OxywellTM

User Manual

Oxywell
2 — 4805-C3_01 A

Oxywell

Symbol Definition Symbol Definition

Read user manual before operation. See user manual 

for instructions.

FAA Approved Symbol: The U.S. Federal Aviation 

Administration (FAA) has approved this device for 

use on-board commercial aircraft. 

No Smoking Icon: Do not smoke near unit. Flow Setting Indicator

Warnings / ALERT (Yellow) Indicator Pulse Mode Operation

ALARM (Red) Indicator Device operating normally; power button

External Power Indicator Increase Flow Setting

Power Cartridge (battery) Status Gauge Decrease Flow Setting

Power Cartridge Symbol Use no oil or grease.

A Amperes No open or naked flames.

Alternating Current No serviceable parts inside. Do not open cover.

IEC Icon for DC Power (Input) Warm Surface.

02 Oxygen Output Do not get wet.

Type B Applied Part (degree of protection against

electric shock)

Certified for both the U.S. and Canadian markets, to 

the applicable U.S. and Canadian standards.

Recycle Symbol

This device complies with the requirements of 

Directive 93/42/EEC concerning medical devices. It 

therefore bears the CE marking as shown.

This symbol is to remind the equipment owners 

to return it to a recycling facility at the end of its 

life, per Waste Electrical and Electronic Equipment 

(WEEE) Directive.

IPX1

Drip Proof Equipment-IPX1: The Oxywell Power 

Cartridge (Battery), AC and DC Power Supplies 

provide protection against the harmful effects of the 

ingress of liquids. (IPX1, per IEC 60529)

Drip Proof Equipment-IPX0: The Oxywell enclosure 

does not provide protection against the harmful 

effects of the ingress of liquids. (IPX0, per IEC 

60529)

Name and address of manufacturer

Authorized representative in the European 

Community

User Controls & System Status Indicators
Oxywell User Manual — 3

Oxywell

Contents

User Controls & System Status Indicators ..............................................................................................2

Quick Start Guide ......................................................................................................................................4

Indications for Use ....................................................................................................................................7

Contraindications for Use.........................................................................................................................7

Safety Guidelines ......................................................................................................................................7

Recommended Operating Environments ................................................................................................9

Operating Instructions............................................................................................................................10

First Time Oxywell Operation.................................................................................................................14

User Troubleshooting Table ...................................................................................................................16

Cleaning, Care & Routine Maintenance.................................................................................................17

General Airline Travel Information ........................................................................................................19

Power Cartridge & Airline Travel Table .................................................................................................19
4 — 4805-C3_01 A

Oxywell

Quick Start Guide

AC Power Supply 

WITH NEMA POWER CORD

DC Cable

Oxywell Cart

Power 

Cartridge 

(battery)

1 Unpack Your Oxywell 2 Getting to Know Your Oxywell

Review all Warnings, Cautions and additional device information 

in the rest of this manual.

Become familiar with the key features of the Oxywell and the 

User Control Panel.

BACK

Power 

Cartridge 

Latch

Exhaust 

Vent

Power

Cartridge

(Battery)

Service 

Cabinet

FRONT

Control 

Panel Handle

External Power

Receptacle

Oxygen

Outlet Port

FAA 

Approval

Icon

Rating Label & 

Serial Number 

Location

Air inlet

Filter

Service Port

Filter
Oxywell User Manual — 5

Oxywell

ON/OFF Button (Green) Indicator

Press and hold the “ON/OFF” button for two (2) seconds to 

power ON your Oxywell. A brief audible and visual power on 

self-test will occur. Verify that all the indicators are illuminated 

and the buzzer sounds for three (3) seconds. Press and hold 

the “ON/OFF” button for two (2) seconds to power OFF your 

Oxywell.

To connect to AC power source: Insert the AC Power Supply cord 

securely into the External Power Receptacle on the right side of 

the device. Plug the AC power cord into a grounded AC outlet. 

When the Oxywell is plugged in properly, a green indicator on 

the AC Power Supply will light and the External Power Indicator, 

will appear on the user control panel. 

Install the Battery: Slide the battery into the empty compartment 

until it locks into position and is flush with the front of the 

device.

When the battery is properly installed and the Oxywell is ON, the 

Power Cartridge Status Gauge will appear. 

step 2

3 The Oxywell User Control Panel displays important operating Powering on the Oxywell

information.

ON/OFF Button (Green) Indicator: This button powers 

the device ON or OFF. The Green Indicator is illuminated 

when the device is ON and functioning properly.

Increase or Decrease Flow Setting Buttons: Use these 

buttons to set the flow to your prescribed setting.

Delivery Mode Button and Indicator: The button toggles 

between Continuous Flow and Pulse Dose Mode. The 

Pulse Dose Mode activates autoSAT Technology and 

allows a significant increase in the operating time 

while powered by the battery. When the Pulse Dose 

Mode is activated, the green Pulse Dose Mode Indicator 

illuminates and a pulse of oxygen is delivered with 

each inspiratory effort.

ALERT (Yellow) Indicator—Low and Medium Priority 

Alerts: When illuminated, this indicates a low priority 

awareness condition or Caution. Continue to use your 

system and refer to the Troubleshooting Table for the 

proper response, or contact your Home Care Provider. 

A flashing yellow indicates a medium priority alert. A 

prompt response is necessary. (Typically low oxygen or 

low battery.)

ALARM (Red) Indicator—High Priority Alarms: Indicates 

a high priority alarm condition. An immediate response 

is necessary. Refer to the Troubleshooting Table and 

contact your home care provider. (Typically loss of 

power, low oxygen, or malfunction.)

45m

C 3LPM

Information Display. 

• C or P – “C” indicates Continuous Flow mode and 

“P” indicates Pulse Dose mode.

• Time on battery – indicates estimated time remaining 

on battery

• Setting (continuous flow or pulse dose) – each time 

you power the device ON, the previous mode and/or 

setting has been saved and will be used at start-up.

Power Cartridge (battery) Status Gauge: This indicator 

displays the charge remaining in the battery. Each of 

the five horizontal gray bars represents approximately 

20% of the total battery charge. When the battery is 

being charged, the charge indicator bars will blink in 

a waterfall-type fashion. If the battery is not installed, 

or if it is improperly installed, the Power Cartridge 

(battery) Status Gauge will not be illuminated or will 

flash on and off.

External Power Indicator: When the Oxywell is properly 

plugged in and is using the AC or 12VDC Cable, this 

indicator will appear on the User Control Panel.

Buzzer: An audible alarm (or buzzer) alerts you to the operating condition 

of the device, either a warning or failure, and confirms a valid key press.
6 — 4805-C3_01 A

Oxywell

Connect the oxygen supply tubing to the oxygen outlet and 

connect the oxygen cannula per the instructions provided with the 

cannula, or per the instructions from your home care provider.

Breathe normally through the nasal cannula.

When operating on battery power, utilizing the Pulse Dose Mode 

will achieve longer duration-of-use.

Align the locater holes on the bottom of the device to the tabs on 

the Universal Cart. Align the large threaded screw to the insert at 

the back of the Oxywell. Then hand-tighten the knob on the cart 

until the Universal Cart is tightly secured to the device. Depress 

the push-button on the telescoping Universal Cart handle to 

adjust the height of the handle.

Your unit is equipped with an Air Inlet Filter (extra filter 

provided). Check daily and clean once a week.

Air Inlet Filter

6 Begin Using the Oxywell

7 Universal Cart Operation

Flow Mode Button and Indicator

Press the Flow Mode button to select your desired flow mode. 

Pressing this button repeatedly will toggle you back and forth 

between continuous flow and pulse dose modes. In Continuous 

Flow Mode the oxygen is provided at a constant flow rate 

between 0.5 and 3.0 LPM. In Pulse Dose Mode, oxygen is 

supplied in a bolus at the beginning of each inspiration, providing 

a selectable range setting of 16mL to 192mL, and can be adjusted 

for sensitivity and breath bolus delivery. Flow settings in both 

modes should be prescribed for you by your physician.

Increase or Decrease Flow Setting Button

Using the + or - buttons, adjust the setting to the prescription 

supplied by your physician.

Continuous Flow Mode Display

Pulse Dose Mode Display 

4 Select Flow Delivery Mode

5 Adjust Flow Setting to Prescribed Level

WARNING: It is very important to select only the prescribed level of oxygen. Do not change the flow

selection unless you have been directed to do so

by a licensed clinician. The portable Oxygen Concentrator

may be used during sleep under the recommendation of a

qualified clinician.

Air inlet

Filter
Oxywell User Manual — 7

Oxywell

Important! 

Safety Instructions are defined as follows:

Indications for Use

Contraindications for Use

Safety Guidelines

WARNING: IMPORTANT SAFETY INFORMATION FOR HAZARDS THAT MIGHT CAUSE SERIOUS INJURY.

CAUTION: Important information for preventing damage to 

the Oxywell.

Note: Information needing special attention.

WARNING: It is very important to select only the

prescribed level of oxygen. Do not change the

flow selection unless you have been directed to do

so by a licensed clinician. The portable Oxygen Concentrator may be used during sleep under the recommendation

of a qualified clinician.

WARNING: Federal (USA) law restricts this device to sale or

rental by order of a physician or other licensed health

care provider.

WARNING: This unit is not to be used for life support. Geriatric, pediatric, or any other patient unable to communicate

discomfort while using this device may require additional

monitoring. Patients with hearing and/or sight impairment(s) may need assistance with monitoring alarms.

WARNING: No modification of this equipment is

permitted

WARNING: IN CERTAIN CIRCUMSTANCES, THE USE OF 

NON-PRESCRIBED OXYGEN CAN BE HAZARDOUS. THIS 

DEVICE SHOULD ONLY BE USED WHEN PRESCRIBED BY A 

PHYSICIAN.

WARNING: NOT FOR USE IN THE PRESENCE OF FLAMMABLE ANESTHETICS.

WARNING: AS WITH ANY ELECTRICALLY POWERED DEVICE, THE 

USER MAY EXPERIENCE PERIODS OF NON-OPERATION AS A RESULT 

OF ELECTRICAL POWER INTERRUPTION, OR THE NEED TO HAVE THE 

Oxywell SERVICED BY A QUALIFIED TECHNICIAN. THE Oxywell

IS NOT APPROPRIATE FOR ANY PATIENT WHO WOULD EXPERIENCE 

ADVERSE HEALTH CONSEQUENCES AS THE RESULT OF SUCH TEMPORARY INTERRUPTION.

WARNING: The Manufacturer recommends an alternate source of supplemental oxygen in the event

of a power outage, alarm condition, or mechanical failure. Consult your physician or Equipment Provider

for the type of reserve system required. 

WARNING: This device supplies high-concentration oxygen

that promotes rapid burning. Do not allow smoking or

open flames within the same room of (1) this device, or (2) 

any oxygen-carrying accessory. Failure to observe this

warning can result in severe fire, property damage, and/or

cause physical injury or death.

WaRNING: Do not operate unit in a restricted or confined

space (i.e., a small case or handbag) where ventilation can

be limited. This can cause the Oxygen Concentrator to overheat and impair performance.

WARNING: The concentrator should be located as to avoid

smoke, pollutants or fumes.

WARNING: The use of some oxygen administrating accessories not specified for this Oxygen concentrator may impair

its performance. Recommended accessories are referenced

within this manual. 

WARNING: If the oxygen concentrator has been dropped, 

damaged or exposed to water please contact your home

care provider for inspection or possible repair of the

device. Do not use the oxygen concentrator if it has a damaged power cord or plug.

WARNING: Properly secured, belt or otherwise restrain the

oxygen concentrator when in a vehicle during transport

to prevent damage or injury.

WARNING: Do not allow either the air intake or the air

outlet vents to become blocked. DO NOT drop or insert any

objects into any openings on the device. This can cause the

Oxygen Concentrator to overheat and impair performance.

WARNING: DO NOT overfill the optional humidifier. Fill the

optional humidifier with water only to the level shown by

the manufacturer of the humidifier.

WARNING: DO NOT operate the Oxywell and ambulate while a

humidifier is attached. Remove the humidifier bottle before

walking. DO NOT lay the Oxywell down while attached to a

humidifier bottle.

WARNING: The US Department of Transportation (DOT) and

United Nations (UN) Regulations require the removal of the

battery from the device for all international airline travel when the oxygen concentrator is checked as luggage. 

When shipping the oxygen concentrator, the battery must

also be removed from the device and packaged properly.

WARNING: DO NOT OPERATE THIS EQUIPMENT WITHOUT FIRST 

READING AND UNDERSTANDING THIS MANUAL. IF YOU ARE UNABLE 

TO UNDERSTAND THE WARNINGS AND INSTRUCTIONS, CONTACT 

YOUR EQUIPMENT PROVIDER BEFORE ATTEMPTING TO USE THIS 

EQUIPMENT; OTHERWISE, INJURY OR DAMAGE MAY RESULT.
8 — 4805-C3_01 A

Oxywell

WARNING: If you feel discomfort or are experiencing a medical emergency, seek medical assistance

immediately.

WARNING: Operating the oxygen concentrator outside of

the operational temperature specifications can limit the

concentrator’s ability to meet Oxygen Concentration specification. Refer to the specification section of this manual

for temperature limits.

WARNING: Use no oil, grease, or petroleum-based or other

flammable products with the oxygen-carrying accessories

or the Oxygen Concentrator. Only water based, oxygen

compatible lotions or salves should be used. Oxygen accelerates the combustion of flammable substances.

WARNING: The incorrect use of the battery can cause the

battery to get hot, ignite, and may cause serious injury. Be

sure not to pierce, strike, step on, drop the battery, or otherwise subject the battery to strong impacts or shocks. The

use of a damaged battery may cause personal injury.

WARNING: While using the portable Oxygen Concentrator

outdoors with the AC power supply, connect the power supply into a Ground Fault Interrupted (GFI) outlet only.

WARNING: The Oxygen Concentrator should not be used adjacent to or stacked with other equipment. If adjacent or

stacked use is unavoidable, the device should be observed

to verify normal operation.

CAUTION: In the event of an alarm or you observe the 

Oxygen Concentrator is not working properly; consult 

the Troubleshooting section in this manual. If you cannot 

resolve the problem, consult your Equipment Provider.

CAUTION: To prevent a void warranty, follow all manufacturers’ 

instructions.

WARNING: Smoking while using oxygen is the number one cause of fire injuries and related deaths. 

You must follow these safety warnings:

WARNING: Do not allow smoking, candles, or open flames

in the same room with the device or the oxygen-carrying

accessories. 

WARNING: Smoking while wearing an oxygen cannula may

cause facial burns and possibly death.

WARNING: If you smoke, these 3 steps may save your life: 

turn off the oxygen concentrator, take off the cannula, 

and leave the room where this device is located. 

WARNING: “No Smoking – Oxygen in Use” signs must be

prominently displayed in the home, or where oxygen is in

use. Patients and their caregivers must be informed about

the dangers of smoking in the presence of, or while using, 

medical oxygen.

WARNING: Do not use your oxygen concentrator in the

presence of flammable gases. This can result in rapid

burning causing property damage, bodily injuries or

death.

WARNING: Removing the cannula and putting it on clothing, bedding, sofas, or other cushion material will cause

a flash fire when exposed to a cigarette, heat source, or

flame. 

WARNING: Do not leave a nasal cannula on or under clothing, bed coverings or chair cushions. If the unit is turned

on but not in use, the oxygen will make the material flammable. Set the I/0 power switch to the 0 (Off) position when

the Oxygen Concentrator is not in use.

WARNING: Do not attempt any maintenance other

than the possible solutions listed within this

manual. Do not remove covers, only your Equipment Provider or a qualified service technician should

remove the covers or service this device. 

WARNING: Only accessories recommended by the manufacturer. Use of any other may be hazardous, cause serious

damage to your oxygen concentrator and will void the

warranty .

CAUTION: Always disconnect AC power supply from the 

wall before disconnection the AC power supply from the 

oxygen concentrator.

CAUTION: Always place oxygen supply tubing and power cords in a 

manner that prevents a trip hazard.

CAUTION: When using the Oxygen Concentrator in an automobile, 

boat, or on other DC sources with the 12VDC Cable, make sure that 

the vehicle is started and running before connecting the Oxygen 

Concentrator. If the 12VDC Cable does not illuminate and requires 

resetting, disconnect the 12VDC Cable from the DC outlet, restart 

your vehicle, and then reconnect your 12VDC Cable into the DC 

outlet. Failure to follow these instructions can result in the power 

supply not supplying power to the Oxygen Concentrator.

CAUTION: When the automobile in which you are using the Oxygen 

Concentrator unit is turned off, disconnect and remove the device 

from the automobile. Do not store the Oxygen Concentrator in a 

very hot or cold automobile or in other similar, high-or low-temperature environments.

Note: Portable and mobile RF communications equipment can effect 

medical electrical equipment.

WARNING: The Manufacturer recommends an alternate source of supplemental oxygen in the event

of a power outage, alarm condition, or mechanical failure. Consult your physician or Equipment

Provider for the type of reserve system required. 

WARNING: Do not allow either the air intake or the air

outlet vents to become blocked. DO NOT drop or insert any

objects into any openings on the device. This can cause the

Oxygen Concentrator to overheat and impair performance.
Oxywell User Manual — 9

Oxywell

Recommended Operating Environments

For proper use of the device, the following chart provides 

important information concerning the recommended operating 

environments, or operating conditions.

Specifications

Operating 

Temperature

32o F to 104o F (0o C to 40o C)

Operating 

Humidity

10% - 95% at an 82.4° F (28° C) dew point

Transport/

Storage 

Temperature

 -4o F to 140o F (-20o C to 60o C) 

Humidity: Up to 95% Non-Condensing

Electrical Use no extension cords. Use no electrical outlets controlled by a switch.

Altitude 0 – 13,123 feet (0 - 4,000 meters)

Placement DO NOT block the air inlet (top back cover) or the exhaust 

vent (back bottom left). Place device a minimum of 3 inches (7.5 cm) away from walls, draperies, furniture, etc.

Environment Must be smoke, pollutant, and fume free.

Operating 

Time

24-hours a day when connected to an external AC or DC 

power source.

Proper Placement of the Oxywell

Select a location for the device that avoids the intake of smoke, 

fumes and pollutants. Correct placement of the device should 

allow intake of air through the air inlet filter at the top rear of the 

cabinet and allow exhaust air to freely leave the exhaust vent at 

the bottom right of the device. 

Place the device such that the alarms can be heard.

Position the oxygen supply tubing in such a way that it does not 

kink or occlude.

Note: If the Oxygen Concentrator has been stored for an extended period 

of time outside its normal operating temperature range, the unit should 

be allowed to return to normal operating temperature before being turned 

on. (Refer to the Specifications section in this manual.) Note: When the automobile in which you are using the Oxygen Concentrator unit is turned off, disconnect and remove the device from the 

automobile. Do not store the Oxygen Concentrator in a very hot or cold 

automobile or in other similar, high-or low-temperature environments. 

DO NOT leave the Oxygen Concentrator or the Power Supply plugged into 

the vehicle if the ignition is in the off position. Doing so may drain the 

vehicle’s battery.

WARNING: Do not use your oxygen concentrator in

the presence of flammable gases. This can result

in rapid burning causing property damage, bodily

injuries or death. Use no oil, grease, or petroleum-based

or other flammable products with the oxygen-carrying

accessories or the Oxygen Concentrator. Only water

based, oxygen compatible lotions or salves should be used. 

Oxygen accelerates the combustion of flammable substances.

WARNING: This device supplies high-concentration oxygen

that promotes rapid burning. Do not allow smoking or

open flames within the same room of (1) this device, or (2) 

any oxygen-carrying accessory. Failure to observe this

warning can result in severe fire, property damage, and/

or cause physical injury or death.

CAUTION: Do not operate unit in a restricted or confined 

space (i.e., a small case or handbag) where ventilation 

can be limited. This can cause the Oxygen Concentrator to 

overheat and impair performance. Do not allow either the air intake 

or the air outlet vents to become blocked. This can cause the Oxygen 

Concentrator to overheat and impair performance.
10 — 4805-C3_01 A

Oxywell

AC POWER SUPPLY

The Oxywell includes a universal AC Power Supply for use at 

home or wherever standard AC power is available. To connect 

to an AC power source, be sure the AC Power Supply cord is 

securely inserted into the recessed power receptacle on the side 

of the device and that the cord from the AC Power Supply is 

plugged into a grounded AC outlet. When the Oxywell is plugged 

in properly, a green indicator on the AC Power Supply will light 

and the External Power Indicator, will appear on the 

user control panel. 

Operating Instructions

Before Operating

This users manual serves as your reference to help you operate 

and maintain the device. If you have any questions or concerns 

please call your home care provider.

Important! DO NOT attempt to operate the Oxywell without 

first reading the Safety Guidelines section of this manual. 

Please follow all of the operating instructions. Please observe 

all Warnings on the device and in the Users Manual. In order to 

reduce the risk of fire, personal injury and serious damage to the 

Oxywell, please observe all of the safety precautions. 

Powering on your Oxywell 

The device is capable of being operated directly from three (3) 

different power sources:

• AC Power Supply with NEMA Power Cord

• 12VDC Cable

• Rechargeable Power Cartridge (Battery)

General Charging Information

The Oxywell operates from external AC power, DC power, or 

a rechargeable battery. The system includes a “smart battery 

charger” that recharges the internal battery whenever the Oxywell 

is connected to external power. The system monitors and controls 

the power source and the battery charger.

The Oxywell charges the battery when using AC power and 

when the battery temperature is within safe charging temperature. 

When external power is disconnected, the device will 

automatically switch over to the battery, as long as the battery 

is installed and charged. When external power is restored by 

plugging into an AC outlet, the device will accept power from the 

external power source and replenish the battery. 

Proper connection of the AC Supply or 12VDC cable requires 

aligning the power plug with the recessed external power 

receptacle on the Oxywell. The recessed external power 

receptacle is located on the right side of the device when 

facing the Control Panel. Push the power plug into the recessed 

receptacle.

WARNING: Clean the cabinet, control panel, and

power cord only with a mild household cleaner

applied with a damp (not wet) cloth or sponge, 

and then wipe all surfaces dry. Do not allow any liquid to

get inside the device. Pay special attention to the oxygen

outlet for the cannula connection to make sure it remains free of dust, water, and particles.

CAUTION: Do not allow either the air intake or the air 

outlet vents to become blocked. This can cause the Oxygen 

Concentrator to overheat and impair performance. Do not 

operate the Oxygen Concentrator without the air intake filter in 

place. If a second filter is provided, insert the “replacement” filter 

before you clean the dirty filter. Clean the dirty filter in a warm soap 

and water solution then dry thoroughly prior to use.

AC Power Supply

with NEMA Power Cord

CAUTION: Use only accessories recommended by the 

manufacturer. Use of any other may be hazardous, 

cause serious damage to your oxygen concentrator 

and will void the warranty.

CAUTION: Do not use extension cords with this unit or connect too 

many plugs into the same electrical outlet. The use of extension 

cords could adversely affect the performance of the device. Too 

many plugs into one outlet can result in an overload to the electrical 

panel causing the breaker/fuse to activate or fire if the breaker or 

fuse fails to operate.

CAUTION: Use of cables and adapters other than those specified, 

with the exception of cables and adapters sold by the manufacturer of the medical electrical equipment as replacement parts for 

internal components, may result in increased emissions of decreased 

immunity of the Oxygen Concentrator.

Note: Use only electrical voltage specified on the specification label 

affixed to the device.

Step 2
Oxywell User Manual — 11

Oxywell

12VDC Cable

12VDC Cable

A 12VDC Cable allows the system to operate from DC outlets, 

such as those found in motor vehicles. 

1. Start your vehicle. 

2. Insert the 12 Volt DC cord into the recessed power 

receptacle on the side of the device. 

3. Insert the 12VDC Cable plug into the DC power outlet in 

your motor vehicle.

When the device is properly connected and receiving power from 

the DC power source, a green indicator light on the Power Supply 

and the External Power Indicator on the control panel will 

illuminate.

Oxywell DC Supply Connection

When the Oxywell is connected to the 12VDC Cable in a vehicle, 

the following applies:

• The Oxywell will run at 3.0LPM on continuous flow while 

being operated with the DC Supply in a car.

• The Oxywell will charge the battery while on 12VDC Cable in 

a car while running on continuos flow at 2LPM or less.

• If the Oxywell battery is being charged, the battery symbol will 

waterfall. If the Oxywell battery is not charging, the battery 

symbol will be displayed solid.

Settings Available When Operating on DC for Oxywell 5

Flow Setting Continuous Bolus Size (mL) Pulse

0.5 Yes - -

1.0 Yes 16 Yes

1.5 Yes 24 Yes

2.0 Yes 32 Yes

2.5 Yes 40 Yes

3.0 Yes 48 Yes

- - 56 Yes

- - 64 Yes

- - 72 Yes

- - 80 Yes

- - 88 Yes

- - 96 Yes

- - 128 Yes

- - 160 Yes

- - 192 Yes

Note: While operating the device from a 12VDC Cable, the Battery may 

not charge.

Note: When operating the Oxywell from a 12VDC Cable, it is recommended that the Pulse Dose Mode be used if prescribed to use a pulse dose 

device.

Note: To ensure that the device is utilizing the DC power, check that the 

External Power Indicator is illuminated. If a beep is heard, this could indicate a loss of external power. Check if the green Battery Power Verification 

LED is on or off. If the battery is removed on DC power, it will ensure no 

loss of charge while the unit is running. If the indicator is not illuminated, 

the battery is powering the device.

WARNING: Do not allow either the air intake or the

air outlet vents to become blocked. This can cause

the Oxygen Concentrator to overheat and impair

performance.

WARNING: If the vehicle’s DC power source drops below 11.5 

Volts, the Oxywell will revert to the Battery operation (if

the battery is present AND CHARGED). (The External Power

Indicator may remain on.)

Note: Actual performance of the Oxywell may vary depending upon usage 

conditions of the motor vehicle.
12 — 4805-C3_01 A

Oxywell

Initial Battery Charging

The new Battery supplied with your Oxywell is not fully charged 

when it is shipped from the factory. Before using your Oxywell 

for the first time, you must first fully charge the Battery. 

With the AC Power Supply plugged in, and the Battery correctly 

installed in the Power Compartment of the device, allow the 

Battery to completely charge. The Battery is fully charged when 

the Power Cartridge Status Gauge, on the Control Panel, is no 

longer cascading.

The Oxywell can be used while either discharging or recharging 

the Battery. It may take up to 4 hours, dependent upon the flow 

setting, to achieve 80% capacity from a fully discharged Battery. 

It will charge when the Oxywell is plugged into AC power and 

turned off.

To remove the Battery, pinch the black release lever. While 

holding DOWN the release lever, pull the Battery away from the 

Oxywell.

Battery Operating Times: The battery operating time is displayed 

on the Control Panel, in both symbol and alpha-numeric formats. 

A variety of factors, such as flow setting, Pulse or Continuous 

Flow Mode, and breath rate will impact the operating time. The 

table Typical New Power Cartridge Operating Times provides 

operating time estimates for the Oxywell using a new, fully 

charged Battery based on flow settings and operating conditions 

as indicated. 

WARNING: For oxygen concentrators equipped with

batteries: Store in a cool and dry location to help

ensure the longevity of your battery. Storing your

Oxygen Concentrator for extended periods of time at high

temperatures or with a fully charged or completely discharged battery can degrade its overall battery life. Do

not attempt to open the battery; there are no serviceable

parts inside the battery. Keep batteries away from children.

In the event of an AC or DC power interruption, the Oxywell will 

beep and automatically switch to Battery operation, as long as 

the Battery is present and charged. When AC power is restored, 

the Oxywell will beep and the Battery will automatically start 

recharging, if not fully charged or too warm. If the Battery is not 

present, or fully discharged, during an AC power interruption, 

the Oxywell will shut down and alarm. The Battery is fully 

discharged when the Power Cartridge Status Gauge on the 

control panel is no longer illuminated. The device will display a 

yellow light and sound a low Battery alert beep until the device 

completely shuts down with a red light and constant alarm. This 

audible alert may continue for up to 5 minutes until the device 

shuts down.

* Batteries are recyclable. *

CAUTION: When the automobile in which you are using 

the Oxygen Concentrator unit is turned off, disconnect 

and remove the device from the automobile. Do not store 

the Oxygen Concentrator in a very hot or cold automobile or in 

other similar, high-or low-temperature environments. DO NOT leave 

the Oxygen Concentrator or the Power Supply plugged into the 

vehicle if the ignition is in the OFF position. Doing so may drain the 

vehicle’s battery.

CAUTION: Only use the Manufacturer’s provided batteries. For proper battery disposal, contact your Equipment Provider or your local 

government agency for disposal requirements.

Note: Press and hold the Oxywell ON/OFF button for a minimum of 5 

seconds, to silence the Loss of Power alarm. Connecting the Oxywell to an 

external AC power supply for 2 minutes will also eliminate this alarm.

Rechargeable Power Cartridge (Battery)

The Oxywell can be powered by the 

rechargeable Power Cartridge (Battery), which 

is supplied with the device. 

Installing the Battery: Align the Battery 

so that it will slide into the empty Power 

Compartment. Push the Battery into the empty 

Power Compartment until it locks (clicks) 

into position and is flush with the front of the 

device. When the Battery is properly installed 

and the Oxywell is ON, the Battery Status 

Gauge will appear on the Control Panel. 

(If you DO NOT have a Battery installed, 

the Power Cartridge Status Gauge will not 

illuminate). The unit will still operate on 

external power.

Rechargeable Power 

Cartridge (Battery)
Oxywell User Manual — 13

Oxywell

Typical Battery Recharge Time

The typical time to recharge your Battery to achieve 80% 

capacity from a fully discharged Battery is between 2 and 5 

hours, depending upon the device flow setting. When the battery 

is fully charged, the battery gauge will cease cascading. The 

battery can be charged in the unit when the Oxywell is turned off 

and plugged in.

If the Battery becomes too warm during discharging, recharging 

will not begin until the Battery sufficiently cools. Removing the 

Battery and allowing it to cool may expedite this cooling process.

The Battery will not recharge when operating from the 12VDC 

Cable.

The Battery charges when in the device is operating on AC power 

only. The Battery will also charge from the optional Desktop 

Charger.

Typical New Power Cartridge Operating Times

Operating Type Battery Type

12-cell 24-cell

Continous Mode

1 LPM 1.96 hrs 4.4 hrs

2 LPM 1.1 hrs 2.5 hrs

3 LPM 0.73 (44 min) 1.67 hrs

Pulse Mode (12 BPM)

Bolus 16 mL 2.38 hrs 5.4 hrs

Bolus 48 mL 2.16 hrs 4.9 hrs

Bolus 96 mL 1.54 hrs 3.5 hrs

Note: Operating the Oxygen Concentrator outside of its normal operating 

temperature range can affect performance and decrease battery run time 

and/or increase battery charge time. (Refer to the Specifications section in 

this manual.) You will achieve longer operating time on the Battery if you 

operate your Oxywell in the Pulse Mode (Physician approved mode).
14 — 4805-C3_01 A

Oxywell

Step 1: Positioning Your Oxywell for Use 

Place the Oxywell in a well-ventilated, well lit area. Be sure the 

air inlet and exhaust vents are not obstructed.

Position the Oxywell so that all 

audible and visual indicators or 

alarms can be easily seen and 

heard.

Be sure the air inlet filter is in 

place before operating your 

Oxywell. If the air inlet filter is 

missing, contact your home care 

provider. If the air inlet filter is 

dirty, wash with warm soapy 

water, rinse with clear water and 

allow it to dry before placing the 

air inlet filter back in the device. 

If necessary, replace it with a new, 

clean air inlet filter.

Plug the device into a grounded 

AC Power outlet, a DC Power source, or be sure there is a fully 

charged Battery installed. 

Step 2: Power ON the Device and Allow it to 

Warm-Up

Press and hold the “ON/OFF” Button for one (1) second to power 

ON your Oxywell. A brief visual power-on self-test will occur. 

Step 3: Connect the Oxygen Supply Tubing or Nasal 

Cannula to the Oxygen Outlet

Replace the cannula and oxygen supply tubing regularly, as recommended by 

your home care provider. Consult your home care provider for cannula and 

supply tubing cleaning and replacement information.

Note: Ensure the cannula is fully inserted and secure. This ensures that the 

Oxygen Concentrator can properly detect inspiration for oxygen delivery. 

During inhalation, you should hear or feel oxygen flow to the prongs of 

the nasal cannula.

Step 4: Select the Flow Mode

Press the Flow Mode button to select the desired mode. Pressing 

this button repeatedly will toggle you back and forth between 

Pulse Dose Mode and Continuous Flow Mode. The proper 

mode and proper setting should be prescribed for you by your 

physician.

Continuous Flow Mode Operation:

When operating in the Continuous Flow Mode, a continuous 

supply of oxygen measured in liters per minute (LPM) will flow 

through your tubing and nasal cannula. You may use a humidifier 

when in Continuous Flow mode.

Pulse Dose Mode Operation:

When operating in Pulse Mode, a bolus of oxygen, measured 

in milliliters (mL), is delivered upon inspiration. Remove the 

humidifier bottle before walking. DO NOT use a humidifier in 

the Pulse Dose Mode. DO NOT lay the Oxywell down while 

attached to a humidifier bottle.

To ensure proper saturations, secure a prescription for pulse dose 

from your physician with consideration to your needs at rest, 

during exercise, and when traveling at altitude. Contact your care 

provider for prescription and boarding forms.

When operating in pulse dose mode, the Oxywell supplies a 

continuous flow of oxygen if a breath is not detected. The device 

will continually check for an inspiratory effort every 15 seconds. 

Once an inspiration is detected, the device reverts to measured 

bolus delivery. To exit Pulse Dose Mode and engage Continuous 

Flow Mode, push the flow mode button.

First Time Oxywell Operation

The Oxywell has an oxygen concentration status indicator 

(OCSI) built in to the device. The OCSI continually monitors 

the oxygen output of the device and incorporates a green light 

(normal), yellow light (alert), red light (alarm) and a buzzer. 

These indicators will all illuminate during the power-on self-test 

at start-up. After initially powering ON the device, please allow 

up to five (5) minutes for the device to reach its performance 

specifications. When only the green indicator light is illuminated, 

the device has reached its performance specifications.

Note: If there are three beeps at start-up, this indicates a low battery 

condition for the 9-volt battery inside the cabinet of the Oxywell. Please 

contact your home care provider and request that the 9-volt battery be 

replaced. 

Air inlet Filter

Exhaust Vent
Oxywell User Manual — 15

Oxywell

Pulse Dose Mode and autoSat Features

Your Oxywell has a feature called autoSAT® that delivers a 

consistent-sized bolus of oxygen up to 40 breaths per minute (see 

Max Breath Rate table below). As your breath rate increases, ie. 

during ambulation, the autoSAT feature servo-controls the device, 

delivering the set bolus size. autoSAT Technology is proprietary 

to CAIRE and is utilized during Pulse Mode only.

Mas Breath Rate Bolus 

Volume (+/- 15%)

Pulse Dose Setting

Bolus Size

(± 15%) mL

Max Breath Rate

1.0 16 40

2.0 32 40

3.0 48 40

4.0 64 40

5.0 80 37

6.0 96 31

- 128 23

- 160 18

- 192 15

Step 5: Adjust the Flow Setting to the Prescribed 

Level

Using the Flow Setting Buttons, adjust the flow setting to the 

prescribed setting supplied by your physician.

Note: Bolus volume decreases as breath rate exceeds published range.

Note: The pulse dose setting (bolus) is not equal to LPM.

WARNING: Pulse dose Mode settings must be determined for each patient individually for their needs

at rest, during exercise and when traveling.

WARNING: The Oxywell may not be able to detect all inspiratory efforts in Pulse Mode. if the oxywell does not

respond to your inspiration efforts, ask your home care

provider to check the sensitivity setting.

Note: To Equipment Provider: The following oxygen administration accessories are recommended for use with the Oxygen Concentrator: 

• Nasal Cannula with 7 feet (2.1 m) of tubing: PN 5408-SEQ

• For any additional recommended accessories, please see the Accessories 

Catalog (PN ML-LOX0010) available on www.CAIREmedical.com.

WARNING: It is very important to select only the

prescribed level of oxygen. Do not change the

flow selection unless you have been directed to

do so by a licensed clinician.

Step 6: Begin Using Your Oxywell

Breathe normally through the nasal cannula.

Note: You will achieve longer operating time on the Battery if you operate 

your Oxywell in the Pulse Mode (Physician approved mode).

Step 7: Oxywell Cart Operation

The Oxywell Cart is designed for use on pedestrian-type surfaces.

Align the locater holes on the bottom of the device to the tabs on 

the Universal Cart.

Align the large threaded screw to the 

insert at the back of the Oxywell. Then 

hand-tighten the knob on the cart until the 

Universal Cart is tightly secured to the 

device.

Depress the push-button on the Universal 

Cart handle to adjust the height of the 

handle.

Step 8: Power Off the Device

Press and hold the “ON/OFF” Button for two (2) seconds to 

power OFF your Oxywell.

Store the Oxywell in a cool and dry location for a period no 

longer than 1 month. The Oxywell must run for a minimum of 2 

hours per month to ensure proper function of the unit.

Note: DO NOT operate the Oxywell and ambulate while a humidifier is 

attached. Remove the humidifier bottle before walking. DO NOT lay the 

Oxywell down while attached to a humidifier bottle.
16 — 4805-C3_01 A

Oxywell

User Troubleshooting Table 

Do Not Ignore Alarms

Symptom Possible Cause Your Action

Oxywell does not power 

on when ON/OFF button is 

pressed

No battery installed Install battery or plug into external power 

Battery is discharged or no 

external power is present

Plug into external power

AC Power Supply or 12v 

DC cable is not plugged in 

correctly

Check connections at outlet, Power Supply box, and at the Oxywell to ensure secure 

connections.

Insufficient power from the 

outlet

Ensure there is adequate power from the outlet. (Outlet may be controlled by a wall 

switch or the circuit breaker may have engaged.)

Other Contact your dealer/provider

No Oxygen

Filter blocked Clean Air Inlet Filter

Humidifier Check humidifier attachment and tubing

Oxywell not ON Power Oxywell ON

Tubing or cannula is kinked or 

not properly connected 

Check tubing, cannula, and connections

Other Contact your dealer/provider

Low Oxygen 

Concentration

Restriction in tubing or 

humidifier

Repair or replace tubing or humidifier. Position your Oxywell to ensure adequate air 

flow.

Filter restricted Clean or replace Air Inlet Filter. Position your Oxywell to ensure adequate air flow.

Exceeding maximum breath 

rate in Pulse Dose mode. See 

Pulse Dose mode Breath Rate 

Chart

Switch to Continuous Flow mode until respiratory rate reaches an acceptable rate.

Other Contact your dealer/provider

Low Oxygen Flow

Restriction in humidifier or 

tubing

Repair or replace humidifier or tubing

Filter Blocked Clean or replace Air Inlet Filter.

Other Contact your dealer/provider

No Oxygen Delivered in Pulse 

Flow Mode

Tubing/cannula longer than 7 

feet (2.1m)

Attach 7 foot (2.1m) tubing/cannula

Humidifier attached Remove humidifier

No inspiration detected Contact your dealer/provider

Unit in Continuous Flow mode Switch to Pulse Dose mode

Nasal cannula not in the nose, 

or mouth breathing

Place nasal cannula in nostrils and breathe through nose

Power Cartridge Status Gauge 

never indicates fully charged

Power Cartridge (battery) is 

aging

Contact your dealer/provider to replace Power cartridge (battery)

WARNING: The Manufacturer recommends an alternate source of supplemental oxygen in the event of a power outage, alarm condition, or mechanical failure. Consult your physician or Equipment Provider for the type of reserve

system required.
Oxywell User Manual — 17

Oxywell

Air Inlet Filter

Ambient air is drawn into the device through the air inlet located 

on the top rear of the device. Cleaning the air inlet filter is the 

most important maintenance activity you will perform as a user. 

Cleaning the air inlet filter will keep your Oxywell performing 

properly. Check this air inlet filter daily and clean it at least once 

a week using the following procedure:

1. Remove the air inlet filter 

2. Wash the filter in warm tap water using a mild soap 

detergent solution

3. Rinse the filter thoroughly with clear tap water and squeeze 

out the excess water

4. Allow the filter to air dry

5. Reinsert the filter in the cabinet

Cleaning tip: Keep a clean second filter as a replacement for use 

when one filter is drying.

Routine Maintenance

Servicing of the internal components inside the cabinet of the 

Oxywell must be conducted by a CAIRE trained and qualified 

service technician as needed. 

Preventative Maintenance (PM) to be completed on an annual 

or yearly basis. Consult your provider to arrange PM schedule. 

Your home care provider or qualified technician will perform 

inspections of the alarms, internal 9-volt battery, battery, internal 

filters and other internal parts only as needed.

User Care and Cleaning of the Device

Cannula Replacement

Replace your supply tubing and cannula on a regular basis as 

recommended by your home care provider. Your physician or 

home care provider will provide you with cleaning, disinfection 

and replacement information.

The air inlet filter should be replaced annually. If the Oxywell is 

used in a dusty environment, the filter may need to be replaced 

more frequently. You should not operate the Oxywell without an 

air inlet filter installed for more than 5 minutes.

Cleaning, Care & Routine Maintenance

CAUTION: Do not operate the Oxygen Concentrator without the air intake filter in place. If a second filter is provided, insert the “replacement” filter before you clean the 

dirty filter. Clean the dirty filter in a warm soap and water solution 

then dry thoroughly prior to use.

Note: The Manufacturer does not recommend the sterilization of this 

equipment.

Note: Do not operate the Oxygen Concentrator without the air intake 

filter in place. If a second filter is provided, insert the “replacement” filter 

before you clean the dirty filter. Clean the dirty filter in a warm soap and 

water solution then dry thoroughly prior to use.

Note: Always follow the cannula manufacturer’s instructions for proper 

use. Replace the disposable cannula as recommended by the cannula 

manufacturer or your Equipment Provider. Additional supplies are available from your Equipment Provider.

Note: If three beeps sound after the power ON self-test, this indicates the 

9v battery is low in power. Contact your home care provider if you hear 

three beeps.

Air inlet

Filter
18 — 4805-C3_01 A

Oxywell

Optional Humidifier

If used, clean your humidifier daily to reduce the possibility of 

contamination. Follow the cleaning recommendations provided 

by the home care provider, or the manufacturer of the humidifier. 

Replace the humidifier monthly, or as recommended by the 

manufacturer, or your home care provider. (Do not overfill the 

humidifier.)

WARNING: Do not use liquid directly on the unit. 

A list of undesirable chemical agents includes

but is not limited to the following: alcohol and

alcohol-based products, concentrated chlorine-based

products (ethylene chloride), and oil- based products

(Pine-Sol®, Lestoil®). These are NOT to be used to clean the

plastic housing on Oxygen Concentrator, as they can damage the unit’s plastic.

CAUTION: DO NOT operate the Oxywell and ambulate 

while a humidifier is attached. Remove the humidifier 

bottle before walking. DO NOT lay the Oxywell down while 

attached to a humidifier bottle.

Universal Cart

Before cleaning the Universal Cart, remove it from the Oxywell. 

Use a damp (not wet) cloth or sponge and mild detergent solution 

to clean the Universal Cart.

Power Cartridge (Battery)

The Battery in the Oxywell requires special care to assure a 

longer life and the highest level of performance. The SeQual 

Battery is the only approved Battery recommended for use with 

the Oxywell.

Use a damp (not wet) cloth or sponge to clean the Battery. First 

spray the cloth or sponge with a mild detergent then clean the 

Battery case and the latch. 

SeQual Recommended Monthly Battery Service: Run the 

Oxywell on Battery completely down to power loss condition 

(red light and alarm). Plug Oxywell into AC power and recharge 

until full on indicator. The device does not have to be running to 

charge (which allows for shorter charge time).

SeQual recommends replacing the power cartridge after 2 years.

Battery Disposal

Your Battery is rechargeable and can be recycled. Always return 

it to your home care provider for proper disposal. You can also 

contact your local city or town offices for instructions on proper 

disposal of the Battery.

AC Power Supply, 12VDC Cable, Power Cord and Outer Case: 

Turn OFF the Oxywell and disconnect from AC or DC power 

before any cleaning or disinfection activity. DO NOT spray the 

outer case directly. Use a damp (not wet) cloth or sponge. Spray 

the cloth or sponge with a mild detergent solution to clean the 

cabinet and power supplies. To disinfect the Oxywell, use Lysol® 

Brand II disinfectant (or equivalent). Proceed as directed by the 

cleaner manufacturer. 

WARNING: Electrical shock hazard. Turn OFF the

unit and disconnect the power cord from the

electrical outlet before you clean the unit to

prevent accidental electrical shock and burn hazard. Only

your Equipment Provider or a qualified service technician

should remove the covers or service the unit. Do not use

liquid directly on the unit. A list of undesirable chemical

agents includes but is not limited to the following: alcohol and alcohol-based products, concentrated chlorine-based products (ethylene chloride), and oil- based

products (Pine-Sol®, Lestoil®). These are NOT to be used to

clean the plastic housing on Oxygen Concentrator, as they

can damage the unit’s plastic.

WARNING: Electrical shock hazard. Disconnect the power

cord from the electrical outlet before you clean the unit

to prevent accidental electrical shock and burn hazard. 

Only your Equipment Provider or a qualified service technician should remove the covers or service the unit.

WARNING: Care should be taken to prevent the Oxygen Concentrator and the battery from getting wet or allowing

fluids to enter the unit. This can cause a malfunction or

shut down, and cause an increased risk for electrical

shock or burns.

WARNING: Clean the cabinet, control panel, and power

cord only with a mild household cleaner applied with a

damp (not wet) cloth or sponge, and then wipe all surfaces

dry. Do not allow any liquid to get inside the device. Pay

special attention to the oxygen outlet for the cannula

connection to make sure it remains free of dust, water, 

and particles.
Oxywell User Manual — 19

Oxywell

1.0 2.0 3.0 4.0 5.0 6.0 7.0 8.0 9.0 10.0

0.5 1 1 1 2 2 2 3 3 4 4

1.0 1 1 2 2 2 3 3 4 4 5

1.5 1 1 2 2 3 3 4 4 5 5

2.0 1 2 2 3 4 4 5 5 6 7

2.5 1 2 3 4 5 5 6 7 8 9

3.0 2 3 4 5 6 7 9 10 11 12

Continuous Flow

SETTING

FLIGHT LENGTH (in Hrs)*

ESTIMATED # OF POWER CARTRIDGES NEEDED

*Table accounts for the typical 150% duration requirement.

These tables calculate flight times only. 

Please allow for ground time (preflight check 

in, security check in, etc.) and layovers.

Most airlines require that you have an adequate number of 

fully charged batteries to power the device for at least 150% of 

the expected maximum trip duration (flight times, all ground 

time; before and after flight, and during connections and for 

unexpected delays). 

It is suggested to always carry 

an extra battery with you on trips.

Each airline has their own requirements 

for traveling with oxygen.

Contact your airline at least 48 hours prior to departure to inform 

them you will be traveling with the Oxywell.

1.0 2.0 3.0 4.0 5.0 6.0 7.0 8.0 9.0 10.0

16 1 1 1 2 2 2 3 3 4 4

24 1 1 2 2 2 3 3 3 4 4

32 1 1 2 2 3 3 3 4 4 5

40 1 1 2 2 3 3 4 4 5 5

48 1 1 2 2 3 3 4 4 5 5

56 1 2 2 3 3 4 4 5 5 6

64 1 2 2 3 3 4 4 5 6 6

72 1 2 2 3 3 4 5 5 6 6

80 1 2 2 3 4 4 5 6 6 7

88 1 2 2 3 4 5 5 6 7 7

96 1 2 3 3 4 5 5 6 7 8

Table is based on 20 Breaths Per Minute, for Pulse Dose Mode only.

128 1 2 2 3 3 4 5 5 6 6

160 1 2 3 3 4 5 6 6 7 8

192 1 2 3 4 5 6 7 8 8 9

Table is based on 15 Breaths Per Minute, for Pulse Dose Mode only.

Pulse Dose Mode

Power Cartridge & Airline Travel Table mL SETTING

FLIGHT LENGTH (in Hrs)*

During taxi, take-off, and landing, the Oxywell must be turned 

off and stowed under the seat or in another approved stowage 

location so as to not block the aisle way or the entry way into the 

row if the Oxywell will not be used.

During taxi, take-off, and landing, the user must be in a seat 

location that does not restrict any other passenger’s access to, or 

use of, any required emergency or regular exit, or the aisle(s) in 

the passenger compartment of the aircraft if the Oxywell is used.

User’s are not permitted to be seated in an exit row if using the 

Oxywell.

If the Oxywell is used when decompression of the cabin occurs 

and the oxygen cabin oxygen system deploys, then the user is to 

discontinue use of the Oxywell and use the aircraft supplemental 

oxygen. The Oxywell unit is to be turned off after securing the 

aircraft supplemental oxygen. 

Prior to travel, the user needs to inspect the Oxywell to ensure it 

is in good operational condition. 

General Airline Travel Information

CAUTION: The US Department of Transportation (DOT) and 

United Nations (UN) Regulations require the removal of 

the battery from the device for all international airline 

travel when the oxygen concentrator is checked as luggage. When 

shipping the oxygen concentrator, the battery must also be removed 

from the device and packaged properly.
SeQual® and Oxywell® and autoSAT® are trademarks of 

Chart SeQual Technologies Inc., Ball Ground, GA 30107

Salter Labs® is a registered mark of Salter Labs, Arvin, CA 92303

Lysol® is a registered trademark of Reckitt Benckiser, UK.

CAIRE Inc. 

2200 Airport Industrial Drive, Ste. 500

Ball Ground, GA 30107 USA

1-800-482-2473

Worldwide: www.CAIREmedical.com

Medical Product Services GmbH

Borngasse 20

35619 Braunfels, Germany

Worldwide: www.CAIREmedical.com

PN 4805-C3_01 A

Chart SeQual Technologies Inc. reserves the right to discontinue its products, or change the prices, materials, equipment, 

quality, descriptions, specifications and/or processes to its products at any time without prior notice and with no further 

obligation or consequence. All rights not expressly stated herein are reserved by us, as applicable.

Oxywell