.4 0.0 1.0 0.2 0.4 0.0 V 0.0 22.0 37.4 -15.5

7.331 19.5 0.0 1.0 0.2 0.4 0.0 V 0.0 20.1 37.4 -17.4

7.138 19.5 0.0 1.0 0.2 0.4 0.0 V 0.0 20.1 37.5 -17.4

7.089 18.8 0.0 1.0 0.2 0.4 0.0 V 0.0 19.4 37.5 -18.1

7.450 18.6 0.0 1.0 0.2 0.4 0.0 V 0.0 19.2 37.4 -18.2

7.282 18.3 0.0 1.0 0.2 0.4 0.0 V 0.0 18.9 37.4 -18.6

7.258 18.3 0.0 1.0 0.2 0.4 0.0 V 0.0 18.9 37.4 -18.6

7.475 18.1 0.0 1.0 0.2 0.4 0.0 V 0.0 18.7 37.4 -18.7

6.392 18.0 0.0 1.0 0.2 0.4 0.0 V 0.0 18.6 37.7 -19.1

7.619 17.3 0.0 1.0 0.1 0.4 0.0 V 0.0 17.8 37.4 -19.5

7.378 17.2 0.0 1.0 0.2 0.4 0.0 V 0.0 17.8 37.4 -19.7

7.522 17.0 0.0 1.0 0.1 0.4 0.0 V 0.0 17.5 37.4 -19.8

7.762 16.8 0.0 1.0 0.1 0.4 0.0 V 0.0 17.3 37.3 -20.0

RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007

09/29/11

PSA 2011.09.01

EMI 2009.8.29

RADIATED EMISSIONS

2

Pass

8

No deviations.

Back, Internal Battery

Autopulse 17 BPM Setting 4

0

10

20

30

40

50

60

70

80

1 10 100

MHz

dBuV/m
Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

6.754 17.0 0.0 1.0 0.2 0.4 0.0 V 0.0 17.6 37.6 -20.0

6.993 16.9 0.0 1.0 0.2 0.4 0.0 V 0.0 17.5 37.5 -20.1

7.231 16.7 0.0 1.0 0.2 0.4 0.0 V 0.0 17.3 37.5 -20.2

6.609 16.7 0.0 1.0 0.2 0.4 0.0 V 0.0 17.3 37.6 -20.4

7.161 16.5 0.0 1.0 0.2 0.4 0.0 V 0.0 17.1 37.5 -20.4

7.186 16.3 0.0 1.0 0.2 0.4 0.0 V 0.0 16.9 37.5 -20.6
scan

EUT: Inogen 1G-3 Work Order: INOG0018

Serial Number: Prototype 1 Date:

Customer: Inogen Temperature: 23C

Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb

Tested by: Jeremiah Darden Power: Battery Job Site: OC11

TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS

Antenna Height(s) (m) 1 Test Distance (m) 1

COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDARD

Run #

Configuration #

Results Signature

Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

2.132 16.7 0.0 1.0 0.3 0.2 0.0 V 0.0 17.2 39.9 -22.7

2.214 14.3 0.0 1.0 0.3 0.2 0.0 V 0.0 14.8 39.8 -25.0

2.460 13.9 0.0 1.0 0.3 0.2 0.0 V 0.0 14.4 39.6 -25.2

2.952 13.1 0.0 1.0 0.3 0.2 0.0 V 0.0 13.6 39.2 -25.6

2.296 13.4 0.0 1.0 0.3 0.2 0.0 V 0.0 13.9 39.7 -25.8

6.969 10.9 0.0 1.0 0.2 0.4 0.0 V 0.0 11.5 37.5 -26.1

2.378 12.7 0.0 1.0 0.3 0.2 0.0 V 0.0 13.2 39.7 -26.5

2.824 12.1 0.0 1.0 0.3 0.2 0.0 V 0.0 12.6 39.3 -26.7

3.280 11.4 0.0 1.0 0.3 0.2 0.0 V 0.0 11.9 39.0 -27.1

2.050 12.3 0.0 1.0 0.3 0.2 0.0 V 0.0 12.8 40.0 -27.2

7.956 9.4 0.0 1.0 0.1 0.4 0.0 V 0.0 9.9 37.3 -27.3

6.354 9.7 0.0 1.0 0.2 0.4 0.0 V 0.0 10.3 37.7 -27.4

3.116 11.0 0.0 1.0 0.3 0.2 0.0 V 0.0 11.5 39.1 -27.6

RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007

09/29/11

PSA 2011.09.01

EMI 2009.8.29

RADIATED EMISSIONS

2

Pass

9

No deviations.

Left, Internal Battery

Autopulse 17 BPM Setting 4

0

10

20

30

40

50

60

70

80

1 10 100

MHz

dBuV/m
Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

2.787 11.2 0.0 1.0 0.3 0.2 0.0 V 0.0 11.7 39.3 -27.6

3.854 10.4 0.0 1.0 0.3 0.3 0.0 V 0.0 11.0 38.7 -27.7

24.925 6.5 0.0 1.0 0.0 0.8 0.0 V 0.0 7.3 35.0 -27.7

3.198 10.7 0.0 1.0 0.3 0.2 0.0 V 0.0 11.2 39.1 -27.9

2.870 10.9 0.0 1.0 0.3 0.2 0.0 V 0.0 11.4 39.3 -27.9

2.706 11.0 0.0 1.0 0.3 0.2 0.0 V 0.0 11.5 39.4 -27.9
scan

EUT: Inogen 1G-3 Work Order: INOG0018

Serial Number: Prototype 1 Date:

Customer: Inogen Temperature: 23C

Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb

Tested by: Jeremiah Darden Power: Battery Job Site: OC11

TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS

Antenna Height(s) (m) 1 Test Distance (m) 1

COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDARD

Run #

Configuration #

Results Signature

Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

119.998 39.3 41.7 1.0 15.2 1.9 0.0 H 0.0 14.8 34.0 -19.2

131.999 39.6 41.5 1.0 13.5 2.0 0.0 H 0.0 13.6 34.0 -20.4

126.004 38.8 41.7 1.0 14.4 2.0 0.0 V 0.0 13.5 34.0 -20.5

131.999 39.0 41.5 1.0 13.5 2.0 0.0 V 0.0 13.0 34.0 -21.0

126.004 38.1 41.7 1.0 14.4 2.0 0.0 H 0.0 12.8 34.0 -21.2

107.997 38.8 41.7 1.0 13.8 1.9 0.0 V 0.0 12.8 34.0 -21.2

114.003 37.6 41.6 1.0 14.7 1.9 0.0 H 0.0 12.6 34.0 -21.4

119.998 36.8 41.7 1.0 15.2 1.9 0.0 V 0.0 12.3 34.0 -21.7

102.002 39.1 41.8 1.0 12.4 1.9 0.0 V 0.0 11.5 34.0 -22.5

138.005 38.1 41.4 1.0 12.8 2.0 0.0 V 0.0 11.5 34.0 -22.5

144.000 38.4 41.4 1.0 12.4 2.0 0.0 V 0.0 11.4 34.0 -22.6

116.396 36.2 41.6 1.0 14.9 1.9 0.0 V 0.0 11.4 34.0 -22.6

114.003 36.3 41.6 1.0 14.7 1.9 0.0 V 0.0 11.3 34.0 -22.7

RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007

09/29/11

PSA 2011.09.01

EMI 2009.8.29

RADIATED EMISSIONS

2

Pass

30

No deviations.

Front, Internal Battery

Autopulse 17 BPM Setting 4

0

10

20

30

40

50

60

70

80

10 100 1000

MHz

dBuV/m
Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

145.184 38.3 41.5 1.0 12.4 2.0 0.0 V 0.0 11.3 34.0 -22.7

128.556 36.4 41.6 1.0 14.0 2.0 0.0 H 0.0 10.8 34.0 -23.2

107.997 36.5 41.7 1.0 13.8 1.9 0.0 H 0.0 10.5 34.0 -23.5

131.327 36.2 41.6 1.0 13.6 2.0 0.0 H 0.0 10.3 34.0 -23.7

130.314 35.5 41.6 1.0 13.8 2.0 0.0 H 0.0 9.7 34.0 -24.3

128.458 34.9 41.6 1.0 14.0 2.0 0.0 V 0.0 9.3 34.0 -24.7
scan

EUT: Inogen 1G-3 Work Order: INOG0018

Serial Number: Prototype 1 Date:

Customer: Inogen Temperature: 23C

Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb

Tested by: Jeremiah Darden Power: Battery Job Site: OC11

TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS

Antenna Height(s) (m) 1 Test Distance (m) 1

COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDARD

Run #

Configuration #

Results Signature

Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

131.999 40.7 41.5 1.0 13.5 2.0 0.0 V 0.0 14.7 34.0 -19.3

138.030 40.5 41.4 1.0 12.8 2.0 0.0 V 0.0 13.9 34.0 -20.1

126.004 39.0 41.7 1.0 14.4 2.0 0.0 V 0.0 13.7 34.0 -20.3

107.997 38.5 41.7 1.0 13.8 1.9 0.0 V 0.0 12.5 34.0 -21.5

114.003 37.4 41.6 1.0 14.7 1.9 0.0 V 0.0 12.4 34.0 -21.6

119.998 36.8 41.7 1.0 15.2 1.9 0.0 V 0.0 12.3 34.0 -21.7

138.005 38.8 41.4 1.0 12.8 2.0 0.0 V 0.0 12.2 34.0 -21.8

102.002 39.7 41.8 1.0 12.4 1.9 0.0 V 0.0 12.1 34.0 -21.9

109.193 36.5 41.7 1.0 14.1 1.9 0.0 V 0.0 10.8 34.0 -23.2

140.349 36.8 41.4 1.0 12.6 2.0 0.0 V 0.0 10.0 34.0 -24.0

144.000 36.8 41.4 1.0 12.4 2.0 0.0 V 0.0 9.8 34.0 -24.2

123.624 34.4 41.7 1.0 14.8 2.0 0.0 V 0.0 9.4 34.0 -24.6

147.699 35.3 41.5 1.0 12.5 2.0 0.0 V 0.0 8.4 34.0 -25.6

RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007

09/29/11

PSA 2011.09.01

EMI 2009.8.29

RADIATED EMISSIONS

2

Pass

31

No deviations.

Right, Internal Battery

Autopulse 17 BPM Setting 4

0

10

20

30

40

50

60

70

80

10 100 1000

MHz

dBuV/m
Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

116.274 32.8 41.6 1.0 14.9 1.9 0.0 V 0.0 8.0 34.0 -26.0

192.003 44.7 41.3 1.0 17.8 2.4 0.0 V 0.0 23.6 49.7 -26.0

116.506 32.3 41.6 1.0 14.9 1.9 0.0 V 0.0 7.5 34.0 -26.5

109.938 32.7 41.7 1.0 14.3 1.9 0.0 V 0.0 7.2 34.0 -26.8

117.666 31.9 41.6 1.0 15.0 1.9 0.0 V 0.0 7.2 34.0 -26.8

124.710 32.0 41.7 1.0 14.6 2.0 0.0 V 0.0 6.9 34.0 -27.1
scan

EUT: Inogen 1G-3 Work Order: INOG0018

Serial Number: Prototype 1 Date:

Customer: Inogen Temperature: 23C

Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb

Tested by: Jeremiah Darden Power: Battery Job Site: OC11

TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS

Antenna Height(s) (m) 1 Test Distance (m) 1

COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDARD

Run #

Configuration #

Results Signature

Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

126.004 40.0 41.7 1.0 14.4 2.0 0.0 V 0.0 14.7 34.0 -19.3

149.982 41.2 41.5 1.0 12.6 2.1 0.0 V 0.0 14.4 34.0 -19.6

131.999 40.3 41.5 1.0 13.5 2.0 0.0 V 0.0 14.3 34.0 -19.7

119.998 37.8 41.7 1.0 15.2 1.9 0.0 V 0.0 13.3 34.0 -20.7

114.003 37.9 41.6 1.0 14.7 1.9 0.0 V 0.0 12.9 34.0 -21.1

140.422 39.6 41.4 1.0 12.6 2.0 0.0 V 0.0 12.8 34.0 -21.2

138.005 39.0 41.4 1.0 12.8 2.0 0.0 V 0.0 12.4 34.0 -21.6

101.087 40.2 41.8 1.0 12.2 1.9 0.0 V 0.0 12.4 34.0 -21.6

101.160 39.9 41.8 1.0 12.2 1.9 0.0 V 0.0 12.1 34.0 -21.9

107.997 38.1 41.7 1.0 13.8 1.9 0.0 V 0.0 12.1 34.0 -21.9

102.002 39.1 41.8 1.0 12.4 1.9 0.0 V 0.0 11.5 34.0 -22.5

140.374 37.9 41.4 1.0 12.6 2.0 0.0 V 0.0 11.1 34.0 -22.9

144.000 37.8 41.4 1.0 12.4 2.0 0.0 H 0.0 10.8 34.0 -23.2

RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007

09/29/11

PSA 2011.09.01

EMI 2009.8.29

RADIATED EMISSIONS

2

Pass

32

No deviations.

Back, Internal Battery

Autopulse 17 BPM Setting 4

0

10

20

30

40

50

60

70

80

10 100 1000

MHz

dBuV/m
Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

131.999 36.5 41.5 1.0 13.5 2.0 0.0 H 0.0 10.5 34.0 -23.5

144.000 37.2 41.4 1.0 12.4 2.0 0.0 V 0.0 10.2 34.0 -23.8

126.004 35.0 41.7 1.0 14.4 2.0 0.0 H 0.0 9.7 34.0 -24.3

119.680 34.0 41.7 1.0 15.2 1.9 0.0 H 0.0 9.5 34.0 -24.5

138.005 35.9 41.4 1.0 12.8 2.0 0.0 H 0.0 9.3 34.0 -24.7

124.100 34.1 41.7 1.0 14.7 2.0 0.0 H 0.0 9.1 34.0 -24.9
scan

EUT: Inogen 1G-3 Work Order: INOG0018

Serial Number: Prototype 1 Date:

Customer: Inogen Temperature: 23C

Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb

Tested by: Jeremiah Darden Power: Battery Job Site: OC11

TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS

Antenna Height(s) (m) 1 Test Distance (m) 1

COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDARD

Run #

Configuration #

Results Signature

Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

131.999 40.1 41.5 1.0 13.5 2.0 0.0 V 0.0 14.1 34.0 -19.9

126.004 39.3 41.7 1.0 14.4 2.0 0.0 V 0.0 14.0 34.0 -20.0

145.233 40.9 41.5 1.0 12.4 2.0 0.0 V 0.0 13.9 34.0 -20.1

137.993 39.5 41.4 1.0 12.8 2.0 0.0 V 0.0 12.9 34.0 -21.1

140.422 39.6 41.4 1.0 12.6 2.0 0.0 V 0.0 12.8 34.0 -21.2

107.997 38.4 41.7 1.0 13.8 1.9 0.0 V 0.0 12.4 34.0 -21.6

119.998 36.9 41.7 1.0 15.2 1.9 0.0 V 0.0 12.4 34.0 -21.6

102.002 39.5 41.8 1.0 12.4 1.9 0.0 V 0.0 11.9 34.0 -22.1

109.181 37.3 41.7 1.0 14.1 1.9 0.0 V 0.0 11.6 34.0 -22.4

114.003 36.4 41.6 1.0 14.7 1.9 0.0 V 0.0 11.4 34.0 -22.6

130.778 35.5 41.6 1.0 13.7 2.0 0.0 V 0.0 9.6 34.0 -24.4

126.627 33.6 41.7 1.0 14.3 2.0 0.0 V 0.0 8.3 34.0 -25.7

149.725 35.0 41.5 1.0 12.6 2.1 0.0 V 0.0 8.2 34.0 -25.8

RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007

09/29/11

PSA 2011.09.01

EMI 2009.8.29

RADIATED EMISSIONS

2

Pass

33

No deviations.

Left, Internal Battery

Autopulse 17 BPM Setting 4

0

10

20

30

40

50

60

70

80

10 100 1000

MHz

dBuV/m
Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

144.000 34.9 41.4 1.0 12.4 2.0 0.0 V 0.0 7.9 34.0 -26.1

69.935 48.1 42.1 1.0 8.7 1.5 0.0 H 0.0 16.2 42.4 -26.2

101.099 34.8 41.8 1.0 12.2 1.9 0.0 H 0.0 7.0 34.0 -27.0

148.736 33.8 41.5 1.0 12.5 2.1 0.0 V 0.0 6.9 34.0 -27.1

101.050 34.7 41.8 1.0 12.2 1.9 0.0 H 0.0 6.9 34.0 -27.1

132.243 32.9 41.5 1.0 13.5 2.0 0.0 V 0.0 6.9 34.0 -27.1
scan

EUT: Inogen 1G-3 Work Order: INOG0018

Serial Number: Prototype 1 Date:

Customer: Inogen Temperature: 23C

Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb

Tested by: Jeremiah Darden Power: Battery Job Site: OC11

TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS

Antenna Height(s) (m) 1 Test Distance (m) 1

COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDARD

Run #

Configuration #

Results Signature

Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

287.999 53.1 41.3 1.0 16.3 2.9 0.0 V 0.0 31.0 52.4 -21.4

243.755 53.8 40.9 1.0 14.3 2.7 0.0 V 0.0 29.8 51.3 -21.4

246.368 53.6 40.9 1.0 14.4 2.7 0.0 V 0.0 29.8 51.4 -21.6

239.995 53.5 41.0 1.0 14.1 2.6 0.0 V 0.0 29.3 51.2 -21.9

248.175 53.2 40.9 1.0 14.5 2.7 0.0 V 0.0 29.5 51.4 -22.0

246.905 53.2 40.9 1.0 14.5 2.7 0.0 V 0.0 29.4 51.4 -22.0

246.002 53.2 40.9 1.0 14.4 2.7 0.0 V 0.0 29.4 51.4 -22.0

232.450 53.7 41.1 1.0 13.8 2.6 0.0 V 0.0 29.0 51.0 -22.0

249.127 53.1 41.0 1.0 14.6 2.7 0.0 V 0.0 29.4 51.4 -22.0

245.513 53.1 40.9 1.0 14.4 2.7 0.0 V 0.0 29.2 51.3 -22.1

245.074 53.1 40.9 1.0 14.4 2.7 0.0 V 0.0 29.2 51.3 -22.1

246.612 53.0 40.9 1.0 14.4 2.7 0.0 V 0.0 29.2 51.4 -22.2

248.297 52.9 41.0 1.0 14.5 2.7 0.0 V 0.0 29.2 51.4 -22.3

RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007

09/29/11

PSA 2011.09.01

EMI 2009.8.29

RADIATED EMISSIONS

2

Pass

26

No deviations.

Front, Internal Battery

Autopulse 17 BPM Setting 4

0

10

20

30

40

50

60

70

80

100 1000

MHz

dBuV/m
Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

247.296 52.8 40.9 1.0 14.5 2.7 0.0 V 0.0 29.0 51.4 -22.4

244.537 52.7 40.9 1.0 14.3 2.7 0.0 V 0.0 28.8 51.3 -22.5

242.950 52.6 40.9 1.0 14.3 2.6 0.0 V 0.0 28.6 51.3 -22.7

251.032 52.4 41.0 1.0 14.6 2.7 0.0 V 0.0 28.7 51.5 -22.7

243.584 52.5 40.9 1.0 14.3 2.6 0.0 V 0.0 28.5 51.3 -22.8

250.031 52.2 41.0 1.0 14.6 2.7 0.0 V 0.0 28.5 51.5 -22.9
scan

EUT: Inogen 1G-3 Work Order: INOG0018

Serial Number: Prototype 1 Date:

Customer: Inogen Temperature: 23C

Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb

Tested by: Jeremiah Darden Power: Battery Job Site: OC11

TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS

Antenna Height(s) (m) 1 Test Distance (m) 1

COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDARD

Run #

Configuration #

Results Signature

Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

778.049 55.4 42.1 1.0 21.8 5.2 0.0 V 0.0 40.3 59.2 -18.9

287.999 55.3 41.3 1.0 16.3 2.9 0.0 V 0.0 33.2 52.4 -19.2

246.978 54.8 40.9 1.0 14.5 2.7 0.0 V 0.0 31.0 51.4 -20.4

246.905 54.6 40.9 1.0 14.5 2.7 0.0 V 0.0 30.8 51.4 -20.6

245.489 54.5 40.9 1.0 14.4 2.7 0.0 V 0.0 30.6 51.3 -20.7

252.008 54.4 41.0 1.0 14.6 2.7 0.0 V 0.0 30.8 51.5 -20.8

239.995 54.6 41.0 1.0 14.1 2.6 0.0 V 0.0 30.4 51.2 -20.8

246.417 54.2 40.9 1.0 14.4 2.7 0.0 V 0.0 30.4 51.4 -21.0

245.758 54.0 40.9 1.0 14.4 2.7 0.0 V 0.0 30.1 51.3 -21.2

248.321 53.8 41.0 1.0 14.5 2.7 0.0 V 0.0 30.1 51.4 -21.4

245.294 53.6 40.9 1.0 14.4 2.7 0.0 V 0.0 29.7 51.3 -21.6

241.802 53.7 40.9 1.0 14.2 2.6 0.0 V 0.0 29.6 51.2 -21.6

247.394 53.5 40.9 1.0 14.5 2.7 0.0 V 0.0 29.7 51.4 -21.7

RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007

09/29/11

PSA 2011.09.01

EMI 2009.8.29

RADIATED EMISSIONS

2

Pass

27

No deviations.

Right, Internal Battery

Autopulse 17 BPM Setting 4

0

10

20

30

40

50

60

70

80

100 1000

MHz

dBuV/m
Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

246.685 53.5 40.9 1.0 14.4 2.7 0.0 V 0.0 29.7 51.4 -21.7

246.051 53.5 40.9 1.0 14.4 2.7 0.0 V 0.0 29.7 51.4 -21.7

248.810 53.4 41.0 1.0 14.5 2.7 0.0 V 0.0 29.7 51.4 -21.7

245.074 53.3 40.9 1.0 14.4 2.7 0.0 V 0.0 29.4 51.3 -21.9

244.073 53.3 40.9 1.0 14.3 2.7 0.0 V 0.0 29.4 51.3 -21.9

239.800 53.4 41.0 1.0 14.1 2.6 0.0 V 0.0 29.2 51.2 -22.0
scan

EUT: Inogen 1G-3 Work Order: INOG0018

Serial Number: Prototype 1 Date:

Customer: Inogen Temperature: 23C

Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb

Tested by: Jeremiah Darden Power: Battery Job Site: OC11

TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS

Antenna Height(s) (m) 1 Test Distance (m) 1

COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDARD

Run #

Configuration #

Results Signature

Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

287.999 56.0 41.3 1.0 16.3 2.9 0.0 V 0.0 33.9 52.4 -18.5

246.441 55.9 40.9 1.0 14.4 2.7 0.0 V 0.0 32.1 51.4 -19.3

245.343 55.7 40.9 1.0 14.4 2.7 0.0 V 0.0 31.8 51.3 -19.5

247.125 55.1 40.9 1.0 14.5 2.7 0.0 V 0.0 31.3 51.4 -20.1

246.002 55.1 40.9 1.0 14.4 2.7 0.0 V 0.0 31.3 51.4 -20.1

244.610 55.1 40.9 1.0 14.3 2.7 0.0 V 0.0 31.2 51.3 -20.1

245.660 54.9 40.9 1.0 14.4 2.7 0.0 V 0.0 31.0 51.3 -20.3

242.535 54.9 40.9 1.0 14.2 2.6 0.0 V 0.0 30.9 51.3 -20.4

246.197 54.7 40.9 1.0 14.4 2.7 0.0 V 0.0 30.9 51.4 -20.5

249.005 54.5 41.0 1.0 14.6 2.7 0.0 V 0.0 30.8 51.4 -20.6

243.755 54.6 40.9 1.0 14.3 2.7 0.0 V 0.0 30.6 51.3 -20.6

248.077 54.5 40.9 1.0 14.5 2.7 0.0 V 0.0 30.7 51.4 -20.7

246.905 54.5 40.9 1.0 14.5 2.7 0.0 V 0.0 30.7 51.4 -20.7

RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007

09/29/11

PSA 2011.09.01

EMI 2009.8.29

RADIATED EMISSIONS

2

Pass

28

No deviations.

Back, Internal Battery

Autopulse 17 BPM Setting 4

0

10

20

30

40

50

60

70

80

100 1000

MHz

dBuV/m
Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

244.756 54.5 40.9 1.0 14.3 2.7 0.0 V 0.0 30.6 51.3 -20.7

240.703 54.4 40.9 1.0 14.2 2.6 0.0 V 0.0 30.2 51.2 -21.0

241.558 54.3 40.9 1.0 14.2 2.6 0.0 V 0.0 30.2 51.2 -21.0

248.273 54.1 40.9 1.0 14.5 2.7 0.0 V 0.0 30.4 51.4 -21.1

242.290 54.2 40.9 1.0 14.2 2.6 0.0 V 0.0 30.1 51.3 -21.1

244.439 54.1 40.9 1.0 14.3 2.7 0.0 V 0.0 30.2 51.3 -21.1
scan

EUT: Inogen 1G-3 Work Order: INOG0018

Serial Number: Prototype 1 Date:

Customer: Inogen Temperature: 23C

Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb

Tested by: Jeremiah Darden Power: Battery Job Site: OC11

TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS

Antenna Height(s) (m) 1 Test Distance (m) 1

COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDARD

Run #

Configuration #

Results Signature

Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

246.685 56.7 40.9 1.0 14.4 2.7 0.0 V 0.0 32.9 51.4 -18.5

232.792 56.6 41.1 1.0 13.8 2.6 0.0 V 0.0 31.9 51.0 -19.1

244.879 55.8 40.9 1.0 14.4 2.7 0.0 V 0.0 31.9 51.3 -19.4

247.955 55.5 40.9 1.0 14.5 2.7 0.0 V 0.0 31.7 51.4 -19.7

247.638 55.5 40.9 1.0 14.5 2.7 0.0 V 0.0 31.7 51.4 -19.7

287.999 54.8 41.3 1.0 16.3 2.9 0.0 V 0.0 32.7 52.4 -19.7

240.313 55.6 41.0 1.0 14.1 2.6 0.0 V 0.0 31.4 51.2 -19.8

245.806 55.3 40.9 1.0 14.4 2.7 0.0 V 0.0 31.4 51.3 -19.9

231.571 55.8 41.1 1.0 13.7 2.5 0.0 V 0.0 31.0 50.9 -20.0

243.511 55.1 40.9 1.0 14.3 2.6 0.0 V 0.0 31.1 51.3 -20.2

247.003 55.0 40.9 1.0 14.5 2.7 0.0 V 0.0 31.2 51.4 -20.2

244.122 55.0 40.9 1.0 14.3 2.7 0.0 V 0.0 31.1 51.3 -20.2

248.541 54.9 41.0 1.0 14.5 2.7 0.0 V 0.0 31.2 51.4 -20.3

RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007

09/29/11

PSA 2011.09.01

EMI 2009.8.29

RADIATED EMISSIONS

2

Pass

29

No deviations.

Left, Internal Battery

Autopulse 17 BPM Setting 4

0

10

20

30

40

50

60

70

80

100 1000

MHz

dBuV/m
Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

243.340 55.0 40.9 1.0 14.3 2.6 0.0 V 0.0 31.0 51.3 -20.3

240.093 55.1 41.0 1.0 14.1 2.6 0.0 V 0.0 30.9 51.2 -20.3

242.144 55.0 40.9 1.0 14.2 2.6 0.0 V 0.0 30.9 51.2 -20.3

241.924 55.0 40.9 1.0 14.2 2.6 0.0 V 0.0 30.9 51.2 -20.3

244.415 54.8 40.9 1.0 14.3 2.7 0.0 V 0.0 30.9 51.3 -20.4

245.709 54.7 40.9 1.0 14.4 2.7 0.0 V 0.0 30.8 51.3 -20.5
scan

EUT: Inogen 1G-3 Work Order: INOG0018

Serial Number: Prototype 1 Date:

Customer: Inogen Temperature: 23C

Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb

Tested by: Jeremiah Darden Power: Battery Job Site: OC11

TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS

Antenna Height(s) (m) 1 Test Distance (m) 1

COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDARD

Run #

Configuration #

Results Signature

Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

1051.886 42.6 37.7 1.0 23.8 5.0 0.0 H 0.0 33.6 50.0 -16.4

1093.395 41.8 37.6 1.0 24.1 5.0 0.0 H 0.0 33.2 50.0 -16.8

1053.107 42.1 37.7 1.0 23.8 5.0 0.0 H 0.0 33.1 50.0 -16.9

1078.745 41.6 37.6 1.0 24.0 5.0 0.0 H 0.0 32.9 50.0 -17.1

1047.003 41.6 37.7 1.0 23.7 4.9 0.0 V 0.0 32.6 50.0 -17.4

1073.251 41.0 37.7 1.0 23.9 5.0 0.0 H 0.0 32.2 50.0 -17.8

1047.003 41.2 37.7 1.0 23.7 4.9 0.0 H 0.0 32.2 50.0 -17.8

1086.070 40.7 37.6 1.0 24.0 5.0 0.0 H 0.0 32.1 50.0 -17.9

1065.926 40.9 37.7 1.0 23.9 5.0 0.0 H 0.0 32.1 50.0 -17.9

1095.837 40.6 37.6 1.0 24.1 5.0 0.0 V 0.0 32.1 50.0 -17.9

1034.794 40.9 37.7 1.0 23.6 4.9 0.0 V 0.0 31.7 50.0 -18.3

1081.797 40.4 37.6 1.0 24.0 5.0 0.0 V 0.0 31.7 50.0 -18.3

1073.251 40.4 37.7 1.0 23.9 5.0 0.0 V 0.0 31.6 50.0 -18.4

RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007

09/29/11

PSA 2011.09.01

EMI 2009.8.29

RADIATED EMISSIONS

2

Pass

19

No deviations.

Front, Internal Battery

Autopulse 17 BPM Setting 4

0

10

20

30

40

50

60

70

80

1000 10000

MHz

dBuV/m
Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

1088.512 40.2 37.6 1.0 24.0 5.0 0.0 V 0.0 31.6 50.0 -18.4

1070.809 40.3 37.7 1.0 23.9 5.0 0.0 H 0.0 31.5 50.0 -18.5

1028.690 40.6 37.7 1.0 23.6 4.9 0.0 V 0.0 31.4 50.0 -18.6

1056.770 40.3 37.7 1.0 23.8 5.0 0.0 V 0.0 31.4 50.0 -18.6

1031.132 40.5 37.7 1.0 23.6 4.9 0.0 H 0.0 31.3 50.0 -18.7

1026.248 40.5 37.7 1.0 23.6 4.9 0.0 H 0.0 31.3 50.0 -18.7
scan

EUT: Inogen 1G-3 Work Order: INOG0018

Serial Number: Prototype 1 Date:

Customer: Inogen Temperature: 23C

Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb

Tested by: Jeremiah Darden Power: Battery Job Site: OC11

TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS

Antenna Height(s) (m) 1 Test Distance (m) 1

COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDARD

Run #

Configuration #

Results Signature

Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

1034.794 43.2 37.7 1.0 23.6 4.9 0.0 V 0.0 34.0 50.0 -16.0

1054.938 42.2 37.7 1.0 23.8 5.0 0.0 V 0.0 33.2 50.0 -16.8

1072.641 41.4 37.7 1.0 23.9 5.0 0.0 H 0.0 32.6 50.0 -17.4

1056.770 41.5 37.7 1.0 23.8 5.0 0.0 V 0.0 32.6 50.0 -17.4

1028.690 41.6 37.7 1.0 23.6 4.9 0.0 V 0.0 32.4 50.0 -17.6

1078.745 41.0 37.6 1.0 24.0 5.0 0.0 V 0.0 32.3 50.0 -17.7

1065.926 41.1 37.7 1.0 23.9 5.0 0.0 H 0.0 32.3 50.0 -17.7

1021.975 41.5 37.8 1.0 23.6 4.9 0.0 H 0.0 32.2 50.0 -17.8

1089.733 40.5 37.6 1.0 24.0 5.0 0.0 H 0.0 31.9 50.0 -18.1

1046.392 40.9 37.7 1.0 23.7 4.9 0.0 V 0.0 31.9 50.0 -18.1

1020.755 41.1 37.8 1.0 23.5 4.9 0.0 V 0.0 31.8 50.0 -18.2

1024.417 41.0 37.8 1.0 23.6 4.9 0.0 V 0.0 31.7 50.0 -18.3

1079.966 40.4 37.6 1.0 24.0 5.0 0.0 H 0.0 31.7 50.0 -18.3

RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007

09/29/11

PSA 2011.09.01

EMI 2009.8.29

RADIATED EMISSIONS

2

Pass

20

No deviations.

Right, Internal Battery

Autopulse 17 BPM Setting 4

0

10

20

30

40

50

60

70

80

1000 10000

MHz

dBuV/m
Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

1048.834 40.6 37.7 1.0 23.7 4.9 0.0 V 0.0 31.6 50.0 -18.4

1051.886 40.5 37.7 1.0 23.8 5.0 0.0 H 0.0 31.5 50.0 -18.5

1074.472 40.2 37.7 1.0 23.9 5.0 0.0 H 0.0 31.4 50.0 -18.6

1063.484 40.2 37.7 1.0 23.8 5.0 0.0 V 0.0 31.3 50.0 -18.7

1092.785 39.9 37.6 1.0 24.0 5.0 0.0 V 0.0 31.3 50.0 -18.7

1094.616 39.7 37.6 1.0 24.1 5.0 0.0 H 0.0 31.1 50.0 -18.9
scan

EUT: Inogen 1G-3 Work Order: INOG0018

Serial Number: Prototype 1 Date:

Customer: Inogen Temperature: 23C

Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb

Tested by: Jeremiah Darden Power: Battery Job Site: OC11

TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS

Antenna Height(s) (m) 1 Test Distance (m) 1

COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDARD

Run #

Configuration #

Results Signature

Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

1021.975 42.8 37.8 1.0 23.6 4.9 0.0 V 0.0 33.5 50.0 -16.5

1026.859 42.4 37.7 1.0 23.6 4.9 0.0 V 0.0 33.2 50.0 -16.8

1020.755 42.3 37.8 1.0 23.5 4.9 0.0 V 0.0 33.0 50.0 -17.0

1025.027 42.2 37.7 1.0 23.6 4.9 0.0 V 0.0 33.0 50.0 -17.0

1045.782 41.9 37.7 1.0 23.7 4.9 0.0 V 0.0 32.9 50.0 -17.1

1071.420 41.4 37.7 1.0 23.9 5.0 0.0 V 0.0 32.6 50.0 -17.4

1068.368 41.4 37.7 1.0 23.9 5.0 0.0 H 0.0 32.6 50.0 -17.4

1094.006 40.9 37.6 1.0 24.1 5.0 0.0 V 0.0 32.3 50.0 -17.7

1036.626 41.4 37.7 1.0 23.7 4.9 0.0 V 0.0 32.3 50.0 -17.7

1031.132 41.4 37.7 1.0 23.6 4.9 0.0 H 0.0 32.2 50.0 -17.8

1095.226 40.5 37.6 1.0 24.1 5.0 0.0 H 0.0 32.0 50.0 -18.0

1041.509 40.9 37.7 1.0 23.7 4.9 0.0 H 0.0 31.8 50.0 -18.2

1031.132 41.0 37.7 1.0 23.6 4.9 0.0 V 0.0 31.8 50.0 -18.2

RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007

09/29/11

PSA 2011.09.01

EMI 2009.8.29

RADIATED EMISSIONS

2

Pass

21

No deviations.

Back, Internal Battery

Autopulse 17 BPM Setting 4

0

10

20

30

40

50

60

70

80

1000 10000

MHz

dBuV/m
Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

1034.794 40.9 37.7 1.0 23.6 4.9 0.0 H 0.0 31.7 50.0 -18.3

1073.862 40.5 37.7 1.0 23.9 5.0 0.0 V 0.0 31.7 50.0 -18.3

1083.018 40.4 37.6 1.0 24.0 5.0 0.0 V 0.0 31.7 50.0 -18.3

1055.549 40.6 37.7 1.0 23.8 5.0 0.0 V 0.0 31.7 50.0 -18.3

1063.484 40.5 37.7 1.0 23.8 5.0 0.0 V 0.0 31.6 50.0 -18.4

1059.211 40.5 37.7 1.0 23.8 5.0 0.0 V 0.0 31.6 50.0 -18.4
scan

EUT: Inogen 1G-3 Work Order: INOG0018

Serial Number: Prototype 1 Date:

Customer: Inogen Temperature: 23C

Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb

Tested by: Jeremiah Darden Power: Battery Job Site: OC11

TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS

Antenna Height(s) (m) 1 Test Distance (m) 1

COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDARD

Run #

Configuration #

Results Signature

Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

1072.641 45.4 37.7 1.0 23.9 5.0 0.0 H 0.0 36.6 50.0 -13.4

1037.236 45.1 37.7 1.0 23.7 4.9 0.0 H 0.0 36.0 50.0 -14.0

1047.003 44.0 37.7 1.0 23.7 4.9 0.0 H 0.0 35.0 50.0 -15.0

1053.718 43.9 37.7 1.0 23.8 5.0 0.0 H 0.0 34.9 50.0 -15.1

1070.199 43.0 37.7 1.0 23.9 5.0 0.0 H 0.0 34.2 50.0 -15.8

1021.365 43.2 37.8 1.0 23.5 4.9 0.0 H 0.0 33.9 50.0 -16.1

1067.757 42.7 37.7 1.0 23.9 5.0 0.0 H 0.0 33.9 50.0 -16.1

1058.601 42.4 37.7 1.0 23.8 5.0 0.0 H 0.0 33.5 50.0 -16.5

1084.849 42.1 37.6 1.0 24.0 5.0 0.0 H 0.0 33.4 50.0 -16.6

1040.899 42.4 37.7 1.0 23.7 4.9 0.0 H 0.0 33.3 50.0 -16.7

1024.417 42.2 37.8 1.0 23.6 4.9 0.0 H 0.0 32.9 50.0 -17.1

1050.665 41.8 37.7 1.0 23.8 5.0 0.0 H 0.0 32.8 50.0 -17.2

1032.353 41.8 37.7 1.0 23.6 4.9 0.0 H 0.0 32.6 50.0 -17.4

RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007

09/29/11

PSA 2011.09.01

EMI 2009.8.29

RADIATED EMISSIONS

2

Pass

22

No deviations.

Left, Internal Battery

Autopulse 17 BPM Setting 4

0

10

20

30

40

50

60

70

80

1000 10000

MHz

dBuV/m
Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

1054.938 41.4 37.7 1.0 23.8 5.0 0.0 H 0.0 32.4 50.0 -17.6

1065.316 41.2 37.7 1.0 23.9 5.0 0.0 H 0.0 32.4 50.0 -17.6

1081.797 41.0 37.6 1.0 24.0 5.0 0.0 V 0.0 32.3 50.0 -17.7

1023.196 41.5 37.8 1.0 23.6 4.9 0.0 V 0.0 32.2 50.0 -17.8

1079.355 40.9 37.6 1.0 24.0 5.0 0.0 V 0.0 32.2 50.0 -17.8

1070.809 40.8 37.7 1.0 23.9 5.0 0.0 V 0.0 32.0 50.0 -18.0
scan

EUT: Inogen 1G-3 Work Order: INOG0018

Serial Number: Prototype 1 Date:

Customer: Inogen Temperature: 23C

Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb

Tested by: Jeremiah Darden Power: 110VAC/60Hz Job Site: OC11

TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS

Antenna Height(s) (m) 1 Test Distance (m) 1

COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDARD

Run #

Configuration #

Results Signature

Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

2.231 34.1 0.0 1.0 0.3 0.2 0.0 V 0.0 34.6 39.8 -5.2

2.292 32.4 0.0 1.0 0.3 0.2 0.0 V 0.0 32.9 39.7 -6.8

2.313 32.3 0.0 1.0 0.3 0.2 0.0 V 0.0 32.8 39.7 -6.9

2.067 32.4 0.0 1.0 0.3 0.2 0.0 V 0.0 32.9 39.9 -7.0

2.484 31.8 0.0 1.0 0.3 0.2 0.0 V 0.0 32.3 39.6 -7.3

2.103 31.8 0.0 1.0 0.3 0.2 0.0 V 0.0 32.3 39.9 -7.6

3.255 30.9 0.0 1.0 0.3 0.2 0.0 V 0.0 31.4 39.0 -7.6

3.696 30.5 0.0 1.0 0.3 0.3 0.0 V 0.0 31.1 38.8 -7.7

2.544 31.3 0.0 1.0 0.3 0.2 0.0 V 0.0 31.8 39.5 -7.7

2.181 31.4 0.0 1.0 0.3 0.2 0.0 V 0.0 31.9 39.8 -7.9

2.403 31.2 0.0 1.0 0.3 0.2 0.0 V 0.0 31.7 39.6 -7.9

2.710 30.9 0.0 1.0 0.3 0.2 0.0 V 0.0 31.4 39.4 -8.0

2.565 30.9 0.0 1.0 0.3 0.2 0.0 V 0.0 31.4 39.5 -8.1

RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007

09/29/11

PSA 2011.09.01

EMI 2009.8.29

RADIATED EMISSIONS

1

Pass

10

No deviations.

Top, DC Power Supply: Digital Power Model ADP90-119 SN ADP901195210A0010592

Autopulse 17 BPM Setting 4

0

10

20

30

40

50

60

70

80

1 10 100

MHz

dBuV/m
Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

2.927 30.4 0.0 1.0 0.3 0.2 0.0 V 0.0 30.9 39.2 -8.3

2.248 30.9 0.0 1.0 0.3 0.2 0.0 V 0.0 31.4 39.8 -8.4

2.005 31.1 0.0 1.0 0.3 0.2 0.0 V 0.0 31.6 40.0 -8.4

2.827 30.4 0.0 1.0 0.3 0.2 0.0 V 0.0 30.9 39.3 -8.4

2.656 30.5 0.0 1.0 0.3 0.2 0.0 V 0.0 31.0 39.4 -8.4

2.053 31.0 0.0 1.0 0.3 0.2 0.0 V 0.0 31.5 39.9 -8.4
scan

EUT: Inogen 1G-3 Work Order: INOG0018

Serial Number: Prototype 1 Date:

Customer: Inogen Temperature: 23C

Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb

Tested by: Jeremiah Darden Power: 110VAC/60Hz Job Site: OC11

TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS

Antenna Height(s) (m) 1 Test Distance (m) 1

COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDARD

Run #

Configuration #

Results Signature

Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

11.334 27.4 0.0 1.0 0.1 0.5 0.0 V 0.0 28.0 36.6 -8.6

2.569 30.4 0.0 1.0 0.3 0.2 0.0 V 0.0 30.9 39.5 -8.6

11.366 27.0 0.0 1.0 0.1 0.5 0.0 V 0.0 27.6 36.6 -9.0

11.331 26.9 0.0 1.0 0.1 0.5 0.0 V 0.0 27.5 36.6 -9.1

11.250 26.2 0.0 1.0 0.1 0.5 0.0 V 0.0 26.8 36.6 -9.8

11.392 26.1 0.0 1.0 0.1 0.5 0.0 V 0.0 26.7 36.6 -9.9

2.463 29.2 0.0 1.0 0.3 0.2 0.0 V 0.0 29.7 39.6 -9.9

11.344 26.0 0.0 1.0 0.1 0.5 0.0 V 0.0 26.6 36.6 -10.0

2.998 28.7 0.0 1.0 0.3 0.2 0.0 V 0.0 29.2 39.2 -10.0

11.260 25.9 0.0 1.0 0.1 0.5 0.0 V 0.0 26.5 36.6 -10.1

2.203 29.1 0.0 1.0 0.3 0.2 0.0 V 0.0 29.6 39.8 -10.2

2.544 28.8 0.0 1.0 0.3 0.2 0.0 V 0.0 29.3 39.5 -10.2

2.827 28.5 0.0 1.0 0.3 0.2 0.0 V 0.0 29.0 39.3 -10.3

RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007

09/29/11

PSA 2011.09.01

EMI 2009.8.29

RADIATED EMISSIONS

1

Pass

11

No deviations.

Bottom, DC Power Supply: Digital Power Model ADP90-119 SN ADP901195210A0010592

Autopulse 17 BPM Setting 4

0

10

20

30

40

50

60

70

80

1 10 100

MHz

dBuV/m
Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

12.316 25.4 0.0 1.0 0.1 0.5 0.0 V 0.0 26.0 36.4 -10.4

2.142 28.9 0.0 1.0 0.3 0.2 0.0 V 0.0 29.4 39.9 -10.5

2.188 28.8 0.0 1.0 0.3 0.2 0.0 V 0.0 29.3 39.8 -10.5

11.265 25.4 0.0 1.0 0.1 0.5 0.0 V 0.0 26.0 36.6 -10.6

2.116 28.7 0.0 1.0 0.3 0.2 0.0 V 0.0 29.2 39.9 -10.7

11.347 25.3 0.0 1.0 0.1 0.5 0.0 V 0.0 25.9 36.6 -10.7
scan

EUT: Inogen 1G-3 Work Order: INOG0018

Serial Number: Prototype 1 Date:

Customer: Inogen Temperature: 23C

Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb

Tested by: Jeremiah Darden Power: 110VAC/60Hz Job Site: OC11

TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS

Antenna Height(s) (m) 1 Test Distance (m) 1

COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDARD

Run #

Configuration #

Results Signature

Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

147.198 54.6 41.5 1.0 12.5 2.0 0.0 H 0.0 27.7 34.0 -6.3

147.503 54.2 41.5 1.0 12.5 2.0 0.0 H 0.0 27.3 34.0 -6.7

146.637 54.2 41.5 1.0 12.5 2.0 0.0 H 0.0 27.2 34.0 -6.8

146.270 54.2 41.5 1.0 12.5 2.0 0.0 H 0.0 27.2 34.0 -6.8

146.148 53.8 41.5 1.0 12.4 2.0 0.0 H 0.0 26.8 34.0 -7.2

145.819 53.8 41.5 1.0 12.4 2.0 0.0 H 0.0 26.8 34.0 -7.2

147.357 53.6 41.5 1.0 12.5 2.0 0.0 H 0.0 26.7 34.0 -7.3

145.098 53.6 41.5 1.0 12.4 2.0 0.0 H 0.0 26.6 34.0 -7.4

146.759 53.5 41.5 1.0 12.5 2.0 0.0 H 0.0 26.5 34.0 -7.5

145.550 53.5 41.5 1.0 12.4 2.0 0.0 H 0.0 26.5 34.0 -7.5

145.855 53.4 41.5 1.0 12.4 2.0 0.0 H 0.0 26.4 34.0 -7.6

144.525 53.4 41.4 1.0 12.4 2.0 0.0 H 0.0 26.4 34.0 -7.6

145.281 53.4 41.5 1.0 12.4 2.0 0.0 H 0.0 26.4 34.0 -7.6

RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007

09/29/11

PSA 2011.09.01

EMI 2009.8.29

RADIATED EMISSIONS

1

Pass

13

No deviations.

Top, DC Power Supply: Digital Power Model ADP90-119 SN ADP901195210A0010592

Autopulse 17 BPM Setting 4

0

10

20

30

40

50

60

70

80

10 100 1000

MHz

dBuV/m
Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

147.674 53.2 41.5 1.0 12.5 2.0 0.0 H 0.0 26.3 34.0 -7.7

149.908 53.1 41.5 1.0 12.6 2.1 0.0 V 0.0 26.3 34.0 -7.7

149.628 53.1 41.5 1.0 12.6 2.1 0.0 H 0.0 26.3 34.0 -7.7

147.039 53.2 41.5 1.0 12.5 2.0 0.0 H 0.0 26.3 34.0 -7.7

145.245 53.2 41.5 1.0 12.4 2.0 0.0 H 0.0 26.2 34.0 -7.8

144.769 53.2 41.5 1.0 12.4 2.0 0.0 H 0.0 26.2 34.0 -7.8
scan

EUT: Inogen 1G-3 Work Order: INOG0018

Serial Number: Prototype 1 Date:

Customer: Inogen Temperature: 23C

Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb

Tested by: Jeremiah Darden Power: 110VAC/60Hz Job Site: OC11

TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS

Antenna Height(s) (m) 1 Test Distance (m) 1

COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDARD

Run #

Configuration #

Results Signature

Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

149.652 58.5 41.5 1.0 12.6 2.1 0.0 H 0.0 31.7 34.0 -2.3

149.530 58.5 41.5 1.0 12.6 2.1 0.0 H 0.0 31.7 34.0 -2.3

148.041 58.3 41.5 1.0 12.5 2.1 0.0 H 0.0 31.4 34.0 -2.6

149.384 57.7 41.5 1.0 12.6 2.1 0.0 H 0.0 30.9 34.0 -3.1

148.651 57.4 41.5 1.0 12.5 2.1 0.0 H 0.0 30.5 34.0 -3.5

148.505 57.4 41.5 1.0 12.5 2.1 0.0 H 0.0 30.5 34.0 -3.5

149.957 57.3 41.5 1.0 12.6 2.1 0.0 H 0.0 30.5 34.0 -3.5

149.860 57.1 41.5 1.0 12.6 2.1 0.0 H 0.0 30.3 34.0 -3.7

149.188 57.0 41.5 1.0 12.6 2.1 0.0 H 0.0 30.1 34.0 -3.9

147.577 57.0 41.5 1.0 12.5 2.0 0.0 H 0.0 30.1 34.0 -3.9

148.334 56.8 41.5 1.0 12.5 2.1 0.0 H 0.0 29.9 34.0 -4.1

148.871 56.7 41.5 1.0 12.6 2.1 0.0 H 0.0 29.8 34.0 -4.2

148.981 56.6 41.5 1.0 12.6 2.1 0.0 H 0.0 29.7 34.0 -4.3

RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007

09/29/11

PSA 2011.09.01

EMI 2009.8.29

RADIATED EMISSIONS

1

Pass

14

No deviations.

Bottom, DC Power Supply: Digital Power Model ADP90-119 SN ADP901195210A0010592

Autopulse 17 BPM Setting 4

0

10

20

30

40

50

60

70

80

10 100 1000

MHz

dBuV/m
Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

148.370 56.6 41.5 1.0 12.5 2.1 0.0 H 0.0 29.7 34.0 -4.3

147.821 56.5 41.5 1.0 12.5 2.1 0.0 H 0.0 29.6 34.0 -4.4

149.310 56.4 41.5 1.0 12.6 2.1 0.0 H 0.0 29.6 34.0 -4.4

149.066 56.4 41.5 1.0 12.6 2.1 0.0 H 0.0 29.5 34.0 -4.5

148.712 56.4 41.5 1.0 12.5 2.1 0.0 H 0.0 29.5 34.0 -4.5

148.773 56.2 41.5 1.0 12.6 2.1 0.0 H 0.0 29.3 34.0 -4.7
scan

EUT: Inogen 1G-3 Work Order: INOG0018

Serial Number: Prototype 1 Date:

Customer: Inogen Temperature: 23C

Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb

Tested by: Jeremiah Darden Power: 110VAC/60Hz Job Site: OC11

TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS

Antenna Height(s) (m) 1 Test Distance (m) 1

COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDARD

Run #

Configuration #

Results Signature

Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

287.999 57.9 41.3 1.0 16.3 2.9 0.0 V 0.0 35.8 52.4 -16.6

277.231 54.3 41.2 1.0 15.3 2.8 0.0 V 0.0 31.2 52.2 -21.0

293.981 52.0 41.3 1.0 16.8 2.9 0.0 V 0.0 30.5 52.6 -22.1

291.637 51.5 41.3 1.0 16.6 2.9 0.0 V 0.0 29.8 52.5 -22.7

263.387 52.5 41.1 1.0 14.9 2.8 0.0 V 0.0 29.0 51.8 -22.8

259.333 52.4 41.1 1.0 14.8 2.7 0.0 V 0.0 28.8 51.7 -22.9

262.532 52.3 41.1 1.0 14.9 2.8 0.0 V 0.0 28.8 51.8 -23.0

260.774 52.1 41.1 1.0 14.8 2.7 0.0 V 0.0 28.6 51.8 -23.2

261.043 52.1 41.1 1.0 14.8 2.7 0.0 V 0.0 28.6 51.8 -23.2

261.678 52.0 41.1 1.0 14.8 2.7 0.0 V 0.0 28.5 51.8 -23.3

262.605 52.0 41.1 1.0 14.9 2.8 0.0 V 0.0 28.5 51.8 -23.3

263.924 51.9 41.1 1.0 14.9 2.8 0.0 V 0.0 28.4 51.8 -23.4

246.002 51.7 40.9 1.0 14.4 2.7 0.0 H 0.0 27.9 51.4 -23.5

RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007

09/29/11

PSA 2011.09.01

EMI 2009.8.29

RADIATED EMISSIONS

1

Pass

16

No deviations.

Top, DC Power Supply: Digital Power Model ADP90-119 SN ADP901195210A0010592

Autopulse 17 BPM Setting 4

0

10

20

30

40

50

60

70

80

100 1000

MHz

dBuV/m
Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

265.218 51.8 41.2 1.0 14.9 2.8 0.0 V 0.0 28.3 51.9 -23.6

260.115 51.7 41.1 1.0 14.8 2.7 0.0 V 0.0 28.2 51.7 -23.6

263.533 51.7 41.1 1.0 14.9 2.8 0.0 V 0.0 28.2 51.8 -23.6

257.991 51.6 41.1 1.0 14.8 2.7 0.0 V 0.0 28.0 51.7 -23.7

259.871 51.6 41.1 1.0 14.8 2.7 0.0 V 0.0 28.0 51.7 -23.7

261.995 51.6 41.1 1.0 14.8 2.8 0.0 V 0.0 28.1 51.8 -23.7
scan

EUT: Inogen 1G-3 Work Order: INOG0018

Serial Number: Prototype 1 Date:

Customer: Inogen Temperature: 23C

Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb

Tested by: Jeremiah Darden Power: 110VAC/60Hz Job Site: OC11

TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS

Antenna Height(s) (m) 1 Test Distance (m) 1

COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDARD

Run #

Configuration #

Results Signature

Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

287.999 57.9 41.3 1.0 16.3 2.9 0.0 V 0.0 35.8 52.4 -16.6

845.587 52.6 42.3 1.0 23.2 5.5 0.0 H 0.0 39.0 59.7 -20.7

289.196 53.2 41.3 1.0 16.4 2.9 0.0 V 0.0 31.2 52.5 -21.2

261.726 52.6 41.1 1.0 14.8 2.7 0.0 V 0.0 29.1 51.8 -22.7

260.457 52.5 41.1 1.0 14.8 2.7 0.0 V 0.0 29.0 51.7 -22.8

480.064 51.3 41.2 1.0 18.7 4.1 0.0 V 0.0 32.8 55.9 -23.1

261.800 52.2 41.1 1.0 14.8 2.8 0.0 V 0.0 28.7 51.8 -23.1

263.704 52.2 41.1 1.0 14.9 2.8 0.0 V 0.0 28.7 51.8 -23.1

260.945 52.1 41.1 1.0 14.8 2.7 0.0 V 0.0 28.6 51.8 -23.2

263.558 52.1 41.1 1.0 14.9 2.8 0.0 V 0.0 28.6 51.8 -23.2

262.264 52.0 41.1 1.0 14.8 2.8 0.0 V 0.0 28.5 51.8 -23.3

262.776 52.0 41.1 1.0 14.9 2.8 0.0 V 0.0 28.5 51.8 -23.3

246.881 51.8 40.9 1.0 14.5 2.7 0.0 H 0.0 28.0 51.4 -23.4

RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007

09/29/11

PSA 2011.09.01

EMI 2009.8.29

RADIATED EMISSIONS

1

Pass

15

No deviations.

Bottom, DC Power Supply: Digital Power Model ADP90-119 SN ADP901195210A0010592

Autopulse 17 BPM Setting 4

0

10

20

30

40

50

60

70

80

100 1000

MHz

dBuV/m
Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

259.968 51.8 41.1 1.0 14.8 2.7 0.0 V 0.0 28.2 51.7 -23.5

261.336 51.8 41.1 1.0 14.8 2.7 0.0 V 0.0 28.3 51.8 -23.5

262.654 51.5 41.1 1.0 14.9 2.8 0.0 V 0.0 28.0 51.8 -23.8

264.144 51.5 41.1 1.0 14.9 2.8 0.0 V 0.0 28.0 51.8 -23.8

256.184 51.3 41.0 1.0 14.7 2.7 0.0 V 0.0 27.7 51.6 -23.9

257.282 51.3 41.1 1.0 14.7 2.7 0.0 V 0.0 27.7 51.7 -23.9
scan

EUT: Inogen 1G-3 Work Order: INOG0018

Serial Number: Prototype 1 Date:

Customer: Inogen Temperature: 23C

Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb

Tested by: Jeremiah Darden Power: 110VAC/60Hz Job Site: OC11

TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS

Antenna Height(s) (m) 1 Test Distance (m) 1

COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDARD

Run #

Configuration #

Results Signature

Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

1089.733 42.2 37.6 1.0 24.0 5.0 0.0 H 0.0 33.6 50.0 -16.4

1025.027 42.0 37.7 1.0 23.6 4.9 0.0 H 0.0 32.8 50.0 -17.2

1092.785 40.9 37.6 1.0 24.0 5.0 0.0 H 0.0 32.3 50.0 -17.7

1051.886 41.2 37.7 1.0 23.8 5.0 0.0 H 0.0 32.2 50.0 -17.8

1096.447 40.6 37.6 1.0 24.1 5.0 0.0 V 0.0 32.1 50.0 -17.9

1045.782 41.1 37.7 1.0 23.7 4.9 0.0 V 0.0 32.1 50.0 -17.9

1049.444 41.0 37.7 1.0 23.7 4.9 0.0 V 0.0 32.0 50.0 -18.0

1048.834 41.0 37.7 1.0 23.7 4.9 0.0 H 0.0 32.0 50.0 -18.0

1092.785 40.3 37.6 1.0 24.0 5.0 0.0 V 0.0 31.7 50.0 -18.3

1081.797 40.4 37.6 1.0 24.0 5.0 0.0 H 0.0 31.7 50.0 -18.3

1083.628 40.3 37.6 1.0 24.0 5.0 0.0 H 0.0 31.6 50.0 -18.4

1053.107 40.6 37.7 1.0 23.8 5.0 0.0 V 0.0 31.6 50.0 -18.4

1042.119 40.7 37.7 1.0 23.7 4.9 0.0 H 0.0 31.6 50.0 -18.4

RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007

09/29/11

PSA 2011.09.01

EMI 2009.8.29

RADIATED EMISSIONS

1

Pass

17

No deviations.

Top, DC Power Supply: Digital Power Model ADP90-119 SN ADP901195210A0010592

Autopulse 17 BPM Setting 4

0

10

20

30

40

50

60

70

80

1000 10000

MHz

dBuV/m
Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

1020.755 40.8 37.8 1.0 23.5 4.9 0.0 H 0.0 31.5 50.0 -18.5

1656.818 42.1 37.9 1.0 25.2 5.0 0.0 V 0.0 34.4 53.0 -18.6

1076.914 40.1 37.6 1.0 23.9 5.0 0.0 V 0.0 31.4 50.0 -18.6

1085.460 40.0 37.6 1.0 24.0 5.0 0.0 V 0.0 31.4 50.0 -18.6

1070.199 40.1 37.7 1.0 23.9 5.0 0.0 H 0.0 31.3 50.0 -18.7

1056.159 40.2 37.7 1.0 23.8 5.0 0.0 H 0.0 31.3 50.0 -18.7
scan

EUT: Inogen 1G-3 Work Order: INOG0018

Serial Number: Prototype 1 Date:

Customer: Inogen Temperature: 23C

Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb

Tested by: Jeremiah Darden Power: 110VAC/60Hz Job Site: OC11

TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS

Antenna Height(s) (m) 1 Test Distance (m) 1

COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDARD

Run #

Configuration #

Results Signature

Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

1026.859 42.0 37.7 1.0 23.6 4.9 0.0 H 0.0 32.8 50.0 -17.2

1063.484 41.3 37.7 1.0 23.8 5.0 0.0 V 0.0 32.4 50.0 -17.6

1070.809 41.1 37.7 1.0 23.9 5.0 0.0 V 0.0 32.3 50.0 -17.7

1041.509 41.3 37.7 1.0 23.7 4.9 0.0 V 0.0 32.2 50.0 -17.8

1078.745 40.9 37.6 1.0 24.0 5.0 0.0 V 0.0 32.2 50.0 -17.8

1021.975 41.4 37.8 1.0 23.6 4.9 0.0 H 0.0 32.1 50.0 -17.9

1090.343 40.5 37.6 1.0 24.0 5.0 0.0 H 0.0 31.9 50.0 -18.1

1098.279 40.4 37.6 1.0 24.1 5.0 0.0 H 0.0 31.9 50.0 -18.1

1036.015 41.0 37.7 1.0 23.7 4.9 0.0 V 0.0 31.9 50.0 -18.1

1028.690 41.0 37.7 1.0 23.6 4.9 0.0 H 0.0 31.8 50.0 -18.2

1061.653 40.6 37.7 1.0 23.8 5.0 0.0 H 0.0 31.7 50.0 -18.3

1037.846 40.8 37.7 1.0 23.7 4.9 0.0 H 0.0 31.7 50.0 -18.3

1082.407 40.2 37.6 1.0 24.0 5.0 0.0 V 0.0 31.5 50.0 -18.5

RTCA DO-160F Category M: 2007 RTCA DO-160F: 2007

09/29/11

PSA 2011.09.01

EMI 2009.8.29

RADIATED EMISSIONS

1

Pass

18

No deviations.

Bottom, DC Power Supply: Digital Power Model ADP90-119 SN ADP901195210A0010592

Autopulse 17 BPM Setting 4

0

10

20

30

40

50

60

70

80

1000 10000

MHz

dBuV/m
Freq Amplitude Preamp

Antenna 

Height Transducer Cable

External 

Attenuation

Polarity/ 

Transducer 

Type Detector

Distance 

Adjustment Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (m) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) (dB) dBuV/m dBuV/m (dB)

1081.187 40.2 37.6 1.0 24.0 5.0 0.0 H 0.0 31.5 50.0 -18.5

1065.926 40.3 37.7 1.0 23.9 5.0 0.0 V 0.0 31.5 50.0 -18.5

1091.564 40.0 37.6 1.0 24.0 5.0 0.0 H 0.0 31.4 50.0 -18.6

1039.678 40.5 37.7 1.0 23.7 4.9 0.0 H 0.0 31.4 50.0 -18.6

1077.524 40.1 37.6 1.0 23.9 5.0 0.0 H 0.0 31.4 50.0 -18.6

1065.316 40.2 37.7 1.0 23.9 5.0 0.0 H 0.0 31.4 50.0 -18.6
photo

re (9).JPG

re (1).JPG

 RADIATED EMISSIONS

PSA 2011.09.01
photo

re (2).JPG

re (3).JPG

 RADIATED EMISSIONS

PSA 2011.09.01
photo

re (4).JPG

re (5).JPG

 RADIATED EMISSIONS

PSA 2011.09.01
photo

re (6).JPG

re (7).JPG

 RADIATED EMISSIONS

PSA 2011.09.01
photo

re (8).JPG

 RADIATED EMISSIONS

PSA 2011.09.01
description

MODES OF OPERATION

POWER SETTINGS INVESTIGATED

CONFIGURATIONS INVESTIGATED

FREQUENCY RANGE INVESTIGATED

Start Frequency 150 kHz Stop Frequency 30 MHz

SAMPLE CALCULATIONS

Conducted Emissions: Adjusted Level = Measured Level + Transducer Factor + Cable Attenuation Factor + External Attenuator

TEST EQUIPMENT

ID Interval

OCT 12

CAA 0

CAC 0

LIJ 12

LIL 12

NCE 24

AFG 12

MEASUREMENT BANDWIDTHS

MEASUREMENT UNCERTAINTY

TEST DESCRIPTION

Autopulse 17 BPM Setting 4

Description Manufacturer Model Last Cal.

Battery

110VAC/60Hz

INOG0018 - 1

OC11 Cables N/A Disturbance Power Cables 8/25/2011

10uF Feedthrough Capacitor Solar 6512-106R NCR

10uF Feedthrough Capacitor Solar 6512-106R NCR

LISN Fischer Custom FCC-LISN-DO-160 7/27/2011

LISN Fischer Custom FCC LISN DO-160 7/27/2011

Current Probe Fischer Custom F-33-4 6/3/2011

A measurement uncertainty estimation has been performed for each test per our internal quality document WP 342. The estimation 

is used to compare the measured result with its "true" or theoretically correct value. The expanded measurement uncertainty for 

radiated emissions measurements is less than +/- 4 dB, and for conducted emissions measurements is less than +/- 2.7 dB. Our 

measurement data meets or exceeds the measurement uncertainty requirements of CISPR 16-4; therefore, the test data can be 

compared directly to the specification limit to determine compliance. The calculations for measurement uncertainty are available 

upon request.

400.0 - 1000.0

1000.0 - 6000.0

100.0

1000.0

0.15 - 30.0

30.0 - 400.0

1.0

10.0

Frequency Range

(MHz)

BWI

(kHz)

PSA 2011.09.01

Testing was performed using the mode(s) of operation and configuration(s) noted within the report. The individuals and/or the 

organization requesting the test provided the modes, configurations and settings used to complete the evaluation. The actual test 

parameters are specified in the test data, this includes items such as investigated frequency range (scanned) and test levels. The 

testing methods and performance specifications, as well as the test site used for the evaluation are indicated in the test data.

CONDUCTED EMISSIONS

Spectrum Analyzer Agilent E4440A 4/28/2011

Using the mode of operation and configuration noted within this report, a conducted emissions test was performed. The frequency 

range tested, is also noted on the data sheets in this report. Conducted emissions measurements were made with the test setup 

as defined in RTCA DO160 E. A calibrated current probe was positioned within 5 cm of the EUT to measure the conducted 

disturbances while the EUT was powered. Where applicable the conducted disturbances on both the power leads and 

interconnecting cable bundles were measured.
cond

EUT: Ambient Work Order: INOG0018

Serial Number: None Date:

Customer: Inogen Temperature: 23C

Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb

Tested by: Jeremiah Darden Power: 110VAC/60Hz Job Site: OC11

TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS

Cable or Line Tested Ambient

COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDARD

Run #

Configuration #

Results Signature

Freq Amplitude Transducer Cable

External 

Attenuation Detector Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) dBuA dBuA (dB)

1.932 2.1 0.1 0.2 0.0 2.4 40.4 -38.0

2.000 0.2 0.1 0.2 0.0 0.5 40.0 -39.5

4.398 0.1 0.1 0.3 0.0 0.5 40.0 -39.5

3.222 0.1 0.1 0.2 0.0 0.4 40.0 -39.6

12.146 -0.3 0.1 0.5 0.0 0.3 40.0 -39.7

2.768 0.0 0.1 0.2 0.0 0.3 40.0 -39.7

6.373 -0.2 0.1 0.4 0.0 0.3 40.0 -39.7

2.021 -0.2 0.1 0.2 0.0 0.1 40.0 -39.9

6.289 -0.4 0.1 0.4 0.0 0.1 40.0 -39.9

6.839 -0.4 0.1 0.4 0.0 0.1 40.0 -39.9

23.706 -0.9 0.1 0.8 0.0 0.0 40.0 -40.0

2.499 -0.3 0.1 0.2 0.0 0.0 40.0 -40.0

16.643 -0.8 0.1 0.7 0.0 0.0 40.0 -40.0

3.620 -0.4 0.1 0.3 0.0 0.0 40.0 -40.0

4

No deviations.

None

None

1

Pass

RTCA DO-160F Category M: 2007 RTCA DO-160F Interconnecting Cables: 2007

09/29/11

PSA 2011.09.01

EMI 2009.8.29

CONDUCTED EMISSIONS

-10

0

10

20

30

40

50

60

70

80

0.1 1 10 100

MHz

dBuA
Freq Amplitude Transducer Cable

External 

Attenuation Detector Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) dBuA dBuA (dB)

2.719 -0.4 0.1 0.2 0.0 -0.1 40.0 -40.1

3.037 -0.4 0.1 0.2 0.0 -0.1 40.0 -40.1

3.784 -0.5 0.1 0.3 0.0 -0.1 40.0 -40.1

23.010 -1.1 0.1 0.8 0.0 -0.2 40.0 -40.2

3.900 -0.6 0.1 0.3 0.0 -0.2 40.0 -40.2
cond

EUT: Inogen 1G-3 Work Order: INOG0018

Serial Number: Prototype 1 Date:

Customer: Inogen Temperature: 23C

Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb

Tested by: Jeremiah Darden Power: 110VAC/60Hz Job Site: OC11

TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS

Cable or Line Tested DC Cable

COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDARD

Run #

Configuration #

Results Signature

Freq Amplitude Transducer Cable

External 

Attenuation Detector Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) dBuA dBuA (dB)

11.989 7.3 0.1 0.5 0.0 7.9 40.0 -32.1

9.507 6.6 0.1 0.5 0.0 7.2 40.0 -32.8

11.906 6.5 0.1 0.5 0.0 7.1 40.0 -32.9

13.382 6.1 0.1 0.6 0.0 6.8 40.0 -33.2

13.386 5.9 0.1 0.6 0.0 6.6 40.0 -33.4

11.816 4.9 0.1 0.5 0.0 5.5 40.0 -34.5

13.713 4.6 0.1 0.6 0.0 5.3 40.0 -34.7

13.056 4.5 0.1 0.6 0.0 5.2 40.0 -34.8

12.075 4.5 0.1 0.5 0.0 5.1 40.0 -34.9

11.733 4.3 0.1 0.5 0.0 4.9 40.0 -35.1

12.729 4.2 0.1 0.6 0.0 4.9 40.0 -35.1

12.483 4.2 0.1 0.6 0.0 4.9 40.0 -35.1

12.975 4.0 0.1 0.6 0.0 4.7 40.0 -35.3

11.333 4.0 0.1 0.5 0.0 4.6 40.0 -35.4

1

No deviations.

Probe position on DC In Side of PS, DC Power Supply: Digital Power Model ADP90-119 SN ADP901195210A0010592

Autopulse 17 BPM Setting 4

1

Pass

RTCA DO-160F Category M: 2007 RTCA DO-160F Interconnecting Cables: 2007

09/29/11

PSA 2011.09.01

EMI 2009.8.29

CONDUCTED EMISSIONS

-10

0

10

20

30

40

50

60

70

80

0.1 1 10 100

MHz

dBuA
Freq Amplitude Transducer Cable

External 

Attenuation Detector Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) dBuA dBuA (dB)

12.562 3.9 0.1 0.6 0.0 4.6 40.0 -35.4

11.579 3.9 0.1 0.5 0.0 4.5 40.0 -35.5

12.236 3.7 0.1 0.5 0.0 4.3 40.0 -35.7

11.646 3.7 0.1 0.5 0.0 4.3 40.0 -35.7

12.153 3.6 0.1 0.5 0.0 4.2 40.0 -35.8
cond

EUT: Inogen 1G-3 Work Order: INOG0018

Serial Number: Prototype 1 Date:

Customer: Inogen Temperature: 23C

Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb

Tested by: Jeremiah Darden Power: 110VAC/60Hz Job Site: OC11

TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS

Cable or Line Tested DC Cable

COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDARD

Run #

Configuration #

Results Signature

Freq Amplitude Transducer Cable

External 

Attenuation Detector Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) dBuA dBuA (dB)

9.761 4.2 0.1 0.5 0.0 4.8 40.0 -35.2

11.661 4.0 0.1 0.5 0.0 4.6 40.0 -35.4

12.154 3.7 0.1 0.5 0.0 4.3 40.0 -35.7

12.975 3.5 0.1 0.6 0.0 4.2 40.0 -35.8

12.730 3.4 0.1 0.6 0.0 4.1 40.0 -35.9

13.302 3.3 0.1 0.6 0.0 4.0 40.0 -36.0

7.620 3.3 0.1 0.4 0.0 3.8 40.0 -36.2

11.744 3.2 0.1 0.5 0.0 3.8 40.0 -36.2

12.401 3.1 0.1 0.6 0.0 3.8 40.0 -36.2

13.220 3.0 0.1 0.6 0.0 3.7 40.0 -36.3

13.445 2.9 0.1 0.6 0.0 3.6 40.0 -36.4

13.468 2.9 0.1 0.6 0.0 3.6 40.0 -36.4

13.795 2.9 0.1 0.6 0.0 3.6 40.0 -36.4

11.908 2.6 0.1 0.5 0.0 3.2 40.0 -36.8

2

No deviations.

Probe position on DC Out Side of PS, DC Power Supply: Digital Power Model ADP90-119 SN ADP901195210A0010592

Autopulse 17 BPM Setting 4

1

Pass

RTCA DO-160F Category M: 2007 RTCA DO-160F Interconnecting Cables: 2007

09/29/11

PSA 2011.09.01

EMI 2009.8.29

CONDUCTED EMISSIONS

-10

0

10

20

30

40

50

60

70

80

0.1 1 10 100

MHz

dBuA
Freq Amplitude Transducer Cable

External 

Attenuation Detector Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) dBuA dBuA (dB)

13.139 2.4 0.1 0.6 0.0 3.1 40.0 -36.9

9.772 2.4 0.1 0.5 0.0 3.0 40.0 -37.0

11.086 2.4 0.1 0.5 0.0 3.0 40.0 -37.0

11.417 2.4 0.1 0.5 0.0 3.0 40.0 -37.0

13.271 2.3 0.1 0.6 0.0 3.0 40.0 -37.0
cond

EUT: Inogen 1G-3 Work Order: INOG0018

Serial Number: Prototype 1 Date:

Customer: Inogen Temperature: 23C

Attendees: Mike Pollack Humidity: 42%

Project: None Barometric Pres.: 1012mb

Tested by: Jeremiah Darden Power: 110VAC/60Hz Job Site: OC11

TEST SPECIFICATIONS TEST METHOD

TEST PARAMETERS

Cable or Line Tested DC Cable

COMMENTS

EUT OPERATING MODES

DEVIATIONS FROM TEST STANDARD

Run #

Configuration #

Results Signature

Freq Amplitude Transducer Cable

External 

Attenuation Detector Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) dBuA dBuA (dB)

8.236 6.7 0.1 0.4 0.0 7.2 40.0 -32.8

11.047 4.1 0.1 0.5 0.0 4.7 40.0 -35.3

11.004 3.5 0.1 0.5 0.0 4.1 40.0 -35.9

12.159 3.3 0.1 0.5 0.0 3.9 40.0 -36.1

11.818 3.3 0.1 0.5 0.0 3.9 40.0 -36.1

8.240 3.2 0.1 0.4 0.0 3.7 40.0 -36.3

9.610 3.0 0.1 0.5 0.0 3.6 40.0 -36.4

8.790 2.8 0.1 0.4 0.0 3.3 40.0 -36.7

11.302 2.7 0.1 0.5 0.0 3.3 40.0 -36.7

12.588 2.6 0.1 0.6 0.0 3.3 40.0 -36.7

11.331 2.6 0.1 0.5 0.0 3.2 40.0 -36.8

7.064 2.6 0.1 0.4 0.0 3.1 40.0 -36.9

7.533 2.6 0.1 0.4 0.0 3.1 40.0 -36.9

10.841 2.5 0.1 0.5 0.0 3.1 40.0 -36.9

3

No deviations.

Probe position on DC Out Side of EUT, DC Power Supply: Digital Power Model ADP90-119 SN ADP901195210A0010592

Autopulse 17 BPM Setting 4

1

Pass

RTCA DO-160F Category M: 2007 RTCA DO-160F Interconnecting Cables: 2007

09/29/11

PSA 2011.09.01

EMI 2009.8.29

CONDUCTED EMISSIONS

-10

0

10

20

30

40

50

60

70

80

0.1 1 10 100

MHz

dBuA
Freq Amplitude Transducer Cable

External 

Attenuation Detector Adjusted Spec. Limit

Compared to 

Spec.

(MHz) (dBuV) (dB) (dB) (dB) (blank equal peaks 

[PK] from scan) dBuA dBuA (dB)

2.055 2.7 0.1 0.2 0.0 3.0 40.0 -37.0

13.221 2.3 0.1 0.6 0.0 3.0 40.0 -37.0

11.076 2.3 0.1 0.5 0.0 2.9 40.0 -37.1

10.910 2.2 0.1 0.5 0.0 2.8 40.0 -37.2

11.577 2.2 0.1 0.5 0.0 2.8 40.0 -37.2
photo

ce (4).JPG

ce (1).JPG

 CONDUCTED EMISSIONS

PSA 2011.09.01
photo

ce (2).JPG

ce (3).JPG

 CONDUCTED EMISSIONS

PSA 2011.09.01